Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
The steel tube is mounted in a sensor pocket (immersion pipe) which has an inner diameter of 6 mm and an outer diameter of 8 mm. Sensor pockets are available with R½ (conical ½”) connection in stainless steel i lengths of 65, 90 and 140 mm. The sensor construction with separate immersion pipe permits replacement of sensors without having to switch off the flow. Furthermore, the wide range of immersion pipe lengths ensures that the sensors can be mounted in all existing pipe dimensions.The plastic tube on the sensor cable is placed opposite the sealing screw and the screw is tightened lightly by hand before sealing.Figure 16Figure 17The stainless steel pockets can be for mounting in PN25 systems!Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201679MULTICAL® 8018.4 Pt500 short direct sensor pair