Title: DOCA0005EN-12.pdf
0 = digital output disabled1 = for Alarm (digital output is associated with the overload alarm)BI05 - Alarm Status50 = Alarm is inactive1 = Alarm is activeBI06 - Unacknowledged status60 = historic alarm is acknowledged1 = historic alarm is unacknowledged116DOCA0005EN-12iEM3000 series user manualChapter 9 SpecificationsElectrical characteristicsPower system inputs: iEM31•• metersCharacteristicValueMeasured voltageWye: 100 - 277 V L-N, 173 - 480 V L-L ±20%Delta: 173 - 480 V L-L ±20%Maximum current63 AMeasured current0.5 A to 63 AOverload332 V L-N or 575 V L-LVoltage impedance3 MΩCurrent impedance< 0.3 mΩFrequency50 / 60 Hz ±10%Measurement categoryIIIMinimum wire temperature rating required 90 °C (194 °F)Burden< 10 VA at 63 AWire16 mm2 / 6 AWGWire strip length11 mm / 0.43 inTorque1.8 Nm / 15.9 in•lbWithstand10 A continuous, 20 A at 10 sec/hrPower system inputs: iEM33•• metersCharacteristicValueMeasured voltage