Title: 370.01-RT601_RH601 - Produal - KLH100 (Fukt_Temp.giver).pdf
Technical data Supply Humidity measurement24 Vac/dc (22…28 V), < 2 WHD-R: Relay (24 Vac, 1 A) that switches according to the one measurement value or according to the all values. The relay switching point can be adjusted by using ML-SER tool.Wiring:Range Accuracy (25 °C)Temperature measurementRange Accuracy (25 °C)Outputs Operating conditions Temperature HumidityHousing MountingDimensions (w x h x d)0…100 %rH ±2 %rH0...50 °C ±0.5 °C 0…10 V < 2 mA0...+50 °C 0...100 % RH (non cond.) IP20, ABS plastic on the wall surface or on the standard flush mounting box (60 mm hole distance) 87 x 86 x 30 mmOrdering guide: ModelProduct number 1132210 1132211 1135001 1135002 1135003 1139010KLH100 KLH100-N HD-P HD-PU HD-R ML-SERDescription%RH and °C room transmitter room transmitter with display option, passive potentiometer option, 0…10 V potentiometer option, relay 24 Vac 1 A transmitter commissioning tool