<div class="description">
<div>INDEX</div><div>HOW TO USE YOUR MYO-MONITOR</div><div>USE OF TEST CIRCUIT</div><div>ELECTRODE CONNECTION</div><div>PULSE CIRCUIT</div><div>AMPLITUDE DIAL ADJUSTMENTS</div><div>THRESHOLD</div><div>FREEWAY SPACE</div><div>OVERCLOSURE CIRCUIT</div><div>INFLUENCE OF POSTURE</div><div>BALANCE CONTROL</div><div>CLINICAL INDICATIONS</div><div>TMJ DISFUNCTION SYNDROME</div><div>TMJ TREATMENT</div><div>OCCLUSAL ADJUSTMENT</div><div>FULL MOUTH RECONSTRUCTION</div><div>TECHNIC FOR COMPLETE PROSTHODONTICS</div><div>REDUCTION OF TRISMUS AND SWELLING</div><div>DO'S &amp; DONT'S</div><div>POWER SUPPLY</div>
<br>Subjects: Dentistry; Instruction manual</div>
<div class="fn value">
Myotronics, Inc</div>
<dl><dd><a rel="nofollow" class="external free" href="https://archive.org/details/MyoMonitorInstructionManualModelJ2">https://archive.org/details/MyoMonitorInstructionManualModelJ2</a></dd>
<dd><a rel="nofollow" class="external free" href="https://archive.org/download/MyoMonitorInstructionManualModelJ2/Myo-Monitor%20Instruction%20Manual%20Model%20J-2.pdf">https://archive.org/download/MyoMonitorInstructionManualModelJ2/Myo-Monitor%20Instruction%20Manual%20Model%20J-2.pdf</a></dd></dl>