Title: Data tester Smestad skole - Detaljert.pdf
Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 67,0 68,0 348,0 19,0 11,5Limit 100,0 - 555,0 50,0 25,0Margin 33,0 - 207,0 31,0 13,5Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 1,4 3,1 1,7 3,10Worst Value: Local: 45 28,0 43,7 15,7 499,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 78 14,8 8,0 6,8 469,00Remote: 78 22,7 18,4 4,3 32,50Worst Value: Local: 78 14,8 8,0 6,8 469,00Remote: 12 15,7 8,5 7,2 358,00dB -2030800Local250500 MHzdB -2030800Remote250500 MHzNEXT: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36-45 28,4 26,7 1,7 499,00