Used Fluke 87V multimeter working correctly?
Yes it is perfectly normal. Most measuring instruments need a little warm up time. This time allows electronics component to stabilise at a certain working temperature where the calibration is made. 10 seconds warm up seems about right.

As others said, it is impossible to measure a zero capacitance as there will always be a parasitic capacitance between leads and between the pcb traces of the instrument.