Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Formulak-factorProg.Hot pipeCold pipeInstallation:k-factor for T1 inInlet tableA=3 (Flow sensor in inlet pipe)V1 and T1T2Heat meterE1=V1(T1-T2)kk-factor for T2 inOutlet tableA=4 (Flow sensor in outlet pipe)T1V1 and T2k-factor for T1 inOutlet tableA=3 (Flow sensor in inlet pipe)T2V1 and T1Cooling meterE3=V1(T2-T1)kk-factor for T2 inInlet tableA=4 (Flow sensor in outlet pipe)V1 and T2T1Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201631MULTICAL® 8015.2 EMC conditions MULTICAL® 801 has been designed and CE-marked according to EN 1434 Class A and Class C (corresponding to Electromagnetic environment: Class E1 and E2 of the Measuring Instruments Directive) and can thus be installed in both domestic and industrial environments.