Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
P = q (T1 – T2) x k [kW] or [MW]”k” being the heat coefficient of water, which is currently calculated by MULTICAL 801 according to EN 1434:2007.Example: - -Current water flow, q = 316 l/h and flow meter mounted in outlet pipeT1 = 70.00°C and T2 = 30.00°C, k-factor is calculated at 1.156 kWh/m3/KP = 0.316 (70-30) x 1.156 = 14.6 [kW]Current power in V1Both heat and cooling power is displayed numerically (without signs)42Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8016.7 Min. and max. flow and power, V1MULTICAL 801 registers minimum and maximum flow and power on both monthly and yearly basis. The complete registration can be read via data communication. Furthermore, a few monthly and yearly registers can be read from the display, depending on the selected DDD-code.The min. and max. registrations include the following flow and power values with indication of date:Type of registrationMax. dataMin. dataYearly data Monthly data