OWON VDS1022i USB oscilloscope reading incorrect DC (and probably AC) voltages
I had the same thing Voltage was reading 40-50% high.
Solution:- use dedicated USB supply ( was sharing with phone ).
I assume the USB was not supplying a clean 5V, so the OWON was struggling with a low or dirty power supply.
Running perfectly now within the +/-3% now.