Title: +038=370.104-OE004T_01-Multical 801 Energimåler Kamstrup - Installasjons- og brukerguide.pdf
8.9 ZigBee® + pulse inputs,type 67-00-60198.10 Metasys N2 + pulse inputs,type 67-00-62208.11 SIOX module (Auto detectBaud rate), type 67-00-64/67-0M208.12 BACnet® + pulse inputs,type 67-00-66208.13 Modbus RS485 RTU Slave Module + pulse inputs, type 67-00-67218.14 High Power Radio Router + 2 pulse inputs (VA, VB), type 67-00-848.15 Module overview 8.16 Insertion of modules21 22 239Information codes ”INFO” 2410 Terminal Overview25561 General informationPlease read this guide before installing the energy meter. If the meter is installed incorrectly, Kamstrup’s warranty obligations will no longer apply.Please note that the following installation conditions must be obeyed:Pressure stage ULTRAFLOW®:PN16/PN25/PN40, see marking. Marking of flow part does not cover included accessories.Pressure stage Kamstrup sensor set type DS:Pressure stage Kamstrup sensor set type DS:PN16Pressure stage Kamstrup pocket sensors type PL:PN25/PN40 - depending on type