Title: +175=574.001_OE001_Manual - Varmemåler Hydrocal M3 - På rør.Produktinformasjon.pdf
The configuration screens are divided into 10 sub-levels. Here below there is a “step by step” guide for the correct configuration of the heat meter:1) The first parameter that has be configured is the pulse value "k" of the C1 cold/hot water meter.The value can be selected by using the button T1: 0.1–0.25–1–2.5–10–25–100–250–1000 L/impConfirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.2) The second parameter that has be configured is the update of the consumption data reported on the C1 hot/cold water meter dial. Select the digit using the T2 button and change the single numbers with the T1 button.Hydrocal M3 Manuale utente – User Manual - v1.438Confirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.3) The third parameter that has be configured is the pulse value "k" of the C2 cold/hot water meter.