Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
This technical description has been written with a view to enabling operations managers, meter installers, consulting engineers and distributors to utilize all functions comprised in MULTICAL® 801. Furthermore, the description is directed to laboratories performing tests and verification.MULTICAL® 801 is based on the platform used for MULTICAL® 601. However, many extra facilities such as back illuminated display, back up of energy metering during power failure, extra communication channels and the option of four analog outputs have been added.1.1 Block diagram6Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8012 Technical data2.1 Approved meter dataApprovalDK-0200-MI004-009StandardEN 1434:2007 and OIML R75:2002EU-directivesMeasuring Instrument Directive, Low Voltage Directive, Electromagnetic Compatibity DirectiveTemperature range Differential rangeθ: 2 °C…180 °C ∆Θ: 3 K…170 KAccuracyE C ± (0.5 + ∆Θ min/∆Θ) %Temperature sensors