Title: +135.01=370.003-OE401T_02-320 Datablad Energimåler UH50.._.pdf
These modules have no impact on consumption metering and can therefore be replaced at any time without having to destroy the security seal. Within 30 seconds after mounting (in case of the analog module up to 2 minutes), the meter automatically detects the module used and is then ready for  communication or for delivering pulses. The type of module used can be displayed within the service loop (loop 4) based on the display parameters.Permissible module combinationsAMAnalog moduleStep 1:MBM-bus moduleSlot for module 2 is equipped with...MB G4M-bus moduleMB MIM-bus moduleCLCL-moduleRFRF moduleGPRSGPRS module485E- BACBACnet module485E-MOD Modbus moduleAM (5) Pulse module MBMB G4CLGPRSRF 868 MHzBACne tModb us"Stan dard“"Fast" *)Step 2: Slot for module #1 can be equipped with...AMPulse module **) "Standard"YesYesYesYes (3)YesYes (4)Yes (2) Yes (4)YesYesYesYesNoYesYesYesNoNoNoNoMBYesYesYesYes (4)Yes