Title: Kapasitetsmåler Mecon.pdf
Fig. 1 Inlet path and outflow zoneFor some flow ranges a special version for shortened inlet path of 5 x D and shortened outflowzone of min. 2 x D is available (please see page 12). Before installation following conditions have tobe observed:A straight inlet path of 5 X D is required. A longer inlet path is not allowed.The inlet path has to be connected directly to a one quarter pipe.Fig. 2 Shortened inlet path and outflow zoneThe installation can be in any line routing - horizontal to vertical - place (Fig. 3). However, it isimportant to ensure that the flow direction of the arrow marked on the device and corresponds tothe differential pressure sampling tube (Fig. 6/7, pos. 7) is in the horizontal position. Forattachment of the bypass meter, sufficient clearance must be provided. Important for thecompliance of the measuring tolerance is the central mounting of the pipeline. The center offsetmust not exceed 0.5 mm.Page 6 / 16www.mecon.de