Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
14.3.2 Verification equipment, e.g. item no. 66-99-370 for verification of 67-F/K (4-W/Pt100) or item no. 66-99-371 for verification of 67-G/L(4-W/Pt500) is used for verification of the calculator, MULTICAL 801. The verification includes energy verification of ”E1” and ”E3”, test of volume inputs ”V1”, ”V2”, ”VA” and ”VB” as well as test of temperature input ”T3”.Different temperatures are simulated for the two sensor inputs ”T1” and ”T2”. Together with the volume simulation (autointegration), these temperatures form the basis of the verification of the energy calculation.The equipment was primarily constructed for use in laboratories, which test and verify heat meters, but can also be used for performance testing the meter.The computer program ”METERTOOL HCW” is used for configuration, test and verification.