Title: 370.011 Siemens UC50 Kalkulator.pdf
Multipolar terminals are used for connecting external cables to the modules.5 mmStrip-back length: Connection capacity  Rigid or flexible, 0.2 – 2.5 mm²  Flexible with end ferrules, 0.25 – 1.5 mm²  Conductor sizes 26 - 14 AWG Multiple-conductor connection (2 conductors of same cross-section)  Rigid or flexible, 0.2 – 0.75 mm²  Flexible with end ferrules without plastic sleeve, 0.25 – 0.34 mm²  Flexible with TWIN ferrules with plastic sleeve, 0.5 – 0.75 mm²Recommended screwdriver: 0.6 x 3.5 mmTightening torque:0.4 NmInterfaces of the electronic unit18/2912.2 Permissible combinations of modulesAM = analog module MB, MB G4, MB MI = M-Bus module CL = CL module RF = Radio module GPRS = GPRS module GSM = GSM module ZB = ZigBee module) 5 (M APulse module“ d r a d n a) *“ t s at s „f „Slot for module #2 is equipped with...B M4 G B M) * * *IM B ML CS R P GM S Gz H M 8 6 8 F RAMs e ys e ys e y) 4 (s e ys e ys e ys e yo no ns e y. . .