Title: Kapasitetsmåler Mecon.pdf
and heat. Keep away from direct Avoid external load to the device. The storage temperature range is – 20… + 60 °C.Operating instructions FO Turbo-Lux 3 FM VdSwww.mecon.dePage 7 / 1605/2015 OI_FO_Turbo-Lux 3 FM VdSStart-up3 Start-upRead the exact value when a consistent flow has been attained and the float has reached a stableposition. The pipeline must always be filled.Read the value at the greatest diameter of the float:Fig. 4 FloatFor the bypass meter it is important that when starting up the pump, the shut-off-/control valve ofthe bypass orifice is opened min. 30% to avoid hydraulik shocks or pressure surges that coulddamage the bypass meter.Fig. 5 Rotation bypass meterWhen the bypass meter is commissioned or set into operation, bubbles of air will initiallyaccumulate at the top part, which must be removed. For this purpose, the union nut (Fig. 6, pos.9) must be somewhat loosened during operation and the device must be rotated by 360°, so that