Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
401Totalizer 1 change date Lastchange date40674410Flow direction totalizer 2Reverse40678411Totalizer 2 change date Lastchange date78Value rangeAccess levelDescription10 ... 10 m/s 0 ... 210 ... 10 m/s 0 ... 2HW key Velocity added to measured flowvelocityR/WThis calibration factor will be calculated at the factory with wet-calibration.For FUS080-SONOKIT (retrofit) installations this parameter should be calculated by the pipe geometric tool (see 'Pipe geo. assistant' in 'Device' menu)2 ... 2R/WMeter correction factor for customer adjustment of the calculated flow value. Can be used at installations where a reference meter is used to get reference to the true flow rate0 ... 200 mR/WCable length (m) from sensor to transmitter0 ... 100 s R/WGeneral time filter for flow measurement. Filter constant is the time that must pass before the filter's ouput value is at 70% of the change of its input value. Higher number gives a slower and more stable flow signal.