Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
MULTICAL® 80113 Calibration and verification13.1 High-resolution energy readingShould you need high-resolution energy reading during test and verification, it can be initialized as follows: -Switch off the supply voltage and remove the plug from the backup battery. Wait until the display is blankPress both pushbuttons at a time whilst connecting the supply voltage (or the plug of the backup battery) and keep pressing both buttons until the display becomes activeThe display now shows energy with 0.1 [Wh] resolution until one of the pushbuttons is activatedThe above display example showing 345.4 [Wh] corresponds to the amount of energy accumulated at inlet = 43.00°C and outlet = 40.00°C as well as a return volume of 0.1 m3.The high-resolution energy reading is displayed in Wh at a volume resolution of 0.01 m³ (qp 1,5 m³/h). For bigger meters the energy indication must be multiplied by 10 or 100.m3 0.001 0.01 0.1 1Wh x 0.1 x 1 x 10 x 100