Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
No, YesReadProtection of the linearization parameters. It is decided by the order number and cannot be changed.The linearization parameters are listed in the device menu 'Linearization'Forward, Reverse, NetR/WCalculation principle on flow direction for forward, reverse or net flowdd-mm- yyyy; hh:mm:ssReadDate and time when totalizer 1 function was changedForward, Reverse, NetR/WCalculation principle on flow direction for forward - reverse or net flowdd-mm- yyyy; hh:mm:ssReadDate and time when totalizer 2 function was changedSITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-02Modbus AddressParameterDefault valuePipe data40181300Number of pathsProduct dependent40149302 Max sample frequency 1540147303Sample frequency0.540167304Rn for path 1Product dependent40169305Rn for path 2Product dependent40171306Inner pipe diameterProduct dependentSITRANS FUS080 transmitter