Title: +135.01=370.001-OE401T_02-320 Datablad Energimåler UH50.._.pdf
The application-related technical data is only guaranteed together with the products mentioned in this data sheet. Siemens rejects any and all warranties in the event that third- party products are used.CE2N5324en 2018-08-18Technical dataBuilding TechnologiesProcessor unitFunction dataMeasuring range (national approvals may differ)0…180 °CRange of temperature differential ΔΘ3…120 KTemperature response threshold0.2 KThermal coefficientShifting compensatedTemperature-measuring error without sensor (0.5 + ΔΘmin./ΔΘ) %, max. 1.5% at ΔΘ = 3 KTemperature sensor.Sensing elementPt500 per EN 60751Temperature range0…150 °C (up to 45 mm mounting length) 0…180 °C (from 100 mm mounting length)Optical interfaceBasic designSimilar to EN 62056-21ProtocolPer EN 62056-21Cable lengthControl cable1.5, 3, and 5 mSensor typesDS M10x1 mm, sensor length 27.5 mm DS M10x1 mm, sensor length 38 mm Pocket short Ø 5.2x45 mm Pocket long Ø 6x100 mm Pocket long Ø 6x150 mm