Title: Kapasitetsmåler Mecon.pdf
Operating Instructions FO Turbo-Lux 3 FM VdS05/2015 OI_FO_Turbo-Lux 3 FM VdSInstallation and mode of operationMounting the bypass meterThe bypass meter is only compatible with the delivered orifice plate. Before loosening the cap (Fig.6/7, pos. 10), the pipeline must be emptied to prevent the escape of liquids. The meter is pluggedand screwed with a nut (Fig. 6, pos. 9). It must always be mounted vertically so that the float (Fig.6, pos. 4) can move freely in the tube. About foreign bodies that have come behind the filter mustbe removed. The tightening of the nut or the cap should be done by hand as possible. The threadsmust - for example be slippery - by fat. To avoid air strikes, the tube should be slowly filled withwater.Fig. 3 Examples of installation2.3 Mode of operationThe orifice plate flowmeter FO Turbo-Lux 3 consists of an orifice plate (Fig. 6/7, pos 1) forstationary installation and a portable bypass meter (Fig. 6, pos. 2). The bypass meter contains a