Title: +135.01=370.004-OE401T_02-320 Datablad Energimåler UH50.._.pdf
In Germany, the temperature sensors must be installed submerged for meters to qp 6 and to PN16. Immersion sleeves only have to be used in case of higher stress levels.In any case, the end of the sensors’ probe must extend to at least the pipe center. Temperature sensors and fittings must be sealed to prevent tampering. Overpressure must prevent cavitation across the entire measuring range, that is, at least 1 bar with qp and approximately 3 bar with qs (at 80 °C). Only authorized personnel are allowed to remove and replace calibration seals when carrying out service work.CE2N5324en 2018-08-18Building TechnologiesBall valveExample of mounting with a ball valve (recommended up to qp 6)Welded sleeve with  protection pocketExample of mounting with protection pockets (recommended above qp 10)