Title: Data tester Smestad skole - Detaljert.pdf
Unspecified-Building Unspecified-Floor Unspecified-Room Unspecified-Rack Unspecified-PanelOverall Result:Local Ser. No.: Local Adapter: Local Calibration Date: Device Software:pw20201689 Cat 6A Link Apr 17 2013 7.4Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 35,7 68,0 185,0 10,0 6,6Limit 100,0 - 555,0 50,0 25,0Margin 64,3 - 370,0 40,0 18,4Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 0,6 3,0 2,4 2,35Worst Value: Local: 36 15,3 43,8 28,5 500,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 78 17,1 9,9 7,2 256,00Remote: 78 25,8 18,5 7,3 32,25Worst Value: Local: 12 16,5 8,0 8,5 493,00