Title: +038=370.104-OE004T_01-Multical 801 Energimåler Kamstrup - Installasjons- og brukerguide.pdf
The passive contact output of the flow sensor on terminals 56 and 57 is connected directly to the flow sensor input which is not galvanically separated. This permits a cable length of max. 10-20 m between flow meter and calculator.Auxiliary voltage from terminals 97A and 98A is added to the passive contact output of the flow sensor on terminals 56 and 57 before the signal is connected to the galvanically separated flow sensor input. This permits a cable length of up to 100 m between flow sensor and calculator.Auxiliary voltage from terminals 97A and 98A is added to the passive contact output on terminals 24 and 25 before the signal is connected to the galvanically separated flow sensor input. This permits a cable length of up to 100 m between flow sensor and calculator.16KROHNE ALTOFLUXMULTICAL® 80110B (69B)P PV1 (V2)11B (79B)E+ E-PEBAILEY FISCHER & PORTER COPA/MAG-XE COPA/MAG-XM/CM V3 24V+ V1(V9) (V8)10 9V1 (V2)MULTICAL® 80110B (69B)11B (79B)PEPESITRANS