Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
4. Click on "Apply pipe geometric settings" and then on "OK" to transfer the data into appropriate "Value" and "Unit" cells. Values are transferred and shown in metric units only. Close the pipe geometric tool dialog box by clicking on "OK". Click "OK" to confirm the data entered.4. Click on "Apply pipe geometric settings" and then on "OK" to transfer the data into appropriate "Value" and "Unit" cells. Values are transferred and shown in metric units only. Close the pipe geometric tool dialog box by clicking on "OK". Click "OK" to confirm the data entered.Figure 6-4Changed values successfully transferredSITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-0239Rn for path 1 and path 2Cal. factorFor further information, see:Dimension dependent settings, SONOKIT 2-path (Page 86) or SONOKIT 1-path (Page 89).Measurement report, SONOKIT 2-path (Page 85) or SONOKIT 1-path (Page 88).6.4.5Optimizing the system