Title: 370.011 OE501 Siemens UC50 Kalkulator.pdf
19/2912.3 Pulse moduleCE, C2, CV, CT or RIDisplay on LCD Consider the appropriate instructions manual (attached to the meter/module)!The following can be set: • pulse type: • pulse output: • pulse duration, if only 1 pulse module is fitted: from 2 ms to 100 ms in steps of 1 ms; Pulse duration, if 2 pulse modules are fitted: from 5 ms to 100 ms in steps of 5 ms; The pulses can be received and evaluated with a suitable device. The settings must be made in accordance with the information on the data sheet of the controller. *) Linear pulses are output proportionally to the measured value. In the case of scaled pulses, the number of pulses at the upper and lower end of the range can be defined. In this case, the pulse receiving device can detect, for example, a connection error.„linear“ or „scaled“ pulses *) energy or volume12.4 CL moduleCL-ModulRTX RTX2120-+CL moduleCL (Current Loop)Display on LCD Consider the appropriate instructions manual (attached to the meter/module)!