Title: 370.011 Siemens Sitrans FUE080 Flowmåler.pdf
In wall mounting kit:4 x M20 polyamide cable glands (2 for mains and outputcables, 2 for the transducer coaxial cables)Ambient temperature5 °C …+60 °C (23 °F…140 °F)Storage temperature−40 °C…+85 °C (−40 °F…+185 °F)Transducer / sensor cablesCoaxial cable, impedance 75 Ohm, ∅ app. 6mm;Cable length: Max. 30 meter between transmitter and transducer (factory sets 15 or 30 m (49.21 or 98.43 ft)Mechanical vibration2 g, 1…800 Hz sinusoidal in all directions according to IEC 68-2-6Power supplyBattery:Replaceable 3.6 V LiSOCl (Lithium Thionyl Chloride)Double battery pack 32 Ah or single back-up battery 16 Ah;Battery change interval: 6 years at 60 °C (140 °F) operation(depending on used functions)Mains:87 … 265 V AC (50 … 60 Hz)DisplayLCD, 8 digits, additional 2 digits and symbols for statusinformationUnits:– Volume unit: m3 (default display unit) – Flow unit: m3/h (default display unit)Alarm codes: F 1 to F 9Push button