Title: Pos. 12A Kjolebenker ProffG2_Bruksanvisning.pdf
format 32 x 74 x50 mm format is a single stage temperature controller suitable for blinking). Consider the maximum current which can be applied to each relay (see Technical Data). then turned off when the temperature NOTE2: the set value is stored even when the procedure is exited by waiting the time-out to expire. Push the SET key for more than 2 seconds to change the Set point value; 1. The hidden menu includes all the parameters of the instrument. Fit the probe where it is not accessible by the End User. The instrument must not be opened. HOW TO CHANGE THE SETPOINT provided with 1 NTC probe input and one configurable digital input. The instrument is fully configurable To change the Set value push the o or n arrows within 10s. 3. keys for 3s (“°C” or “°F” LED starts 1. Enter the Programming mode by pressing the SET+ address) with a detailed description of the fault. HOW TO MOVE A PARAMETER FROM THE HIDDEN MENU TO THE FIRST The regulation is performed according to the temperature measured by probe. The