<div class="description">
<p>lected points. Incorporated instrumentation includes pressure transducers attached to a pitot-static tube, wall static pressure taps, and a pressure rake; a hotwire anemometry system, and a linear traverse system. These were integrated with a data acquisition (DAQ) processor with analog to digital conversion and digital I/O boards, and controlled using in-house developed LabVIEW software. Testing showed a maximum axial velocity of 38 m/s, which is 84% of the tunnel?s rated speed. The 2-D flow uniformity was within Â±7% by pressure rake, and Â±3% with a turbulence intensity ?0.11% at full speed using a CTA, affirming the tunnel?s viability as a demonstration platform. Spectral density plots in the boundary layer exhibit typical behavior of fully developed equilibrium turbulent flow with an intertial sub-range present. Future testing of a flat-plate wake for drag modification is planned.
</p>
<br>Subjects: Boundary layer; Mechanical engineering</div>
<div class="fn value">
Harvey, Scott A.</div>
Internet Archive identifier