Title: DOCA0005EN-12.pdf
1001101151201 - 10000000 - 170000 - 20005060DD-MMM-YYYYhh:mmDisableby Communicationby Digital Inputby Internal ClockDisableEnableFront panel display and meter setupDescriptionSelect the power system type the meter is wired to.Select how many voltage transformers (VT) are connected to the electrical power system.Define how many current transformers (CT) are connected to the meter and which terminals they are connected to.Select the size of the CT secondary, in Volts.Enter the size of the CT primary, in Amps.Select the size of the VT secondary, in Volts.Enter the size of the VT primary, in Volts.Enter the phase angle compensation, in rad (radian).For negative phase shift:Formula = 10000 - (Angle in rad*1000) Example: For -30° negative phase shift, the value in rad is -0.524 Value to be entered = 10000 - (-0.524*1000), which is equal to 10524For positive phase shift: