Minimed Paradigm
Introduction The Medtronic Minimed Paradigm 515 pump (RF receiver), Silhouette infusion set, and UltraLink blood sugar meter (RF transmitter). '''MiniMed Paradigm''' is a series of insulin pumps manufactured by Medtronic for patients with diabetes mellitus. The pump operates with a single AAA battery and uses a piston-plunger pump to infuse a programmed amount of insulin into the patient through a length of tubing. The Paradigm uses a one-way wireless radio frequency link to receive blood sugar measurements from select glucose meters. The Paradigm RT (Real Time) series adds the ability to receive data from a mated continuous blood-glucose monitor. Although the pump can use these measurements to assist in calculating a dose of insulin, no actual change in insulin delivery occurs without manual user-intervention.In the United States, the device is regulated by a branch of the Food and Drug Administration.
Description Pump and infusion set (catheter) placement.Insulin basal bolus profile.Insulin pumps are drug delivery devices used to treat patients with type 1 and type 2 diabetes. The Minimed Paradigm REAL-Time and Continuous Glucose Monitoring (CGM) system, which received FDA clearance in 2006, uses tubing and a reservoir with rapid-acting insulin. This "infusion set" is patient-connected via a catheter to the abdomen region. The infusion set can remain in the place for three days while the pump is clip-belt worn. There is a quick-disconnect feature for the tubing. The pump delivers insulin in two modes. In Basal rate mode, the delivery is continuous in small doses similar to a pancreas, for example 0.15 units per hour throughout the day. Basal rates are set to meet individual metabolic rates. In Bolus mode, the delivery is programmed to be a one-time delivery prior to eating or after an unexpected high, for example 18 units spread out to several hours. This type of continuous treatment is in contrast to traditional multiple daily injections (MDI) that use slower-acting insulin. Continuous treatment reduces glucose variability.The Paradigm system consists of two basic parts: an insulin pump and an optional glucose sensor CGM worn for up to 3 days. The disposable sensor is subcutaneously-placed to make glucose measurements in interstitial fluid every 5 minutes and transmit the reading via low power radio frequency (ISM band) to the pump for realtime display. However, insulin therapy may be conducted without CGM and although there is not yet an automated insulin-regulation feedback mechanism between measure and infusion to control the amount and timing of insulin, this is clearly a future objective. So any change in basal or bolus is patient-driven by programming the pump using the Bolus Wizard. The latest model pumps are the MiniMed Paradigm 522 and 722 which differ in reservoir size, 176 versus 300 units, respectively. In 2007 the FDA approved a pediatric model for patients 7 to 17 years old.
History '''The development history of the Minimed pump goes back to the 1980s. '''* 1983 – 1st Pump MiniMed 502 (Eli Lilly makes synthetic insulin)* 1985 – MiniMed 504 Insulin Pump* 1992 – Launch Of MiniMed 506 Insulin Pump* 1996 – Introduction of MiniMed 507 Pump* 1999 – Launch of the Model 507C* 1999 – Introduction of MiniMed 508 Insulin Pump* 2002 – Inauguration of the MiniMed Paradigm 511* 2003 – 1st Wireless MiniMed Paradigm 512/712 (followed by 515/715)* 2006 – MiniMed Paradigm REAL-Time 522/722* 2010 – MiniMed Paradigm REAL-Time Revel 523/723
FDA Classification '''The Food and Drug Administration has at least six classifications for the various parts of the Minimed Paradigm System. '''+ Product code  Generic part name  Regulation number  Device classification code   Description MDS  CGM  null   Premarket approval   Invasive glucose sensor FMF  Drug reservoir  21CFR888.5860  standards   Piston syringe NBW  Glucose meter  21CFR862.1345 standards   Glucose test system LZG/FRN  Infusion/insulin pump  21CFR888.5725 standards   Infusion pump FPA Infusion set  21CFR888.5440 standards   Intravascular administration set KZH  Infusion set insertion system  21CFR880.6920 standards   Syringer needle introducer
Competitors devices Competitive devices includes Deltec Cozmo, Animas Ping, Tandem Diabetes Care, Inc., Insulet OmniPod, Accu-chek Spirit Combo, and Sooil DiabecareIIS.
Future devices Currently, Medtronic has the following research and development projects in its pipeline: Next Generation REAL-Time Continuous Glucose Monitoring System; Next Generation Insulin Pump; Pre-filled Insulin Reservoirs; Implantable Insulin Pump; and Artificial pancreas (Semi-Automated System & Closed-Loop System). The industry trend in portable devices has piggybacked on the success of wireless technology but not on the success of other disciplines, such as dynamical system, Cybernetics and adaptive systems, for root cause solutionsto close the "true" loop. While the slow-responding dynamics in the physiology of glucose regulation is not beyond the mathematics of PID controllers, 50 years of patient-perspective "advancement" says it is beyond the corporate, congressional, and lobbying leaders. Also, the accuracy of existing continuous glucose monitoring systems poses a problem for an artificial pancreas.
References * ADA - CSII Diabetes Care 2004; 27: S110. * ADA - Implications of the DCCT study ''Diabetes Care'' 2002; 25: 25–27. * ADA - Implications of the UKPD Study ''Diabetes Care'' 2002; 25: 28–32. * * * *  ''Diabetes Technol Ther.'' 2004;6(2):105-13. * * * * * The DCCT Research Group. JAMA. 1997;227(5):374-4 Skyler J.S. Diabtes Technol Ther. 2000. * The DCCT Research Group. Progression of retinopathy with intensive versus conventional treatment in the diabetes control and complications trial. 1994. * The DCCT/EDIC Study Research Group. Intensive diabetes treatment and cardiovascular disease in patients with type 1 diabetes. ''NEJM'' 2005; 353(25): 2643–2653. * The Expert Committee on the Diagnosis and Classification of Diabetes Mellitus. Report of the expert committee on the diagnosis and classification of diabetes mellitus ''Diabetes Care'' 2001; 24: 1. * * * * * * * *
Notes