Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Max. cable length 25 m Requirements to ext. contact Leak current at function open < 1 µAPulse outputs CE and CV Energy (16-17) Volume (18-19)TypeOpen collector (OB)Pulse durationProgrammable 32, 100 or 247 ms via METERTOOLExternal voltage5…30 VDCCurrent1…10 mAResidual stressU CE ≈ 1 V at 10 mAElectrical isolation2 kVMax. cable length25 m2.3 Mechanical data Environmental classFulfils EN 1434 class A and CAmbient temperature5…55 °C non-condensing, closed location (installation indoors)Protection classIP67Storage temperature20…60 °C (drained flow meter)Weight1.4 kgs excl. sensors and flow meterCable adapters6 pcs. D 3…6 mm and 3 pcs. D 4…8 mm2.4 Material Top coverPCConnection basePC + 10 %GFSealing cover, topABSSealing cover, bottomPCPrism behind displayPMMA10Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_20162.5 AccuracyFigure 1MULTICAL® 801 typical accuracy compared to EN 1434.