Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
The time tariff is suitable for billing in housing areas close to industrial areas with large district heating consumption as well as billing industrial customers.The adjustment of the clock ought to be checked in order to secure correct time as a basis for the time tariff.EE=20 Heat/cooling volume tariffHeat/cooling volume tariff is used for division of volume into heat and cooling consumption. TA2 accumulates the volume consumed together with E1 (heat energy) and TA3 accumulates the volume consumed together with E3 (cooling energy).T1 ≥ T2Volume is accumulated in TA2 and V1T2 > T1 and T1 < T1 limitVolume is accumulated in TA3 and V1TL2 and TL3 are not usedT2 > T1 and T1 > T1 limitVolume is accumulated in TA2 and V1For combined heat/cooling metering the total volume is accumulated in the register V1, whereas heat energy is accumulated in E1 and cooling energy in E3. The heat/cooling tariff is used for dividing the consumed volume into heat and cooling volume.