Reasons for strain gauge drift
Mounting strain gauges is an art. Your long term variation seems to me that the gauges are mounted in a way that the adhesive is not keeping them in the same position over an extended period of time. At least this is what I would consider after making sure to rule out temperature and voltage variations.