Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Meter type Heat meter, closed systems (MID) Heat meter, closed systems Cooling meter Heat/cooling meter Volume meter, hot water Volume meter, cooling water Energy meter, open systems Delivery code (language on label etc.)2 4 5 6 7 8 9If MULTICAL® 801 has been supplied as a combined heat/cooling meter, heat energy (E1) is measured at positive temperature difference (T1 > T2) whereas cooling energy (E3) is measured at negative temperature difference (T2 > T1). Temperature sensor T1 (with a red type sign) must be installed in the hydraulic inlet pipe whereas T2 is installed in the outlet pipe.The temperature point “T1 limit” is used as a ”filter” for cooling measurement in the way that cooling is only measured when the current inlet temperature T1 is below T1 limit.T1 limit is configurable in the temperature range 0.01…180.00°C. T1 limit is configured via METERTOOL.