Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 45] [Pair 45]20.6 90 2 3.31Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 36] [Pair 36] [Pair 36]32.1 499.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 9.4 466.0 29.5 45 10.0 410.0 28.3SR 36-45 6.1 466.0 29.5 36 7.6 467.0 26.6MAIN 36-45 10.4 500.0 27.8 36 10.9 500.0 25.0SR 36-45 6.3 488.0 28.4 36 7.8 491.0 25.410080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 45-78 15.4 4.0 53.2 45 14.3 3.4 51.7SR 45-78 15.4 4.0 53.2 45 14.2 2.6 53.9MAIN 36-45 17.5 468.0 11.8 45 19.8 498.0 8.3