Title: +038=320.103-OE009T_02-Multical 801 Energimåler Kamstrup - Produktdatablad.pdf
T4 N/A 0.01…180.00 °C N/A 0.01…180.00 °CMax. cable lengthPt100, 2-wire 2 x 0.25 mm²: 2.5 m 2 x 0.50 mm²: 5 mPt500, 2-wire 2 x 0.25 mm²: 10 m 2 x 0.50 mm²: 20 mPt500, 4-wire 4 x 0.25 mm²: 100 m -Flow measuring V1 and V2 EN 1434 pulse class Pulse input Pulse ON Pulse OFF Pulse frequency Integration frequency Electrical isolation Max. cable lengthULTRAFLOW® V1: 9-10-11 and V2: 9-69-11 IC 220 kΩ pull-up to 3.6 V < 0.4 V for > 0.5 msec. > 2.5 V for > 10 msec. < 128 Hz < 1 Hz No 10 mReed switches V1: 10-11 and V2: 69-11 IB 220 kΩ pull-up to 3.6 V < 0.4 V for > 50 msec. > 2.5 V for > 50 msec. < 1 Hz < 1 Hz No 25 m24 V active pulses V1: 10B-11B and V2: 69B-79B (IA) 12 mA at 24 V < 4 V for > 0.3 msec. > 12 V for > 10 msec. < 128 Hz < 1 Hz 2 kV 100 mPulse inputs VA and VB VA: 65-66 and VB: 67-68 Pulse input Pulse ON Pulse OFF Pulse frequency Electrical isolation Max. cable length Requirements to external contact