Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Max. 50 mA/230 VACMax. 450 mA/24 VAC3.65 VDC, 2 batteries A-cell lithium(Type No. 66-99-619)10 years’ normal operation (with mains supply)1 year (without supply)The replacement interval is reduced at high ambient temperatureFulfils EN 1434 class A and C (MID class E1 and E2)Kamstrup A/S · Technical Description · 5512-571_R1_GB_11_2016MULTICAL® 801Temperature measurementT1T2T3T4Type 67-F and 67-K 4-W Pt100Measuring range 0.00…185.00 °C 0.00…185.00 °C 0.00…185.00 °CN/APreset range0.01…180.00 °C 0.01…180.00°C 0.01…180.00 °C 0.01…180.00 °CType 67-G and 67-L 4-W Pt100Measuring range 0.00…185.00 °C 0.00…185.00 °C 0.00…185.00 °CN/APreset range0.01…180.00 °C 0.01…180.00 °C 0.01…180.00 °C 0.01…180.00 °CMax. Cable lengthsPt100, 2-wirePt500, 2-wirePt500, 4-wire(Max Ø6mm cable)2 x 0.25 mm2: 2.5 m 2 x 0.50 mm2: 5 m 2 x 1,00 mm2: 10 m2 x 0.25 mm2: 10 m 2 x 0.50 mm2: 20 m4 x 0.25 mm2: 100 mKamstrup A/S · Technical Description · 5512-571_R1_GB _11.20169