Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
SITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-02SettingsBB.1Factory settingsThe FUS080/SONOKIT is from factory configured according to the order specification.For the calibration / configuration of the flowmeter the application specific pipe dimensions and the specific transducer locations are used.Table B- 1Factory settingsIdentification1-path factory settings (DN (100 to DN 1200)2-path factory settings DN (DN 200 to DN 1200)Possible settingsMeter identificationFree text 15 charactersApplication locationFree text 15 charactersDisplay menuMenu 1- 5 (all)Menu 1 – 5 (all)Menu 1 – 5, min one of the fivePipe dataNumber of paths121 or 2Inner pipe diameter (m)See "Dimension dependent settings"See "Dimension dependent settings"Input to pipe geometry programTransducer distance (m)See "Dimension dependent settings"See "Dimension dependent settings"Input to pipe geometry program