PWM signal not visible upon increasing the scope time base
The scope is running out of acquisition memory for the larger timebase and therefor has to reduce the sample rate. The reduced sample rate does not allow to capture the complete waveform of the fast PWM signal, because the Nyquist-Theorem is not fullfilled anymore. That's why the signal is displayed wrong.