Title: 370.011 Siemens Sitrans FUE080 Flowmåler.pdf
DirectionNo influence, while "Alarm"No influence, while "Alarm"Forward/Reverse/Forward net/Reverse netAmount per pulseNo influence, while "Alarm"No influence, while "Alarm"Unit: The same as Totalizer unitValue freely selectablePulse widthNo influence, while "Alarm"No influence, while "Alarm"5, 10, 20, 50, 100, 200, 500 msB.2Factory settings for MODBUS communicationParameterDefault settingSlave device address1Data transmission rate19 200Vertical parity positionE-8-1 (0 - even)Response timeout10000 msResponse delay1 msInterframe space35The settings can be changed via SIMATIC PDM or via the MODBUS.SITRANS FUS080 transmitter84Operating Instructions, 07/2012, A5E03059912-02Settings B.3 Measurement report, SONOKIT 2-pathB.3Measurement report, SONOKIT 2-pathSensor geometry measurement reportCompanyNameApproved byDateSensor sizePressure (PN)Tag no.:1A (2A)1A1B2A2B1B (2B)Note→ = Flow directionMeasureCalculation