Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
The verification equipment is available in three different types, depending on the MULTICAL 801 type used and the temperature points to be tested. The two most common types can be seen below.66-99-370 Standard (EN1434/MID) Type 67-F/K (4-wire Pt100)T1 [°C] 160 80 43T2 [°C] 20 60 40T3 [°C] 566-99-371 Standard (EN1434/MID) Type 67-G/L (4-wire Pt500)T1 [°C] 160 80 43T2 [°C] 20 60 40T3 [°C] 5For other equipment variants (types or temperature points), please contact Kamstrup A/S.112Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 801Function14.3.3 Verification equipment, e.g. item no. 66-99-370 or 66-99-371 is mounted in a standard MULTICAL base and includes battery, verification PCB with connection terminals, interface for calculator, microprocessor, control relays and precision resistors.The connection between verification equipment and MULTICAL 801 consists of a 14-pole test connector.