Title: Data tester Smestad skole - Detaljert.pdf
Local Ser. No.: Local Adapter: Local Calibration Date: Device Software:pw20201689 Cat 6A Link Apr 17 2013 7.4Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 44,6 68,0 231,0 12,0 7,7Limit 100,0 - 555,0 50,0 25,0Margin 55,4 - 324,0 38,0 17,3Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 0,8 3,0 2,2 2,05Worst Value: Local: 36 19,1 43,6 24,5 497,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 17,3 9,2 8,1 301,00Remote: 36 16,3 8,0 8,3 472,00Worst Value: Local: 12 17,3 9,2 8,1 301,00Remote: 36 16,3 8,0 8,3 472,00dB -2030800Local250500 MHzdB -2030800Remote250500 MHzNEXT: Pass*