Can I make UV-Vis calibrant in the lab?
The last time I calibrated an absorbance spectrometer, it was one I made from parts (for an undergraduate Instrumental Analysis teaching laboratory) and I used a 0.0509 M holmium chloride solution I prepared. The visible region spectrum I obtained is here:



Holmium solutions are well known as UV-Vis calibration standards and you can purchase from, e.g., www.starnacells.com. Below is a screenshot from the Starna Cells web site, indicating that their ready-made sealed holmium reference standard cuvette cells have NIST traceability. Evidently, they are good down to 241.0 nm. I would have purchased one, if I had had the money at the time, but the solution I prepared worked well and was not particularly expensive. Erbium and praseodymium solutions also work well.

Sharp peaks in an aqueous solution at room temperature are always nice. Maybe this will help you!