Title: 40 Produktark nettanalysator 4x-serie.pdf.pdf
For devices with pulse counting feature, the minimum time necessary for the pulse counting can be selected from following options:1. 40 milliseconds 2. 60 milliseconds 3. 80 milliseconds 4. 100 milliseconds 5. 150 milliseconds 6. 200 milliseconds 7. 300 milliseconds 8. 400 milliseconds 9. 500 millisecondsThe device will ignore pulses that have width less than the selected value.371. Press the SET button while on Pulse Inputs menu. 2. On the displayed selection screen, press the SET button to start the selection. 3. Select a pulse width by using UP/DOWN buttons. 4. When exiting from the menu with the BACK or ESC button, don’t forget to save the changes.Output Parameter SettingsWhen an output or input/output module is connected to the device, the purpose of the output is selected from this menu. The number of appearing menus depends on how many outputs are available on the connected module. Digital output of the device can be used for one of the following applications: