Title: +038=370.101-OE002T_02-Multical 801 Energimåler Kamstrup - Teknisk beskrivelse.pdf
Number of bytes in each field Field designation1Start byte1Destination address1CID0-?Data2CRC1Stop byteOSI – layerApplication layerData link layerPhysical layerThe protocol is based on half-duplex serial synchroneous communication with setup: 8 data bits, no parity and 2 stop bits. The data bit rate is 1200 or 2400 baud. CRC16 is used in both request and response.Data is transferred byte for byte in a binary data format, of which the 8 data bits represent one byte of data.Byte Stuffing is used for extending the value range.12.1.1 The register IDs of MULTICAL 801Kamstrup A/S · Technical Description · 5512-571_R1_GB _11.201699MULTICAL® 801ID 1003 60 94 63 61 62 95 96 97 110 64 65 68 69 84 85 72 73 1004 113 1002 99 86 87 88 122 89 91 92 74 75 80 123 124 125 126 127 128 129 130 138 139 140 141 142 143 144 145 146 147 149 150 66 67 98 152 153 168 1001 112 1010 114 104 1005 154 155 157 158