Title: +038=522.000-NI002._05-Testrapport EKOM-anlegg, plan 03 (Uniconsult).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 78] [Pair 45] [Pair 12] [Pair 45]4.2 19 1 1.08Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 45] [Pair 45] [Pair 45]37.6 498.0 41.5Worst Case Margin Worst Case ValueN/A Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 5.1 474.0 29.1 36 4.9 475.0 26.2SR 36-45 4.6 474.0 29.1 36 5.1 475.0 26.2MAIN 36-45 5.1 474.0 29.1 36 4.9 475.0 26.2SR 36-45 4.7 475.0 29.0 36 5.1 475.0 26.210080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 45-78 15.0 3.5 54.4 45 14.3 2.6 53.9SR 45-78 15.0 3.4 54.7 45 14.3 2.5 54.3MAIN 45-36 16.3 488.0 11.5 45 15.4 497.0 8.3