Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Current flow indication (l/h or m³/h) is calculated based on volume pulses/10 s (see paragraph 6.5)Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201619MULTICAL® 8013.3.5 High-resolution CCC-codes for ULTRAFLOW (for cooling meters etc.)Number of decimals in displayCCC No.184 107 136 138183 185 186 187 188 189 191 192Pre- count er300 100 500 250150 100 500 250 150 100 400 250Flow factor78642 235926 471852 9437041572840 2359260 471852 943704 1572840 2359260 589815 943704kWh1 1 0 00 0MWh Gcal3 33 3 2 2 2 2 1 1GJ3 3 2 22 2 1 1 1 1 0 0m³[tons] 3 3 2 22 2 1 1 1 1 0 0l/h0 0 0 00 0m³/h2 2 2 2 1 1kW1 1 1 11 1 0MW3 3 3 2 2Imp./l300 100 50.0 25.015.0 10.0 5.0 2.5 1.5 1.0 0.4 0.25qp [m³/h]0.6 1.5 3.5 6.0 10 10 15 40 60 100 150 400 600 1000 1000Type No.Flow sensor1-2-7-8 1-2-7-8-N 1-2-7-8-N 1-2-7-8-N1-2-7-8 1-2-7-8-N 1-2-7-8-N 1-2-7-8-N 1-2-7-8 1-2-7-8-N 1-2-7-8-N 1-2-7-8-N1931501572840100120.15