Title: DOCA0005EN-12.pdf
FormatbinhexNo data0000x08-bit integer0001x116-bit integer0010x224-bit integer0011x332-bit integer0100x432-bit real0101x548-bit integer0110x664-bit integer0111x7Variable length1101xDData function types used by the meter (DIF bits 5-4)Function typebinInstantaneous00Maximum01Primary VIF used by the meterNOTE: E denotes the extension bit; x in the hex value is determined by bits 7-4 of the VIF.Primary VIFbinhexDescriptionEnergyE000 0011x3Wh with a resolution of 100 in int64 kWh with a resolution of 103 in float32PowerE000 1110xEkW with a resolution of 103Time pointE110 1101xDDate and time in data type F, as detailed in the M-Bus protocol documentationBus addressE111 1010xAData type C (unsigned integer), as detailed in the M-Bus protocol documentationPrimary VIFE1111 1101FDIndicates that the first VIFE is a primary VIF extensionManufacturer-specific VIFE1111 1111FF