Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
– Calibration Factor (Flow-Profile-Factor for retrofit, line 310)– Diameter (line 306)2. Define sensor-specific parameters based on data from the sensor measurement report and the selection made at ordering:2. Define sensor-specific parameters based on data from the sensor measurement report and the selection made at ordering:– Select metric (m, m/s) or imperial (in, ft/s) units for data entry. Internal calculation inthe device is made in metric units only.– Enter outer pipe diameter (Du) and average pipe wall thickness (t).– Flow velocity (V) is automatically calculated using entered pipe data and given Qmax(PDM parameter row 10).– Enter media viscosity in cSt (e.g. 1 cSt for water at 20 °C).– Enter the pipe roughness value.The pipe-roughness default value is automatically set according to pipe material or can be entered manually by selecting "User defined" (range: 0.0000 to 0.0100 m).Pipe materialTypical roughnessSmooth plastic pipe (select PVC)0.0001 m