Title: +038=522.000-NI002._05-Testrapport EKOM-anlegg, plan 03 (Uniconsult).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 36] [Pair 45]29.2 127 2 4.41Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]27.7 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-78 9.5 498.0 29.3 36 8.9 498.0 26.4SR 36-45 4.8 499.0 29.3 36 6.0 499.0 26.4MAIN 36-78 9.5 498.0 29.3 36 8.9 498.0 26.4SR 36-45 4.9 500.0 29.2 36 6.0 499.0 26.410080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 36-45 15.0 4.9 51.5 45 13.7 2.9 53.1SR 36-45 15.0 5.9 49.9 45 13.7 2.6 53.9MAIN 45-78 15.4 500.0 11.3 45 16.0 500.0 8.3