Peculiar input current
This is a meter error. If you look at the data sheet you'll see that the accuracy for AC current is 1.5% + 3 counts. However, if you look at the bottom of the page, you'll see that this is only specified over the range of 1% to 100% of the span. The smallest range in AC current is 4 amps, so any current less than 40 mA has no accuracy spec at all, and you simply can't trust it. 1.19 mA is less than 3% of 40 mA. 

Don't believe your meter when you are misusing it.