Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
4. Enter a value in "Amount per pulse A", e.g. 100, to define the volume/pulse with respect to the "Minimum amount per pulse value".4. Enter a value in "Amount per pulse A", e.g. 100, to define the volume/pulse with respect to the "Minimum amount per pulse value".5. Press "Apply Change of Pulse" to apply the settings.6. If needed, proceed with setting Pulse Output B according to application specific requirements (Default setting = Alarm).6. If needed, proceed with setting Pulse Output B according to application specific requirements (Default setting = Alarm).SITRANS FUS080 transmitter40Operating Instructions, 07/2012, A5E03059912-026.4.6Checking the operation readinessAll parameters are now set and defined according to the application.1. Select: "Device" → "Download to device..." to download the parameters to the device.NoteBefore downloading the parameters, check that all listed data are loaded or changed and in accordance with the application requirements.