Title: +038=67.005-##003._04-Ø6 Trykktransmitter.pdf
Operation Pressure applied to the P499 transducer acts on a piezoresistive strain gauge, which is bonded with high- temperature glass onto a stainless steel diaphragm. The strain gauge translates media pressure into an electrical signal. An Application-Specific Integrated Circuit (ASIC) conditions and amplifies the signal. The output signal varies in direct linear proportion to the sensed pressure.DimensionsSee wire harness ordering information.n e m i d 9 9 4 P G I F:Packard Connector Plug with Wire Harness5/16 (8)Maximumo1/4 in. SAE 45 Flare Internal Thread (7/16-20 UNF) with Depressor (Style 47)P499 transducers provide several output options:0.5 VDC to 4.5 VDC ratiometric0 VDC to 10 VDC4 mA to 20 mA2-11/16 (69)Pressure connections are available in two standard styles:1/2 (12)7/8 (22)Hex1/4 in. SAE 45° flare (7/16-20 UNF) internal thread fitting with Schrader® valve depressor (Style 47)1/8-27 NPT external thread fitting (Style 49)