Title: Data tester Smestad skole - Detaljert.pdf
Unspecified-Building Unspecified-Floor Unspecified-Room Unspecified-Rack Unspecified-PanelOverall Result:Local Ser. No.: Local Adapter: Local Calibration Date: Device Software:pw20201689 Cat 6A Link Apr 17 2013 7.4Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 65,6 68,0 341,0 19,0 11,5Limit 100,0 - 555,0 50,0 25,0Margin 34,4 - 214,0 31,0 13,5Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 45 1,4 3,1 1,7 2,95Worst Value: Local: 45 28,5 43,8 15,3 500,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 78 14,0 8,8 5,2 329,00Remote: 78 21,1 15,1 6,0 76,75Worst Value: Local: 78 14,0 8,8 5,2 329,00Remote: 36 16,2 8,0 8,2 475,00