Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
dd-mm-yyyyRead Previously stored value of totalizer 1Value rangeAccess levelDescriptiondd-mm- yyyyR/W*Latest service date (can also be used for date of installation)3.4E+38 h ReadTotal operation hours since last power up0 ... 65535 ReadTotal number of power-ups since first power-up.One or more of the following values: 1, 2, 3, 4, ... up to 16ReadFault status1: Path 1 error2: Path 2 error3: Dual slope calibration error4: Database checksum error5: Low power warning6: Flow overload warning7: Pulse A overload warning8: Pulse B overload warning9: Consumption interval error10: Not used11: Not used12: Not used13: Not used14: Not used15: Not used16: Not useddd-mm- yyyy, hh:mm:ssReadDate of last fault log resetNo, YesR/WReset the fault log and faultsNo, YesR/WSelect 'Yes' to reset active call-up73Parameter lists A.3 DiagnosticModbus AddressParameterDefault valueValue rangeAccess level21209Reset consumption log faultNo