Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Copper @ 20°CCopper @ 20°C0.252.50.45012.50.0900.505.00.20025.00.0400.757.50.13337.50.0271.5015.00.06775.00.013Table 68.2.2 4-wire sensor pairFor installations, requiring longer cables than listed in the table above, we recommend the use of 4-wire sensor sets.MULTICAL 801 has a ”real” 4-wire construction, which uses two conductors for measuring current and the two conductors for measuring signal, which means that the construction is in theory uninfluenced by long sensor cables. In practice, cables ought not to be longer than 100 m and we recommend the use of 4 x 0.25 mm2.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201677MULTICAL® 801