Title: 370.011 Siemens Sitrans FUE080 Flowmåler.pdf
3. Remount connection board in wall/pipe mounting kit.SITRANS FUS080 transmitter24Operating Instructions, 07/2012, A5E03059912-024. Ground transmitter via grounding terminal (PE) on PCB.Note Sensor groundingTo ensure identical potential for sensor and transmitter, a direct grounding of transmitter and sensor is recommended.5. Tighten both transducer cable glands.6. Make sure that transducer cables are not overbent. Minimum cable bend radius is 50 mm.6. Make sure that transducer cables are not overbent. Minimum cable bend radius is 50 mm.5.2Wiring the power supply and the outputThe wiring of the power supply and the output is done in three steps:1. Wiring the power supply2. Wiring the pulse output (if relevant)3. Mounting the transmitter5.2.1Wiring the power supplyWARNINGMake sure the power supply requirements stated on the nameplates are met!SITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-0225