Title: +175=574.001_OE001_Manual - Varmemåler M3 - Løs fra rør.Produktinformasjon.pdf
The value can be selected by using the button T1: 0.1–0.25–1–2.5–10–25–100–250 L/impConfirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.7) The seventh parameter that has be configured is the update of the consumption data reported on the C2 hot/cold water meter dial. Select the digit using the T2 button and change the single numbers with the T1 button.Confirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.8) The eighth parameter that has be configured is the enabling/disabling of the anti-tampering contact of the C2 cold/hot water meter. By using the T1 button, proceed with the enabling [1] or disabling [0].Confirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.