A sector ICP-MS will commonly have four sections: an extraction acceleration region, steering lenses, an electrostatic sector and a magnetic sector. The first region takes ions from the plasma and accelerates them using a high voltage. The second uses may use a combination of parallel plates, rings, quadrupoles, hexapoles and octopoles to steer, shape and focus the beam so that the resulting peaks are symmetrical, flat topped and have high transmission. The electrostatic sector may be before or after the magnetic sector depending on the particular instrument, and reduces the spread in kinetic energy caused by the plasma. This spread is particularly large for ICP-MS, being larger than Glow Discharge and much larger than TIMS. The geometry of the instrument is chosen so that the instrument the combined focal point of the electrostatic and magnetic sectors is at the collector, known as Double Focusing (or Double Focussing).
A sector ICP-MS will commonly have four sections: an extraction acceleration region, steering lenses, an electrostatic sector and a magnetic sector. The first region takes ions from the plasma and accelerates them using a high voltage. The second uses may use a combination of parallel plates, rings, quadrupoles, hexapoles and octopoles to steer shape and focus the beam so that the resulting peaks are symmetrical, flat topped and have high transmission. The electrostatic sector may be before or after the magnetic sector depending on the particular instr,ument, and reduces the spread in kinetic energy caused by the plasma. This spread is particularly large for ICP-MS, being larger than Glow Discharge and much larger than TIMS. The geometry of the instrument is chosen so that the instrument the combined focal point of the electrostatic and magnetic sectors is at the collector, known as Double Focusing (or Double Focussing).