text:
Single-Needle Cross-Over System. Brief description
- Alyson Shepherd
- 3 years ago
- Views:
1 Single-Needle Cross-Over System Brief description
2 Introduction Brief description of the Single-Needle Cross-Over system from B. Braun Unique pressure-pressure controlled SN process Constant blood flow as a result of simultaneously operating pumps Virtually constant pressure in the dialyzer Minimum recirculation Low impact on shunt as a result of indirect blood transport More than 15 years of market experience The Single-Needle Cross-Over (SN-CO) name is derived from the control s special feature: When reaching a certain control pressure, the opposite tube clamp closes and the respective other phase begins. Hence, the clamps control is carried out crosswise: Cross-Over. 2
3 Function description We distinguish between the arterial and venous phases: Arterial phase The arterial clamp is open, the venous clamp is closed. Blood flows from the patient into the arterial expansion chamber and then on through the dialyzer into the venous expansion chamber. Blood collects in the venous expansion chamber as the venous tube clamp is closed; the venous pressure rises. When a certain venous control pressure is reached, the clamps are switched: the arterial clamp is closed and the venous clamp is opened. The venous phase begins. Function description Venous phase The venous pressure falls again. Blood is released from the arterial expansion chamber and flows through the dialyzer into the venous expansion chamber and back to the patient. As the arterial clamp is closed, the arterial pressure falls. When a certain arterial control pressure is reached, the clamps are switched: the venous clamp is closed and the arterial clamp is opened. The arterial pressure rises again. A new arterial phase begins. 3
4 Function description Setup Insert the arterial and venous tube system: Feed the lines leading to the patient through the corresponding clamps. Do not yet insert the tube segment into the second (venous) blood pump. Ensure that the second pump segment is above the pump. Connect the PBS pressure sensor and the other pressure sensors. The PBS pressure sensor should always be connected prior to the blood-side self test. If the pressure sensor needs to be connected following the start of treatment, the PBS values displayed must be confirmed. Fill PV PA Level in the venous expansion chamber Level in the arterial expansion chamber Set the saline level in the arterial and venous chambers as shown. Enter patient's parameters as usual. 4
5 Handling Connect patient Switch to treatment mode via the Connect patient icon. Confirm patient data! Start the blood pump, fill the system with blood and finally connect the venous line to the patient. Device settings Touch the Parameter icon and the SN icon. Handling Select SN-CO. The pre-set limit values for the arterial and venous control pressures set themselves automatically. Recommendations for the control pressures Optimum control pressures: Arterial mmhg Venous mmhg Central catheter / up to to 390 Good fistula Delicate fistula up to Initial punctures -120 to to 300 Insert the second pump segment into the venous pump. Start the blood pump. 5
6 Handling Recommendation for the blood pump speed When connecting to a fistula, start the blood pump with ml/min and increase slowly. Final speed depending on the condition of the blood vessel. When connecting a central venous catheter, start with 150 ml/min, increase slowly. Final speed ml/min. Target value for phase volume appr ml. Checking the phase volume, the blood flow and the pressure values. You also have the option to select SN-CO even before connecting the patient. End During end of therapy leave the second pump segment in the venous blood pump. 6
7 Alarms and warnings Pressures Monitoring the pressures in the blood vessels is carried out using a time window. Reaching the pressure compensation in the expansion chambers must take place within a specific time; the system will otherwise trigger an alarm. To this end, the blood pump speed has an influence on the time window. "Arterial pressure lower limit" Does the patient s blood access offer insufficient conditions? Check the blood lines (kink?). Reduce blood pump speed. Possibly measure the blood pressure / change patient s position. If the measures do not result in the desired effect, a mechanical blockade (i.e. suction of the cannula / of the catheter) may be the cause. Increase the arterial control pressure toward zero (e.g. from -180 to -150 mmhg) to reduce the suction on the tip of the cannula / catheter. Check, and if necessary, correct the position of the cannula / catheter. Monitor the phase volume. If the phase volume increases, increase the blood pump speed step-by-step again. "Venous pressure upper limit" Reduce blood flow. Check the position of the patient / the catheter / the cannula. If necessary, reduce the venous control pressure. Alarms & warnings 7
8 Alarms and warnings Phase volume Monitoring the phase volume is carried out using relative and absolute limit values. A warning highlighted in yellow appears on reaching the relative limit values. The limit values can be readjusted by activating the blood pump (1x +, 1x -). Reaching an absolute limit value triggers an alarm (red). "Phase volume too low" Reduce blood flow and, if necessary, extend control pressures. If necessary, correct the cannula / catheter position. "Phase volume too high" Important: check the tube system connections for air intake! Check whether the arterial patient line is inserted in the arterial clamp. Increase blood flow and, if necessary, decrease control pressures. Additional alarms "PBS too high" / "PBS too low" Short-term change of the blood flow. If necessary, end single-needle mode. If necessary, contact the technical service. 8
9 FAQs How does the blood transport work? The system autonomously adapts to the pressure and flow conditions in the blood vessel. The blood transport is carried out exclusively by pressure compensation in the expansion chambers. This is the result of the generated underpressures (arterial) or overpressures (venous) caused by the closed clamps. This ensures that the blood is not directly transported from the blood vessel, hence correspondingly relieving the shunt. What function does the PBS carry out? The PBS controls the venous blood pump by monitoring the pressure. This prevents pressure fluctuations in the dialyzer, permitting virtually constant TMP. The pressure sensor measures the pressure at the dialyzer outlet and sets the blood pump speed taking the ultrafiltration into account so that there is no pressure drop at the dialyzer membrane. This means that, depending on the ultrafiltration volume, the venous blood pump runs more slowly than the arterial blood pump and hence counteracts a pressure drop in the dialyzer. What effect do the control pressures have? Upon reaching the set control pressures, the clamps are opened or closed. Here, the arterial control pressure closes the venous clamp and vice-versa. IMPORTANT: the pressure value displayed does not as in doubleneedle mode show the pressure conditions in the blood vessel, it shows the control pressure. FAQs 9
10 FAQs What is the phase volume and what value should it reach? The phase volume is the volume that is transported into the expansion chambers per cycle. To achieve sufficient dialysis efficiency, the volume should reach at least 30 ml. The phase volume sets itself automatically, but is dependent of - the blood flow / blood pump stop - the control pressures - the height of the levels in the chambers - pressure changes in the shunt. What are the benefits of the system? As both pumps operate at the same time, a constant flow is achieved in the dialyzer and pressure fluctuations are extensively avoided. The shunt is relieved as a result of indirect pressure compensation. Minimisation of the tube system compliance as a result of the arterial and venous clamps and hence extremely low recirculation. Generally a higher cumulated blood volume than in the case of other systems. 10
11 Notes 11 Notes
12 These brief instructions are not a substitute for reading the detailed instructions B. Braun Avitum AG Schwarzenberger Weg Melsungen Germany Tel Fax W /1 No A Edition: 02/2010
The Hemodialysis Machine Case Study
The Hemodialysis Machine Case Study Atif Mashkoor Software Competence Center Hagenberg GmbH Hagenberg, Austria firstname.lastname@example.com 1 Introduction This documents presents a description of a case studyMore information
biologic RR Comfort Creating dimensions in intelligent blood pressure stabilization Haemodialysis
biologic RR Comfort Creating dimensions in intelligent blood pressure stabilization Haemodialysis Intelligent Dialysis Technology Creating dimensions in treatment quality With the Dialog+ dialysis systemMore information
Dialog + Dialysis Machine. Instructions for Use: Software Version 9.1x
Dialog + Dialysis Machine Instructions for Use: Software Version 9.1x CE marking according to directive 93/42/EEC Technical alterations reserved Rx only Manufacturer: B. Braun Avitum AG 34209 Melsungen,More information
B. Braun Space GlucoseControl (SGC)
B. Braun Space GlucoseControl (SGC) Quick Reference Guide Valid for software I9305_C Automated Infusion Systems Starting a new insulin therapy SpaceControl cannot be switched on and off by itself. It isMore information
Ultrafiltration Devices
6 21 CFR 876.5820, Hemodialysis system and accessories. 7 21 CFR 876.5860, High permeability hemodialysis system. 10 21 CFR 876.5540, Blood access device and accessories. 8 21 CFR 876.5540, Blood accessMore information
CARDIOHELP INTERFACE PRESENTER (CIP) USER GUIDE GENERAL INFORMATION ABOUT THE CIP: The CARDIOHELP Interface Presenter is a sales support tool. It gives sales staff the chance to get to know the new CARDIOHELPMore information
The MEDOS VAD System
The MEDOS VAD System On the market since 1994 more than 500 operations in about 80 heart centres short and medium therm use one of the leading systems in Europe Application field not as a matter of routine,More information
Protocol for New AVF Cannulation
Protocol for New AVF Cannulation Protocol for New AVF Cannulation Define successful cannulation Cannulation guidelines New AVF Mature AVF Unsuccessful cannulations Detailed instructions for complicationsMore information
The Dialog+ Dialysis System
The Dialog+ Dialysis System Creating new dimensions in treatment quality Haemodialysis B. Braun Avitum B. Braun has been a trusted partner in the supply of high quality, cost effective dialysis productsMore information
CRRT with Prismaflex LEADS TO More Flexibility, Ease of Use and Safety
CRRT with Prismaflex LEADS TO More Flexibility, Ease of Use and Safety Leading the way Science is changing and so are your requirements For many years, Gambro has focused its efforts on development ofMore information
Thopaz Quick Reference Guide. Precious life Progressive care
Thopaz Quick Reference Guide Precious life Progressive care How to prepare Thopaz for use single double 0.3 L 0.8 L Select single or double tubing and ONE of the following canisters (0.3 L, 0.8 L) in OpenMore information
Joline - a modern medical technology company built on tradition. For further information please contact us!
Long Term Joline - a modern medical technology company built on tradition has built on a long tradition in the development and production of dialysis catheters and has been manufacturing medical productsMore information
with FLASH TIPS PLUM A+ INFUSION SYSTEM OPERATING INSTRUCTIONS REFER TO SYSTEM OPERATING MANUAL FOR COMPLETE OPERATING INSTRUCTIONS
with FLASH TIPS CONTENTS Definitions....................................2, 3, 4 Setup PlumSet...................................5 Clinical Care Area...................................5 Simple DeliveryMore information
Module 3 Training for the HomeChoice SmartCare (version 10.4) Software. Participant s Workbook
Module 3 Training for the HomeChoice SmartCare (version 10.4) Software Participant s Table of Contents Learning Objectives...2 Topics Covered in this Module...3 Chapter 1: Software Update Overview...4More information
Wyoming STATE BOARD OF NURSING
David D. Freudenthal Governor Wyoming STATE BOARD OF NURSING Mary Kay Goetter, PhD, RNC, NEA-BC Executive Director 1810 Pioneer Avenue Cheyenne, Wyoming 82002 Phone: 307-777-7601 FAX: 307-777-3519 http://nursing.state.wy.usMore information
Design of an Insulin Pump. Purpose of an Insulin Pump:
Design of an Insulin Pump Purpose of an Insulin Pump: Insulin is a hormone central to regulating carbohydrate and fat metabolism in the body. It is secreted regularly within the body and aids in convertingMore information
PICC & Midline Catheters Patient Information Guide
PICC & Midline Catheters Patient Information Guide medcompnet.com 1 table of contents Introduction 4 What is a PICC or Midline Catheter? 4 How is the PICC or Midline Catheter Inserted? 6 Catheter CareMore information
Laerdal Patient Monitor Help Page 1 June 14, 2012, Rev E
Laerdal Patient Monitor Help Page 1 Using the Laerdal Patient Monitor The Laerdal Patient Monitor software is used to simulate a typical Patient Monitor found in hospitals and ambulances. It is made availableMore information
Title/Description: Nursing Care of the Patient on Continuous Renal Replacement Therapy (CRRT)
University of Kentucky / UK HealthCare Policy and Procedure Policy # NE08-06 Title/Description: Nursing Care of the Patient on Continuous Renal Replacement Therapy (CRRT) Purpose: To provide directionMore information
Medications or therapeutic solutions may be injected directly into the bloodstream
Intravenous Therapy Medications or therapeutic solutions may be injected directly into the bloodstream for immediate circulation and use by the body. State practice acts designate which health care professionalsMore information
Prismaflex Schematic
Prismaflex Machine Prismaflex Schematic Communication Unit Give a general indication of operating conditions. Status Lights Green Indicates all parameters are normal during treatment YellowIndicates advisoryMore information
Understanding Your. Hemodialysis. Access Options
Understanding Your Hemodialysis Access Options aakp American Association of Kidney Patients 3505 E. Frontage Road, Suite 315 Tampa, FL 33607 Toll-Free: 800-749-2257 Fax: 813-636-8122 www.aakp.org OverviewMore information
How to design an insulin pump
How to design an insulin pump Learn about the purpose of an insulin pump, its overall workings, and the requirements needed for its design as well as implementation. By Asha Ganesan Applications EngineerMore information
Heat. Investigating the function of the expansion valve of the heat pump. LD Physics Leaflets P22.214.171.124. Thermodynamic cycle Heat pump
Heat Thermodynamic cycle Heat pump LD Physics Leaflets P188.8.131.52 Investigating the function of the expansion valve of the heat pump Objects of the experiment g To study the operational components of theMore information
fabian +ncpap evolution
fabian +ncpap evolution Quick guide TO THE OPERATOR AND PERSON IN CHARGE OF MAINTENANCE AND CARE OF THE UNIT: This Quick Guide is not a substitute for the Operation Manual. Read the Operation Manual carefullyMore information
Within the Scope of Practice/Role of APRN RN _ X_LPN CNA
Wyoming State Board of Nursing 130 Hobbs Avenue, Suite B Cheyenne, WY 82002 Phone (307) 777-7601 Fax (307) 777-3519 E-Mail: firstname.lastname@example.com Home Page: https://nursing-online.state.wy.us/ OPINION:More information
Fluid Management DYONICS 25. DYONICS 25 Tube Sets. Fluid Management System. Easy set up and use. High flow. Consistent performance
DYONICS 25 Fluid Management Fluid Management System Easy set up and use High flow Consistent performance DYONICS 25 Tube Sets A variety of tubing options are available, including cost-effective day tubingMore information
Infusion Nurse Training Manual For the Invisiport Product Line
Stealth Therapeutics, Inc. Infusion Nurse Training Manual For the Invisiport Product Line Stealth Therapeutics, Inc. 101 Nob Hill Road, Suite 100 Madison, WI 53713 (608) 217-2685 firstname.lastname@example.com information
Intravenous Infusion Lines Enhancing Patient Safety During IV Therapy
Intravenous Infusion Lines Enhancing Patient Safety During IV Therapy Key to product features Feature Benefit Key ASV Minimises the risk of downloading ARV Minimises the risk of backtracking Clamp ForMore information
Policy and Procedure Flushing and / or Blood withdrawal Aspiration Procedure For PICC Line and Midline Catheters
Policy and Procedure Flushing and / or Blood withdrawal Aspiration Procedure For PICC Line and Midline Catheters Purpose: Blood Withdrawal: To obtain blood samples for laboratory evaluation, eliminatingMore information
ELECTRIC CENTRAL HEATING FLOW BOILER EKCO.T Used product can t be treated as general communal waste. Disassembled appliance has to be delivered to the collection point of electrical and electronic equipmentMore information
Quick Reference Guide. Control Panel
v.a.c.simplicity TM KCI Active Healing by Design TM Quick Reference Guide Power Connection Battery Charging Indicator Light Control Panel Canister Please consult the V.A.C. Simplicity User Manual for furtherMore information
Patient Safety by Fresenius Medical Care Where new benchmarks are set
Cardioprotective Haemodialysis Patient Safety by Fresenius Medical Care Where new benchmarks are set Cardioprotective Haemodialysis Protect your Patient Cardioprotective Haemodialysis Wide-ranging cardioprotectionMore information
with O 2 Controller Instruction Manual
14500 Coy Drive, Grass Lake, Michigan 49240 734-475-2200 E-mail: email@example.com www.coylab.com Hypoxic Cabinets (In-Vivo / In-Vitro) with O 2 Controller Instruction Manual Index Page Warranty 2 WarningsMore information
For Treatment of Protein Malnutrition in Peritoneal Dialysis Patients, in a Glucose- Free, Phosphate-Free Exchange.
PD SOLUTIONS Solutions for Life... A unique combination of solutions designed to optimize patient therapy outcomes Dianeal Baxter s Premier Solution used Globally for over 25 Years by Millions of Patients.More information
Nursing Management of Venous Access Devices: Peripheral IV lines
Nursing Management of Venous Access Devices: Peripheral IV lines Mimi Bartholomay, RN, MSN, AOCN Denise Dreher,, RN, CRNI, VA-BC Theresa Evans, RN, MSN Susan Finn, RN, MSN, AOCNS Debra Guthrie, RN, CRNIMore information
Electronic Pressure Switch EDS 300
Electronic Pressure Switch EDS 300 User manual Page 2 of 16 Content 1. Functions of the EDS 300...3 2. Mounting...3 3. Operating keys on the membrane keypad...4 4. Digital display...4 5. Output function...5More information
Automation and Systems Technology. LABCONTROL TFM/TPM User Manual TroxMConnect Operating Software. 1. Introduction 1. 2. Software Installation 2
1. Introduction 1 2. Software Installation 2 2.1 System Requirements 2 2.2 TroxMConnect Installation 2 2.3 Connection to the PC 3 3. TroxMConnect Description 4 3.1 Starting the Software 4 3.2 Layout 4More information
Procedure 36 Assembling And Priming A Basic Administration Set. Procedure 42 Applying A Transparent Film Dressing To An Intravenous Infusion Site
Chapter 7 Intravenous Therapy Procedure 36 Assembling And Priming A Basic Administration Set Procedure 37 Inserting A Peripheral IV In An Adult Procedure 38 Inserting A Heparin Lock Procedure 39 InsertingMore information
Everest Healthcare Services Corporation August 4, 2000 WHAT TO DO WHEN THE URR IS UNDER 65 (See Model: Low URR Checklist adapted from Midwest Kidney Centers, Peoria, IL and presented in an Excel spreadsheetMore information
SIGMA Spectrum Infusion System Compatible Baxter IV Sets
SIGMA Spectrum Infusion System Compatible Baxter IV Sets (For use with SIGMA Spectrum Pumps that have been calibrated for Baxter IV set tubing). NOTE: The superscript numbers referenced in the CompatibleMore information
02 DEPARTMENT OF PROFESSIONAL AND FINANCIAL REGULATION 380 BOARD OF NURSING Chapter 10: REGULATIONS RELATING TO ADMINISTRATION OF INTRAVENOUS THERAPY BY LICENSED NURSES SUMMARY: This chapter identifiesMore information
LEARNING MODULE FOR PERIPHERAL I.V. THERAPY INITIATION AND MAINTENANCE CC 80-019 Developed by: Jackie Jayasinghe, Nurse Educator Medicine/Ambulatory Care Nursing Date: April, 1998 Krista Newlands, NurseMore information
5.2. MEASUREMENT OF BLOOD PRESSURE Purpose of experiment To record the pressure changes in the vascular system. To set the acoustic signal voltage and time parameters. To calculate the systolic and diastolicMore information
Electrical connection
(I)GSU 14B Ultrasonic Label Fork Dimensioned drawing Part No. 501 09234 4mm 12-30 V DC 4 m/s Ultrasonic forked sensor for universal application Large mouth width, hence also suitable for booklets or fan-foldMore information
ARROW AutoCAT 2 WAVE IABP Abbreviated Operation and Troubleshooting Guide
ARROW AutoCAT 2 WAVE IABP Abbreviated Operation and Troubleshooting Guide ARROW 24-HOUR INTRA-AORTIC BALLOON PRODUCT SUPPORT: US and Canada (800)-447-IABP (800)-447-4227 Modes of Operation AutoPilot PumpMore information
Suffering from varicose veins? Patient Information. ELVeS Radial Minimally invasive laser therapy of venous insufficiency
Suffering from varicose veins? Patient Information ELVeS Radial Minimally invasive laser therapy of venous insufficiency Do you suffer from heavy legs or visible veins? This makes diseases of the veinsMore information
Tunneled Hemodialysis Catheters: Placement and complications
Tunneled Hemodialysis Catheters: Placement and complications Arif Asif, M.D. Director, Interventional Nephrology Associate Professor of Medicine University of Miami, FL Tunneled Hemodialysis Catheters:More information
3-2. Using the air conditioning system and defogger Automatic air conditioning system
Automatic air conditioning system Airflow and outlets are automatically adjusted according to the temperature setting. Press to display the air conditioning operation screen. Air conditioning operationMore information
Measuring central venous pressure
Elaine Cole Senior lecturer ED/Trauma, City University Barts and the London NHS Trust 1 Learning outcomes That the clinician can: Describe the sites of central venous catheterisation Understand why centralMore information
Regents of the University of Michigan 5/7/2015. Review pump functionality, sets and accessories Understand key process steps for
April 2015 Goals Review pump functionality, sets and accessories Understand key process steps for Starting an infusion Stopping an infusion Changing a container Re-programming the pump TroubleshootingMore information
Introcan Safety. 1 billion times protection. Vascular Access
Introcan Safety 1 billion times protection Vascular Access The B. Braun Introcan Safety IV Catheter Reduces Needlestick Injuries Passive Safety Technology Established worldwide: B. Braun has minimizedMore information
Nursing Education and Research
Melissa Meloche Meloche, RN RN, MSN MSN, CCRN Nursing Education and Research Describe the purpose p of common clinical equipment found in the Intensive Care Unit and how this equipment could impact a patientMore information
Power Injectors for Diagnostic Imaging
Power Injectors for Diagnostic Imaging Contrast under Control A resource for all your contrast delivery needs Contrast under Control 2 1 3 Mallinckrodt provides contrast delivery systems for ComputedMore information
2008K Hemodialysis Machine Operator s Manual
2008K Hemodialysis Machine Operator s Manual P/N 490042 Rev O 2008K Hemodialysis Machine Operator s Manual Copyright 2000-2015 Fresenius USA, Inc. All Rights Reserved This document contains proprietaryMore information
Plasmax Plasma Concentration System BIOLOGICS
Plasmax Plasma Concentration System BIOLOGICS Plasmax Plasma Concentration System Plasmax Plasma Concentrate System Contains Plasmax concentrator GPS III separator Provides 16ml of autologous output, 6mlMore information
RD547 water leak detector
Quick Start User Guide RD547 water leak detector 90/UG100INT/01 ISSUE 1 11/2014 Important: This guide provides basic operation instructions and is intended as a field reference only. It also contains importantMore information
EL-USB-2-LCD FEATURES WINDOWS CONTROL SOFTWARE DISPLAY FUNCTIONS. www.lascarelectronics.com Page 1 of 5
This standalone data logger measures and stores up to 16,379 relative humidity and 16,379 temperature readings over 0 to 100%RH and -35 to +80 C (-31 to +176 F) measurement ranges. The user can easilyMore information
Paediatric Intensive Care Unit Nursing Procedure: Care of Arterial Lines.
Paediatric Intensive Care Unit Nursing Procedure: Care of Arterial Lines. Definition: Arterial Line Placement of an indwelling arterial catheter for the purpose of continuous monitoring of intra arterialMore information
ECMO (Extracorporeal Membrane Oxygenation)
ECMO (Extracorporeal Membrane Oxygenation) ECMO stands for Extracorporeal Membrane Oxygenation. It is a type of life support that uses a machine to pump blood rich in oxygen to support the heart or lungs,More information
All About Your Peripherally Inserted Central Catheter (PICC)
All About Your Peripherally Inserted Central Catheter (PICC) General Information Intravenous (IV) therapy is the delivery of fluid directly into a vein. An intravenous catheter is a hollow tube that isMore information
Instruction and Maintenance Manual
Instruction and Maintenance Manual GRYF OXY P 0/100/2 PM Contact GRYF HB, spol. s r.o. Cechova 314 Havlickuv Brod 580 01 tel.: +420 569 426 627 fax: +420 569 426 627 Czech Republic www.gryf.eu TechnicalMore information
COOLING THERAPY SYSTEM Q U I C K R E F E R E N C E G U I D E Garment Models Description and Operation Controller Troubleshooting The Quick Reference Guide is a guideline only and does not replace the User'sMore information
GYN 46 5.0 01/2016-E HYSTEROMAT E.A.S.I. Endoscopic Automatic System for Irrigation
GYN 46 5.0 01/2016-E HYSTEROMAT E.A.S.I. Endoscopic Automatic System for Irrigation NEW Easy The HYSTEROMAT E.A.S.I. is an intelligent, pressure-controlled double roller pump that maintains a constantMore information
Treatment Options: Hemodialysis
Patient Education Chapter 3 Page 1 Treatment Options: Hemodialysis Objectives: 1. Understand purpose and care of blood access. 2. Understand purpose and basic principles of hemodialysis. 3. UnderstandMore information
Christian In better control with his pump since 2012 WHAT CAN I DO TO REDUCE MY RISK OF DEVELOPING THE COMPLICATIONS OF TYPE 1 DIABETES? Many people with Type 1 diabetes worry about potential long-termMore information
The PRISMAFLEX System. Making possible personal.
The PRISMAFLEX System Making possible personal. TPE on the PRISMAFLEX System The PRISMAFLEX System can help you meet the demands of multiple therapies with a versatile platform that can be customized toMore information
THE PRISMAFLEX SYSTEM The Prismaflex Control Unit is intended for: Continuous Renal Replacement Therapy (CRRT) for patients weighing 20 kilograms or more with acute renal failure and/or fluid overload.More information
GE Healthcare. Aespire 7100. Essential performance Compact design
GE Healthcare Aespire 7100 Essential performance Compact design Clinician-inspired solutions help improve patient care Optimized work environment The Aespire 7100 was developed using our unique approachMore information
Understanding Your Hemodialysis Access Options UNDERSTANDING YOUR HEMODIALYSIS ACCESS OPTIONS
Understanding Your Hemodialysis Access Options UNDERSTANDING YOUR HEMODIALYSIS ACCESS OPTIONS What is a vascular access? A vascular access is a place on your body close to a vein and artery. This placeMore information
VAG 1687 Diagnostic Tool, Checking Intake System for Leaks. Condition. Service 00-A354 00-A353
Subject: Model(s): VAG 1687 Diagnostic Tool, Checking Intake System for Leaks All with 1.8L Turbo All with 2.7L BiTurbo 1997 2000 Group: Number: Date: 01 01 01 Oct. 26, 2001 Condition VAG 1687 DiagnosticMore information
AKCess Pro Server Acknowledge Alerts & Sensor Status Manual
www.akcp.com AKCess Pro Server Acknowledge Alerts & Sensor Status Manual Copyright 2015, AKCP Co., Ltd. Table of Contents Introduction... 3 The example SEC configuration... 4 Acknowledging sensor alerts...More information
Section 6: Your Hemodialysis Catheter
Section 6: Your Hemodialysis Catheter What you should know about your dialysis catheter How to change your catheter TEGO connectors Starting dialysis using a catheter End of dialysis using a catheter ChangingMore information
INTERVENTIONAL RADIOLOGY: VASCULAR DIAGNOSTIC AND THERAPEUTIC PROCEDURES An important skill for the coder of vascular diagnostic and therapeutic procedures to possess is the ability to read and understandMore information
Understanding Treatment Options for Renal Therapy
Understanding Treatment Options for Renal Therapy Kidney failure happens to many different people for many different reasons 1 What causes kidney failure? Diabetes Chronic infection High blood pressureMore information
Alarm Monitoring Basics
Alarm Monitoring is a BASIS application designed for the monitoring of Access Control devices, their status and usage. It can be used to check if a device is having problems, if there is an alarm, suchMore information
Checklist for Insertion and Setup of an ICP Monitoring Device
Checklist for Insertion and Setup of an ICP Monitoring Device Before proceeding, contact neurosurgery to identify the type of catheter to be inserted. The catheter type will define the equipment requirementsMore information
Automatic air conditioning system
Automatic air conditioning system Airflow and outlets are automatically adjusted according to the temperature setting. With navigation system Owners of models equipped with a navigation system should referMore information
1. To provide the patient with an open air passage by preventing encrustation and obstruction.
I. Care of the Patient with A Tracheostomy A. Purpose: 1. To provide the patient with an open air passage by preventing encrustation and obstruction. 2. To aid in bronchial hygiene by facilitating removalMore information
Insulin Delivery System. Reference Guide
Insulin Delivery System Reference Guide Reference Guide As an important member of our family, we would like you to know we understand the importance of managing your diabetes. We care about your healthMore information
CONTENTS. What is ROBOTC? Section I: The Basics
BEGINNERS CONTENTS What is ROBOTC? Section I: The Basics Getting started Configuring Motors Write Drive Code Download a Program to the Cortex Write an Autonomous Section II: Using Sensors Sensor SetupMore information
Home Hemodialysis & Peritoneal Dialysis
Home Hemodialysis & Peritoneal Dialysis What do you already know about kidney disease? Kidney disease Non institutionalised adults: 4.4 million people world wide who suffer from kidney disease, or, 1.9%More information
Device Equivalence Evaluation Form
Comparison of the Beurer (Sanitas) SBM 45 with the Andon KD 5915 Devices Beurer (Sanitas) SBM 45 Andon KD 5915 Pictures Display Validation ESH Device 1 Criteria Buttons/Switches Touch Screen ActivationMore information
CFMRI Physiological Monitoring System. Operator s Manual
CFMRI Physiological Monitoring System Operator s Manual This manual provides a quick guidance to using the physiological monitoring system during an MRI session. It is not intended as a substitute forMore information
VersaCare Bed. Quick Reference Guide
VersaCare Bed Quick Reference Guide Bed Features Item Description Page A Emergency CPR and Emergency 3 Trendelenburg a Control B Battery Back-Up 4 C Foot Controls 4 D Point-of-Care Caregiver Control PanelMore information
ENAiKOON ultrasonic fuel-level sensor
ultrasonic fuel-level sensor Product Summary The ultrasonic fuel-level sensor is the perfect solution for monitoring the fuel level and fuel consumption of trucks, boats, construction equipment, generators,More information
Intra-aortic Balloon Pump (IABP)
Intra-aortic Balloon Pump (IABP) Information for patients of the TGH Coronary Intensive Unit and their families Read this information to learn: how your heart works how an IABP can help your heart whatMore information
X-Plain Subclavian Inserted Central Catheter (SICC Line) Reference Summary
X-Plain Subclavian Inserted Central Catheter (SICC Line) Reference Summary Introduction A Subclavian Inserted Central Catheter, or subclavian line, is a long thin hollow tube inserted in a vein under theMore information
Insulin Pump Management and Continuous Glucose Monitoring Systems (CGMS)
Insulin Pump Management and Continuous Glucose Monitoring Systems (CGMS) Faith Daily, RN, BSN, CDE, CPT Certified Diabetes Educator/Insulin Pump Trainer August 16, 2014 Why Pump Therapy? Mimics normalMore information
Maxi Pressurisation Units (Maxi, Maxi Plus & HP)
Maxi Pressurisation Units (Maxi, Maxi Plus & HP) Installation and Maintenance Instructions Contents 1 Dimensions 3 1.1 Installation 4 1.1 Mains/boosted cold water inlet 4 1.2 Connection to the system 5More information
9.0 Components Description. 10.0 SENTIO MMG System Quick Setup. 10.1 Plugging in Control Unit. 10.2 Attaching Sensor Harness
9.0 Components Description The main components of the SENTIO MMG System consist of the SENTIO MMG Control Unit with preinstalled SENTIO MMG Software and the SENTIO MMG Sensor Harness. The SENTIO MMG ControlMore information
Peripheral Long Line Catheter or Long Line Peripherally Inserted Central Catheter or P.l.C.C.
Peripheral Long Line Catheter or Long Line Peripherally Inserted Central Catheter or P.l.C.C. A Peripheral Long Line Catheter ( Long Line ) and a Peripherally Inserted Central Catheter ( P.I.C.C. ) areMore information
Before installing the Zebra printer with USB connector, download the drivers from www.zebra.com.
Download the Zebra Drivers 1 Download the Zebra Drivers Before installing the Zebra printer with USB connector, download the drivers from www.zebra.com. Note: You must first create an account on Zebra.comMore information
SELEMION SELEMION ION EXCHANGE MEMBRANES Safety Warning Before using the products, please be sure to read and observe the cautions printed in the user's manual. Sole Agent Membrane Division WBG MariveMore information
Sterilization & Infection Control Products
Sterilization & Infection Control Products L a b o r a t o r y L i n e A u t o c l a v e s Introduction High performance. Attractive design. Favorable price. Vertical and benchtop autoclaves, 7.5 to 60More information
Patient Transport. with the Automated Impella Controller
Patient Transport with the Automated Impella Controller TABLE OF CONTENTS PURPOSE OF THESE INSTRUCTIONS... 2 INDICATION FOR USE... 2 FAA ADVISORY... 2 ABOUT THE IMPELLA SYSTEM... 3 PATIENT TRANSPORT WITHMore information
User Tutorial on Changing Frame Size, Window Size, and Screen Resolution for The Original Version of The Cancer-Rates.Info/NJ Application
User Tutorial on Changing Frame Size, Window Size, and Screen Resolution for The Original Version of The Cancer-Rates.Info/NJ Application Introduction The original version of Cancer-Rates.Info/NJ, likeMore information
PPM Users Manual Signature Software 01-12-00
PPM Users Manual Signature Software 0-2-00 PPM User Manual /8/02 Software Versions: 0.0.27 Contents. Introduction 2 2. Parameters 3 2. Overload Limit...4 2.2 Relative Upper Limit...4 2.3 Relative LowerMore information
Haemodialysis. Bloodlines, Tubing Systems, Accessories Product Range
Haemodialysis Bloodlines, Tubing Systems, Accessories Product Range 2010/2011 5008 Therapy System the innovation Enabling the application of ONLINE Haemodiafi ltration in the most practical way. ONLINEMore information
SUBCUTANEOUS THERAPY A. ADMINISTERING SUBCUTANEOUS MEDICATIONS INTERMITTENTLY/CONTINUOUSLY B. (SUBCUTANEOUS INFUSION) HYDRODERMOCLYSIS PARTS I. Purposes II. General Information III. Responsibilities IV.More information


text:
This post is dedicated to my best friend Elvira who is the designer of this blog and who also happens to have her birthday today! Happy Birthday my beloved!!!
We had some really spontaneous moments the other day and aren't these photos quite the proof? :p
T-shirt: H&M, Shorts and Necklace: Zara, Flip flops: Haralas

subset: commoncrawl