Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
Polished stainless steel pipe0.0001 mStandard carbon steel pipe (select STEEL)0.0004 mRusty carbon steel pipe0.001 ... 0.002 mConcrete pipe (Select CONCRETE)0.002 ... 0.005 mSITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-0237Commissioning 6.4 Commissioning via PDM3. Depending on the SONOKIT type (1-path or 2-path), enter additional sensor data:Parameter1-path2-pathNumber of paths12Outer pipe diameterDU DUPipe wall thicknessttAverage path distance*HHPath 1 angleA1A1Path 2 angleA2Average transducer 1 distanceL1L1Average transducer 2 distanceL2Velocity (auto calculated with Qmax)VVFor crossed paths solutions with no path displacement: Enter "0".Figure 6-21-path sensor dataSITRANS FUS080 transmitter38Operating Instructions, 07/2012, A5E03059912-02Figure 6-32-path sensor data