Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 45] [Pair 36]38.8 169 3 5.74Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 36] [Pair 36] [Pair 36]24.6 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 6.8 500.0 29.2 45 7.3 495.0 26.5SR 36-45 7.8 473.0 29.8 36 7.1 500.0 26.4MAIN 36-45 6.8 500.0 29.2 45 7.3 495.0 26.5SR 36-45 8.0 500.0 29.2 36 7.1 500.0 26.410080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 45-36 15.4 32.3 35.1 45 14.1 2.6 53.9SR 36-45 15.4 32.3 35.1 45 14.2 5.0 48.3MAIN 12-45 16.5 460.0 12.0 45 16.3 420.0 9.8