Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
The high-resolution energy can be used for both heat energy (E1) and cooling energy (E3).Note: Hour counter and info event counter are always reset when HighRes is provoked by pressing both buttons in connection with reset.13.1.1 Data reading of high-resolution energyThe register ”HighRes” can be data read with ID = 155.In connection with data reading, measuring unit and value will be correct irrespective of meter size.102Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 80113.2 Pulse interfaceDuring test and verification of MULTICAL 801, where high-resolution energy pulses are required verification adapter type 66-99-461, placed as module 1, can be used.The pulse interface collects serial data from MULTICAL 801 every 7 s and converts these high-resolution data to high-resolution energy pulses with the same resolution as the high-resolution register of the display (see section 12.1)