Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
All control cables must be drawn separately and not parallel to e.g. power cables or other cables with the risk of inducing electromagnetic interference. There must be a distance of min. 25 cm between signal cables and other installations.5.3 Climatic conditions MULTICAL® 801 has been designed and approved for indoor installation in non-condensing environments with ambient temperatures from 5…55°C.Furthermore, MULTICAL® 801 can also be installed in unheated rooms as the instrument is protected by self- heating.Protection class IP67 allows short-term submergence, provided that all cable unions have been correctly mounted and that the plastic cover has been properly fastened.5.4 Electrical installationsSee paragraph 105.5 Terminal Overview MULTICAL® 801 has many connection options. The terminals are placed at the bottom of the meter. Additional information can be found in Section 7 (Flow Meter Connection), Section 8 (Temperature Sensors) and Section 9 (Other connections).32