Title: 370.011 OE501 Siemens UC50 Kalkulator.pdf
Display on LCD: Consider the appropriate instructions manual (attached to the meter/module)!Tariff control (optional)22/2913 Tariff control (optional)The following options are available for tariff control.The summation of the quantity of energy and volume in the standard registers is performed independently of the tariff situation!Threshold value tariff (tariffs T2, 3, 4, 5, 6)Tariff registers 1, 2 and 3 can be controlled via up to 3 threshold values.Either the quantity of energy or the volume can be summated in the tariff registers.The threshold values can be derived from the flow rate (tariff T2), the power (tariff T3), the return temperature (tariff T4), the flow temperature (tariff T5) or the temperature difference (tariff T6).Supplied quantity of energy (tariff T7)In tariff register 1, a quantity of energy is summated that is calculated from the flow temperature (instead of from the temperature difference).Returned quantity of energy (tariff T8)