Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
The CCC-code states the calculator’s adaption to a specific flow sensor type to the effect that calculating speed and display resolution are optimized for the selected flow sensor at the same time as type approval regulations about minimum resolution and maximum register overflow are obeyed. We have divided the CCC-codes into smaller tables in order to obtain a faster overview.CCC(V1) states the CCC-code of the flow sensor connected to flow sensor input V1 on terminals 9-10-11 (or 10B- 11B). In most applications, this is the flow sensor used for energy calculation.CCC(V2) states the CCC-code of a possible extra flow sensor, which can be connected on terminals 9-69-11 (or 69B- 79B). If V2 is not used, CCC(V2) is equal to CCC(V1). For leak surveillance CCC(V2) must be equal to CCC(V1).14Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016Prog. numberAFlow meter positionk-factor tableInlet (at T1) - Outlet (at T2)3 4Measuring unit, Energy - x10 GJ - GJ - kWh - MWh - Gcal