Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
EE=20 ought always to be selected together with heat/cooling meters, type 67-xxxxxxx-6xx.EE=21 PQ tariffThe PQ tariff is a combined power and flow tariff. TA2 functions as power tariff and TA3 functions as flow tariff.P ≤ TL2 and q ≤ TL3Accumulation in main register onlyP > TL2Accumulation in TA2 and main registerTL2 = power limit (P)q > TL3Accumulation in TA3 and main registerTL3 = flow limit (q)P > TL2 and q > TL3Accumulation registerin TA2, TA3 and mainThe PQ tariff can e.g. be used for customers paying a fixed charge based on max. power and max. flow.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201657MULTICAL® 8016.12 Data loggersMULTICAL 801 includes a permanent memory (EEPROM), in which the values from various data loggers are saved. The meter includes the following data loggers:Data logging intervalData logging depthLogged valueYearly logger15 yearsCounter registerMonthly logger36 monthsCounter registerDaily logger