Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
MULTICAL 801 currently calculates the average temperatures of inlet and outlet pipes (T1 and T2) in °C without decimals, and the background calculations E8 and E9 (m3 x T1 and m3 x T2) are carried out with every energy calculation (e.g. with every 0.01 m3 if the meter size is qp 1.5), whereas the display is updated every 24 hours. The average temperatures are thereby volume weighted and can, therefore, be used for check purposes directly.Type of registrationAverageYearly data Monthly dataYear-to-date average (since latest target date)Month-to-date average (since latest target date)Year-to-date average for T1(Current date with a stipulated line under year or month is shown immediately BEFORE this reading)6.8.3 Preprogrammed temperatures