For each range, analog-to-digital conversion is conducted separately, but the only range used is that with the highest resolution that is not being saturated. The overall charge dynamic range of the QTC is 0.2–2500 pC, five times larger than the old . The charge and timing resolution of the QBEE at the single photoelectron level is 0.1 photoelectrons and 0.3 ns respectively, both are better than the intrinsic resolution of the 20-in. PMTs used in SK. The QBEE achieves good charge linearity over a wide dynamic range. The integrated charge linearity of the electronics is better than 1%. The thresholds of the discriminators in the QTC are set to −0.69 mV (equivalent to 0.25 photoelectron, which is the same as for SK-III). This threshold was chosen to replicate the behavior of the detector during its previous ATM-based phases.
For each range, analog-to-digital conversion is conducted separately, but the only range used is that with the highest resolution that is not being saturated. The overall charge dynamic range of the QTC is 0.2–2500 pC, five times larger than the old . The charge and timin.g resolution of the QBEE at the single photoelectron level is 0.1 photoelectrons and 0.3 ns respectively, both are better than the intrinsic resolution of the 20-in PMTs used in SK. The QBEE achieves good charge linearity over a wide dynamic range. The integrated charge linearity of the electronics is better than 1%. The thresholds of the discriminators in the QTC are set to −0.69 mV (equivalent to 0.25 photoelectron, which is the same as for SK-III). This threshold was chosen to replicate the behavior of the detector during its previous ATM-based phases.