Title: +038=522.000-NI002._05-Testrapport EKOM-anlegg, plan 03 (Uniconsult).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 12] [Pair 45] [Pair 36] [Pair 36]27.8 120 1 4.20Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]28.4 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 12-45 8.5 344.0 32.7 12 9.6 471.0 27.0SR 36-45 7.2 485.0 29.5 45 7.9 493.0 26.5MAIN 12-45 8.7 475.0 29.7 36 10.3 500.0 25.0SR 36-45 7.4 499.0 29.3 36 9.0 493.0 25.310080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 45-36 14.1 3.5 54.4 45 14.1 4.3 49.7SR 36-45 14.0 3.3 55.0 45 14.1 3.8 50.8MAIN 36-45 15.1 472.0 11.8 45 16.3 476.0 8.7