Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
T2 ≤ TL2Accumulation in main register onlyTL3 ≥ T2 > TL2Accumulation in TA2 and main registerTL3 > TL2T2 > TL3Accumulation in TA3 and main registerSetting up data TL3 must always be bigger than TL2.The outlet temperature tariff can be used as a basis for weighted user charge. A high outlet temperature indicates insufficient heat utilization, which is uneconomical for the heat supplier.56Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 801EE=19 Time-controlled tariffThe time-controlled tariff is used for time division of the heat consumption. If TL2 = 08:00 and TL3 = 16:00, the daily consumption from 8 a.m. to 4 p.m. is accumulated in TA2, whereas the consumption during the evening and night from 16:01 to 07:59 will be accumulated in TA3.TL2 must include a lower hour value than TL3.TL 3 ≥ Clock ≥ TL2Accumulation in TA2 and main registerTL3 > TL2TL 2 > Clock > TL3Accumulation in TA3 and main register