Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
0.04 0.004≤ 3,0 1…30 10…300 100…3000 ≤ 6 3…60 30…600Current flow indication (l/h or m³/h) is calculated based on measured duration between two volume pulses. (see paragraph 6.5)Selecting one of the above-mentioned CCC-codes, both CCC (V1) and CCC (V2) must be selected from this table.Note: CCC=9XX cannot be used for MULTICAL® 801, but only for MULTICAL® 602.Note: Continuous maximum water flow and permanent ∆Θ > 75 K may cause overflow in the daily data logger at CCC=010-011-012-013-150-202-205-206. With these combinations, we recommend you to use the built Prog. data logger.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.2016Flow sensorL L L L L L L23MULTICAL® 8013.4 Display codingDisplay code ”DDD” indicates the active readings of each meter type. ”1” is the first primary reading, whereas e.g. ”1A” is the first secondary reading. The display automatically returns to reading ”1” after 4 minutes.p m a t S e t a Dr e t e mt a e H0 1 2 = D D Dr e t e mt a e H