Title: +038=310.101-RF022T_06-multical 62 Vannmåler Kamstrup - Installasjons- og brukerveiledning.pdf
Carry out a function control when the meter has been fully mounted. Press the button MULTICAL® 62 and check that the displayed values for water flow etc. are credible values.on5 Plug-in modulesA number of extra functions can be added to MULTICAL® 62 by means of plug-in modules. The individual modules are briefly described below.5.1 Base modules 5.1.1 Data + pulse inputs, type 67-00-10The data terminals are used for connection of e.g. a PC. The signal is passive and galvanically separated by means of optocouplers. Conver- sion into RS232 level requires connection of data cable 66-99-106 (D-Sub 9F) or 66-99-098 (USB) with the following connections: (DAT) Brown 62 (REQ) White 63 (GND) Green 64The pulse inputs can be used for connection of electricity and water meters. Please note the maximum pulse frequency as well as correct pulse coding (l/imp. and Wh/ imp.), which are selected by means of the FF and GG configurations. Input A 65 - 66 Input B 67 - 68Kamstrup A/S • 55121068_B3_GB_03.20157