Title: 370.011 Siemens UC50 Kalkulator.pdf
The flow sensor must be installed at the same circuit as the temperature sensors. When a polarity dependent pulse transmitter is used, take care of the correct orientation. The reference potential must be connected on the right site (-), the positive voltage must be connected on the left spring-type terminal (+).The cable must be fixed to the outer sheath with a clamp for stain relief. If the line has a shielding braid, the shielding braid covering can be pushed over the counter covering of the cable and be connected by a clamp. The cage clamp terminals can be used for cable cross-section of 0.5-1.5mm² (solid or stranded). In order to preserve the IP protection class of the housing, the outside diameter of the cable sheath shall be between 3.7 and 4.4 mm.black +green -