Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
11.1.20 High Power Radio Router + 2 pulse inputs (VA, VB) (67-00-84) (PCB - 5550-1221) The High Power RadioRouter module has built-in router functionality and is thus optimized to form part of a Kamstrup radio network, the read data being automatically transferred to system software via the network unit RF Concentrator.Furthermore, the module can be read by Kamstrup’s hand-held reading systems, e.g. USB Meter Reader and MT Pro. The RadioRouter module is available for operation in both licence-free and licence demanding frequences permitting a transmitting strength of up to 500 mW. The module is by default fitted with internal antenna, connection for external antenna, and two extra pulse inputs.See paragraph 7.3 Pulse inputs VA and VB regarding the function of the pulse inputs.96Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 80111.2 Retrofitting modules