Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 12] [Pair 45]16.1 70 1 2.70Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]33.8 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 7.8 451.0 30.2 36 8.0 472.0 26.3SR 36-45 9.5 465.0 29.5 36 8.8 465.0 26.7MAIN 36-45 7.9 472.0 29.2 36 8.7 495.0 25.2SR 36-78 9.7 480.0 28.8 36 9.9 495.0 25.210080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 45-36 15.5 3.8 53.8 45 14.3 2.9 53.1SR 36-45 15.5 4.3 52.7 45 14.4 3.5 51.4MAIN 36-45 17.0 470.0 11.8 45 19.4 470.0 8.8