text: Spare 6V20W Halogen Bulb 
  1 
  
  1  Lens Cleaning Tissue 
  
  1  Cedar Oil 
  
  1  1A Fuse (spare) 
  
  1  Specification 
  
  Inspection Certificate  1 
  
  Packing List  1 
  
  **OPERATION**
  
  Note: For oil immersion, please use the index of refraction 1.515 oil 
  **Eyepiece Specifications**
  
  1. Remove all components from package. Identify all parts before 
   assembling instrument. 
  2. Attach 4x, 10x and 40x objectives by screwing into revolving 
   turret. Tighten and secure to maximum finger pressure only. 
  3. Place the specimen on the stage and secure with spring clips. 
   NOTE: The cover glass must face upward (the thinner glass is 
   the cover glass), otherwise when the 40x objective is used the 
   specimen cannot be observed. Observation is best when the 
   thickness of the cover glass is 0.1-1.1mm and the cover glass 
   is 0.17mm. 
  4. Plug power cord into an electrical outlet. Turn microscope 
   lamp ON. 
  5. Observe the specimen using the lowest magnification objective 
   first. The 10x objective provides a larger field of view making it 
   easier to search the specimen. 
  
  **Classification Magnification**
  **Field of View (FOV)**
  **Diameter**
  
  Plain Field 
  Eyepiece 
  10x  18mm 