Title: 91290FDV Campus F13 433.04 West elektro.pdf
70.1–700 (CT x VT)7.1–70 (CT x VT)< 7.1 (CT x VT)90mswrong current direction (only with PFB option or with “B” measurement selection in case of X option)6A, @ 50Hz 5 In1.2 Un 2 Un1.2Mohm < 1.25VA Installation guide to indicate if connections are correctly carried out. Can be disabled. Indicates if the phase sequence is not the correct one (L1-L2-L3) Indicates if the current direction is not the right one (only with PFB option or with type “B” measurement selection in case of XEnergy additional errors Influence quantities Temperature drift Sampling rateAccording to EN62053-21 ≤200ppm/°C 4096 samples/s @ 50Hz 4096 samples/s @ 60HzDisplay and touch key-pad3Input specifications (cont.)Load conditionsoption). The wrong connection detection works in case of loads with: - PF>0.766 (<40°) if inductive or PF>0.996 (<5°)if capacitive - a current at least equal to 10% rated currentDigital input specificationsDigital inputs Function