Title: +135.01=544.001-XS001T_01-Pasientsignal operating_and_installation_instruction_2011.pdf
DIN 18024-2 “Barrier-free buildings” further requires that operating elements for people in wheelchairs be installed at a height of 0.85 m.In WCs and wet rooms, the special provisions of DIN VDE 0100 must be observed. Only correspondingly suitable devices may be installed in these rooms. Observe the instructions of the manufacturer that specially designate the devices as suitable for installation in WCs and wet room areas.Pull switches, etc. in showers must be installed at least 20 cm above the highest possible position of the shower head, and the pull cord should not be farther than 20 cm away from the top edge of the floor to ensure that people lying on the floor can still reach the pull cord.In its standard design, the compact call set is intended for use in dry, indoor rooms. The range of applications of the corresponding components can be extended to damp rooms through use of the IPx4 seal inserts (Part No. 88160A).The devices are not suitable for installation in locations at risk to explosion.