Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 36] [Pair 45]26.0 113 2 4.02Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]29.8 498.0 41.5Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-78 7.9 490.0 28.3 36 8.3 490.0 25.5SR 36-45 8.8 465.0 29.5 36 9.2 461.0 26.9MAIN 36-78 8.1 494.0 28.1 36 8.4 494.0 25.3SR 36-45 9.7 492.0 28.2 36 10.1 498.0 25.110080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 36-45 15.9 4.3 52.7 45 14.4 3.0 52.7SR 45-36 15.9 4.3 52.7 45 14.3 2.6 53.9MAIN 78-45 16.4 500.0 11.3 45 17.5 500.0 8.3