Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
14.4.1 Introduction and installation Regarding ”Introduction”, ”Interface” and ”Installation” see paragraph 14.1 Introduction METERTOOL HCW since it is similar for LogView HCW.14.4.2 General information ”LogView HCW” (6699-725) is used for read-out of logging data from MULTICAL 801 calculator and modules (e.g. ”Prog. data logger + RTC + 4…20 mA inputs + pulse inputs” (67-00-22)) as well as for carrying out interval logging. The read out data can be used for analysis and diagnostic test of the heating installation. Data can be presented as table or graphics. Tables can be exported to ”Windows Office Excel”.For available logging data see paragraph 6.12 Data loggers.14.4.3 ”Log” Select the required data function.Data Logger ”Internal KMP-Logger” makes it possible to read data from the ”Programmable KMP-Logger”, which saves data in the calculator.