{'fraction_non_alphanumeric': 0.02481902792140641, 'fraction_numerical': 0.0, 'mean_word_length': 5.088050314465409, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Continuous sampling air-ICP for metals emission monitoring

An air-inductively coupled plasma (air-ICP) system has been developed for continuous sampling and monitoring of metals as a continuous emission monitor (CEM). The plasma is contained in a metal enclosure to allow reduced-pressure operation. The enclosure and plasma are operated at a pressure slightly less than atmospheric using a Roots blower, so that sample gas is continuously drawn into the plasma. A Teflon sampling chamber, equipped with a sampling pump, is connected to the stack that is to be monitored to isokinetically sample gas from the exhaust line and introduce the sample into the air-ICP. Optical emission from metals in the sampled gas stream is detected and monitored using an acousto-optic tunable filter (AOTF)-echelle spectrometer system. A description of the continuous sampling air-ICP system is given, along with some preliminary laboratory data for continuous monitoring of metals.