Title: 370.011 Siemens UC50 Kalkulator.pdf
Connect the conductors as shown on the circuit diagram printed on the unit. The 2-wire connection is always made at terminals 5/6 and 7/8 (also when a meter is provided for 4-wire connection). No cable shield must be connected to the meter. Then insert the sensors in the pockets, ball valves, or T-elements and seal against tamper. The end of the sensors must suffice at least by the middle of the pipe cross section. Replace the housing cover and press it in gently until you hear all the tabs latch.Connection of flow sensor7/294Connection of flow sensor