ECG cardiac monitoring
An Ag/AgCl electrode is covered in AgCl.  It is not shiny and bright.  If you shine it, you are removing the AgCl coating from the electrode, and changing it from a non-polarizable electrode to a polarizable electrode, incredibly changing its properties, likely to the point of uselessness.