Title: +038=522.000-NI002._04-Testrapport EKOM-anlegg, plan 11 (Pareto).pdf
Software Version: V6.1 Build 3 Limits Version: V7.1 Calibration Date: Main (Module): 19-07-2017 Remote (Module): 19-07-2017Length (m), Limit 90.0 Prop. Delay (ns), Limit 496 Delay Skew (ns), Limit 43 Resistance (ohms), Limit 20.60[Pair 12] [Pair 45] [Pair 36] [Pair 45]11.7 51 1 2.05Wire Map (T568B) PASSInsertion Loss Margin (dB) Frequency (MHz) Limit (dB)[Pair 45] [Pair 45] [Pair 45]35.5 500.0 41.6Worst Case Margin Worst Case ValuePASS Worst Pair NEXT (dB) Freq. (MHz) Limit (dB) Worst Pair PS NEXT (dB) Freq. (MHz) Limit (dB)MAIN 36-45 9.2 447.0 30.3 36 9.4 465.0 26.7SR 36-45 8.8 288.0 34.3 45 8.8 288.0 31.7MAIN 12-36 11.0 494.0 28.1 36 10.2 496.0 25.2SR 36-45 9.4 466.0 29.5 36 10.6 496.0 25.210080604020NEXT (dB)PASS Worst Pair ACR-F (dB) Freq. (MHz) Limit (dB) Worst Pair PS ACR-F (dB) Freq. (MHz) Limit (dB)MAIN 78-45 15.7 3.9 53.5 45 14.4 2.3 55.2SR 45-78 15.8 4.3 52.7 45 14.4 2.3 55.2MAIN 45-36 16.7 472.0 11.8 45 18.9 499.0 8.3