Looking for a temperature probe connector
For a Pt100 in 2-wire or 3-wire configuration, connector resistance directly affects accuracy (specifically variation between contacts in connector resistance in the 3-wire case).  

Screw terminal blocks are commonly used, eg, 5mm or 5.08mm pitch. SMT is not really a good idea for something that invites an idiot with a screwdriver to torque on it, unless  you like field failures. Use through-hole parts, would be my recommendation.