Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Application no. 7Open system with 2 flow metersForwarded energy: E4 = V1 (T1-T3)k T1:InletReturned energy: E5 = V2 (T2-T3)k T2:Inlet(∆E = E4-E5 cannot be calculated by MULTICAL® 801)Heat energy: E2 = V2 (T1-T2)k T2:OutletT3 is measured or programmedMass: M1 = V1 (Kmass t1) Mass: M2 = V2 (Kmass t2)Application no. 8Hot water boiler with circulationTotal consumption E1 = V1(T1-T2)k T2:OutletCirculated consumption: E7 = V2(T1-T3)k T3:OutletKamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 801Application no. 92 cooling circuits with joint inlet pipeCooling energy #1: E4 = V1 (T1-T3)k T1:InletCooling energy #2: E5 = V2 (T2-T3)k T2:InletApplication no. 10Two-stage boiler system with 1 flow meterBoiler energy „B“: E3 = V1 (T2-T1)k T1:OutletBoiler energy „A“: E4 = V1(T1-T3)k T1:InletM2 = V2 (Kmass t3)* only with delivery codes (930…939)!Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201637MULTICAL® 801E8 and E9