Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
q = (88 x 235926)/65535 = 316.8, which is displayed as 316 [l/h]Current water flow in V1Slow volume pulses (CCC = 0XX)The current water flow of slow volume pulses (typically from flow meters with reed contact) is calculated without average determination as a scaling factor divided by the duration between two volume pulses.q = flow factor/(256 x period of time in s) [l/h] or [m3/h]Example: - Mechanical flow meter Qn 15 qp m3/h with 25 l/imp. (CCC=021), flow factor = 230400 -Current water flow = 2.5 m3/h, which corresponds to 36 s of the duration between 2 pulsesq = 230400 /(256 x 36) = 25 which is displayed as 2.5 [l/h]V1 and V2 must be the same type (either quick (CCC > 100) or slow (CCC=0XX)) but can have different qp-codings (CCC).The actual flow rate on the display will be shown a ”0”, when the period between pulses exceed 15 min.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201641MULTICAL® 8016.6 Power measurement, V1