Title: +038=310.101-RF022T_06-multical 62 Vannmåler Kamstrup - Installasjons- og brukerveiledning.pdf
Possible reason Power supply missingProposal for correction Change battery or check mains supply Change battery or check mains supply. Is there 3.6 VDC on terminals 60(+) and 61 (-)? The info code is corrected automatically Check the error indicated by the info code There is air in the flow sensor? Bleed the system and check the meter again.Power supply missingIf “INFO” = 1Supply voltage has been interrupted Read “INFO” in the displayNo accumulation of volume (m3) If “INFO” = 16Communication errorIf “INFO” = 2048If “INFO” = 4096If “INFO” = 16384Flow sensor programmed with wrong pulse figure Signal too weakFlow sensor mounted in wrong directionCheck that the flow direction matches the arrow on the flow sensor Please contact Kamstrup A/SThere is air in the flow sensor? Bleed the system and check the meter again Check that the flow direction matches the arrow on the flow sensor16Kamstrup A/S • 55121068_B3_GB_03.2015MULTICAL® 62 • Water meter9 Setup via front keys