Title: +038=370.104-OE004T_01-Multical 801 Energimåler Kamstrup - Installasjons- og brukerguide.pdf
The module has connection possibility for two pressure transmitters on terminals 57, 58 and 59 and can be adjusted for current reading or pressure range 6, 10 or 16 bar. The module is prepared for remote reading, data from meter/module being transferred to the system software via the external GSM/GPRS modem connected on terminals 62, 63 and 64. Furthermore, the module has two extra pulse inputs, VA and VB. The module must be powered by 24 VAC. 8.7 Lon Works, type 67-00-24/67-0Y Re mounting of Lon Works type 67-00-24, see installation guide 5512-396 (DK) or 5512-403 (GB). 8.8 Wireless M-Bus, type 67-00-30/67-00-31/67-00-35/67-00-38 The radio module has been designed to form part of the hand-held Wireless M-Bus Reader systems of Kamstrup A/S at license-free radio frequency (868 MHz). The module fulfils the C-mode specifications of prEN13757-4 and can thus form part of other systems using Wireless M-Bus C-mode communication. The radio module comes with internal antenna and external antenna connection as well as