Title: +038=310.101-MO039._02-F-INNV med spyling, pumper, sondestyring.pdf
30 meterLading for Charge for blybatteri:Maks 80mA, 13.7 VDCABS Norge AS, Industriveien 40, Postboks 473, 1302 Sandvika, Tel +47 67 55 47 00, Fax +47 67 55 47 50, www.absgroup.no. s t n e m p o l e v e d l a c n h c e t o t e u d s n o i t a c fi c e p siir e t l a o ts t h g i r e h t e v r e s e r e W| 9 2 - 1 0 - 9 0 0 2 O N 2 1 2 P C / 2 1 1 P Cl e n a p l o r t n o c S B AABS hydrostatic level sensor MD 126Features: * Dry ceramic sensor * Accuracy ≤ 0.3 % * Cable according Bg V V-1.12.96 specificationsTechnical specificationsDescriptionMaterialCablePElength 5m + measuring rangeHousingStainless steel 1.4404SensorOutput signalCeramic AL203, gold coated 4...20 mA, 2-wire, fixed to thesensors measuring rangeAccuracy≤ ± 0.3 % FS @ 25 °CMax overload8 barResponse time200 msSupply voltage9...30 V DCAmbient temperature25...80 °CTemperature coeff.≤ ± 0.015 % FS/K (Zero)Long tem stability≤ ± 0.3 % FS p.a.ProtectionIP 68