Graphing using vector signal analyzer
The measurement in the last picture is in the time domain, whilst the first two are in frequency domain. To have a signal analyzer checking signals in the time domain you have to set the SPAN of the measurement to 0. So, set the center frequency of the measurement to 83.685 kHz and the SPAN to 0 Hz and you should get the measurement you want.