Title: Data tester Smestad skole - Detaljert.pdf
Local Ser. No.: Local Adapter: Local Calibration Date: Device Software:pw20201689 Cat 6A Link Apr 17 2013 7.4Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 34,3 68,0 178,0 10,0 6,2Limit 100,0 - 555,0 50,0 25,0Margin 65,7 - 377,0 40,0 18,8Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 0,5 3,0 2,5 1,75Worst Value: Local: 36 14,7 43,8 29,1 500,00dB -2020600Local250500 MHzReturn Loss: Pass*Pair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 20,6 15,2 5,4 76,50Remote: 12 13,1 12,3 0,8 146,50Worst Value: Local: 78 15,0 8,0 7,0 484,00Remote: 12 13,1 12,3 0,8 146,50dB -2020600Local250500 MHzdB -2020600Remote250500 MHzNEXT: Pass