Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 36] [Pair 45]30.9 134 2 4.70Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]27.4 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 10.3 497.0 29.3 36 9.3 494.0 26.5SR 36-45 6.1 478.0 29.7 45 7.4 470.0 27.0MAIN 36-45 10.3 497.0 29.3 36 9.4 500.0 26.4SR 36-45 6.1 478.0 29.7 45 7.5 494.0 26.510080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 78-45 16.0 483.0 11.6 45 14.7 2.5 54.3SR 78-45 16.0 488.0 11.5 45 14.7 2.6 53.9MAIN 78-45 16.3 500.0 11.3 45 16.7 488.0 8.5