Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8018 Temperature sensorsMULTICAL 801 uses either Pt100 or Pt500 temperature sensors according to EN 60751 (DIN/IEC 751). A Pt100 or Pt500 temperature sensor respectively is a platinum sensor, of which the nominal ohmic resistance is 100.000 Ω and 500,000 Ω at 0.00°C and 138.506 Ω and 692,528 Ω at 100.00°C respectively. All ohmic resistance values are determined in the international standard IEC 751, applying to Pt100 temperature sensors. The ohmic resistance values of Pt500 sensors are five times higher. The tables below include resistance values for each degree celcius in [Ω] for both Pt100 and Pt500 sensors:Pt100°C01234567890100.000 100.391 100.781 101.172 101.562 101.953 102.343 102.733 103.123 103.51310103.903 104.292 104.682 105.071 150.460 105.849 106.238 106.627 107.016 107.40520107.794 108.182 108.570 108.959 109.347 109.735 110.123 110.510 110.898 111.28630