Title: +038=320.103-OE009T_02-Multical 801 Energimåler Kamstrup - Produktdatablad.pdf
Approval Standard EU-directives – MID (Measuring Instruments Directive) – LVD (Low Voltage Directive) – EMC (Electromagnetic Compatibility Directive) Temperature range Differential range Accuracy Temperature sensors – Type 67-F and 67-K – Type 67-G and 67-L Flow meter types – ULTRAFLOW® – Electronic meters with active and passive pulse output – Mechanical meters with electronic pick-up – Mechanical meters with reed switch Flow meter sizes – [kWh] – [MWh] – [GJ] EN 1434 designation MID designation – Mechanical environment – Electromagnetic environment – Non condensing, closed location (indoor installation), 5…55 °Cθ: 2...180 °C ∆Θ: 3...170 K EC ±(0,5 + ∆Θmin/∆Θ)%Pt100 – EN 60 751, 4-wire connection Pt500 – EN 60 751, 4-wire connectionqp 0,6 m3/h...qp 15 m3/h qp 0,6 m3/h...qp 15000 m3/h qp 0,6 m3/h...qp 30000 m3/h Environmental class A and CClass M1 Class E1 and E2Kamstrup A/S • 5810625_K1_GB_05.2016MULTICAL® 801Backup-batteryModule 2Module 1 Two optional pulse inputs VA and VB