Title: +038=310.101-RF022T_06-multical 62 Vannmåler Kamstrup - Installasjons- og brukerveiledning.pdf
The module has connection possibility for two pressure transmitters on terminals 57, 58 and 59 and can be adjusted for current reading or pressure ranges of 6, 10 or 16 bar.The module is prepared for remote reading, data from meter/module being transferred to the sys- tem software via the connected external GSM/GPRS modem on terminals 62, 63 and 64.Furthermore, the module has two extra pulse inputs, VA and VB.The module must be powered by 24 VAC.5.1.5 Analog output, type 67-00-23The module has two active analog outputs, which can be individually configured at 0…20 mA or 4…20 mA. Furthermore, the outputs can be configured for a specific measuring value as well as the required range scaling.All output values are updated every 10 seconds.The module must be mounted in MULTICAL® 62 and is powered by 24 VAC. Configuration to be carried out via the “Base module” menu of METERTOOL.See installation instructions 5512-369 (DK-GB-DE).8Kamstrup A/S • 55121068_B3_GB_03.2015MULTICAL® 62 • Water meter