Title: Simployer - Data testrapport =514.204-FS001.pdf
SR 1,2-3,6 26.1 247.0 -2.5 3,6 27.9 249.0 -5.7PASS Worst Pair RL (dB) Freq. (MHz) Limit (dB)MAIN 4,5 3.3 143.5 10.4SR 4,5 1.1 152.5 10.2MAIN 4,5 4.7 249.5 8.0SR 4,5 1.4 247.5 8.1Measurement is within the accuracy limits of the instrument.LinkWare™ PC Version 10.4Project: DEFAULT Simployer - Data testrapport =514.204-FS001.flwSite: SIMPLOYERCable ID: 521.204-UD89 Date / Time: 28/01/2022 14:04:43 Headroom 7.1 dB (NEXT 1,2-3,6) Test Limit: EN50173 Channel Class E Cable Type: Cat 6 F/UTP NVP: 70.0%Operator: TORKIL GRIMSMO Software Version: 2.7800 Limits Version: 1.9500 Calibration Date: Main (Tester): 20/01/2009 Remote (Tester): 20/01/2009Length (m) Prop. Delay (ns), Limit 555 Delay Skew (ns), Limit 50 Resistance (ohms), Limit 25.0[Pair 1,2] [Pair 3,6] [Pair 3,6] [Pair 4,5]4.6 23 1 2.2Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 3,6] [Pair 3,6] [Pair 3,6]33.8 250.0 35.9Worst Case Margin Worst Case Value