Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 45] [Pair 45]41.4 180 3 6.13Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]23.3 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 12-45 6.7 473.0 29.8 45 7.8 485.0 26.7SR 36-45 7.7 472.0 29.8 36 8.3 472.0 26.9MAIN 12-45 6.7 473.0 29.8 45 7.9 500.0 26.4SR 36-45 8.1 500.0 29.2 36 8.5 498.0 26.410080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 36-78 15.7 415.0 12.9 45 15.0 17.8 37.3SR 78-36 15.7 415.0 12.9 45 14.9 8.9 43.3MAIN 36-78 16.0 429.0 12.6 78 16.2 415.0 9.9