Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
The configuration of MULTICAL® 801 can be read out directly. The program is self-explanatory as to most coding numbers (see text in ”combo-boxes”). Further details can be found in the respective paragraphs of the technical description.There are two programming options ”Partial programming” and ”Total programming”.”Partial programming” does not allow change of coding which is important to energy calculation, e.g. Type No. and Prog. No.By means of ”Total programming” it is possible to change the remaining values too. Programming is only possible if the programming seal is broken (see picture below) and the internal programming lock is closed (short-circuit pen 66-99-278). In order to carry out verification, the jumper connection must remain closed throughout the verification. It is not possible to change the serial number as it is a unique number allocated to the meter during production.”V2(CCC)”, ”T1”, ”T2” and ”Max T1 for cooling” can be disabled, depending on the meter type in question.