Title: 370.011 Siemens Sitrans FUE080 Flowmåler.pdf
– Ambient temperature: -5 to +60 °C (23 to 140 °F)The enclosure rating is IP67 (NEMA 4X/6) or better.NOTICEMake sure that the pressure and temperature specifications indicated on the device nameplate/label will not be exceeded.NOTICEDirect sunlightDo not expose the device to direct sunlight!4.1Installing FUS080Note Sensor installationIt is required that the sensor has been installed according to the relevant operating instructions and that the sensor measurement report has been completed!SITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-02419Installing/Mounting 4.1 Installing FUS0801. Mount wall/pipe mounting bracket in an appropriate placePipe mountingWall mountingNoteTake coaxial cable length into consideration, and allow adequate space for cable inlets underneath and on both sides.SITRANS FUS080 transmitter20Operating Instructions, 07/2012, A5E03059912-021. Wiring the transducer cables (Page 22)