Title: Manual - Varmemåler M3 - Løs fra rør.pdf
Confirm the selected parameter by holding the T2 button for more than 3 seconds, thus moving to the next parameter.15) Finally to exit the menu and save the configured parameters, select the number [1] in the ESC panel by using the button T1. On the contrary, select the number [0] to return to the first sublevelWarning: before confirming the settings saving, make sure to properly set all parameters, in particular the input pulse value of the flow sensor as it is no longer editable.CONSULTING MENUThe consulting menu is divided in six levels by a numeric index, always shown in the left-upper part of the display.By pressing the T1 button it is possible to choose the desired level, and by pressing the T2 button it is possible to consult the sublevels of the preset level.Here below, it is shown the navigation map of the consulting menu.LEVEL 1: CUMULATIVE VALUES1.1 Accounted energy (Heating) – cumulative value1.2 Display test – All segments on