Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
2. Press push button again within six seconds in order to reset the internal battery counter. The battery indicator now shows full. If the push button is not pressed again, the battery indicator will show "Low". (cid:13)(cid:21)(cid:17)(cid:19)(cid:22)(cid:20) (cid:20)(cid:20) (cid:24) (cid:47) (cid:26) (cid:27)Figure 6-1Reset internal battery counterSoftware versionCommissioning via PDMSIMATIC PDM (Process Device Manager) is a software package for configuring, parameterizing, commissioning, and maintaining field devices (e.g. transducers).Among other features, SIMATIC PDM contains a simple process for monitoring process values, interrupts and status/diagnosis signals of a field device.NoteFor instructions on installation and operation of SIMATIC PDM, refer to the SIMATIC PDM Getting Started (included in the documentation package that comes with PDM).NoteMinimum PDM version required is SIMATIC PDM V6.0 + SP5Note Default password