Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Figure 3The short-circuit pen functions in >back-up mode< as ”Total reset” and >with supply voltage< as ”Total Prog”Do not forget to switch on the supply voltage again.Note: ”Date” is after reset set to 2000.01.01. Therefore, do not forget to adjust date/time via hand-held terminal or PC with METERTOOL if correct time is important for the application in question.6.15 SMS commands MULTICAL® 801 can be read by means of an SMS. In order to do so, a GSM-module fitted with a SIM-card must be mounted in the meter (see paragraph 11.1.5). You send an SMS from a mobile phone direct to the meter. Subsequently, you receive a reply with the following values:Acc. energy: [kWh], [MWh], [GJ] or [Gcal] • Current power: [kW] or [MW] • Hour counter • Meter number