Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
The connection cable ought to have an outer diameter of 5-6 mm in order to obtain optimum tightness of both MULTICAL 801 and the screw-joint for the 4-wire sensor. The isolation material/cover of the cable ought to be selected based on the maximum temperature in the installation. PVC cables are normally used up to 80°C and for higher temperatures, silicone cables are often used.Kamstrup’s 4-wire sensor pair has a replaceable sensor insert and is available in lengths of 90, 140 and 180 mm.78Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8018.3 Pocket sensorsThe Pt500 cable sensor is constructed with 2-wire silicone cable and closed with a D 5.8 mm shrunk on stainless steel tube, which protects the sensor element.