Title: 91290FDV Campus F13 433.04 West elektro.pdf
For pulse output proportional to the active energy (kWh) Selectable in multiple of 100 Max 500 or 1500 kWh according to pulse ON duration4Pulse ON durationOutput typeSelectable: 30ms or 100 ms according to EN62052-31 Open collector PNPLoadVON 1 V dc max. 100mA VOFF 80 V dc max.General specificationsOperating temperatureStorage temperatureOvervoltage category Insulation (for 1 minute)25 to +65 °C (-13 to 149° F), indoor, (R.H. from 0 to 90% non-condensing @ 40°C)30°C to +80°C (-22 to 176° F) (R.H. < 90% non condensing @ 40°C) Cat. III30°C to +80°C (-22 to 176° F) (R.H. < 90% non condensing @ 40°C) Cat. III4000 V ac RMS between measuring inputs and digital/serial output (see table) 4000 V ac RMSRadio frequencyStandard compliance Safety Metrology Approvals Connections Cable cross-section areaOther terminalsAccording to CISPR 22EN62052-11 EN62053-21, EN50470-3 CE, MID (PF option only)