Regarding Passive Probes 1:1 and 10:1
You need to establish that the ratio of series resistor to scope input resistance is the same for the capacitors. If series resistor is 10x bigger than scope resistance then series cap needs to be one tenth of cable and input capacitance. If using microcap look at the spectrum of Vout and it should be flat.