Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
is he measured temperature difference, e.g. ∆Θ = inlet temperature – outlet temperature. Please note that different temperatures are used for the calculation of ∆Θ as MULTICAL 801 can calculate various different energy types. Both in the display and during data reading each energy type is uniquely defined, e.g.Heat energy: E1 = V1(T1-T2)kCooling energy: E3 = V1 (T2-T1)kkis the heat coefficient of water which is calculated on the basis of the formula stated in EN 1434-1:2007 (identical with the energy formula of OIML R75-1:2002). For checking the measurement, Kamstrup can supply an energy calculator:Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201633MULTICAL® 8016.2 Application typesMULTICAL 801 operates with 9 different energy formulas, E1…E9, which are all calculated parallel with each integration, no matter how the meter is configured.Formula∆ΘExample of an applicationIncluded in Application No.Register type