Title: Pos. 12A Kjolebenker ProffG2_Bruksanvisning.pdf
current which can be applied to each relay (see Technical Data). Fit the probe where it is not accessible by the End User. The instrument must not be opened. Push the SET key for more than 2 seconds to change the Set point value; 1. Defrost in progress On HOW TO CHANGE THE SETPOINT Programming mode Flashing 2. Released the keys, then push again the SET+ Lyser The instrument shall not be used for purposes different from those described hereunder. It cannot be displayed immediately followed from the Hy parameter. To change the parameter’s value operate as follows: Check the application limits before proceeding. Programming mode Flashing through special parameters that can be easily programmed through the keyboard or the by HOTKEY. Measurement unit On provided with 1 NTC probe input and one configurable digital input. The instrument is fully configurable to change its value. 3. Use or Ensure that the wires for probes, loads and the power supply are separated and far enough from each In case of failure or faulty