Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Type 67-F and 67-K Pt100 – EN 60 751, 4-wire connection -Type 67-G and 67-L Pt500 – EN 60 751, 4-wire connectionCompatible flow meter typesULTRAFLOW -Electronic meters with active or passive pulse output -Mechanical meters with electronic pick-up -Mechanical meters with reed contactFlow meter sizes[kWh] [MWh] [GJ]qp 0.6 m3/h…15 m3/h qp 0.6 m3/h…15000 m3/h qp 0.6 m3/h…30000 m3/hEN 1434 designationEnvironmental class A and CMID designationMechanical environment: Class M1Electromagnetic environment: Class E1 and E2Non-condensing environment, closed location 5…55 °C (indoors)Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.20167MULTICAL® 8012.2 Electrical dataCalculator dataTypical accuracyDisplayResolutionEnergy unitsData logger (Eeprom)Clock/calendarData communicationPower of temperature sensorsMains supplyInsulation voltagePower consumptionCurrent consumptionBattery backupReplacement intervalBackup periodEMC data8