Title: Data tester Smestad skole - Detaljert.pdf
Unspecified-Building Unspecified-Floor Unspecified-Room Unspecified-Rack Unspecified-PanelOverall Result:Local Ser. No.: Local Adapter: Local Calibration Date: Device Software:pw20201689 Cat 6A Link Apr 17 2013 7.4Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 64,9 68,0 336,0 18,0 11,5Limit 100,0 - 555,0 50,0 25,0Margin 35,1 - 219,0 32,0 13,5Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 1,3 3,0 1,7 2,80Worst Value: Local: 45 27,4 43,8 16,4 500,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 15,9 8,8 7,1 332,00Remote: 12 22,9 16,6 6,3 55,25Worst Value: Local: 12 15,9 8,8 7,1 332,00