Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Ball valves with sensor connecting piece are available in G½, G¾, G1, G1¼ and G1½No.6556-474 G½6556-475 G¾6556-476 G16556-526 G1¼6556-527 G1½Recommended temperature sensorDS 27.5 mm DS 27.5 mm DS 27.5 mm DS 38 mmDS 38 mmMax. 130 °C and PN16Figure 2080Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8019 Other connections 9.1 Pulse outputs CE and CV [16-19] MULTICAL® 801 has pulse outputs for energy and volume pulses respectively. CE on terminals 16-17 releases one pulse per least significant digit in the energy count of the display. CV on terminals 18-19 releases one pulse per least significant digit in the volume count of the display.