Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
443 Amount per pulse B1.000000 (m3)0.000001 ... 1000000000R/WVolume per pulse.Valid if pulse output B is enabled and pulse function selected40827444 Pulse width for pulse B5 ms5, 10, 20, 50, 100, 200, 500 msR/WPulse length when the pulse is active.Valid if pulse output B is enabled and pulse function selectedCommunicationFor editing the communication settings, use PDM device menu "Communication Setting"40833591 Device CommunicationAddress11 ... 247Read Meter has default address value 1 with selectable address up to 24740834592 Baudrate40 ... 5Read Communication port speed40835593 Parity00 ... 3Read Communication port parity40836594Interframe space3535 ... 255Read Minimum space between twomessages (bytes x 10)40837595 Response delay51 ... 50Read Minimum time from receiving a request to its responseData logger40619600 Log intervalMonthlyDaily, Weekly, MonthlyR/WLog interval40620