Title: +135.01=370.003-OE401T_02-320 Datablad Energimåler UH50.._.pdf
Log functions In the internal logbook, meteorologically relevant events (errors, states, actions) are stored in chronological order with the time of occurrence. The acquired events are predefined. Logbook data cannot be deleted. Each event is stored in a separate 4-level shift register; the overflows are transferred to a 25- level circulating buffer. So that at least the last 4 times can be traced for each event. In a monthly register, the error states are stored for the current month and the past 18 months (without time stamp).Serial no.Description1F0 = Air in measuring tube2F1 = Interruption in temperature sensor heat side3F2 = Interruption in temperature sensor cold side4F3 = Error temperature electronics5F5 = Short circuit in temperature sensor heat side6F6 = Short circuit in temperature sensor cold side7F8 = Sensor error >8 hours8F9 = ASIC error9Above max. temperature in the flow measuring section10Below min. temperature in the flow measuring section11