Title: 370.011 Siemens Sitrans FUE080 Flowmåler.pdf
2. Smoothly push cables one by one from underneath base through glands and adjust all cable ends approximately 100 mm from upper frame of base.2. Smoothly push cables one by one from underneath base through glands and adjust all cable ends approximately 100 mm from upper frame of base.Figure 5-1Electrical connection to SONOKIT 1-pathSITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-0223Connecting 5.1 Wiring the transducer cablesFigure 5-2Electrical connection to SONOKIT 2-pathNoteEnsure that the right cable entries are used.  1-path:– Terminal 1A is connected to transducer 1A – Terminal 1B is connected to transducer 1B2-path:As 1-path and additionally: – Terminal 2A is connected to transducer 2A – Terminal 2B is connected to transducer 2BNoteFor sensor (transducer) connection, see SONOKIT 1-path or SONOKIT 2-path Operating Instructions.3. Remount connection board in wall/pipe mounting kit.SITRANS FUS080 transmitter24