Title: +135.01=370.004-OE401T_02-320 Datablad Energimåler UH50.._.pdf
Standard parameters The UH50.. comes programmed as follows: ● Set day [TT.MM]: 01.01Power supply The UH50.. can be powered by a power supply or battery.Automatic mains detection The power supply detects whether line voltage is present. The signal is provided to the UH50... The device automatically detects whether it is powered by a battery or a power supply.Power packsPower pack AC/DC 24 VPower pack AC 110/230 VWZU- ACDC24-00WZU-AC110/230-15WZU-AC110/230-50BatteriesStandard battery (2 AA)Universal battery (D-cell)WZU-BA+GUMWZU-BDS9 CE2N5324en 2018-08-1810 Building TechnologiesBattery life Battery life depends on the type of battery and requirements (e.g. short timebase, analog module, etc.).Requirements6 years11 years16 yearsPower pack(with timebase  Q = 4 s and T =AC 110/230 V;30 s)AC / DC 24 VStandard pulses,2x AA cellCDYesM-bus readout (max. every 15minutes),  CL moduleRF module 868 MHzDDYes(mobile, 16 sec. transmission