Title: +038=360.000-OU045T_04-PCD2.m4160 PLS SBC - IO Manual.pdf
For Pt1000 (PCD3.W340)Validity:Temperature range - 50…+ 400°CComputational error: 450 • DV 2474T= - 366.5 +± 1.5°C+ 18.291 • 10-6 • (DV - 2821)2Resistance measurement up to 2.5 kΩ (PCD3.W340)Special temperature sensors or any other resistances up to 2.5 kΩ can be connected to the PCD3.W340. The digital value can be calculated as follows:DV=16 380 • R(7 500 + R)where 0 ≤ DV ≤ 4095 and R = the resistance to be measured in Ω.For Ni 100 (PCD3.W350)Validity:Temperature range - 50…+ 250°CComputational error:± 1.65°CT= - 28.7 +300 • DV 36287.294 • 10-6 • (DV - 1850)2For Pt100 (PCD3.W350)Validity:Temperature range - 50…+ 600°CComputational error:± 1°CT= - 99.9 +650 • DV 3910+ 6.625 • 10-6 • (DV - 2114)2For Pt1000 (PCD3.W360)Validity:Temperature range - 50…+ 150°CComputational error: 200 • DV 2509T= - 178.1 +± 0.25°C+ 3.873 • 10-6 • (DV - 2786)2T = temperatureDV = digital value6-71