Title: 40 Bender Datablad.pdf.pdf
RCMS460-D Device version RCMS460-D utilises a backlit graphical display. This version is applied when detailed information about all devices in the switchboard cabinet, connected to the bus, are to be displayed locally. This device is capable of assigning parameters to all RCMS devices connected to the BMS bus and displaying all measurement details. Several RCMS-D devices can be used in one system.Overview of device typesDistinctive device featuresParameter setting function Master/Slave Address range Measuring channels per device W…, WR…, WS…, W…AB, W…F series measuring current transformers CT monitoringMeasuring circuitRated residual operating current I∆n2 (Alarm)AC/DC sensitive 0…2000 Hz (Type B) pulsed DC sensitive 42…2000 Hz (Type A) pulsed DC sensitive 42…2000 Hz (Type A) for the channels 9…12 (RCMS4x0-D4/-L4)Switching elementsTime response