Title: DOCA0005EN-12.pdf
The meter supports up to 14 COV subscriptions. You can add COV subscriptions to Analog Input and Binary Input objects using your BACnet-compatible software.Related topicsSee “Hardware and installation” on page 17 for the location of the communications LED.See “Hardware and installation” on page 17 for the location of the communications LED.See “Device configuration” on page 29 for information on configuring the meter using the front panel.See “Device configuration” on page 29 for information on configuring the meter using the front panel.BACnet object and property informationThe following sections outline the supported objects and properties available on the meter.Device objectThe following table outlines the properties of the Device object, whether a property is read-only or read-write, and if the value of the property is stored in the meter’s non- volatile onboard memory.DOCA0005EN-12109110Communications via BACnetiEM3000 series user manualDevice object propertyR/WStore d