A surface reflection-based method for measuring heart rate and saturation has been introduced as one having a significant advantage over legacy methods in that it lends itself for use in special applications such as those where a person's mobility is of prime importance (e.g. during a miner's work) and excluding the use of traditional clips. Then, a complete ATmega1281-based microcontroller platform has been described for performing computational tasks of signal processing and wireless transmission. In the next section remarks have been provided regarding the basic signal processing rules beginning with raw voltage samples of converted optical signals, their acquisition, storage and smoothing. This chapter ends with practical remarks demonstrating an exponential dependence between the minimum measurable heart rate and the readout resolution at different sampling frequencies for different cases of averaging depth (in bits). The following section is devoted strictly to the heart rate and hemoglobin oxygenation (saturation) measurement with the use of the presented platform, referenced to measurements obtained with a stationary certified pulsoxymeter.