Title: +135.01=370.001-OE401T_02-320 Datablad Energimåler UH50.._.pdf
Flow measuring sectionFunction dataTemperature range (national approvals may differ)5…130 °CRecommended temperature range ● Heating ● Cooling10…130 °C 5…50 °CRated pressure1.6 MPa (PN 16; PS 16)/2.5 MPa (PN 25; PS 25)Mounting positionAnyMounting locationReturn or flow37 CE2N5324en 2018-08-1838 Building TechnologiesNominal flow qpOverall lengthConnecti onMaximum flow qsMinimum flow qim3/hmmG/DNm3/hl/h0.6110G3/41.260.6190G 11.260.6190DN201.261.5110G3/43151.5130G 13151.5190G 13151.5190DN203152.5130G 15252.5190G 15252.5190DN205253.5260G11/47353.5260DN257356150G11/412606260G11/412606260DN25126010200G22010010300G22010010300DN402010015200DN503015015270DN503015025300DN655025040300DN808040060360DN100120600150500DN1503001500