Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 36] [Pair 45]40.2 175 3 5.99Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]24.1 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 12-45 8.7 478.0 29.7 12 9.8 478.0 26.8SR 36-45 10.1 500.0 29.2 36 9.4 500.0 26.4MAIN 12-45 9.0 490.0 29.4 12 9.8 478.0 26.8SR 36-45 10.1 500.0 29.2 36 9.4 500.0 26.410080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 78-45 14.8 391.0 13.4 45 14.6 3.9 50.5SR 45-78 14.9 391.0 13.4 45 14.4 2.5 54.3MAIN 78-45 15.3 420.0 12.8 45 16.3 443.0 9.3