Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
Bad inlet conditions (the higher the delta, theworse the inlet conditions)Disturbances in the pipe (bad flow profile)Unstable/fluctuating valuesAir bubbles or solids in mediaBad inlet conditionsThe actual velocities for the paths should be stable, constant and smoothly changing values between 0 and 10 m/s.SITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-0259Troubleshooting/FAQs 9.2 Diagnosing with PDM60SITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-02Technical data1010.1SITRANS FUS080Table 10- 1 Technical data, FUS080DescriptionSpecificationEnclosure design/materialFiber glass reinforced polyamide in light-gray colorEnclosureIP67 according to EN 60529 and DIN 40050 (NEMA 4X/6)Wall mounting kitTerminal box made of fiber glass reinforced polyamide in light-gray color with 2 + 2 cable glands (output, supply, and transducer cables) and a stainless steel bracket for wall or pipe mountingCable glands