MQ135 Gas Sensor Calibration I have bought a MQ135 gas sensor. I have read online from multiple sources that a calibration/warm-up period of between 12-24hrs is required for the sensor to give its most accurate readings. I currently have it attached to one of my UNO's just spitting out Analog readings to the Serial monitor and seems (just after an hour!) to have settled at a reading of 50 which it has stayed at now for a period of around 20 minutes. My questions surrounding this process really are:


What is the point in doing this process if when a project that uses this sensor is powered off for some number of hours? Will you have to let it warm-up/calibrate again before having faith in the readings?
How do I actually identify PPM for a gas such as C02? Is there some form of constant that I should be comparing the reading obtained to or something similar?
By my opinion, it is impossible to calibrate these solid electrolyte based sensors. I achieved best results when I measured the CO2 level in 5 minute intervals, by heating the sensor for 100 seconds. I cant remember these times precisely, but by heating and allowing it to cool down I was able to get quite reliable results. But then I realized that the sensor is getting some offset. If you want to do a two-point calibration, you can put it to fresh air, it should read about 0.04%, or you can breathe directly to the sensor for a short while and it shoud read 4%. But I highly recommend using NDIR CO2 sensor, those are used in home air monitoring devices.