Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
1-2-7-8Current flow indication (l/h or m³/h) is calculated based on volume pulses/10 s (see paragraph 6.5)20Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8013.3.6 CCC-codes for other electronic meters with passive or active outputNumber of decimals in displayCCC No.Pre- counterFlow factor MWh GcalGJm³m³/hkWMWl/impImp./lQmax [m³/h]TypeFlow sensor147 148 149 150 175 176 1771000 400 100 20 7500 4500 25002359260 5898150 2359260 11796300 314568 524280 9437041 1 1 1 1 1 10 0 0 0 0 0 0[tons] 0 0 0 0 0 0 02 2 1 1 2 2 2-3 3 2 2 3 3 31 2.5 10 50 - - -- - - 7.5 4.5 2.518...75SC-18 120…300 SC-120 450…1200 SC-450 1800…3000 SC-180015…30 25…50 40…80DF-15 DF-25 DF-40N N N N N N NNumber of decimals in displayCCC No.Pre- counterFlow factor MWh GcalGJm³m³/h MWl/impImp./lQp range [m³/h]Qs[m³/h]TypeFlow sensor20120220320420520610040400100201002359265898155898152359261179630