Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 12] [Pair 45] [Pair 45] [Pair 45]30.2 131 2 4.63Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]27.9 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 12-45 9.1 480.0 29.6 36 8.8 471.0 27.0SR 36-45 8.3 497.0 29.3 45 8.3 485.0 26.7MAIN 12-45 9.1 480.0 29.6 36 8.9 487.0 26.6SR 36-45 8.3 497.0 29.3 45 8.5 496.0 26.510080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 45-78 16.5 469.0 11.8 45 15.5 11.0 41.4SR 78-45 16.5 500.0 11.3 45 15.5 15.3 38.6MAIN 45-78 16.7 500.0 11.3 78 17.8 500.0 8.3