Title: +175=574.001_OE001_Manual - Varmemåler M3 - Løs fra rør.Produktinformasjon.pdf
The log of all faults is shown at level 6 of the consulting menu, where the abbreviation Err followed by three digits identifies the anomaly.Hydrosplit M3 Manuale utente – User manual - v1.6.5_ITA_ENG53Here is a list of all the error codes:ErrorDescriptionPrescriptionErr101Err102Err103Err104Err105One of the two temperature sensor cables is cut; At least one cable is disconnected One of two temperature sensor measurement out of range; at least one sensor is damaged. The error resets after the sensor replacement. Reading/writing error; Memory of historical data damaged. Calendar Error (time bases corrupted); Faulty time bases oscillator. MBUS communication interface errorCheck the integrity and the connection of the temperature sensor cables. Check the integrity and the functioning of the temperature sensors.Return the device to the manufacturer. Return the device to the manufacturer. Return the device to the manufacturer.Err106 Err107Err108Err201 Err202 Err203