Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Heat energy: E1 = V1(T1-T2)k T1:Inlet or T2:OutletCooling energy: E3 = V1(T2-T1)k T1:Inlet or T2:OutletFlow meter V1 is placed in inlet or outlet as selected during PROG.Mass: M1 = V1 (Kmass t1) or Mass: M1 = V1 (Kmass t2) depending on Inlet/Outlet programming.Application no. 2Closed thermal system with 2 identical flow metersBilling energy: E1 = V1(T1-T2)k T1:InletControl energy: E2 = V2 (T1-T2)k T2:OutletT3 can be used for checking the measurement of either inlet for outlet temperature, but T3 is not used for calculation.Mass: M1 = V1 (Kmass t1) Mass: M2 = V2 (Kmass t2)34Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.2016MULTICAL® 801Application no. 32-string system with 2 flow metersHeat energy: E1 = V1(T1-T2)k T1:Inlet or T2:OutletTap water energy: E6 = V2 (T3-T4)k T3:InletT3 is measured or programmed T4 is programmedFlow meter V1 is placed in inlet or outlet as selected during PROG.