Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
9.2 Analog outputs [80-87] MULTICAL® 801 is available with 4 analog outputs. The outputs are active 0-20 mA or 4-20 mA, can be loaded with 0…500 Ω and are optoisolated in relation to the supply. The 4 analog outputs, however, are not mutually isolated. All values of the four analog outputs are updated every 10 seconds. The total response time however, may be up to 30-40 seconds including the response time for the flow sensor, the calculator and the digital to analog conversion. This response time has to be considered when using the analog outputs for other purposes than remote displaying.Example of configuration of the analog outputs: