Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 12] [Pair 45] [Pair 36] [Pair 45]28.3 122 1 4.34Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 45] [Pair 45] [Pair 45]28.4 497.0 41.5Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 8.9 482.0 29.6 45 9.5 482.0 26.7SR 36-78 7.7 498.0 29.3 36 8.2 451.0 27.4MAIN 36-45 8.9 482.0 29.6 36 10.3 498.0 25.1SR 36-78 7.7 498.0 29.3 36 8.7 498.0 25.110080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 36-45 14.4 3.5 54.4 45 14.0 2.9 53.1SR 45-36 14.4 3.5 54.4 45 14.0 2.6 53.9MAIN 45-78 17.0 500.0 11.3 45 17.7 500.0 8.3