Title: +038=310.101-RF022T_06-multical 62 Vannmåler Kamstrup - Installasjons- og brukerveiledning.pdf
Lon Works + pulse inputs, type 67-00-245.1.6 The LonWorks module is used for data transfer from MULTICAL® 62 either for data reading/registra- tion or regulation purposes via the Lon-Bus.The module must be powered by 24 VAC.A list of network variables (SNVT) and further details about the LonWorks module appear from data sheet. Regarding mounting we refer to installation instructions 5512-396.See installation instructions 5512-396 (DK) or 5512-403 (GB).5.1.7 Wireless M-Bus + 2 pulse inputs, type 67-00-30 The radio module has been designed to form part of the hand-held Wireless M-Bus Reader sys- tems of Kamstrup A/S at license-free radio frequency (868 MHz).The module fulfils the C-mode specifications of prEN13757-4 and can thus form part of other systems using Wireless M-Bus C-mode communication.