Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Example: CCC=012 is suitable for a mechanical flow meter with 100 litres/imp. Flow sensors with Qmax. in the range of 10…300 m3/h can use this CCC-code.66Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8017.1.3Flow sensor with active output, supplied through MULTICAL This connection is used together with both Kamstrup’s ULTRAFLOW and Kamstrup’s electronic pick-up units for vane wheel meters. The current consumption of these units is very low and furthermore adapted to the battery lifetime of MULTICAL.A suitable CCC-code with the same number of imp/litre as the flow sensor must be selected and for this flow meter type the CCC-code must be CCC > 100.Example: CCC=119 suits an electronic meter with 100 imp/litre and normally qp 1.5 m3/h.V1 and V2 are connected as shown in the table below.V1V2Red (3.6 V)99Yellow (signal)1069Blue (GND)1111Table 27.1.3.1 Use of Pulse Transmitter between ULTRAFLOW® and MULTICAL®