Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
2 tariff conditions, TL2 and TL3, which are always used in the same tariff type, are connected to each tariff function. However, it is not possible to “mix” two tariff types.Example: EE=11 (Power tariff)TA2 shows energy consumed……above power limit TL2 (but below TL3)54Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8016.11.1 Tariff typesThe below-mentioned table lists the tariff types, for which MULTICAL 801 can be configured:EE=TARIFF TYPEFUNCTION001112No active tariffPower tariffFlow tariffNo function Energy is accumulated in TA2 and TA3 based on the power limits programmed for TL2 and TL3. Energy is accumulated in TA2 and TA3 based on the flow limits programmed for TL2 and TL3.13141519T1-T2 tariffInlet temperature tariffOutlet temperature tariffTime controlled tariff