An ultrasonic flow meter is a type of flow meter that measures the velocity of a fluid with ultrasound to calculate volume flow. Using ultrasonic transducers, the flow meter can measure the average velocity along the path of an emitted beam of ultrasound, by averaging the difference in measured transit time between the pulses of ultrasound propagating into and against the direction of the flow or by measuring the frequency shift from the Doppler effect. Ultrasonic flow meters are affected by the acoustic properties of the fluid and can be impacted by temperature, density, viscosity and suspended particulates depending on the exact flow meter. They vary greatly in purchase price but are often inexpensive to use and maintain because they do not use moving parts, unlike mechanical flow meters. Means of operation There are three different types of ultrasonic flow meters. Transmission (or contrapropagating transit-time) flow meters can be distinguished into in-line (intrusive, wetted) and clamp-on (non-intrusive) varieties. Ultrasonic flow meters that use the Doppler shift are called reflection or Doppler flow meters. The third type is the open-channel flow meter. Principle Time transit flow meter Ultrasonic flow meters measure the difference between the transit time of ultrasonic pulses propagating with and against the flow direction. This time difference is a measure for the average velocity of the fluid along the path of the ultrasonic beam. By using the absolute transit times and , both the averaged fluid velocity and the speed of sound can be calculated. Using these two transit times, the distance between receiving and transmitting transducers and the inclination angle , if we assume that sound has to go against the flow when going up and along the flow when returning down, then one can write the following equations from the definition of velocity: and By adding and subtracting the above equations we get, and where is the average velocity of the fluid along the sound path and is the speed of sound. Doppler shift flow meters Another method in ultrasonic flow metering is the use of the Doppler shift that results from the reflection of an ultrasonic beam off sonically reflective materials, such as solid particles or entrained air bubbles in a flowing fluid, or the turbulence of the fluid itself, if the liquid is clean. Doppler flowmeters are used for slurries, liquids with bubbles, gases with sound-reflecting particles. This type of flow meter can also