Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Includes a double-chamber safety transformer, which fulfils the double-isolation requirements. The power consumption is lower than 3 W (without analog outputs) or less than 9 W with analog outputs.regulationsfor electricNational installations must be observed. The 24 VAC module can be connected/disconnected by the heating station’s personnel, whereas the fixed 230/24 V installation into the meter panel must be carried out by an authorized electrician.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201685MULTICAL® 801MULTICAL® 801 is specially suited for installation together with a 230/24 V safety transformer, e.g. type 6699-403, which can be installed in the meter panel in front of the safety relay. When the transformer is used, the power consumption will be lower than 3 W (without analog outputs) or lower than 9 W with analog outputs, for the complete meter incl. 230/24 V transformer.