Title: +038=310.101-RF022T_06-multical 62 Vannmåler Kamstrup - Installasjons- og brukerveiledning.pdf
If a longer distance (up to 10 m) is required between flow sensor and display unit, Pulse Transmitter (type number 66-99-618) can be used.See instructions 5512-587 for further information.Smaller cold water meters 1.6 to 10 m3/h (except for G¾Bx110 and G1Bx110) can be fitted with a strainer (filter) and/or nonreturn valve (backflow preventer). The enclosed special polyethylene gaskets must be used. Strainer, nonreturn valve and special gaskets must only be used in cold water meters.1.1 Permissible operating conditions / measuring range Temperature of medium in flow sensor:Cold water meter: 0.1...50 °CHot water meter: 0.1...90 °CPressure stage:Threaded meters PN16 Flange meters PN25Mechanical environment:M1 (MID). Fixed installation with minimum vibra- tion.Electromagnetic environment:E1 (MID). Housing and light industry. The meter’s control cable must be drawn at min. 25 cm distance from other installations.Climatic environment: