Title: Data tester Smestad skole - Detaljert.pdf
Unspecified-Building Unspecified-Floor Unspecified-Room Unspecified-Rack Unspecified-PanelOverall Result:Local Ser. No.: Local Adapter: Local Calibration Date: Device Software:pw20201689 Cat 6A Link Apr 17 2013 7.4Remote Ser. No.: Remote Adapter: Remote Calibration Date: Reporting Software:pw20201690 Cat 6A Link Apr 17 2013 Build_#972_7.4_2018-04-02_10-26-38Wiremap: PassLength (m): Cable NVP: Propagation Delay (ns): Delay Skew (ns): Resistance (Ohms):Value 47,0 68,0 244,0 14,0 7,9Limit 100,0 - 555,0 50,0 25,0Margin 53,0 - 311,0 36,0 17,1Wiremap: T568BInsertion Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 36 0,9 3,0 2,1 2,80Worst Value: Local: 45 20,5 43,8 23,3 500,00dB -2020600Local250500 MHzReturn Loss: PassPair: Value (dB): Limit (dB): Margin (dB): Frequency (MHz):Worst Margin: Local: 12 24,7 17,1 7,6 48,75Remote: 12 15,5 12,4 3,1 143,50Worst Value: Local: 78 16,2 8,0 8,2 484,00