Every time I connect ADE measuring chip , it blows up
If you look closely at the datasheet, you'll see that it plainly says the maximum signal you can apply to the amplifier inputs with respect to analog ground is +/- 1V (+/- 6V over voltage tolerance).

Your high-side current transducer is going to put a lot more common-mode stress than that on those pins, hence the blow-ups.

Consider a low-side sensor or an isolated transducer like a current transformer that can be referenced to the IC return.

Be safe - playing with the mains can be lethal. (I'm glad to see a fuse in your circuit!)