Title: Data tester Smestad skole - Detaljert.pdf
Unspecified-Building Unspecified-Floor Unspecified-Room Unspecified-Rack Unspecified-PanelOverall Result:Local Ser. No.: Local Adapter: Local Calibration Date: Device Software:pw20201689 Cat 6A Link Apr 17 2013 7.4Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 41,1 68,0 214,0 12,0 7,0Limit 100,0 - 555,0 50,0 25,0Margin 58,9 - 341,0 38,0 18,0Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36 0,8 3,0 2,2 2,80Worst Value: Local: 36 17,4 43,6 26,2 498,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 78 15,1 10,0 5,1 250,00Remote: 12 24,2 18,4 5,8 32,50Worst Value: Local: 78 14,1 8,0 6,1 497,00