Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
If either power or flow tariff is used you obtain an overview of the total consumption compared to the part of the consumption used above tariff limit.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201655MULTICAL® 801EE=13 T1-T2 tariff (∆t)If the current T1-T2 (∆t) is lower than TL2 but exceeds TL3, heat energy is counted in TA2 parallel to the main register. If the current cooling falls below or is equal to TL3, heat energy is counted in TA3 parallel with the main register.∆t ≥ TL2Accumulation in main register onlyTL3 < ∆t < TL2Accumulation in TA2 and main registerTL3 < TL2∆t ≤ TL3Accumulation in TA3 and main registerSetting up tariff limits TL3 must always be lower than TL2.The T1-T2 tariff can be used as a basis for weighted user charge. Low ∆t (small difference between inlet and outlet temperatures) is uneconomical for the heat supplier.EE=14 Inlet tariff