Title: +135.01=370.003-OE401T_02-320 Datablad Energimåler UH50.._.pdf
The number of decimal places of a value depends on the selected measuring path and the selected unit. The meter’s displays are arranged in 5 loops: ● 1 user loop ● 4 service loops123451Display2Button 13Button 24Optical interface5Calibration seal5 CE2N5324en 2018-08-186 Building TechnologiesA short press on button 1 enables selection of the required loop. The user loop reappears after the last loop. Press button 2 to read the value on the respective loop. The first value reappears after the last loop.NOTICEThe range of display and the displayed data may deviate from this description and certain button functions may be disabled, depending on the parameter settings on the meter.User loop LOOP 0Error message with error code number (only in case of error) Accumulated energy with tariff state Tariff register 1 (optional) Accumulated volume Segment testService loop LOOP 1Current flow rate Current power