Title: +175=574.001_OE001_Manual - Varmemåler Hydrocal M3 - På rør.Produktinformasjon.pdf
112 x 78 x 76.5mmWeights635g (DN15); 708g (DN20)IP protection gradeIP54Approval2004/22/CE MID (Module B)Certificate nr. IT-002-15-MI004-NB2213EN1434Environmental classA (E1, M1)Temperature measurement range, MID approved (heating applications)Θ: 5÷90°CTemperature difference range, MID approved (heating applications)ΔΘ: 3÷70KTemperature measurement range (cooling)Θ: 2÷24°CTemperature difference range (cooling)ΔΘ: 3÷20KAccuracy class2RangeDepending on diameter and nominal flow rateDIAMETERQp (m3/h)Range (H/V)DN150.625/25DN151.550/50DN202.550/50Operative counting conditionsHeating: ΔΘ≥1K and fluid temperature ≥5°C (MID approved), ΔΘ≥1K and fluid temperature ≥5°C (accounting enabling conditions)Cooling: ΔΘ≤0.2K and fluid temperature ≤24°CMaximum measurable power650 kWDisplayLCD, 8 digits + iconsMeasurement unitMWh (standard), GJ (optional)Temperature sensorsPt1000Cable length1.5 mPulse inputs