Title: DOCA0005EN-12.pdf
Select the power system type the meter is wired to.3PH3W1PH4W Multi L-NFrequencyFrequency5060Select the frequency of the electrical power system, in Hz.DateDateDD-MMM-YYYY Set the current date using the specified format.TimeTimehh:mmSet the time using the 24-hour format.DOCA0005EN-123334Front panel display and meter setupiEM3000 series user manualSectionParameterOptionsDescriptionSlave Address1 - 247Set the address for this device. The address must be unique for each device in a communications loop.CommunicationBaud Rate19200384009600Select the speed for data transmission. The baud rate must be the same for all devices in a communications loop.ParityEvenOddSelect None if the parity bit is not used. The parity setting must be the same for all devices in a communications loop.NoneNOTE: Number of stop bits = 1.ContrastContrast1-9Increase or decrease the value to increase or decrease the display contrast.Reset Config