Title: DOCA0005EN-12.pdf
1TypeType 1 (IEC 61131-2)Programmable digital inputMaximum inputVoltage 40 V DCCurrent 4 mAVoltage OFF0 - 5 V DCVoltage ON11 - 40 V DCNominal voltage24 V DCIsolation3.75 kV rmsWire1.5 mm2 / 16 AWGWire strip length6 mm / 0.23 inTorque0.5 Nm / 4.4 in•lbSpecificationsMetersiEM3135 / iEM3155 / iEM3165 / iEM3235/ iEM3255 / iEM3265 / iEM3335 / iEM3355 / iEM3365 / iEM3455 / iEM3465 / iEM3555 / iEM3565iEM3110 / iEM3210 / iEM3310iEM3115 / iEM3215iEM3135 / iEM3155 / iEM3165 / iEM3175 / iEM3235 / iEM3255 / iEM3265 / iEM3275 / iEM3335 / iEM3355 / iEM3365 / iEM3375/ iEM3455 / iEM3465 / iEM3555 / iEM3565iEM3115 / iEM3135 / iEM3155 / iEM3165 / iEM3175 / iEM3215 / iEM3235 / iEM3255 / iEM3265 / iEM3275 / iEM3335 / iEM3355 / iEM3365 / iEM3375/ iEM3455 / iEM3465 / iEM3555 / iEM3565119Specifications iEM3000 series user manualMechanical characteristics CharacteristicValueMetersFront panelIP40iEM31•• / iEM32•• / iEM33•• / iEM34•• / iEM35••