Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
A suitable CCC-code with the same number of litres/imp as the flow sensor must be selected and for this flow meter type the CCC-code must be in the area 010 ≤ CCC ≤ 022.Example: CCC=012 is suitable for a mechanical flow meter with 100 litres/imp. Flow sensors with Qmax. in the range of 10…300 m3/h can use this CCC-code.66