Title: +038=522.000-NI002._05-Testrapport EKOM-anlegg, plan 03 (Uniconsult).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 36] [Pair 36] [Pair 45]35.3 153 2 5.34Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 12] [Pair 12] [Pair 12]25.3 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 12-45 7.6 467.0 29.9 45 8.3 465.0 27.1SR 36-45 6.0 456.0 30.1 36 7.4 475.0 26.9MAIN 12-45 8.2 500.0 29.2 45 8.5 500.0 26.4SR 36-45 6.7 500.0 29.2 36 7.8 499.0 26.410080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 45-36 14.3 469.0 11.8 45 13.9 2.6 53.9SR 36-45 14.1 469.0 11.8 45 13.8 2.6 53.9MAIN 78-45 14.5 500.0 11.3 45 14.8 500.0 8.3