Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Setting up data TL3 must always include a higher value than TL2. The power-controlled tariff is e.g. used as a basis for the individual heat consumer’s connection fee. Furthermore, this tariff type can provide valuable statistical data if the heating station considers new construction activities.EE=12 Flow controlled tariffIf the current water flow exceeds TL2 but is lower than or equal to TL3, heat energy is counted in TA2 parallel to the main register. If the current water flow exceeds TL3, heat energy is counted in TA3 parallel to the main register. Setting up data TL3 must always include a higher value than TL2.q ≤ TL2Accumulation in main register onlyTL3 ≥ P > TL2Accumulation in TA2 and main registerTL3 > TL2q > TL3Accumulation in TA3 and main registerThe flow controlled tariff is e.g. used as a basis for the individual heat consumer’s connection fee. Furthermore, this tariff type can provide valuable statistical data if the heating station considers new construction activities.