Title: 370.011 Siemens Sitrans FUE080 Flowmåler.pdf
275.2.25.2.328Wiring the pulse outputCAUTIONPulse outputThe pulse output must be connected to equipment complying with Low Voltage Directive in order to be considered safe. The isolation within the FUS080 pulse output is only a functional isolation.1. Prepare cable ends as shown above.2. Connect output cables to terminals 56 and 57 (Output A) and to 66 and 67 (Output B) according to use. The output functions are used as shown above.2. Connect output cables to terminals 56 and 57 (Output A) and to 66 and 67 (Output B) according to use. The output functions are used as shown above.3. Ground cable shielding by fixing it with clamp.CAUTIONEMC performanceIncorrect fixing of the cable shield will affect the EMC performance!Mounting the transmitter1. Tighten power supply and pulse output cable glands.2. Mount transmitter.Figure 5-5 Mounting the transmitter on terminal boxSITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-02Commissioning6