Title: DOCA0005EN-12.pdf
Dismounting the meter from a DIN rail1. Use a flat-tip screwdriver (≤ 6.5 mm / 0.25 in) to lower the locking mechanism andrelease the meter.2. Lift the meter out and up to free it from the DIN rail.Considerations for iEM31•• and iEM33•• devices associated with a contactorConnection requirements for iEM3100 / iEM3110 / iEM3115 / iEM3135 / iEM3150 / iEM3155 / iEM3165 / iEM3175 / iEM3300 / iEM3310 / iEM3335 / iEM3350 / iEM3355 / iEM3365 / iEM3375:When the meter is associated with a contactor, connect the meter upstream of the contactor.When the meter is associated with a contactor, connect the meter upstream of the contactor.The meter must be protected by a circuit breaker.LLiEMiEMLoadLoadDOCA0005EN-12iEM3000 series user manualSplit-core LVCT & Rogowski Coil recommendationsSplit-core LVCTPart NumberSensing CurrentFrequencyLVCT00102S100A50/60HzLVCT00202S200A50/60HzLVCT00302S300A50/60HzLVCT00403S400A50/60HzLVCT00603S600A50/60HzLVCT00803S