Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 12] [Pair 36] [Pair 12] [Pair 45]8.2 35 0 1.59Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]36.6 494.0 41.4Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 5.2 467.0 29.4 36 5.3 467.0 26.6SR 36-45 5.6 466.0 29.5 36 4.8 466.0 26.6MAIN 36-45 5.5 480.0 28.8 36 5.5 481.0 25.9SR 36-78 6.7 496.0 28.0 36 5.9 496.0 25.210080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 36-45 14.6 477.0 11.7 45 14.3 3.8 50.8SR 36-45 14.6 417.0 12.8 45 14.3 4.0 50.2MAIN 36-45 14.6 477.0 11.7 45 15.3 474.0 8.7