Title: +175=574.001_OE001_Manual - Varmemåler M3 - Løs fra rør.Produktinformasjon.pdf
Electrical supply (optional, 3.6 ÷ 5 VDC, 300 mA): in this case, the battery will be used as backup power supply) Li-SoCl2 Lithium-thionyl chloride, 3,6V “size D” 20AhBattery Life10 years +1Operating temperature range5 ÷ 55°CStorage temperature range10 ÷ 55°CDimensions154 x 119 x 51mmIP protection gradeIP52Approval2004/22/CE MID (Module B)Certificate nr.IT-MI004-xxxxEN1434Environmental classA (E1, M1)Temperature measurement range, MID approved (heating applications)Θ: 5÷180°CTemperature difference range, MID approved (heating applications)ΔΘ: 3÷150KTemperature measurement range (cooling)Θ: 2÷24°CTemperature difference range (cooling)ΔΘ: 3÷20KHydrosplit M3 Manuale utente – User manual - v1.6.5_ITA_ENG56Operative counting conditionsHeating: ΔΘ≥1K and fluid temperature ≥5°C (accounting enabling conditions)Cooling: ΔΘ≤0.2K and fluid temperature ≤24°CMaximum measurable power650 kWMaximum measurable flowrate2000 m3\hDisplayLCD, 8 digits + icons