Title: +038=320.103-OE009T_02-Multical 801 Energimåler Kamstrup - Produktdatablad.pdf
Water meter connection FF(VA) and GG(VB) = 01…40 680 kΩ pull-up to 3.6 V < 0.4 V for > 30 msec. > 2.5 V for > 30 msec. < 1 Hz No 25 m Leakage current at function open < 1 µAElectricity meter connection FF(VA) and GG(VB) = 50…60 680 kΩ pull-up to 3.6 V < 0.4 V for > 30 msec. > 2.5 V for > 30 msec. < 3 Hz No 25 mPulse outputs CE and CV Energy (16-17) Volume (18-19) Type Pulse lengthExternal voltage Current Residual voltage Electrical isolation Max. cable lengthOpen collector (OB) Programmable 32 msec., 100 msec. or 247 msec. via METERTOOL HCW 5…30 VDC 1…10 mA UCE ≈ 1 V at 10 mA 2 kV 25 mKamstrup A/S • 5810625_K1_GB_05.201611MULTICAL® 801Mechanical dataEnvironmental class Ambient temperature Protection class Storage temperature Weight Connection cablesMeets EN 1434 Class A and C 5…55 °C non condensing, closed location (indoor installation) IP67 -20…60 °C (drained flow meter) 1.4 kg excluding sensors and flow meter 6 pcs ø3…6 mm and 3 pcs ø4…8 mmMaterials