Title: +038=320.103-OE009T_02-Multical 801 Energimåler Kamstrup - Produktdatablad.pdf
KMP protocol with CRC16 used for optical communication and for base modules < 10 µW RMS+15/-30%, 50/60 Hz (all types) ±50%, 50/60 Hz (Type 67-F/G without analogue outputs) ±25%, 50/60 Hz (Type 67-K/L with analogue outputs) 4 kV < 3 W without analogue outputs < 9 W with analogue outputs Maks. 50 mA/230 VAC Maks. 450 mA/24 VAC 3.65 VDC, 2 pcs A-cell lithium (Type No. 6699-619) 10 years at normal operation (with supply) 1 year (without supply) The replacement interval is reduced at high ambient temperature Meets EN 1434 Class A and C (MID Class E1 and E2)0…20 mA or 4…20 mA 0…12.5 VDC 0…500 Ohm 24 mA 0.15 %Kamstrup A/S • 5810625_K1_GB_05.2016MULTICAL® 801Electrical dataTemperature measurement T1 67-F and 67-K 4-W Pt100 67-G and 67-L 4-W Pt500T2Measuring range 0.00…185.00 °C Preset range 0.01…180.00 °C Measuring range 0.00…185.00 °C 0.01…180.00 °C Preset rangeT3 0.00…185.00 °C 0.01…180.00 °C 0.00…185.00 °C 0.01…180.00 °CT4 0.00…185.00 °C 0.01…180.00 °C 0.00…185.00 °C 0.01…180.00 °C