Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Note: The safery transformer 6699-403 is suitable for MULTICAL® 801 either with analog outputs or with high-power communication.For MULTICAL® 801 with both analog outputs and high-power communication we recommend a stronger transformer, e.g. type 5920-161.Maximum cable length between 230/24 VAC transformer e.g. Kamstrup type 6699-403 and MULTICAL®.Cable typeMaximum length2 x 0.75 m50 m2 x 1.5 mm2100 m86Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 80110.4 Danish regulations for the connection of mains operated metersInstallation to mains connected equipment for registration of consumption (Text from The Danish National Safety Board, 2004-12-06)The consumption of energy and resources (electricity, heat, gas and water) of the individual consumer is to an increasing extent registered by electronic meters, and often equipment for remote reading and remote control of both electronic and non-electronic meters is used.