Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Example 2 The average temperatures must be measured together with the yearly reading. Therefore, E8 and E9 are included in the yearly reading.Date of readingVolumeE8Average of inlet pipeE9Average of outlet pipe2003.06.01534.26 m3 48236186542002.06.01236.87 m3 201237651Yearly consumption297.39 m3 2811328113/297.39 = 94.53°C1100311003/297.39 = 36.99°CTable 138Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 8016.3 Calculator with two flow sensorsMULTICAL 801 can be used in various applications with two flow sensors, e.g. leak surveillance or open systems. When two ULTRAFLOW are direct connected to one MULTICAL 801, a close electric coupling between the two pipes ought to be carried out as a main rule. If the two pipes are installed in a heat exchanger, close to the flow sensors, however, the heat exchanger will provide the necessary electric coupling.Electric coupling