Type Temperature range (°C) Tolerance class (°C) Color code Continuous Short-term One Two IECIEC 60584-3:2007 BS ANSI Low High Low High K 0 +1100 −180 +1370 −40 – 375: ±1.5375 – 1000: ±0.004×T −40 – 333: ±2.5333 – 1200: ±0.0075×T center|56px center|56px center|56px J 0 +750 −180 +800 −40 – 375: ±1.5375 – 750: ±0.004×T −40 – 333: ±2.5333 – 750: ±0.0075×T center|56px center|56px center|56px N 0 +1100 −270 +1300 −40 – 375: ±1.5375 – 1000: ±0.004×T −40 – 333: ±2.5333 – 1200: ±0.0075×T center|56px center|56px center|56px R 0 +1600 −50 +1700 0 – 1100: ±1.01100 – 1600: ±0.003×(T − 767) 0 – 600: ±1.5600 – 1600: ±0.0025×T center|56px center|56px Not defined S 0 +1600 −50 +1750 0 – 1100: ±1.01100 – 1600: ±0.003×(T − 767) 0 – 600: ±1.5600 – 1600: ±0.0025×T center|56px Not defined B +200 +1700 0 +1820 Not available 600 – 1700: ±0.0025×T No standard No standard Not defined T −185 +300 −250 +400 −40 – 125: ±0.5125 – 350: ±0.004×T −40 – 133: ±1.0133 – 350: ±0.0075×T center|56px center|56px center|56px E 0 +800 −40 +900 −40 – 375: ±1.5375 – 800: ±0.004×T −40 – 333: ±2.5333 – 900: ±0.0075×T center|56px center|56px center|56px Chromel/AuFe −272 +300 Reproducibility 0.2% of the voltage.Each sensor needs individual calibration.
typE TemPeraTuRe RaNgE (°c) tOleraNce clasS (°c) coLor CODe contInuous sHort-term onE two ieCieC 60584-3:2007 bs anSi Low hIGH low High K 0 +1100 −180 +1370 −40 – 375: ±1.5375 – 1000: ±0.004×t −40 – 333: ±2.5333 – 1200: ±0.0075×T CenTer|56pX cENteR|56px CeNtEr|56px J 0 +750 −180 +800 −40 – 375: ±1.5375 – 750: ±0.004×T −40 – 333: ±2.5333 – 750: ±0.0075×T cenTeR|56px center|56pX center|56Px N 0 +1100 −270 +1300 −40 – 375: ±1.5375 – 1000: ±0.004×t −40 – 333: ±2.5333 – 1200: ±0.0075×T CenteR|56pX CeNteR|56px ceNTer|56px r 0 +1600 −50 +1700 0 – 1100: ±1.01100 – 1600: ±0.003×(t − 767) 0 – 600: ±1.5600 – 1600: ±0.0025×t CeNtEr|56PX center|56px nOt DefiNED s 0 +1600 −50 +1750 0 – 1100: ±1.01100 – 1600: ±0003×(T − 767) 0 – 600: ±1.5600 – 1600: ±0.0025×t ceNter|56pX Not dEfiNed b +200 +1700 0 +1820 not avaILabLe 600 – 1700: ±0.0025×t no staNdaRd No sTAndaRd noT definED T −185 +300 −250 +400 −40 – 125: ±0.5125 – 350: ±0.004×t −40 – 133: ±1.0133 – 350: ±0.0075×T cenTer|56pX Center|56px cENTer|56pX E 0 +800 −40 +900 −40 – 375: ±1.5375 – 800: ±0.004×T −40 – 333: ±2.5333 – 900: ±0.0075×t center|56Px CEnter|56PX ceNteR|56pX chromEl/auFe −272 +300 RepRoducibiLiTY 0.2% oF the vOlTaGe.eaCh senSOr nEeDs IndiviDuaL CaLIBratioN.