Title: Trykktransmitter luft.pdf
Complete and verify all wiring connections before applying power to the controller to which the module is connected.Output 0...10VOutput 4...20mA0-10V4-20mAVA+Power supply 24 VDC / 24 VAC24VGNDSelecting Pressure Range To adjust the pressure range, set the jumpers as shown.JumpersLEDZERO0-10VJumper 1 Jumper 2 Jumper 3 Jumper 4 Jumper 5 Jumper 6Jumper 1Jumper 2Jumper 34-20mA24VDCGNDZeroing It is recommended to adjust the zero point every 12 months during normal operation.How to do: Note! Supply voltage must be connected one hour before the 0-point adjustment is carried out.1) Loose both tubes from the pressure inlets + and –.2) Push zero button >4 seconds and the red led turns ON.3) Wait until LED turns off and then install tubes again to the pressure inlets.