Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
When the info logger is read in the display the latest 36 changes including dates can be read.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201659MULTICAL® 8016.13 Leak surveillance6.13.1 District heating systemThe leak surveillance system is primarily used for direct connected district heating systems, i.e. systems without exchangers between the district heating network and the heating system of the house. The surveillance equipment consists of two ultrasonically based water meters placed in inlet and outlet pipe respectively as well as temperature sensors in both pipes. Furthermore, the electronic unit of MULTICAL 801, which calculates the heat energy, also monitors the mass difference (temperature corrected volume) which can be found between inlet and outlet pipe.Main tapTap water meter with pulse outputCold water- connectionTapMULTICAL heat meter with remote reading(e.g. integral radio module)District heating connectionShut-off valves