Title: DOCA0005EN-12.pdf
List of objects in the meter:Object_listR—variesiEM3165 / iEM3365: DE1, AI0-AI48, AV0, BI0-BI6iEM3265 / iEM3465 / iEM3565: DE1, AI0-AI55, AV0, BI0-BI6Max_APDU_Length_ AcceptedR—480The maximum packet size (or application protocol data unit) that the meter can accept, in bytesSegmentation_Suppo rtedR—0x03The meter does not support segmentation.DateLocal_DateR—configurableNOTE: You must use the front panel to set the meter’s date.TimeLocal_TimeR—configurableNOTE: You must use the front panel to set the meter’s time.APDU_TimeoutR/W √1000 - 30000The amount of time (in milliseconds) before the meter tries to resend a confirmed message that has not been answered.Number_Of_APDU_R etriesR/W √1 - 10The number of times the meter tries to resend an unanswered confirmed request.DOCA0005EN-12DOCA0005EN-12Communications via BACnetDevice object propertyR/WStore dPossible valuesDescriptionMax_MasterR/W √1 - 127