Title: +038=370.104-OE004T_01-Multical 801 Energimåler Kamstrup - Installasjons- og brukerguide.pdf
Carry out an operational check when the energy meter has been fully mounted. Open thermo-regulators and cocks in order to establish a water flow through the heating system. Activate the upper push button on the MULTICAL® 801 and check that the display values for temperature and water flow are reliable.Kamstrup A/S • 5512603_C1_GB_03.201613MULTICAL® 8017 Electrical connectionThe temperature sensors are mounted in the terminals of the calculator as shown above. Jumpers are used when mounting 2-wire sensors. In connection with flow sensors V1 and V2, the below-mentioned colours are used for connection of ULTRAFLOW® and electronic pick-up units. Flow sensors with Reed switch output must be connected to terminals 11-10 and 11-69, respectively.4-wire51 51A 52A 52 3 78 4Pt T3Pt T22-wire51 51A 52A 52 3 78 41155Pt T16 26 2+ SIGV1 11 9 10V2 11 9 69Blue Red YellowPt T3Pt T2Pt T1Terminal No.T11–5–6–2T2 3–7–8–4V111–9–10V2 11–9–69T3 51–51A– 52A–52