Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
About 35 s.13.3 True energy calculationDuring test and verification the heat meter’s energy calculation is compared to the ”true energy” calculated according to the formula of EN 1434-1:2004 or OIML R75:2002.An energy calculator like the one shown below can be ordered from Kamstrup A/S:The true energy at the most frequently used verification points is indicated in the table below.T1 [°C]T2 [°C]∆Θ [K]Inlet [Wh/0.1 m3]Outlet [Wh/0.1 m3]42 43 53 50 70 80 160 160 17540 40 50 40 50 60 40 20 202 3 3 10 20 20 120 140 155230,11 345,02 343,62 1146,70 2272,03 2261,08 12793,12 14900,00 16270,32230,29 345,43 344,11 1151,55 2295,86 2287,57 13988,44 16390,83 18204,78104Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 80114 METERTOOL HCW14.1 IntroductionThe Kamstrup Software product “METERTOOL HCW” (66-99-724) is used for the configuration of MULTICAL 801 as well as the configuration of other Kamstrup heat, cooling and water meters.System requirements