{'fraction_non_alphanumeric': 0.04678362573099415, 'fraction_numerical': 0.042884990253411304, 'mean_word_length': 4.908045977011494, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[Evaluation of two automated methods for blood lactate levels].

We evaluated two automated methods for measuring blood levels of lactates with Integra 800 (Roche Diagnostics) and ABL 725 (Radiometer). Our evaluation had shown a within run imprecision of 1% (Intégra 800) and 2% (ABL 725) and a between-assay imprecision of 2% (Intégra 800) and 5% (ABL 725). The two methodologies appeared very well associated. The selective electrode remains expensive but it is very interesting because it saves a blood sample.