Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Calculator EC ± (0.15 + 2/∆Θ) %Sensor pair: ET ± (0.4 + 4/∆Θ) %LCD – 7 (8) digits with digit heigth 7.6 mm and back illumination9999.999 – 99999.99 – 999999.9 – 9999999 - 99999999MWh – kWh – GJ – GcalStandard: 460 days, 36 months, 15 years, 50 info codesStandard: Programmable data logger with logging depth 1080 registersStandard: Clock, calendar, leapyear compensation, target dateStandard: Real time clock with battery backupStandard: Battery backup of energy measurement incl. ULTRAFLOWStandard: KMP protocol with CRC16 used for optical communication as well as base modules< 10 µW RMS230 VAC +15/-30 %, 50/60 Hz (all types)24 VAC ±50 %, 50/60 Hz (Type 67-F/G without analog outputs)24 VAC ±25 %, 50/60 Hz (Type 67-F/G with analog outputs)4 kV< 3 W without analog outputs < 9 W with analog outputsMax. 50 mA/230 VACMax. 450 mA/24 VAC3.65 VDC, 2 batteries A-cell lithium(Type No. 66-99-619)10 years’ normal operation (with mains supply)1 year (without supply)