PMC6699011_01
Digital impression and framework digital design, framework trial, and printed
model with teeth arrangement.
10.1177_2050313X19871131-fig1.jpg
PMC6699011_01_10.1177_2050313X19871131-fig1.jpg
fig1-2050313X19871131
The intraoral metal framework try-on was performed, according to clinical criteria (Figure 1).
Intra-oral photographs of the patient with the removable partial denture.
10.1177_2050313X19871131-fig2.jpg
PMC6699011_01_10.1177_2050313X19871131-fig2.jpg
fig2-2050313X19871131
The RDP was then placed in the mouth and adjusted for comfort and function (Figure 2).
(a) 3D deviation of the superimposed digital framework design (CAD) and the
metal framework produced (CAM) and (b) 3D deviation colour map of the
superimposed digital impression and the printing model.
10.1177_2050313X19871131-fig3.jpg
PMC6699011_01_10.1177_2050313X19871131-fig3.jpg
fig3-2050313X19871131
This file was superimposed onto the STL file of the digital model, to a tolerance limit of 100 microns (Figure 3).