Title: Manual - Varmemåler M3 - Løs fra rør.pdf
Temperature sensorsPt1000 (two-wired)Cable length3 mPulse inputs1 dedicated for the heating circuit flow meter2 dedicated for the cold/hot sanitary water volume meterPulse input classClass IA (default): Open Collector or electromagnetic contact (reed), max 5HzPulse input maximum frequency (MID approved)5 HzMaximum length pulse emitter cable3 mInstallation site of flowmeter (Flow in)Return flow pipe (Default), Inlet flow pipe (optional).Always refer to the indication on the cover of the calculator.Supported vector fluidWaterHydrosplit M3 Manuale utente – User manual - v1.6.5_ITA_ENG57FUNCTIONING BEYOND THE LIMITATIONS DECLAREDIt is recommended to ensure that the measurement conditions are within the exposed limits of certification. The unit doesn’t disable the functioning outside these ranges, its use is not covered by the certification if the measurement conditions don’t meet the certification validity conditions.