Title: 370.011 Siemens Sitrans FUE080 Flowmåler.pdf
– Right click on "Modbus Net" and select "Insert New Object" → "Modbus Device"– Click on "Assign" and assign the Modbus device to SITRANS FUS/E080 (Sensors →Flow → Ultrasonic → SIEMENS AG → SITRANS FUS/E080) and click "OK". Rename device according to application requirements (max. 32 characters). Set up communication parameters for SIMATIC MODBUS network– Select "Net" → "Modbus Net", right click on "Modbus net" and select "ObjectProperties"– Select "Connection" to activate IrDAConfiguring the deviceIn the following it is described how to configure the device by defining all sensor specific parameters.Note Measurement accuracyA correct sensor installation and a completed measurement report containing correctly defined sensor-specific data is required to ensure optimum measurement accuracy.  For further information, refer to the sensor operating instructions.  For allowed inaccuracies, refer to "Sensor geometrical data" (Page 63).SITRANS FUS080 transmitter