Title: +175=574.001_OE001_Manual - Varmemåler M3 - Løs fra rør.Produktinformasjon.pdf
9) The ninth parameter that has be configured is the pulse value "k" of the OC pulse heating output (hot out). The value can be selected by using the button T1: 0 - 1 - 10 – 100 - 1000 kWh/impHydrosplit M3 Manuale utente – User manual - v1.6.5_ITA_ENG43Confirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.10) The tenth parameter that has be configured is the pulse value "k" of the OC pulse cooling output (cold out). The value can be selected by using the button T1: 0 - 1 - 10 – 100 - 1000 kWh/impConfirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.11) The eleventh parameter that has be configured is the secondary address of the Wired M-Bus (M-Bus) of the heating data. Select the digit using the T2 button and change the single numbers with the T1 button.Confirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.