Title: DOCA0005EN-12.pdf
Indicates that the next VIFE is manufacturer specific82DOCA0005EN-12iEM3000 series user manualDOCA0005EN-12Communications via M-BusPrimary VIFE codes used by the meterThe primary VIFE codes in the table below are used by the meter when the VIF equals FD hex (1111 1101 bin).NOTE: E denotes the extension bit; x in the hex value is determined by bits 7-4 of the VIFE.Primary VIFE codesbinhexAdditional informationManufacturerE000 1010xA—ModelE000 1100xC—VoltageE100 1001x9Volts with a resolution of 100CurrentE101 1100xCAmps with a resolution of 100Digital outputE001 1010xA—Digital inputE001 1011xB—Cumulation counterE110 0001x1Input metering accumulationError flagE001 0111x7—Manufacturer-specific VIFE codesThe manufacturer-specific VIFE codes in the table below are used by the meter when the VIF equals FF hex (1111 1111 bin).NOTE: E denotes the extension bit; the hex value assumes E = 0.DescriptionbinhexL1 value