Title: Kapasitetsmåler Mecon.pdf
The orifice plate flowmeter FO Turbo-Lux 3 consists of an orifice plate (Fig. 6/7, pos 1) forstationary installation and a portable bypass meter (Fig. 6, pos. 2). The bypass meter contains aconical glass tube (Fig. 6, pos. 3) with float (Fig. 6, pos. 4). The water flows vertically from top tobottom through the flow tube at the upper end of a side panel (Fig. 6, pos. 5) is arranged.A filter (Fig. 6, pos. 13) at the inlet largely prevents the ingress of foreign bodies. Inlet and outletport for the bypass to be measured are arranged concentrically, so that an easy to combine withthe stationary primary element.2.4 MaintenanceIf the filter is blocked by deposits (Fig. 6, pos. 13) the flowmeter must be returned to themanufacturer to be cleaned and tested. Ensure that the O-ring (Fig. 6, pos. 8) and the M 30 x 1,5thread of the orifice plate are lubricated with grease.2.5 StorageStore the emptied device in a dry and dust-free place. Keep away from direct and permanent sun