Multimeter seems to be giving incorrect mA reading
You may have your meter set to the wrong range. The images in your post have been changed around a few times, but I've seen at least one showing a lead plugged into the 10A input, but the dial set to the mA range.

If you are using the 10A current input, you must set the meter to the 10A range. If you are using the general-purpose (V/Ω/Temp/Hz%/mA/µA) input, you must set the meter to the mA or µA range.