Circuit doesn't work when multimeter/current meter is attached I'm working on a battery-powered data logger, using a bare Atmega 328. I'm using sleep between sensor readings, via the Narcoleptic library. Everything seems to work fine, until I connect my multimeter to check on the current.

The multimeter uses manual range-setting (i.e., not autoranging). If I set the multimeter to the 0-20 mA range, I can see that my idle current is about 0.250 mA, but the upper value is off-scale. If I try to get a reading on the 0-200 mA range, the circuit misbehaves, never goes into sleep, and doesn't record any data. The current stays constant somewhere around 14-20mA.

Do I need to do something special to accurately measure current in the range of 0.25-40+mA? Or do I need a different kind/quality of multimeter?

My multimeter is a $15 digital unit from Circuit Test, model DMR-1100A.
Multimeters have something called a "burden voltage". See this PDF: uCurrent - A professional precision current ... - EEVblog.

Also see Dave Jones' video: µCurrent

Meters measure current by dropping voltage over an internal resistor, and measuring the drop. By definition therefore, there is a voltage drop, which is the "burden voltage". 


  the circuit misbehaves, never goes into sleep, and doesn't record any data


If you put a voltmeter on the processor side of the ammeter you will probably find that the voltage going into your processor is somewhat less than you think. That would account for the misbehaving.

One possible approach is to crank up the voltage from your power source, checking the voltmeter to see that it reaches the expected voltage on the processor side of the ammeter, however in a circuit which might turn on and off (ie. sleep) this voltage might jump around (depending on the current) so this should be done with caution (to not exceed the maximum processor voltage).




  Or do I need a different kind/quality of multimeter?


A higher quality meter may well help - watch that video and look at the PDF to see what I am talking about. My meters (which cost more than $15) don't have this problem to the same extent.