qcodes/instrument_drivers/Keysight/keysightb1500/message_builder.py
def dcv(self, chnum: Union[constants.ChNr, int], voltage: float
            ) -> 'MessageBuilder':
        """
        This command forces DC bias (voltage, up to +- 25 V) from the MFCMU.
        When the SCUU (SMU CMU unify unit) is connected, output up to +- 100 V
        is available by using the SMU that can be connected to the
        Force1/Sense1 terminals.

        Execution conditions: The CN/CNX command has been executed for the
        specified channel.

        If you want to apply DC voltage over +- 25 V, the SCUU must be
        connected correctly. The SCUU can be used with the MFCMU and two
        SMUs (MPSMU or HRSMU). The SCUU cannot be used if the HPSMU is
        connected to the SCUU or if the number of SMUs connected to the SCUU
        is only one.

        If the output voltage is greater than the allowable voltage for the
        interlock open condition, the interlock circuit must be shorted.

        Args:
            chnum: MFCMU channel number. Integer expression. 1 to 10 or
                101 to 1001. See Table 4-1 on page 16.

            voltage: DC voltage (in V). Numeric expression.
                0 (initial setting) to +- 25 V (MFCMU) or +- 100 V (with SCUU).
                With the SCUU, the source module is automatically selected by
                the setting value. The MFCMU is used if voltage is below
                +- 25 V (setting resolution: 0.001 V), or the SMU is used if
                voltage is greater than +- 25 V (setting resolution: 0.005 V).
                The SMU will operate with the 100 V limited auto ranging and
                20 mA compliance settings.
        """
        cmd = f'DCV {chnum},{voltage}'

        self._msg.append(cmd)
        return self

        This command forces DC bias (voltage, up to +- 25 V) from the MFCMU.
        When the SCUU (SMU CMU unify unit) is connected, output up to +- 100 V
        is available by using the SMU that can be connected to the
        Force1/Sense1 terminals.

        Execution conditions: The CN/CNX command has been executed for the
        specified channel.

        If you want to apply DC voltage over +- 25 V, the SCUU must be
        connected correctly. The SCUU can be used with the MFCMU and two
        SMUs (MPSMU or HRSMU). The SCUU cannot be used if the HPSMU is
        connected to the SCUU or if the number of SMUs connected to the SCUU
        is only one.

        If the output voltage is greater than the allowable voltage for the
        interlock open condition, the interlock circuit must be shorted.

        Args:
            chnum: MFCMU channel number. Integer expression. 1 to 10 or
                101 to 1001. See Table 4-1 on page 16.

            voltage: DC voltage (in V). Numeric expression.
                0 (initial setting) to +- 25 V (MFCMU) or +- 100 V (with SCUU).
                With the SCUU, the source module is automatically selected by
                the setting value. The MFCMU is used if voltage is below
                +- 25 V (setting resolution: 0.001 V), or the SMU is used if
                voltage is greater than +- 25 V (setting resolution: 0.005 V).
                The SMU will operate with the 100 V limited auto ranging and
                20 mA compliance settings.
        
This command forces DC bias (voltage, up to +- 25 V) from the MFCMU.
When the SCUU (SMU CMU unify unit) is connected, output up to +- 100 V
is available by using the SMU that can be connected to the
Force1/Sense1 terminals.

Execution conditions: The CN/CNX command has been executed for the
specified channel.

If you want to apply DC voltage over +- 25 V, the SCUU must be
connected correctly. The SCUU can be used with the MFCMU and two
SMUs (MPSMU or HRSMU). The SCUU cannot be used if the HPSMU is
connected to the SCUU or if the number of SMUs connected to the SCUU
is only one.

If the output voltage is greater than the allowable voltage for the
interlock open condition, the interlock circuit must be shorted.
This command forces DC bias ( voltage up to + - 25 V ) from the MFCMU . When the SCUU ( SMU CMU unify unit ) is connected output up to + - 100 V is available by using the SMU that can be connected to the Force1 / Sense1 terminals . Execution conditions : The CN / CNX command has been executed for the specified channel . If you want to apply DC voltage over + - 25 V the SCUU must be connected correctly . The SCUU can be used with the MFCMU and two SMUs ( MPSMU or HRSMU ) . The SCUU cannot be used if the HPSMU is connected to the SCUU or if the number of SMUs connected to the SCUU is only one . If the output voltage is greater than the allowable voltage for the interlock open condition the interlock circuit must be shorted .
def dcv(self, chnum: Union[constants.ChNr, int], voltage: float
            ) -> 'MessageBuilder':
        cmd = f'DCV {chnum},{voltage}'
        self._msg.append(cmd)
        return self
def dcv ( self , chnum : Union [ constants . ChNr , int ] , voltage : float ) -> 'MessageBuilder' : cmd = f'DCV {chnum},{voltage}' self . _msg . append ( cmd ) return self
This command forces DC bias (voltage, up to +- 25 V) from the MFCMU.
This command forces DC bias ( voltage up to + - 25 V ) from the MFCMU .
"""
        This command forces DC bias (voltage, up to +- 25 V) from the MFCMU.
        When the SCUU (SMU CMU unify unit) is connected, output up to +- 100 V
        is available by using the SMU that can be connected to the
        Force1/Sense1 terminals.

        Execution conditions: The CN/CNX command has been executed for the
        specified channel.

        If you want to apply DC voltage over +- 25 V, the SCUU must be
        connected correctly. The SCUU can be used with the MFCMU and two
        SMUs (MPSMU or HRSMU). The SCUU cannot be used if the HPSMU is
        connected to the SCUU or if the number of SMUs connected to the SCUU
        is only one.

        If the output voltage is greater than the allowable voltage for the
        interlock open condition, the interlock circuit must be shorted.

        Args:
            chnum: MFCMU channel number. Integer expression. 1 to 10 or
                101 to 1001. See Table 4-1 on page 16.

            voltage: DC voltage (in V). Numeric expression.
                0 (initial setting) to +- 25 V (MFCMU) or +- 100 V (with SCUU).
                With the SCUU, the source module is automatically selected by
                the setting value. The MFCMU is used if voltage is below
                +- 25 V (setting resolution: 0.001 V), or the SMU is used if
                voltage is greater than +- 25 V (setting resolution: 0.005 V).
                The SMU will operate with the 100 V limited auto ranging and
                20 mA compliance settings.
        """
self
chnum
Union[constants.ChNr, int]
MFCMU channel number. Integer expression.
MFCMU channel number . Integer expression .
voltage
float
DC voltage (in V). Numeric expression.
0 (initial setting) to +- 25 V (MFCMU) or +- 100 V (with SCUU).
With the SCUU, the source module is automatically selected by
the setting value. The MFCMU is used if voltage is below
+- 25 V (setting resolution: 0.001 V), or the SMU is used if
voltage is greater than +- 25 V (setting resolution: 0.005 V).
The SMU will operate with the 100 V limited auto ranging and
20 mA compliance settings.
DC voltage ( in V ) . Numeric expression . 0 ( initial setting ) to + - 25 V ( MFCMU ) or + - 100 V ( with SCUU ) . With the SCUU the source module is automatically selected by the setting value . The MFCMU is used if voltage is below + - 25 V ( setting resolution : 0 . 001 V ) or the SMU is used if voltage is greater than + - 25 V ( setting resolution : 0 . 005 V ) . The SMU will operate with the 100 V limited auto ranging and 20 mA compliance settings .