Title: Data tester Smestad skole - Detaljert.pdf
Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 46,9 68,0 244,0 14,0 8,0Limit 100,0 - 555,0 50,0 25,0Margin 53,1 - 311,0 36,0 17,0Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36 0,9 3,0 2,1 2,80Worst Value: Local: 45 20,4 43,5 23,1 496,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 78 16,5 8,9 7,6 324,00Remote: 78 20,2 14,6 5,6 86,75Worst Value: Local: 78 16,5 8,9 7,6 324,00Remote: 78 16,5 8,0 8,5 419,00dB -2020600Local250500 MHzdB -2020600Remote250500 MHzNEXT: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36-45 29,7 26,7 3,0 500,00