Title: 370.011 OE501 Siemens Sitrans FUE080 Flowmåler.pdf
Lay heat-resistant transducer cables if high temperatures can occur on the housing, e.g. due to conduction of heat by the sensor/metering tube. Lay the cables so that they do not come into contact with the hot sensor/metering tube.Lay signal and transducer cables separately from cables with voltages > 60 V.To guarantee the IP67 (NEMA 4X/6) degree of protection, use cables with the required specificationsTo guarantee the IP67 (NEMA 4X/6) degree of protection, use cables with the required specificationsSITRANS FUS080 transmitterOperating Instructions, 07/2012, A5E03059912-0221Connecting 5.1 Wiring the transducer cablesNote Transmitter groundingTo ensure identical potential for sensor and transmitter, a direct grounding of transmitter and sensor is recommended.WARNINGProtective conductor terminalThe required cable is min. AGW16 or 1.5 Cu.WARNINGWire insulation