Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Electric couplingInlet and outlet pipes are closely electrically coupled • No welded joints occurIn installations where the electric coupling cannot be carried out, or where welding in the pipe system can occur, the cable from one ULTRAFLOW must be routed through a Pulse Transmitter with galvanic separation before the cable enters MULTICAL 801.Inlet and outlet pipes are not necessarily closely coupled • Electric welding *) can occur) Electric welding must always be carried out with the earth pole closest to the welding point. Our factory guarantee does not comprise damage to meters due to welding.Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201639MULTICAL® 8016.4 Combined heat/cooling meteringMULTICAL® 801 is available as heat meter (meter type 2xx), cooling meter (meter type 5xx) or combined heat/cooling meter (meter type 6xx).