Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 80115 Approvals15.1 Type approvalsMULTICAL 801 has been type approved based on EN 1434-4:2007 and OIML R75:2002.The test report, project A530573, has been prepared by DELTA and forms the basis of the MID approval.MULTICAL 801 has a national Danish cooling approval, TS 27.02 006, according to BEK 1178 based on EN1434:2007.15.2 The Measuring Instrument DirectiveMULTICAL® 801 is supplied with marking according to MID (2004/22/EF). The certificates have the following numbers:B-module: DK-0200-MI004-009D-module: DK-0200-MID-D-001Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.2016117MULTICAL® 80116 TroubleshootingMULTICAL 801 has been constructed with a view to quick and simple installation as well as long and reliable operation at the consumer.Should, however, an operating problem with the meter occur, the table below can be used for troubleshooting.