Title: +038=320.103-OE009T_02-Multical 801 Energimåler Kamstrup - Produktdatablad.pdf
StandardInfo code 0 1 8 4 32 64 256 512Response time - -Description No irregularities The supply voltage has been interrupted Temperature sensor T1 outside measuring range 1…10 min. Temperature sensor T2 outside measuring range 1…10 min. Temperature sensor T3 outside measuring range 1…10 min. 24 hours Leak in cold water system 24 hours Leak in heating system 120 sec. Burst in heating systemULTRAFLOW® X4 info (enabled when CCC=4XX)Info code 16 1024 2048 128 4096 8192 16384 32768Description Flow meter V1 communication error Flow meter V2 communication error Flow meter V1 wrong pulse value Flow meter V2 wrong pulse value Flow meter V1, signal too weak (air) Flow meter V2, signal too weak (air) Flow meter V1 wrong flow direction Flow meter V2 wrong flow direction