Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
ReadAmplifier setting when measuring path 1ReadAmplifier setting when measuring path 2.Valid for 2-path solutions onlyR/WSelect 'Yes' to enable output AR/WPulse output calculation on pure forward flow or pure reverse flow - or net flow for forward and reverse pulse flow.Valid if pulse output A is enabledR/WVolume per pulse and selected unitValid if pulse output A is enabled.R/WPulse length when the pulse is active.Valid if pulse output A is enabledR/WSelect 'Yes' to enable output BR/WConfiguration of output B as pulse - alarm or call up function.Valid if pulse output B is enabled71Parameter lists A.2 OutputModbus AddressParameterDefault valueValue rangeAccess levelDescription40824442 Pulse B directionReverseForward, Reverse, Forward net, Reverse net.R/WPulse output calculation on pure forward flow or pure reverse flow - or net flow for forward and reverse pulse flow.Valid if pulse output B is enabled and pulse function selected40825