Title: +038=320.103-OE009T_02-Multical 801 Energimåler Kamstrup - Produktdatablad.pdf
Power measurement MULTICAL® 801 calculates current power on the basis of current water flow and the tempera- ture difference measured in connection with the latest integration.Current power is updated in the display simultaneously with the flow update.4Kamstrup A/S • 5810625_K1_GB_05.2016Calculator functionsMin. and max. flow and power MULTICAL® 801 registers minimum and maximum flow and power on a monthly as well as on a yearly basis. The registra- tions which appear from the display or can be read via data communication include max. and min. flow and power values, all with date indication. All max. and min. values are calculated as largest and small- est average respectively of a number of current flow or power measurements. The average period used for all calculations is selected in the interval 1…1440 min.