Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
At the end of a year and a month the max. and min. values are saved in the data logger, and the current max. and min. registers are “reset” according to the selected target date and the meter’s internal clock and calendar.”Reset” is made by setting the max. value to zero and the min. value to 10000,0 kW at e.g. CCC=119.If the max. or min. registration is used for accounting purposes, we recommend that the clock setting is checked in connection with the installation as well as once a year. Furthermore, the back-up battery of MULTICAL 801 ought to be replaced at intervals of max. 10 years.Date of year-to-date max.Value of year-to-date max.Date of this month’s min.Value of this month’s min.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201643MULTICAL® 8016.8 Temperature measurement