PMC5573516_01
Panoramic radiograph of the patient prior to treatment.
jiufd-050-057-e001.jpg
PMC5573516_01_jiufd-050-057-e001.jpg
fig1
Since these findings may be caused by excessive force loads resulting from incorrect prosthetic design (Figure 1, Figure 2), patient was informed about screw retained hybrid prosthesis concept and following clinical and laboratory procedures were performed.
(a) Initial lateral view of the patient (b) Diagnostic casts.
jiufd-050-057-e002.jpg
PMC5573516_01_jiufd-050-057-e002.jpg
fig2
Since these findings may be caused by excessive force loads resulting from incorrect prosthetic design (Figure 1, Figure 2), patient was informed about screw retained hybrid prosthesis concept and following clinical and laboratory procedures were performed.
The metal framework passively seating on the abutments.
jiufd-050-057-e003.jpg
PMC5573516_01_jiufd-050-057-e003.jpg
fig3
Alloys, Eisenbacher Dentalwaren ED GmbH, Bavaria, Germany), recovered and fitted to the implants on the master cast.The patient was recalled for the clinical framework fitting trial. After removing the healing caps, the framework was passively seated on the abutments and each abutment framework interface was checked for an intimate fit (Figure 3).
Delivery of definitive maxillary screw-retained hybrid prosthesis: Lateral profile view of the patient (a) and final occlusion (b).
jiufd-050-057-e004.jpg
PMC5573516_01_jiufd-050-057-e004.jpg
fig4
Hygiene techniques were reviewed, and the patient was scheduled for recall and maintenance (Figure 4).