Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
Note The installation orientation of the battery pack may influence the battery capacity. Optimal battery capacity is achieved with the battery pack in an upright position.10.3Sensor geometrical dataIn order to ensure maximum measurement accuracy, precise sensor-specific data are required.DataSensor sizeRequired accuracyPath angleDN 100-1200 (8" – 48")0.1°Transducer and path distancesDN 100-400 (8 " – 32")0.2 mmOuter pipe diameterDN 400-1000 (32" – 48")0.8 mmPipe wall thicknessDN 1000-1200 (40" – 48")2.0 mmSITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-0263Technical data 10.4 DimensionsProper use of the measurement report normally ensures that measured values come very close to the nominal dimensions listed in the appendix "Settings" "Dimension dependent settings", SONOKIT 2-path and "Dimension dependent settings", SONOKIT 1-pathNote Sensor recalibration