Title: DOCA0005EN-12.pdf
Pulse constant (pulses/unit; applicable when the digital input is used for input metering)Digital output setupNOTE: E denotes the extension bit; the hex value assumes E = 0.SND_UD codeData formatManufacturer- specific VIFERange/optionsDescriptionbinhex0002E001 10101A2, 3, 0xFFFFDigital output control mode 2 = Alarm 3 = Energy (energy pulsing) 0xFFFF = Disable0005E010 11102EiEM3135 / iEM3335: 1, 10, 20, 100, 200, 1000iEM3235: 0.01, 0.1, 1, 10, 100, 500Pulse constantinformationNOTE: This only applies if the digital output control mode is set to for Pulse.Pulse width in ms0002E010 11002C50, 100, 200, 300informationNOTE: This only applies if the digital output control mode is set to for Pulse.101102Communications via M-BusiEM3000 series user manualOverload alarm setup and acknowledgmentUse the information in the table below to configure the overload alarm.NOTE: E denotes the extension bit; the hex value assumes E = 0.SND_UD code