Title: 370.011 Siemens Sitrans FUE080 Flowmåler.pdf
2. Select "Execute even if the device TAG does not match the project data TAG." and click "OK" to download all changes from the table to the device.2. Select "Execute even if the device TAG does not match the project data TAG." and click "OK" to download all changes from the table to the device.View process valuesThe system is now ready for normal operation.1. Select "View" → "Display" to see all measured process values.2. Verify that the fields show the expected values.Note Measured Track ValuesThe "Measured Track Values" should show stable values within the normal range, i.e. the gains should be stable values between 3 and 12 (smaller sizes low values and larger sizes high values) and the actual velocities for the paths should be stable, constant and smoothly changing values between 0 and 10 m/s.SITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-0241Commissioning 6.4 Commissioning via PDMStore settings on the PC