Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
Operating Instructions, 07/2012, A5E03059912-02Value range215153.4E+38 m3.4E+38 m0.05 ... 1.200 mAccess levelR/WReadHW keyReadReadReadParameter lists A.4 Meter setupDescriptionNumber of paths on the sensorThe maximum frequency for flow measurementThe frequency at which the flow is measuredR-factor for path 1.For FUS080/SONOKIT (retrofit) installations this parameter will automatically be calculated by the pipe geometric tool (see 'Pipe geo. assistant' in 'Device' menu).R-factor for path 2.For FUS080/SONOKIT (retrofit) installations this parameter will automatically be calculated by the pipe geometric tool (see 'Pipe geo. assistant' in 'Device' menu).Valid only for 2-path solutionsInner pipe diameter in meters.For FUS080/SONOKIT (retrofit) installations this parameter will automatically be calculated by the pipe geometric tool (see 'Pipe geo. assistant' in 'Device' menu).79Parameter lists A.5 Human InterfaceA.5Human Interface