Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Reserved: These registers are prepared for later extensions of the functionality of the modules. Therefore, they have no actual designations yet.COUNTRY CODESInformation on country codes see 55 14-170MAINTENANCESee instructions no. 55 08-709 concerning update of programming and configuration.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201629MULTICAL® 8014 Dimensioned sketchesFront measurements of MULTICAL® 801Wallmounted MULTICAL® 801 seen from the side30Installation measurements of MULTICAL® 801Cable unions of MULTICAL® 801All measurements in [mm]Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8015 Installation5.1 Mounting in inlet or outlet pipeProg. numberAMULTICAL® 801 is programmed for flow meter mounted in either inlet orFlow sensor position - Inlet (at T1) - Outlet (at T2)k-factor table3 4outlet pipe. The table below indicates installation conditions for: ♦ Heat meters ♦ Cooling meters ♦ Heat/cooling meters