Title: +038=370.104-OE004T_01-Multical 801 Energimåler Kamstrup - Installasjons- og brukerguide.pdf
Leak in the cold-water system24 hours256Leak in the heating system24 hours512Burst in the heating system120 sec.ULTRAFLOW® 54 info (activated when CCC=4XX)16Flow sensor V1, Datacomm errorAfter reset and 1 day (00:00)1024Flow sensor V2, Datacomm errorAfter reset and 1 day (00:00)2048 Flow sensor V1, Wrong meter factorAfter reset and 1 day (00:00)128Flow sensor V2, Wrong meter factorAfter reset and 1 day (00:00)4096 Flow sensor V1, Signal too low (Air)After reset and 1 day (00:00)8192Flow sensor V2, Signal too low (Air)After reset and 1 day (00:00)16384 Flow sensor V1, Wrong flow directionAfter reset and 1 day (00:00)32768 Flow sensor V2, Wrong flow directionAfter reset and 1 day (00:00)24Kamstrup A/S • 5512603_C1_GB_03.2016MULTICAL® 80110 Terminal OverviewMULTICAL® 801 has many connection options. The terminals are placed at the bottom of the meter.97A 98A 16 17 + 12 -18 19 16B 17B 18B 51 51A 52A 52+ CE - + CV - UP COM DNT337T2