Title: Data tester Smestad skole - Detaljert.pdf
Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 49,2 68,0 256,0 15,0 8,5Limit 100,0 - 555,0 50,0 25,0Margin 50,8 - 299,0 35,0 16,5Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 0,9 3,0 2,1 2,80Worst Value: Local: 36 21,0 43,6 22,6 498,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 78 26,4 18,7 7,7 28,75Remote: 12 20,2 14,3 5,9 93,50Worst Value: Local: 12 17,0 8,9 8,1 322,00Remote: 12 15,6 8,0 7,6 419,00dB -2030800Local250500 MHzdB -2030800Remote250500 MHzNEXT: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36-45 30,1 26,7 3,4 500,00