Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
9.5 Auxiliary supply [97A-98A] MULTICAL® 801 comprises a built-in auxiliary supply on terminals 97A-98A. The auxiliary supply is based on an unstabilized power supply. This means that the output voltage varies depending on load. The output current must not exceed 50 mA and the nominal output current is 35 mA.The auxiliary supply is suitable for e.g. supplying a Lon-module or a passive flow meter output.The built in auxiliary supply is available on terminals 97A-98A.The voltage on terminals 97A-98A varies according to load.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201683MULTICAL® 80110 Power supplyMULTICAL® 801 is available for 24 VAC or 230 VAC supply voltage.MULTICAL 801 Type 67-  Supply 230 VAC supply 24 VAC supply7 8As the connection PCB of MULTICAL® 801 is equipped with either a 24 VAC or a 230 VAC transformer, it is not possible to change the supply voltage of a previously supplied meter.10.1 Built in battery backup