Title: DOCA0005EN-12.pdf
Digital outputsProgrammable (energy pulsing or overload alarm)Overload alarmModbusCommunicationsBACnetWidth (18 mm module in DIN rail mounting)DOCA0005EN-125 1 2 3 M Ei1–√––√–√5Overview5 3 2 3 M E0 5 2 3 M E5 5 2 3 M E5 6 2 3 M E5 7 2 3 M Eiiiii–11––––––√√√√√–––––√––––––√––√√√55555 5 4 3 M E5 6 4 3 M E5 5 5 3 M E5 6 5 3 M Eiiii√√√√√√––––√√0.5% 0.5% 0.5% 0.5%√√√√√√√√444422224444333311111111√√√√√–√––√–√555513Overview iEM3000 series user manualTypical applicationsThis range is a cost effective solution to monitor feeder circuits. These meters can monitor energy consumption by usage, by zone or by feeder in the cabinet. They can be used to monitor feeders in a main switchboard or to monitor the main in a distribution cabinet.