Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
For CCC codes with 8-digit counter (e.g. CCC=206) energy pulses (GJ) and volume pulses (m3) will be generated with every least significant digit but one.If a higher resolution of pulse outputs is required, a high-resolution CCC code must be selected.The pulse outputs are passive, optoisolated and tolerate 30 VDC and 10 mA. If active pulse outputs are required, the internal supply on terminals 97A-98A can be used.Passive pulse outputs connected via external supplyActive pulse outputs connected via internal supplyBy means of the PC-programm METERTOOL you can choose between 32, 100 and 247 ms in addition to the option of pulses for combined heat/cooling measurement (CE- and CV-).