Title: +038=522.000-NI002._05-Testrapport EKOM-anlegg, plan 03 (Uniconsult).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 12] [Pair 45] [Pair 36] [Pair 36]19.6 85 1 3.05Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]31.8 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 7.9 456.0 29.9 36 7.5 456.0 27.1SR 36-45 7.0 408.0 31.1 36 7.8 450.0 27.4MAIN 36-45 8.1 467.0 29.4 36 8.1 498.0 25.1SR 36-45 8.2 496.0 28.0 36 9.2 499.0 25.010080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 36-45 15.2 358.0 14.2 45 14.4 33.8 31.7SR 36-45 15.1 358.0 14.2 45 14.4 29.8 32.8MAIN 45-36 16.0 474.0 11.7 45 18.9 499.0 8.3