Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Having entered verification data, the program automatically calculates the true k-factor in accordance with the formula of EN 1434 and OIML R75:2002.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.2016113MULTICAL® 801Verification14.3.5 The verification program menu is opened by activating ”Verification”.Activate ”Start verification” in order to begin test/verification.When the test has been completed, the result will be displayed. If the result can be approved, click on “Save”. The result is now saved in the database under the serial number of the calculator. You can save several results under one serial number without overwriting earlier results.Certificate14.3.6 If you want to print a certificate with saved results, select “Certificate”. The test/verification result can, subsequently, be found according to serial number, and the certificate can be printed.114Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 80114.4 LogView HCW