Title: +175=574.001_OE001_Manual - Varmemåler M3 - Løs fra rør.Produktinformasjon.pdf
Remove the bracket behind of the calculation unit and secure it to the wall. There are two types of wall mounting allowed:1)2)direct wall mounting by using the mounting bracket and the following hooking of the calculation unit on the same bracket; by wall mounting of a Din Rail and following hooking of the mounting bracket (see pictures below).Case 1Case 2Hydrosplit M3 Manuale utente – User manual - v1.6.5_ITA_ENG33Then apply the basis on the bracket and secure it with any anti-removal seals.After the installation, connection, configuration and commissioning, close the cover and apply adhesives and / or lead sealings.TEMPERATURE SENSORS INSTALLATIONFor the temperature sensors installation, see related manual.FLOW METERS INSTALLATIONFor the flow meters installation, see related manual.COMPONENTS CONNECTIONSThe flow/volume meters, temperature sensors, M-Bus network etc. should be connected on the terminal board inside the calculator.