Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
2021Heat/cooling volume tariff (TL2 and TL3 are not used)PQ-tariffVolume (V1) is divided into TA2 for heat (T1>T2) and TA3 for cooling (T1<T2) provided that T1 is below T1 limit. Energy if P>TL2 is saved in TA2 and energy if Q>TL3 is saved in TA3EE=00 No active tariffIf the tariff function is not going to be used, select the setup EE=00.The tariff function can, however, at a later stage be made active by means of reconfiguration with METERTOOL for MULTICAL 801. See section 14 METERTOOL.EE=11 Power controlled tariffIf the current power exceeds TL2 but is lower than or equal to TL3, heat energy is counted in TA2 parallel to the main register. If the current power exceeds TL3, heat energy is counted in TA3 parallel to the main register.P ≤ TL2Accumulation in main register onlyTL3 ≥ P > TL2Accumulation in TA2 and main registerTL3 > TL2P > TL3Accumulation in TA3 and main register