Is this measured rise time inaccurate because of limited oscilloscope bandwidth?
Set your probe to x10. 

At x1 setting, there is a lot more capacitive loading, and the bandwidth is severely reduced.

See the specs for this typical passive probe below, and look at the difference between the risetime/bandwidth at x1 and x10 settings.