Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
The analog outputs can be configured as power, flow (V1, V2), T1, T2, T3 or T1-T2, and the measuring range can be configured. All relevant configurations can be set up from the factory or on site by means of METERTOOL. After reconfiguration of the analog outputs, the meter must be reset. A reset can be effected in two different ways: 1) Switch off the mains supply and remove the plug to the back-up battery. The new values will not be saved in the meter’s memory until back-up battery and mains supply have been reconnected. 2) By means of METERTOOL a ”normal reset” is carried out under ”UTILITY  Reset”. After this, the new values have been stored in the meter’s memory.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201681MULTICAL® 801The analog outputs can also be coupled with common ground.9.3 Data connection [62-64]