Title: Data tester Smestad skole - Detaljert.pdf
Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 36,6 68,0 190,0 10,0 6,4Limit 100,0 - 555,0 50,0 25,0Margin 63,4 - 365,0 40,0 18,6Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36 0,7 3,1 2,4 3,10Worst Value: Local: 45 15,6 43,7 28,1 499,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 15,7 9,2 6,5 305,00Remote: 12 22,2 16,9 5,3 51,75Worst Value: Local: 12 15,7 9,2 6,5 305,00Remote: 45 17,8 8,0 9,8 490,00dB -2020600Local250500 MHzdB -2020600Remote250500 MHzNEXT: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36-45 29,4 26,7 2,7 499,00