Title: +038=310.101-RF022T_07-multical 62 Vannmåler Kamstrup - Produktdatablad.pdf
Open collector (OB) Optionally 32 msec. or 100 msec. 5-30 VDC 1-10 mA UCE ≈ 1 V at 10 mA 2 kV 25 mMechanical data Metrological class Environmental class Mechanical environment Electromagnetic environmental class Ambient temperature Protection class2 Fulfils OIML R49 class B MID class M1Fulfils OIML R49 class E1 5…55 °C, non-condensing, closed location (indoor installation) Calculator IP54 Flow part IP68Temperature of medium – Cold water meter – Cold water meter – Hot water meter Storage temperature Weight Pressure stage – Thread mounted meter – Flange mounted meter Flow meter cable0.1…30 °C (T30) 0.1…50 °C (T50) 0.1…90 °C (T90) -25...60 °C (drained flow part) 0.4 kgs excl. flow partPN16 PN25 2.5 m58101035_G1_GB_01.20185AccuracyMPE according to OIML R49 Meter approved T50 (0.1...50 °C) and T90 (0.1...90 °C) MPE (maximum permissible error)For 0.1 °C < t ≤ 30 °C ± 5 % in range Q1 ≤ Q < Q2 ± 2 % in range Q2 ≤ Q ≤ Q4 For 30 °C < t ≤ 90 °C ± 3 % in range Q2 ≤ Q ≤ Q4)%(