Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Imp./lQp range [m³/h]Qs[m³/h]TypeFlow sensor20120220320420520610040400100201002359265898155898152359261179630235926022111011000x10 *)[tons] 11000x10 *)11100022211112.52.5105010010.40.40.10.020.0110…1007540…200240100…400500150…12001600500…300036001400…18000 36000FUS380 DN50-65 FUS380 DN80-100 FUS380 DN125 FUS380 DN150-250 FUS380 DN300-400 FUS380 DN500- 1200NNNNNNCurrent flow indication (l/h or m³/h) is calculated based on volume pulses/10 s (see paragraph 6.5)) Under this CCC code, the count will display the seven most significant digtes, followed by “0”Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201621MULTICAL® 8013.3.7 CCC-codes for vane-wheel meters with electronic pick-upNumber of decimals in displayCCC No.Pre- count erFlow factorkW hMWh GcalGJm³l/h m³/h kW MWImp./lqp [m³/h]TypeFlow sensor