Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Example: 4Info loggerPushing the bottom pushbutton once more, the data logger for information code is displayed. First, the date of the latest change is shown……then the information code set on this date is displayed. In this case it has been a ”burst alarm” on 1 June 2012.The data logger saves the latest 50 changes. 3The latest 36 changes can be displayed. All 50 changes can be read by means of LogView.Furthermore, the info code is saved in the programmable logger, in the daily logger, in the monthly logger and in the yearly logger for diagnosis purposes.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201651MULTICAL® 8016.10.2 Info code typesInfo CodeDescriptionResponse time0No irregularities1Supply voltage has been interrupted8Temperature sensor T1 outside measuring range1…10 min4Temperature sensor T2 outside measuring range1…10 min32Temperature sensor T3 outside measuring range1…10 min64Leak in cold water system24 hours256