Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
It is also possible to read the modem’s signal strength by means of an SMS. You receive a reply with the modem’s current signal strength on a scale of 0 to 31, the best value being 31. The signal strength must be minimum 12. See the examples on the next page.NOTE: SMS commands must be written in either capital letters or small letters, i.e. an SMS command must not include a mixture of capital and small letters.64Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016READ_HEAT_METER – for reading a MULTICAL ® 801SyntaxReturn reply, errorExample of SMS commandExample of correct replySIGNAL – for reading the signal strengthSyntax, commandReturn reply, errorExample of SMS commandExample of correct replyKamstrup A/S · Technical Description · 5512-571_R1_GB _11.2016=READ_HEAT_METER#NO ANSWER=READ_HEAT_METER#12.067Gj, 120.0kW 6930 Hours, Meter No.: 6055524=SIGNAL#NO ANSWER=SIGNAL#Signal: 16(0-31)MULTICAL® 80165MULTICAL® 8017 Flow meter connection