Title: +038=320.103-OE009T_02-Multical 801 Energimåler Kamstrup - Produktdatablad.pdf
MULTICAL ® 801 is a robust and rugged calculator. It is ideal for buildings and industries using extra communication pos- sibilities, programmable functions and a wide range of other modules. MULTICAL® 801 is used for measurement of both heat and cooling in all water based plants with flow temperatures from 2 °C to 180 °C and with all flow meter sizes between qp 0.6 m³/h and qp 30,000 m³/h. The meter is simple to install, read and verify. Furthermore, MULTICAL® 801 contributes to keeping the annual operating costs at a minimum with its unique combination of high measuring accuracy and long lifetime. If MULTICAL® 801 is connected to flow meters installed in both inlet and outlet pipes, the meter can monitor leaks and burst in the heating/cooling system. Furthermore, leakages in the tap water system can be monitored by means of pulses if a water meter is connected. MULTICAL® 801 receives volume pulses from the connected flow meters and calculates the energy for every predeter- mined water volume. The energy