Increasing Fall/Rise Times for Signal Integrity
The ringing probably isn't real. Note that you're measuring fast signals (keep an eye on the scope's timescale!), and then the inductance of the probe's ground becomes important. It doesn't help to connect the probe's ground a few mm from where you probe the signal if the wire forms a 20 cm loop. You may eliminate possible measurement problems like this using a much shorter ground connection: