Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
11.1.5 The module has connection possibility for two pressure transmitters on terminals 57, 58 and 59 and can be adjusted for current reading or pressure ranges of 6, 10 or 16 bar. The module is prepared for remote reading, data from meter/module being transferred to the system software via the connected external GSM/GPRS modem on terminals 62, 63 and 64. The module has two extra pulse inputs, which can only be used, however, if modules are mounted in module position 1, see paragraph 7.2: Pulse inputs VA and VB as to function. The module must be powered by 24 VAC.