Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
Operating Instructions, 07/2012, A5E03059912-02Settings B.4 Dimension dependent settings, SONOKIT 2-pathPath 1Path 1Path 2Path 2anglelengthanglelength[º][m][º][m]603.3378603.3378603.4384603.4384603.5391603.5391603.6398603.6398603.7404603.7404603.8411603.8411603.9417603.9417604.0424604.042487Settings B.5 Measurement report, SONOKIT 1-pathB.5Measurement report, SONOKIT 1-pathSensor geometry measurement reportCompanyNameApproved byDateSensor sizePressure (PN)Tag no.:1A1A1B1BNote→ = Flow directionMeasureCalculationResultCircumference (C)C / π = Outer diameter of the pipe (Du)DuPipe wall thickness (t)Du - (2 x t) = Internal diameter (Di)DiTransducer distance (L)L1 =L2 =Angle θ transducer 1A (angle 1A = 1B ; +/- 0,1º)Average path angle "(1A + 1B) / 2"Angle θ transducer 1B (angle 1A = 1B ; +/- 0,1º)Cable length (distance from tranducer to converter x 2)Qmax