Title: Data tester Smestad skole - Detaljert.pdf
Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 52,4 68,0 272,0 15,0 8,9Limit 100,0 - 555,0 50,0 25,0Margin 47,6 - 283,0 35,0 16,1Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36 1,0 3,0 2,0 2,20Worst Value: Local: 36 22,3 43,8 21,5 500,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 15,9 8,6 7,3 348,00Remote: 12 15,9 12,7 3,2 134,50Worst Value: Local: 12 15,9 8,6 7,3 348,00Remote: 12 15,2 10,0 5,2 254,00dB -2030800Local250500 MHzdB -2030800Remote250500 MHzNEXT: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36-45 30,2 26,7 3,5 500,00