Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Repairing the meter, if needed, we recommend only replacing battery, temperature sensors and communication modules. Alternatively, the whole meter ought to be replaced.Kamstrup A/S must make major repairs.Before sending in the sensor for repair or check, please use the error detection table below to help you clarify the possible cause of the problem.SymptomPossible reasonProposal for correctionNo function in the display (empty display)Power supply missingChange backup battery or check mains supplyNo energy accumulation (e.g. MWh) and volume (m3)Read “info” in the display Check the error indicated by theinfo code (see paragraph 6.8)If “info” = 000 ⇒Check that the flow direction matches the arrow on the flow sensorIf “info” = 004, 008 or 012 ⇒Check the temperature sensors. If defective, replace the sensor pair.Accumulation of volume (m3), but not of energy (e.g. MWh)Inlet and outlet sensors have been interchanged either in the installation or at the connection