Title: +038=310.101-RF022T_07-multical 62 Vannmåler Kamstrup - Produktdatablad.pdf
Supply No supply Battery, D-cell 230 VAC high power isolated SMPS 24 VAC high power isolated SMPS 230 VAC isolated linear supply 24 VAC isolated linear supply0 2 3 4 7 8Flow part Supplied with one ULTRAFLOW® 241Meter type Hot water meter Cold water meter7 8Country code (language on label etc.)XX858101035_G1_GB_01.2018ULTRAFLOW® 24 flow meter typesType numberNom. Q3 m³/hMax. Flow Q4[m³/h]Min. Flow Q1[l/h]Min. cutoff[l/h]Pressure loss ∆p @ Q3 [bar]Connection on meterLength[mm]Anti-pollu- tion check valve *)Strainer *)65-2-CDAA-XXX 65-2 -CDA1-XXX 65-2-CDAC-XXX **) 65-2-CDAF-XXX 65-2-CEAF-XXX 65-2-CGAG-XXX 65-2-CHAG-XXX 65-2-CJAJ-XXX 65-2-CKCE-XXX 65-2-CLCG-XXX 65-2-CMCH-XXX1.6 1.6 1.6 1.6 2.5 4.0 6.3 10 16 25 402.0 2.0 2.0 2.0 3.1 5.0 7.9 12.5 20 31.3 5016 16 16 16 25 40 63 100 160 250 4003 3 3 3 6 7 12 20 30 50 800.25 0.25 0.25 0.25 0.04 0.09 0.22 0.06 0.16 0.06 0.05