Title: 370.011 Siemens UC50 Kalkulator.pdf
The mounting place and the pulse value of flow sensors with pulse output must correspond to the in the calculator set values (see LOOP 2).Service loop 2 („LOOP 2“)In the service loop 2, the installation details are displayed.Head of the loopMounting place of the flow sensor cold side or hot sidePulse valueAt the heat meter or combined heat/cold meter the mounting place of the flow sensor cold side is equivalent to return flow. At the cold meter the mounting place of the flow sensor hot side is equivalent to return flow.Attention: Calculators with one-time adjustable pulse value and adaptable mounting place being characterized by:At calculators with one-time adjustable pulse value, the pulse value must be adjusted during commissioning in accordance with the flow sensor and the mounting place must be checked! As long as the pulse value has been set, the calculator cannot cumulate energy and volume. (See „Parameterization of pulse value and mounting place of the flow sensor“)Installation