Title: +038=310.101-RF022T_06-multical 62 Vannmåler Kamstrup - Installasjons- og brukerveiledning.pdf
5.1.16 High-Power RadioRouter + 2 pulse inputs (VA, VB), type 602-00-84The High-Power RadioRouter module has built-in router functionality and is thus optimized to form part of a Kamstrup radio network, the read data being automatically transferred to system soft- ware via the network unit RF Concentrator. Furthermore, the module can be read by Kamstrup’s hand-held reading systems, e.g. USB Meter Reader and MULTITERM Pro.The RadioRouter module is available for operation in both licence-free and licence demanding frequences permitting a transmitting strength of up to 500 mW. The module is by default fitted with internal antenna, connection for external antenna, and two extra pulse inputs. The High Power RadioRouter module (602-00-84) must be used together with the High Power mains supply (230 VAC: 602-00-00-3 and 24 VAC: 602-00-00-4).5.2 Top modules5.2.1Type 67-05: RTC + data output + hourly data logger