Title: +038=310.101-RF022T_06-multical 62 Vannmåler Kamstrup - Installasjons- og brukerveiledning.pdf
The module has a galvanically separated data port which functions with the KMP protocol. The data output can be used for e.g. connec- tion of external communication units or other hardwired data communication which it is not expedient to carry out via the optical communi- cation on the meter’s front.T A DQ E RD N GD N G62: DATA (brown) – 63:REQ (white) – 64: GND (green). Use data cable type 66-99-106 with 9-pole D-sub or type 66-99-098 with USB connector.Furthermore, the module includes an hourly data logger.Only current and accumulated data can be read. Hourly/daily/monthly/yearly data loggers cannot be read through the data port of top module 67-05.Kamstrup A/S • 55121068_B3_GB_03.201511MULTICAL® 62 • Water meter5.2.2 Type 67-07: RTC + M-BusM-Bus can be connected in star, ring and bus topology. Depending on M-Bus Master and cable length/cross section up to 250 meters with primary addressing can be connected, and even more using secondary addressing.