Title: +038=360.000-OU045T_04-PCD2.m4160 PLS SBC - IO Manual.pdf
Resistance measurement up to 2.5 kΩ (PCD2.W340) Special temperature sensors or any other resistances up to 2.5 kΩ can be con- nected to the PCD2.W340. The digital value can be calculated as follows:DV=16 380 • R (7 500 + R)where 0 ≤ DV ≤ 4095 and R=the resistance to be measured in Ω.For Ni 100 (PCD2.W350) Validity: Temperature range - 50 … +250 °C Computational error:± 1.65 °CT= - 28.7 +300 • DV 36287.294 • 10-6 • (DV - 1850)2For Pt 100 (PCD2.W350) Validity: Temperature range - 50 … +600 °C Computational error:± 1 °CT= - 99.9 +650 • DV 3910+ 6.625 • 10-6 • (DV -2114)2For Pt 1000 (PCD2.W360) Validity: Temperature range - 50 … +150 °C Computational error:± 0.25 °CT= - 178.1 +200 • DV 2509+ 3.873 • 10-6 • (DV -2786)2T = temperature DV = digital value5-72Analogue input modules with electricaly isolation5.9Analogue input modules with electricaly isolationPCD2.W305 PCD2.W315 PCD2.W325