Title: +038=310.101-RF022T_06-multical 62 Vannmåler Kamstrup - Installasjons- og brukerveiledning.pdf
Cable resistance in network: < 29 ohmCable capacity in network:< 180 nFThe connection polarity of terminals 24-25 is unimportant.Unless otherwise stated in the order, the primary address consists of the last three digits of the customer number, it can be changed, however, via the PC program METERTOOL.Module to be used in mains supplied meters only.5.2.3 Type 67-0B: RTC + pulse output for CV+ prog. data loggerThe RTC and pulse output functions of this top module are identical with the functions described for top module 67-08.The top module is prepared for use in a Kamstrup radio network together with the Radi- oRouter base module 67-00-21-000-3xx, read data being transferred to the system software via the network unit RF Concentrator.5.2.4 Type 602-0C: 2 pulse outputs for CEand CVThis top module has two configurable pulse outputs, which are suitable for volume pulses.