How to measure a 100 ohm differential signal on a spectrum analyzer?
3) The correct way to do your quick-n-dirty measurement is to terminate one lead with 50 ohms to ground, and measure the other with your 50 ohm instrument. In fact, this isn't the quick-n-dirty way, this is the most accurate way, as it avoids the signal degradation of the wound balun.