Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
The insulation between the connected mains supply and the low voltage supply for the flowmeter must be rated with at least double or reinforced insulation at mains voltage.For field wiring installation: Ensure that the National Installation Code of the country in which the flowmeters are installed is met.5.1Wiring the transducer cablesThe system SONOKIT with FUS080 transmitter is supplied with separate transducer coaxial cables:2 cables for 1-pathand4 cables for 2-pathThe cables can be used on all transducers (cables are not paired with a specific transducer).The cables are manufactured with crimp on the cable ends for the connection board.NoteIf the cables are shortened, all cables must be shortened equally, and the factory set cable length must be changed accordingly in the transmitter.SITRANS FUS080 transmitter22Operating Instructions, 07/2012, A5E03059912-02Connecting 5.1 Wiring the transducer cables1. Snap out connection plate and loosen grounding wire.