Title: Data tester Smestad skole - Detaljert.pdf
Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 38,6 68,0 200,0 11,0 6,9Limit 100,0 - 555,0 50,0 25,0Margin 61,4 - 355,0 39,0 18,1Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36 0,7 3,0 2,3 2,80Worst Value: Local: 36 16,9 43,8 26,9 500,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 25,5 18,8 6,7 27,10Remote: 12 18,3 12,5 5,8 141,00Worst Value: Local: 36 14,8 8,0 6,8 497,00Remote: 78 15,2 8,0 7,2 473,00dB -2020600Local250500 MHzdB -2020600Remote250500 MHzNEXT: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36-45 29,9 26,7 3,2 499,00