Title: Data tester Smestad skole - Detaljert.pdf
Local Ser. No.: Local Adapter: Local Calibration Date: Device Software:pw20201689 Cat 6A Link Apr 17 2013 7.4Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 28,8 68,0 149,0 8,0 5,3Limit 100,0 - 555,0 50,0 25,0Margin 71,2 - 406,0 42,0 19,7Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 0,4 3,0 2,6 1,60Worst Value: Local: 36 12,5 43,5 31,0 495,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 15,6 9,0 6,6 313,00Remote: 12 20,0 12,2 7,8 150,00Worst Value: Local: 12 15,6 9,0 6,6 313,00Remote: 45 17,8 8,0 9,8 488,00dB -2030800Local250500 MHzdB -2030800Remote250500 MHzNEXT: Pass