Title: Manual - Varmemåler M3 - Løs fra rør.pdf
Hydrosplit M3 Manuale utente – User manual - v1.6.5_ITA_ENG52COMMISSIONING PROCEDURE1) Ensure that the electrical wirings are carried out correctly;2) Check in the level 3 that all parameters are correctly set on the calculator (especially the pulse value for the flow sensor –flow in-); 3) Verify that the flowmeters and volume meters, the temperature sensors etc. are installed correctly. (always refer to the installation manuals specific to every product) 4) Start the heating system:Check if the registered values are coherent (inlet/return temp.)Check the totalized volume and current flowrate;5) Start the hot/cold sanitary water system:Check if the registered values are coherent (volume)6) Check that no error is active.7) Apply eventual adhesive or lead sealings.ERRORS AND FAULTSwhen one or more anomalies occur, the calculation unit will signal the recorded error and will displaythe following icon:.