Title: +135.01=544.001-XS001T_01-Pasientsignal operating_and_installation_instruction_2011.pdf
76910XE / 04.2011193.2 Wall installation of compact call set (Part No. 76910C1 / 76910D1)The components of the compact call set are installed within a flush wall box and can optionally be supplemented by a sealing kit for standard installation units.Possible cover platesComponents of the wet room for optional installation of the sealing kit1234567Fig. 6: Installation sequence (example with cancel button) Flush wall box Ø 60 mm (installation depth at least 44 mm)  Seal for installation unit (sealing kit for standard installation units)   Seal for frame (sealing kit for standard installation units)  Cover plate frame  Seal for cover plate (sealing kit for standard installation units)  Cover plate for duty room unit, pull switch, electronic module (Part No. 72585B)Installation unit