Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
E1=V1(T1-T2)k T1: Inlet / T2: Outlet T1 > T2 Heat energy (V1 in inlet or outlet pipe)1+2+3+4+5+6+8Legal Display/Data/LogE2=V2(T1-T2)k T2: OutletT1 > T2 Heat energy (V2 in outlet pipe)2+7Display/Data/LogE3=V1(T2-T1)k T2: Inlet / T1: Outlet T2 > T1Cooling energy (V1 in inlet or outlet pipe)1+10Legal Display/Data/LogE4=V1(T1-T3)k T1: InletT1 > T3Forwarded energy7+9+10Display/Data/LogE5=V2(T2-T3)k T2: InletT2 > T3Returned energy or tap from outlet pipe5+7+9Display/Data/LogE6=V2(T3-T4)k T3: InletT3 > T4Tap water energy, separate3+6Display/Data/LogE7=V2(T1-T3)k T3: OutletT1 > T3Returned energy or tap from inlet pipe4+8Display/Data/LogE8=m3 x T1Average temperature in inlet pipeDisplay/Data/LogE9=m3 x T2Average temperature in outlet pipeSee paragraph 6.2.2Display/Data/LogE1…E76.2.1 Energy types E1…E7 are described by application examples below.Application no. 1Closed thermal system with 1 flow meter