What cable standard is needed for UL94V-0 device
In IEC 61010, the requirement is


  Insulated wires and cables shall retard flame propagation.
  
  NOTE 2 Wire with a flammability RATING of UL 2556 VW-1 or equivalent is considered to meet this
  requirement.


This is part (b) of a paragraph where part (a) calls for PWB's to be rated V-2 or better (V-0 being better than V-2).

IEC 61010 is the IEC safety specification for test and measurement equipment. Your application probably requires meeting some other specification. Although many standards have similar requirements, you should check the standards appropriate for your application to see whether they match or not.