Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Tap water energy: E6 = V2 (T3-T4)k T3:InletT3 is measured or programmed T4 is programmedFlow meter V1 is placed in inlet or outlet as selected during PROG.Mass: M1 = V1 (Kmass t1) or Mass: M1 = V1 (Kmass t2) depending on inlet/outlet programming.Mass: M2 = V2 (Kmass t3)*Application no. 42 heating circuits with joint inlet pipeHeat energy #1: E1 = V1(T1-T2)k T2:OutletHeat energy #2: E7 = V2(T1-T3)k T3:OutletT3 is measured or programmed Mass: M1 = V1 (Kmass t1) Mass: M2 = V2 (Kmass t3)*Application no. 5Open system with tapping from outlet pipeHeat energy: E1 = V1(T1-T2)k T1:InletTap water energy: E5 = V2 (T2-T3)k T2:InletT3 is measured or programmedMass: M1 = V1 (Kmass t1) Mass: M2 = V2 (Kmass t2)35MULTICAL® 80136Application no. 6Open system with separate flow meter for tappingHeat energy: E1 = V1(T1-T2)k T2:OutletTap water energy: E6 = V2 (T3-T4)k T3:InletT3 is measured or programmed T4 is programmedMass: M1 = V1 (Kmass t1) Mass: M2 = V2 (Kmass t3)*