ick] is a cylindrical Ge single crystal with a diameter of 70 mm and a thickness of 20 mm. The edges have been beveled at an angle of 45$^o$. The plane surfaces and chamfers have been metalized for ionization measurement. The electrodes are made of 100 nm Al layers sputtered on the surfaces after etching. The top electrode is divided in a central part and a guard ring, electrically decoupled for radial localization of the charge deposition. During data taking, a voltage $V_o =$ 6.37 V was applied to the top electrodes. The electrodes were regularly shorted in order to prevent charge accumulation due to trapping of carriers in the detector volume.

The cross-talk between the centre and guard ring electrode signal is approximately 10%. This does not affect the ionization energy resolution since the cross-talk amplitude is a fixed fraction of the signal amplitude on the other electrode. Its shape is constant in time and this effect is easily taken into account by the simultaneous analysis of the signals recorded on both electrodes.

The thermal sensor consists of a Neutron Transmutation Doped germanium crystal (NTD) of 4 mm$^3$ glued to the beveled part of the surface of the detector. The residual radioactivity of the activated NTD sensor should thus be mostly contained in the region covered by the guard electrode. The resistance of the DC-polarized sensor was approximately 3 M$\Omega$ for a base temperature of 27 mK, stabilized to within $\pm$10 $\mu$K.

The signals from all channels are sent to digitizers triggered by either of the two ionization channels. To generate the trigger, the ionization signals are processed by shaping amplifiers and sent to two discriminators. The trigger is the “or” of the output of the two discriminators. The rise time of the heat signal is of the order of 10 ms, much slower than the $\mu$s-fast ionization channel, and was not used for triggering.

[**Detector Calibration**]{} The heat and ionization responses to $\gamma$ rays were calibrated using $^{57}$Co sources. Over the entire data-taking period, the baseline resolutions on the centre and guard ring ionization signals are better than 2.0 keV FWHM, and it varied between 1.9 and 3.5 keV FWHM for the heat signal. The corresponding resolutions measured at 122 keV for the centre, guard ring and heat signals are approximately 3, 2 and 3.5 keV F
ick ] is a cylindrical Ge single crystal with a diameter of 70   mm and a thickness of 20   mm. The edges have been bevel at an slant of 45$^o$. The airplane surfaces and chamfers have been metalized for ionization measurement. The electrode are made of 100   nm Al layers sputtered on the open after etching. The top electrode is divided in a cardinal character and a guard band, electrically decoupled for radial localization of the accusation deposit. During data taking, a electric potential $ V_o = $ 6.37 V was practice to the top electrode. The electrodes were regularly shorted in order to prevent mission accumulation due to trapping of carrier in the detector book. 

 The cross - lecture between the centre and guard ring electrode signal is approximately 10% . This does not affect the ionization energy resolution since the crabbed - talk amplitude is a fixed fraction of the signal amplitude on the other electrode. Its shape is constant in time and this effect is easily taken into account by the simultaneous analysis of the signals recorded on both electrodes. 

 The thermal sensor consists of a Neutron Transmutation Doped germanium crystal (NTD) of 4 mm$^3 $ glue to the beveled part of the airfoil of the detector. The residual radiation of the activate NTD sensor should thus be mostly contained in the region covered by the guard electrode. The underground of the DC - polarized sensor was approximately 3 M$\Omega$ for a base temperature of 27 mK, stabilized to within $ \pm$10   $ \mu$K. 

 The signals from all channels are send to digitizers triggered by either of the two ionization channels. To generate the gun trigger, the ionization signals are action by shape amplifiers and sent to two discriminators. The trigger is the “ or ” of the end product of the two discriminators. The rise time of the heating system signal is of the order of 10   ms, much slower than the $ \mu$s - fast ionization duct, and was not used for triggering. 

 [ * * Detector Calibration * * ] { } The hotness and ionization responses to $ \gamma$ beam were calibrated practice $ ^{57}$Co sources. Over the integral data - taking period, the service line resolutions on the centre and guard ring ionization signals are better than 2.0 keV FWHM, and it varied between 1.9 and 3.5 keV FWHM for the heat signal. The corresponding resolutions measured at 122 keV for the centre, guard hoop and heat signals are approximately 3, 2 and 3.5 keV degree fahrenheit
ick] is a cylindrical Ge sinnle crystal with a diamxter of 70 mm and x thickness of 20 mm. The edges iave been beveled at an angle ow 45$^o$. The ppane surdacew and chamfxds have been jctalivev for ionizatiok measuremett. The electroger cre made of 100 nm Al layers sputtered og the sirvaces after etshinb. The nok electrode is divided in a centrzl part and a guard ting, electrically decouplef fog radial localizatlon of the xhardw deposition. During daua taking, a boltage $V_o =$ 6.37 V was applied to tfe to' electrodew. Rhe glectrodes wxre redularly shorbvd in osder to prevent chargc accnmulqtion due to trapping of carriers in thg detector vklume.

The cross-talj vetwegn tha cevrre ans juadd rinh emectrode sjgnal is approximately 10%. This dpef not affect tge ionyzwtion energy resolution since the cross-ualk zmplitude is a fixed frqction of the signal wmplitude on the other electrode. Its shape is constant in dime ena tklf wfvect is easily taken into account by the simujfamepus analysis on the signals recotdfd jn both electtodes.

Thz tgermal sensor conslsts of a Neytron Tragsmuyation Doped germanium crysral (NTD) of 4 in$^3$ glued to the bevzled part of the surfsce of the detector. The resjdual radiowctivity kw the activated VTD sansor should thus be mostlr containxd in the reeion coverqd by the huard electrode. The residtancg of tve DC-polarlzed sensor was approximately 3 M$\Omega$ for a bssa tvmperaturz of 27 kK, stabilized to within $\pm$10 $\mo$K.

The siguals ffom all chznnels ere sent to qigitizers trhhgered by eivher of tre tqo iinizatiuv channels. To benerate nht trigger, rhe ionization sigkals xde processed by shqping amplifierx avd fejt tj two discrimhnaturs. Yhe tfigger is tme “ur” og the output of the dwo siscriminators. The rlse time if the hqat signal is of the order of 10 md, muci slowxr tham tre $\mu$s-fast ionization channel, znd was nlt msed for trigderikg.

[**Dejector Caliyration**]{} The heat and ionization responsew to $\gamma$ rays wgre calibrated using $^{57}$Eo sources. Ovxr the entire ddta-taking period, the baseline resolutlons on the centre and guard ring ionization signals are better than 2.0 keV FWHM, and it varied between 1.9 and 3.5 keV FXHI for the hsat xignan. Che correfponviig resolutions messured at 122 keV for the centre, gnard ring dnb heat signals are approximatrlh 3, 2 and 3.5 keV W
ick] is a cylindrical Ge single crystal diameter 70 mm a thickness of been at an angle 45$^o$. The plane and chamfers have been metalized for measurement. The electrodes are made of 100 nm Al layers sputtered on the after etching. The top electrode is divided in a central part and a ring, decoupled radial of the charge deposition. During data taking, a voltage $V_o =$ 6.37 V was applied to top electrodes. The electrodes were regularly shorted in to prevent charge accumulation to trapping of carriers in detector The cross-talk the and ring electrode signal approximately 10%. This does not affect the ionization energy resolution since the cross-talk amplitude is a fixed of the on the electrode. shape constant in time effect is easily taken into account analysis of the signals recorded on both electrodes. thermal sensor of a Neutron Transmutation Doped germanium (NTD) of 4 mm$^3$ glued to the beveled of the surface of the detector. The residual radioactivity of the activated NTD sensor should mostly contained in the covered by the electrode. resistance the sensor was 3 M$\Omega$ for a base temperature of 27 mK, stabilized to $\pm$10 $\mu$K. The signals from all channels are sent to by of the two channels. To generate the the signals are processed by and to trigger the of the output of two discriminators. The rise time the heat signal is ms, much slower than the $\mu$s-fast ionization channel, was not used for triggering. [**Detector Calibration**]{} heat and ionization responses to $\gamma$ rays were calibrated using $^{57}$Co sources. the entire the baseline resolutions on the centre and guard ionization signals are better 2.0 keV FWHM, and it varied between 1.9 and keV for the signal. The corresponding measured at 122 for the centre, and heat are 3, keV F
ick] is a cylindrical Ge single Crystal witH a diaMetEr oF 70 mM and A thiCkness of 20 mm. The eDGes hAve been beveled at an anglE of 45$^o$. THe PLane SUrFaces And chamFErS HAve BeEn MetAlIZeD for iOniZation mEasurement. the ElEctrodes are mADe Of 100 nm Al layeRs sPuttered on thE suRfaces AfTer ETchinG. ThE top eLectroDE is divIded in a ceNtRAl part ANd a guarD RInG, eleCtrically decoupleD FoR Radial localizaTion of ThE ChARGe dEpoSition. DuriNg Data tAKing, a voLTaGE $v_O =$ 6.37 V wAS applied to the Top electrodES. ThE electRoDes WEre regUlarlY sHOrtEd in order to PrevEnt charge AccumuLAtion duE To trappIng of cArrIerS in tHE dEtEctOr VOluME.

THe cROss-Talk betwEeN tHe cenTre aND GUArd rIng ElecTrode Signal is approXimAtelY 10%. thiS does Not afFect ThE ioniZation EnergY rEsolution since tHe crOss-talk amPliTuDe iS a Fixed FRactioN of The Signal aMplitudE On tHe OTHEr Electrode. Its shape iS cONStAnt in timE and thIS eFfECt is easiLy TakEn inTO AccouNt by THe SimultanEous anALySiS of the sIgNals reCoRdeD on Both eLEctrOdes.

ThE thermal SensoR Consists of a NeuTRon TransmutatIOn dOPeD GermAniUm crystal (NTd) of 4 mM$^3$ GlueD to tHE bEveLEd parT of thE sURfACe of the detector. The rEsIdual rAdioaCtivity of the aCtivated NTd SENsor shouLd thUS bE Mostly containeD in thE region covERed by the Guard ElectrodE. The resisTANce of the dC-pOlaRizEd sENSoR was approximaTELy 3 M$\OMeGa$ for a bAse TemperaTurE of 27 MK, sTabIlIzed to witHin $\pm$10 $\mu$K.

thE sIgNaLs fRom alL Channels ArE seNt To dIgitiZErs triGgereD by eItHeR Of tHe two ioNIzATIon cHaNnEls. TO geNeRate tHe trIGgeR, the ionIzation siGnaLS are PrOcEssed by Shaping amplifIeRs and sent tO tWo dIscrimINAtors. The Trigger is the “or” of the outpUT of the tWo dIscriMinaTors. The riSe tIme of tHe hEAt signAl is of The orDeR of 10 MS, Much sLOWeR thAn The $\mu$s-fast IONizAtion ChAnneL, and was Not used for triggeriNG.

[**DeTector CalibraTioN**]{} The HEAt And IOnIZatIoN ResPONses to $\gamma$ rays Were calibrAtED uSing $^{57}$Co sourCEs. OVeR the entIre data-TakinG Period, tHe baselinE resolutiOnS on tHE CenTre and guarD ring ionIzation siGNals aRE bEtter ThaN 2.0 keV FWhM, And It varIed betWEen 1.9 And 3.5 kev FWHM fOr The heaT signAl. the correSponding resolutions measUred at 122 KeV foR thE centre, guArd RIng And heat siGnalS are approxImaTelY 3, 2 and 3.5 kEV F
ick] is a cylindrical Ge s ingle crys tal w ith  adi amet er o f 70 mm and at hick ness of 20 mm. The edg es ha ve been be veled  at ana ng l e  of  4 5$ ^o$ .T he  plan e s urfaces  and chamf ers  h ave been met a li zed for io niz ation measur eme nt. Th eele c trode s a re ma de of1 00 nmAl layers  s p uttere d  on the s ur face s after etching.T he top electrodeis div id e di n  acen tral partan d a g u ard rin g ,e l e ctr i cally decoupl ed for radi a l l ocaliz at ion of the  char ge dep osition. Du ring  data tak ing, a voltage $V_o =$  6.37V w asappl i ed  t o t he top el ect r ode s. The e le ct rodes  wer e r e gula rly  sho rtedin order to p rev entc har ge ac cumul atio ndue t o trap pingof  carriers in th e de tector vo lum e. 

T he  cros s -talkbet wee n the c entre a n d g ua r d ri ng electrode signa li s  a pproxima tely 1 0 %.  T h is doesno t a ffec t the i oniz a ti on energ y reso l ut io n since  t he cro ss -ta lkampli t udeis a f ixed fra ction of the signala mplitude on t h eo t he r  ele ctr ode. Its sh apei s co nsta n tint ime a nd th is ef f ect is easily taken  i nto ac count  by the simul taneous an a l y sis of t he s i gn a ls recorded on  both  electrode s .

The t herma l sensor  consists o f a Neut ron  Tr ans mut a t io n Doped germa n i um c ry stal (N TD)  of 4 m m$^ 3$glu edto  the beve led part  o fth esur faceo f the de te cto r.  Th e res i dual r adioa ctiv it yo f t he acti v at e d  NTD  s en sorsho ul d thu s be mos tly con tained in  th e  reg io ncovered  by the guard  e lectrode.Th e r esista n c e of the  DC-polarized sensor wa s  approx ima tely3 M$ \Omega$ f ora base  te m peratu re of27 mK ,sta b i lized t owit hi n $\pm$10$ \ mu$ K.

T he  sig nals fr om all channels ar e  se nt to digitiz ers  tri g g er edb ye ith er oft h e two ionizatio n channels .T ogenerate t h e t ri gger, t he ioni zatio n  signal s are pro cessed by  s hapi n g  am plifiers a nd sentto two di s crimi n at ors.The  trigg er  is  the“or” o f  th e out put of  t he two  disc ri minators . The rise time of theheat s ignal  is  of the o rde r  of  10 ms, m uchslower tha n t he$\mu$ s-f a st io niza t io n c h annel , an d  was notu se d f o r  t riggering.[ * *De tecto r C a librat ion* *]{} The heat and ionization res pons e s  to  $\ g amma $rays were cali bra te d using $^ {5 7}$Co sourc es. Over  t h e ent ire da ta-tak ing per i o d, the ba seli neresolutio nson the cen tr ea nd gua rd r in g ioni zation sign a l s are better tha n 2.0 k eV FW H M,and i tvariedb etwe en 1.9 and  3.5 keV FW HM for  the  heat  signal .The co rre sp onding res o lutions m easur ed at 1 22  keV  fo r thecent r e , gua rd r in g a nd heat s i g na l sar e  ap prox imate ly  3,2 and 3.5 keV F
ick] is_a cylindrical_Ge single crystal with_a diameter_of_70 mm and_a_thickness of 20 mm._The edges have_been beveled at an_angle of 45$^o$._The_plane surfaces and chamfers have been metalized for ionization measurement. The electrodes are made_of_100 nm Al_layers_sputtered_on the surfaces after etching._The top electrode is divided_in a_central part and a guard ring, electrically decoupled_for_radial localization of_the charge deposition. During data taking, a voltage $V_o_=$ 6.37 V was applied to_the top electrodes._The_electrodes_were regularly shorted in_order to prevent charge accumulation due_to trapping of carriers in the_detector volume.

The cross-talk between the centre and_guard ring electrode signal is approximately_10%. This does not affect_the ionization_energy resolution since the cross-talk_amplitude is a_fixed fraction_of the signal_amplitude on the other electrode. Its_shape is constant_in time and this effect is_easily_taken into account_by_the_simultaneous analysis_of the signals_recorded_on both_electrodes.

The_thermal sensor consists of a Neutron_Transmutation_Doped germanium crystal (NTD) of 4 mm$^3$_glued to the beveled_part_of the surface of_the detector. The residual radioactivity_of the activated NTD sensor should_thus be_mostly contained_in the region covered by the guard electrode. The resistance of_the DC-polarized sensor was approximately 3_M$\Omega$ for a base_temperature of_27_mK, stabilized to_within_$\pm$10 $\mu$K.

The signals_from all channels are sent to digitizers_triggered by_either of the two ionization channels._To generate the trigger,_the_ionization signals are processed by shaping_amplifiers and sent to two discriminators._The trigger is the “or”_of_the_output of the two discriminators._The rise time of the heat_signal is of_the order of 10 ms, much slower than_the_$\mu$s-fast ionization channel, and was not_used_for triggering.

[**Detector Calibration**]{} The heat and_ionization_responses_to $\gamma$ rays were calibrated_using $^{57}$Co sources. Over the entire_data-taking period, the baseline resolutions on the centre and_guard ring ionization_signals are better than 2.0_keV_FWHM,_and it varied between 1.9 and 3.5 keV FWHM for_the heat_signal. The corresponding_resolutions measured at 122 keV for the centre, guard ring_and heat signals are approximately 3, 2_and 3.5 keV F