Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
14Inlet temperature tariff15 Outlet temperature tariff19Time controlled tariff20Heat/cooling volume tariff (TL2 and TL3 are not used)21PQ-tariffEnergy is accumulated in TA2 and TA3 based on the tF-limits in TL2 and TL3. Energy is accumulated in TA2 and TA3 based on the tR-limits in TL2 and TL3. TL2=Start time for TA2 TL3=Start time for TA3 Volume (V1) is divided into TA2 for heat (T1>T2) and TA3 for cooling (T1<T2). (Recommended for heat/cooling applications) Energy if P>TL2 is saved in TA2 and energy if Q>TL3 is saved in TA3See paragraph 6.9 for further details on the tariff registers.26Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8013.6 >FF< Input A (VA), pulse division >GG< Input B (VB), pulse divisionMULTICAL® 801 has 2 pulse inputs, VA and VB, which are placed on base module 1 (see paragraph 7.2 for further details). The inputs are individually configured via the FF and GG codes as shown in the table below.