text:
|Publication number
|Publication type
|Application number
|Publication date
|Oct 11, 1994
|Filing date
|Oct 25, 1993
|Priority date
|Jun 12, 1991
|Fee status
|Also published as
|Publication number
|Donald K. Cohen, Paul J. Caber, Chris P. Brophy
|Original Assignee
|Wyko Corporation
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (5), Non-Patent Citations (6), Referenced by (109), Classifications (9), Legal Events (9)
|External Links: USPTO, USPTO Assignment, Espacenet
This is a continuation of patent application Ser. No. 08/048,747 filed Apr. 16, 1993, by Donald K. Cohen et al., and entitled "ROUGH SURFACE PROFILER AND METHOD", which is a continuation of patent application Ser. No. 07/880,083 filed May 6, 1992 by Donald K. Cohen et al., and entitled "ROUGH SURFACE PROFILER AND METHOD" now U.S. Pat. No. 5,204,734 which issued on Apr. 20, 1993, which is a continuation of patent application Ser. No. 07/714,215 filed Jun. 12, 1991 by Donald K. Cohen et al., and entitled "ROUGH SURFACE PROFILER AND METHOD", now U.S. Pat. No. 5,133,601 which issued on Jul. 28, 1992.
The invention relates to an interferometric measuring device capable of profiling a surface with large height variations.
Conventional phase shifting interferometers require that the surface of an object being profiled be quite smooth, so that continuous interference fringes are produced by it. A large step change (i.e., a quarter of a wavelength of the light used to make the measurement or more) in the height of the surface often destroys the continuity of interference fringes, and consequently conventional phase shifting algorithms executed by a computer in response to fringe intensity data produced by a solid-state imaging array, such as a CCD array, are unable to accurately compute the profile of the surface.
At the present time, measurement of accurate profiles of surface areas is limited to RMS average roughness of approximately one thousand Angstroms using single wavelength interferometric techniques. Using multiple wavelength techniques (such as those described in commonly assigned U.S. Pat. No. 4,832,489, issued May 23, 1989, to Wyant et al.), surfaces with approximately one micron average roughness may be measured. With single wavelength techniques, the present state of the art limits measurement to surface step features of no greater height than approximately 0.16 microns. With multiple wavelength techniques, step height measurements are limited to steps less than approximately 15 microns in height.
U.S. Pat. No. 4,818,110 (Davidson) discloses a Linnik Microscope in combination with a video camera, a wafer transport stage, and data processing electronics, based on the use of an interference microscope to measure height and width of surface features on an integrated circuit. However, this reference does not disclose pixel-by-pixel mapping of the surface of a sample, does not generate a profile, and is incapable of generating an accurate pixel-by-pixel area profile of a surface that is too "rough" to be measured by conventional interferometry.
The article "Profilometry with a Coherent Scanning Microscope", by Byron S. Lee and Timothy C. Strand, Applied Optics, Volume 29, No. 26, Sep. 10, 1990, discloses a "coherence scanning microscope" in which an object is scanned in the z direction. White light interference fringes that result from the scanning are demodulated to find the peak amplitude of an envelope of the fringes to determine the value of z at the peak interference fringe. The Lee and Strand paper discloses no specific way of demodulating the fringes, and indicates that ambiguities introduced by phase change on reflection due to dissimilar materials renders the technique inoperable. No interpolation techniques or curve fitting techniques that might improve accuracy are disclosed. Optical path difference increments apparently are limited by the step size of stepper motors used, as is the speed of incrementing. The disclosed profile data is two-dimensional, rather than three-dimensional. The Lee and Strand reference clearly does not teach a technique to accomplish fast, highly accuracy surface profiling of surfaces having wide ranges of smoothness or roughness, or of dealing with phase ambiguity errors that result from phase change on reflection due, for example, to dissimilar surface materials.
Although phase-shifting techniques can produce measurements of surface roughness of the order of one thousandth of a wavelength, most present methods detect phase modulo 2π, and consequently give rise to errors sometimes referred to as "2π ambiguities" but hereinafter referred to as "phase ambiguities" or "phase ambiguity errors". Various kinds of "phase unwrapping" algorithms are used to track the phase over a large range of surface heights and resolve the phase ambiguity errors. Problems arise when there is a height variation between two adjacent pixels that cannot be unambiguously "unwrapped". The result is an integration error that usually manifests itself as a streak across the field of view.
It is well known that different materials on a surface to be profiled produce a phase shift known as "phase shift on reflection", which introduces phase ambiguity errors when conventional phase shifting techniques are utilized to determine the surface profile. More specifically, it is known that if the material of a surface being interferometrically profiled has optical properties such that the incident ray is delayed in phase by an appreciable amount, there will be a shift, i.e., by the "phase shift on reflection" in the phase of the fringe pattern received at the detector. Phase shifts which can cause phase ambiguity errors also may occur when there is a thin transparent film on the surface being optically profiled, because the film adds delay to the light propagation time therethrough.
There is an unmet need for an accurate, high speed, noncontact profiler capable of profiling a wide variety of rough surfaces.
Accordingly, it is an object of the invention to rapidly produce an accurate area profile of a rough surface having height variations that exceed the focus range of conventional interferometric profilers.
It is another object of the invention to provide a method and apparatus for accurate area profiling of a rough surface composed of differing materials which produce phase changes on reflection.
It is another object of the invention to provide a method and apparatus for interferometrically profiling rough surface areas without the need to use a phase unwrapping algorithm.
It is another object of the invention to provide a method and apparatus for interferometrically profiling rough surface areas very rapidly, without requiring excessive amounts of computer memory and algorithm execution time.
It is another object of the invention to provide an improved apparatus and technique for real time demodulation of interference fringe signals in an interferometric area profiling system.
It is another object of the invention to provide a method and technique for measuring surfaces with RMS average roughness of more than approximately one micron.
Briefly described, and in accordance with one embodiment thereof, the invention provides a method of profiling a rough surface of an object by producing an optical path difference so that initially a highest point of the rough surface is optically aligned with and outside of the focus range of a solid-state imaging array. Then an interferogram of the rough surface is produced by means of an interferometer. The solid-state imaging array is operated to scan the rough surface along x and y axes to produce intensity data for each pixel of the solid-state imaging array for a plurality of frames each shifted in time from the previous one to vary the optical path difference by a preselected phase difference. The contrast or modulation for each pixel is determined from the intensity data. That contrast or modulation is compared with a stored prior value of contrast or modulation of that pixel. The prior value is replaced with the most recently computed contrast or modulation if the most recently computed one is greater than the one previously stored. The corresponding relative height or optical path difference is also stored for that pixel. The optical path difference is either incrementally or linearly varied through a selected distance, and the foregoing procedure is repeated until maximum values of contrast are obtained and stored for each pixel. In one embodiment, the modulation is computed from intensity data obtained during conventional phase-shifting interferometric measurements. In another embodiment, the phase is also computed for each pixel from the intensity data and is used along with the modulation to improve vertical resolution. In another embodiment, intensity data is "amplitude demodulated" using classical communications theory to extract an "envelope" of the intensity data and determine the peak thereof. The envelope signal or modulation signal is "separated" from the "carrier" signal of the intensity waveform produced as the solid-state imaging array passes through focus. The separation is accomplished by a digital low pass filtering operation. The resulting separated modulation signal is input to a digital correlator which detects the peak and correlates it to the surface height of the present pixel.
FIG. 1 is a diagram of the rough surface profiler of the present invention.
FIG. 2 is a diagram showing an output signal that can be produced by detector cells in FIG. 1 as the optical path difference is varied through the best focus point of the objective.
FIG. 3 is a simplified block diagram of another embodiment of the invention.
FIG. 4 is a detailed block diagram of another embodiment of the invention.
FIG. 5 is a flow chart useful in describing the operation of the embodiment of FIG. 4.
FIG. 6 is a diagram useful in explaining a phase interpolation technique to improve resolution of surface height measurements.
Referring to FIG. 1, rough surface profiler 1 includes a Mirau interferometer 2 having a reference mirror 2A on a glass plate 2B, and a beamsplitter 2C. A microscope objective 4 is supported above a test surface 3 by a piezoelectric transducer (PZT) 5. PZT 5 is supported by a frame of Mirau interferometer 2. Mirau interferometer 2 is supported by the frame of microscope section 1A. The vertical or z position of microscope section 1A is controlled by a motor 60, which is connected by a mechanical link 60A to microscope section 1A. Test surface 3 is supported on a stage 66. The optical path difference (OPD) is precisely controlled by PZT 5 in response to PZT driver signal 59 produced by computer 30. Motor 60 is controlled by signal 63 produced by motor controller circuitry 58.
A white light source, which can be a typical quartz halogen lamp, directs white light through a typical commercially available illumination assembly 7, which directs the white light beam onto an ordinary beamsplitter 9. Beamsplitter 9 reflects the white light beam 10 into the upper end of microscope objective 4.
The beams reflected from the reference mirror 2A and the sample surface 3 then pass back up through microscope objective 4, upward through beamsplitter 9, through a collimating lens 31, and through a multilayer-coated beamsplitter 12 which deflects approximately 30 percent of the interference beam 20 into eyepiece assembly 13. Most of the interference beam 20 continues upward through imaging lens 11 to solid-state imaging array 18. Camera electronics 19 processes the signals of the individual CCD (Charge Coupled Device) cells or other solid-state imaging array cells of solid-state imaging array 18 and outputs them via bus 27 to computer 30. Computer 30 communicates via bus 28A with microcontroller 28.
Microcontroller 28 can be an Intel 8098. Block 68 includes a z joystick controller by means of which an operator can manually control motor 60 to control the vertical position of microscope section 1A relative to sample surface 3, and is connected by suitable conductors 68A to inputs of microcontroller 28. Microcontroller 28 communicates by bi-directional bus 28A with computer 30, which can be a desk-top computer in combination with a commercially available WYKO PMI (Phase Measuring Interface). Microcontroller 28 generates outputs 57 which control motor controller circuitry 58. Microcontroller 28 and motor controller circuitry 58 actually are included in the above WYKO PMI unit.
A filter may be positioned under solid-state imaging array 18 and camera scanning electronics 19. The signals produced by detector array 18 represent profile of sample surface 3, and are scanned by camera scanning electronics in block 19 which produce amplified signals on bus 27 that are digitized and input to computer 30 for suitable processing in accordance with the needs of the user.
The reference path and sample path in Mirau interferometer 2 are essentially identical except that the reference path is focused on mirror surface 2A and the sample path is focused on the test surface 3. Numeral 26 designates the vertical z axis of interference microscope 1.
The waveform shown in FIG. 2 is typical of those produced by the individual CCD cells in detector 18 in response to light of the interferogram impinging on detector array 18 as the optical path difference is changed. It is seen that this response is very peaked, and identification of the ideal microscope objective focus point is quite distinct. Dotted line 25 designates an "envelope" of waveform 23, which in accordance with one embodiment of the invention, is extracted from the "carrier signal" using classical communications techniques to obtain the "modulation" signal that is highest when the microscope objective is optimally focused on the rough sample surface at the pixel under consideration.
The image formed by a properly focused interference microscope typically consists of a pattern of light and dark alternating interference fringes. The number of fringes and the orientation of the fringes across the image plane are dependent on the relative tilt between the sample surface and the reference surface. Interference microscopes are assembled such that the brightest fringe occurs at "best focus" i.e. within the depth of focus of the microscope objective.
FIG. 3 shows a simplified diagram of a different embodiment of the invention in which the stage 66, rather than the microscope objective, is vertically moveable in the z direction. Stage 66 is precisely moveable, either linearly or in minute (e.g., 0.1 micron or less) increments in response to increment signals 59 from computer 30.
The specimen to be profiled has a "rough" surface 3, as in FIG. 1, that exceed height variations which can be profiled by the closest prior art interferometer-type surface profilers. Solid-state imaging array 18, which has an objective 4, is supported in a fixed position above stage 66. Camera electronics 19 are connected by bi-directional bus 27 to solid-state imaging array 18 to control scanning of rough test surface 3 by the solid-state imaging array in camera 18 and to receive intensity data from each pixel of the solid-state imaging array. Camera electronics 19 communicate with computer 30, mainly to supply measured pixel intensity data to it. Computer 30 executes software that converts the data into a surface profile and may perform further analysis thereon, and displays the surface area profile and/or analysis results on a screen, or outputs the data to control a plotter that plots the surface profile. In FIG. 3, the elements of the interferometer that produce the fringe pattern (which results from interference between source light reflected from rough surface 3 and a reference surface) viewed by solid-state imaging array 18 are omitted, for simplicity. The interferometer hardware is similar to that shown in FIG. 1.
In accordance with the present invention, either stage 66 is lowered (FIG. 1) or microscope objective section 1A is raised (FIG. 3) so that the image of rough surface 3 is beyond the range of solid-state imaging array 18. Then, computer 30 causes the OPD between solid-state imaging array 18 and the rough surface 3 to be either reduced in minute (e.g., 0.1 micron) increments or reduced linearly and sampled at the same increments by controlling the position of stage 66. For each such increment, solid-state imaging array 18 scans rough surface of sample 3. The intensity data of the interference fringe pattern is sent to computer 30 after the intensity data has been converted to digital form by camera electronics 19.
Computer 30 then, for each pixel (i.e. , x-y location) scanned by the solid-state imaging array 18, computes the modulation (or other suitable parameter) of that pixel, and compares it with the highest modulation stored so far for that pixel. If the new contrast or modulation is higher than the prior highest stored modulation, computer 11 updates the stored highest modulation by replacing it with the new one, and also stores the corresponding value of OPD or relative surface height presently being scanned.
After this has been accomplished for every pixel of solidstate imaging array 30 for the present value of z, the OPD is, in effect, incremented and the entire process is repeated. This happens many times, until the OPD or relative surface height variable z has changed enough to pass the best focus point of objective 4 through the maximum possible height range of all of the rough surface features of rough surface 3 in the field of view.
When the process is completed, computer 30 then has stored both the maximum contrast for each pixel and the value of z constitutes the profile of the surface 3.
The advantage of the forgoing technique is that it does not require any continuity of fringes on steep, high step features of specimen surface 3.
The general interference equation for the intensity measured in a 2-beam interferometer is as follows:
I=I.sub.0 [1+Mcos(φ+α)],
where I0 is the DC bias term, φ is the initial phase angle, α is the incremental phase change, and M is the modulation.
Phase-shifting techniques usually make use of a narrow-band light source and assume that M is constant throughout the measurement. Multiple frames of data are taken while the OPD is varied by a known amount between each frame. The resulting group of above equations (1) then can be solved using least-squares techniques or the like to determine the phase of each pixel at which the intensity is measured. The relative height at each pixel then is given by the formula ##EQU1##
The present invention makes use of the modulation information and the interference data corresponding to each pixel at which the intensity of the sample is measured, and correlates that information to the relative surface height of the sample at that pixel. If a broad bandwidth source, i.e., white light, is used instead of a filtered narrow bandwidth light source, the intensities of the fringes produced fall off sharply as the optical path difference is varied, as indicated in FIG. 2.
If the assumption is made that the maximum value of modulation M occurs when the OPD is near zero, then the position of the peak of the modulation function may be used to map the relative surface height throughout the field of view. The modulation function has the advantage of being non-periodic, and therefore will not have the above mentioned phase ambiguity errors.
It should be noted that measuring the modulation of fringes produced by a white light source is somewhat problematic. If standard phase-shifting methods are used to solve the basic interference equation for modulation M, the phase shift between frames must be very accurately calibrated. This becomes difficult due to spectral variations in the reflectivity oE the sample and variations in the peak wavelength of the light source. The basic interference equation (1) assumes that the modulation is constant with respect to z, which of course is not true for a white light source. The result is a very noisy modulation signal. Even so, respectable results were obtained on "very" rough surfaces using the methods described herein, because the modulation noise was small compared to the roughness of the surface.
In one experimental embodiment of the invention, phase information was used along with modulation information to improve the resolution. In another (presently preferred) embodiment of the invention, "amplitude demodulation" of the fringes was performed using techniques similar to those used in AM radio receivers.
An initial attempt at profiling a rough surface used a stepper head and motor-driven micropositioner as shown in FIG. 3 to produce relative translation of the test surface 3 along the z axis. Standard interferometric phase-shifting techniques were used to obtain intensity data used to compute the modulation. This first attempt proved the general workability of the concept of focusing the microscope on each point of the rough surface to obtain a surface map of values of z at which maximum modulation was computed for each pixel, but worked well only for very rough surfaces. A second attempt used computed phase information, in addition to the computed modulation information, from the measured intensity data to improve measurement resolution. A third attempt, which seemed to produce the best results, used amplitude demodulation techniques somewhat similar to those commonly used in communications systems to decode the modulation function. These three experimental embodiments of the invention will now be described in detail.
The above-mentioned initial attempt to implement the invention used hardware similar to that of FIG. 3, which is a simplified diagram that, for simplicity of illustration, omits the interferometer elements shown in FIG. 1. A vertical translating stage (a KLINGER P/N UZ80 PP) was used to vary the OPD by moving the rough sample surface 3 through "focus" (i.e., the best focus point of microscope objective 4), as controlled by a Hewlett-Packard 330 microcomputer, through a programmable indexer (a KLINGER P/N CC-1.2). The assignee's commercially available TOPO 3D version 4.9 software, revised slightly to provide suitable control of stage 66 and to make calculations described hereinafter, was used. The assignee's standard TOPO five-frame phase-shifting algorithm was used to collect five intensities I1 -I5 at each pixel for each OPD value. The intensities were used to calculate the modulation M at each pixel in accordance with the equation ##EQU2##
11 -I5 are five consecutive frames of intensity data for every pixel, taken with relative phase shifts of π/2 between the frames, I0 being the average light level, which is constant and need not even be used if the purpose of the computations is to find the value of z at which the maximum modulation occurs.
Stage 66 (FIG. 3) was stepped at a 0.1 micron rate for a sufficient total scan length that the entire range of surface height of rough sample surface 3 could pass through focus. More specifically, at each step the piezoelectric transducer (PZT) 5 (e.g., see FIG. 1) was shifted, and five frames of intensity data were taken. The value of M was calculated in accordance with equation (3) for each pixel, and then was compared to a previously stored value. For each pixel, if the new value of M was larger than the previously stored one for that pixel, the new value of M was stored, along with the current z value. Thus, after a complete scan, a complete profile of rough surface 3 consisting of the relative surface height of each pixel at which the maximum modulation or fringe contrast occurred was stored.
The resolution obtained for the above described initial experiment was reasonably good, but not satisfactory. The experiment was repeated, but with the "best focus" of each pixel being selected as the point at which the intensity for that pixel was maximum. However, the results were no better. The same experiments were performed for both filtered light and white light, using a standard GAR P/N S-22 for the purpose of evaluating performance of the device. The amount of time taken to execute the frame shifting algorithm was considered to be too long. Varying degrees of "noise" or error were present on smoother samples, although good results were obtained for the roughest samples, which had peaks of about 30 microns.
In a second attempt to improve the resolution, phase information was used in addition to the modulation M computed in accordance with equation (3), in the hope of obtaining better resolution than the 0.1 to 0.6 micron resolution obtained for the above described embodiment of the invention. The vertical distance between measurements was selected to equal approximately one eighth of the mean wavelength of the broadband light source. The equations used for 3-frame phase shifting with a linear ramping of the OPD with distance or time precisely producing 90 degree phase shifts between frames, are as follows: ##EQU3##
The step size was calibrated for a phase shift of π/2. The algebraic signs (i.e., + or -) of the numerator and denominator of equation (5) are used along with the modulation M of equation (4) calculated at each step, and the new modulation M is stored only if both signs are negative and M is greater than the previous stored value. For this value of z, the phase is also calculated from the measured intensity values and also stored with the current step number. This technique ensures that the peak modulation M is always stored in the same quadrant along with the peak of the fringe and that the phase calculated there will be independent of the phase calculated at adjacent pixels.
The profile of the test surface 3 then is produced at the end of the measurement by taking each stored "step number" (the step number is a variable that is incremented for each phase shift) and multiplying that by the distance per step, and adding or subtracting an incremental distance from that using the corresponding phase data, as indicated by equation (6). This is done for each pixel scanned. The result is a complete three dimensional map which is obtained without the need for phase discontinuity removal techniques, and thus is inherently free from integration errors.
The resulting resolution of this technique produced much higher resolution, almost as good as the assignee's TOPO 3D system, for smooth, flat surfaces. Measurement of rougher samples produced good results similar to those obtained for the first-described technique. Moderately rough surfaces, however, sometimes produce measurements with random spikes that occur at offsets of approximately 2π from surrounding features. It is believed that these phase ambiguity errors resulted from not being able to reliably detect the peak fringe position, causing the calculated z value to jump to the next fringe, a distance of 2π away. Some of these phase shifts are believed to be due to phase shift on reflection. Thin transparent or semi-transparent films (of thickness less than a coherence length of the light source) also produce similar phase shifts. Although the foregoing technique produced very good results on "very" rough surfaces (e.g. , several microns RMS roughness) , on "moderately" rough surfaces (e.g., 100 to 300 nanometers RMS roughness), it was clear that additional processing would be required to detect the peak fringe and that it would be necessary to continually recalibrate the phase shift produced by PZT 5, as the accuracy of such shifting was found to be critical.
Referring to FIG. 6, the above-mentioned improvement in resolution is accomplished by "offsetting" the phase φ between intensity measurement points such as 88, 89 and 90 on the highest intensity fringe 23C of intensity waveform 23. The highest intensity fringe 23C is previously isolated by using the peak of envelope 25 of intensity waveform 23C shown in FIG. 2. Points 88, 89, and 90 indicate already measured intensity points at phases spaced π/2 apart. The above-described resolution problem is caused by the fact that none of intensity measurement points 88, 89, or 90 is located at the value of z at which the peak 23A is located. In order to get an accurate value of the right hand "phase offset term" in equation (6), the present invention involves computing values of M "on the fly" according to equation (4) using intensities measured from points 88, 89, and 90 and storing that value if both 1) it is the highest value of M computed so far, and 2) the terms (I1 -I2) and (I3 -I2) are both negative or one is negative and the other zero. When that condition is met, the phase offset term φ is computed according to equation (5), and is used to "adjust" or "refine" the more approximate value given by the left term of equation (6). This later term represents the peak of the modulation envelope 25 (FIG. 2). It should be noted that the actual phase of the peak modulation relative to the height calculated using equation (6) is 3π/2, but since this relative offset is added to every pixel, the net result is the same. (Those skilled in the art know that the basic interference equation is equation (1), which has three unknown variables. Therefore, at least three measurements of intensity must be measured at different phases to solve for the three unknown variables. Measurement of additional values of intensity at additional phases can, as a practical matter, further improve accuracy of phase computation if the well known least squares technique is used to solve for the unknown variables.)
The above-mentioned third embodiment of the invention involved modifying the system of FIG. 1 so as to produce the system essentially as shown in FIG. 4. In FIG. 4, microcomputer system 30 includes a "frame processor" 36, which can be a digital signal processor (DSP) board plugged into a back plane bus 37 of an IBM AT computer system. Vertical motion control circuitry and PZT control circuitry in block 38 also is connected to the internal bus 37 and to digital signal processor 36. An 80X86 CPU 39 that is connected to internal bus 37 is the main processor of microcomputer system 30. An optional color video monitor 35 can be connected to digital signal processor and video interface circuit 36, for the purpose of effectuating real time video display.
Numeral 41 designates a conventional color monitor of microcomputer system 30, to which keyboard 42, printer 43, and work station bus 40 also are connected. The PZT control circuitry in block 38 produces a linear ramp signal 45 that drives a PZT amplifier 50, which can be a conventional amplifier such as the one used in the assignee's commercially available TOPO 3D system. PZT amplifier 50 produces a high voltage PZT drive signal on conductor 53 to control a PZT (not shown) that is used to increment the fine vertical translator 71, which is shown in FIG. 4 as being "nested" in a coarse vertical translator 70. (OPD variation can, of course, also be produced by incrementing the stage 66 in the z direction.) Coarse vertical translator 70 is driven in response to a circuit shown in block 55, which can be an "autofocus" printed circuit board contained in the assignee's commercially available TOPO A/F system. The coarse position control signal on conductor 61 controls coarse vertical translator 70.
The relative position of test surface 3 and optical system 18 is measured by a conventional linear variable differential transducer (LVDT) 73 or the like, which produces signals on conductor 74 and inputs them to LVDT signal conditioning or buffer circuitry 51. LVDT signal conditioning circuit 51 produces an output signal which is applied as an input to PZT amplifier 50, and also supplies an encoded voltage on conductor 52 to the PZT control circuitry in block 38. The encoded voltage on conductor 52 represents the precise position of the solidstate imaging array 18 relative to the present pixel of rough surface 3. Dotted line 76 designates an optical coupling between solid-state imaging array 18 and rough surface 3.
LVDT signal conditioning circuit 51 produces very precise feedback in the form of encoder voltage 52 indicating the present position of the objective of solid-state imaging array 18. PZT control circuitry in block 38 adjusts the ramp signal on conductor 45 so that solid-state imaging array 18 moves very linearly. The PZT control circuitry in block 38, PZT amplifier 50, PZT included in fine vertical translator 71 of FIG. 4, LVDT 73, and LVDT signal conditioner 51 function as a servo circuit that maintains the translation or variation of the OPD precisely linear.
Waveform 23 in FIG. 2 is of the same general shape for each of the intensity signals produced at the outputs of each of the pixels of solid-state imaging array 18 as the OPD is incrementally or linearly ramped so that the microscope objective 4 scans through the entire range of rough surface features of test surface 3. This waveform can be considered to be analogous to an ordinary AM radio signal. Waveform 23 of FIG. 2 is defined by the above interference equation of equation (1)
I=I.sub.0 +I.sub.O o+Mcos(φ+α). (1)
It is useful to compare this equation to the equation for an amplitude modulated rf signal, used in radio communications:
where s(t) is the product of a modulating signal s(t)=1+m(t) and a sinusoidal carrier signal Ucos(2πft+α), where U is a constant. The two equations (1) and (7) are very similar, with the optical interference signal of equation (1) offset by the DC term I0. Those skilled in the electronic communications art know that amplitude modulation is generally defined as a linear operation where a frequency translation of a modulating signal is performed by multiplication of the signal by a sinewave carrier. Amplitude demodulation is a reverse operation, i.e., the reconstruction of the modulating signal from the modulated signal. The technology for accomplishing this is well known to those skilled in the communications art.
It was hoped that the same demodulation techniques that apply to classical AM detection theory would apply to obtaining the modulation envelope 25 and "detecting" its peak to profile rough surface 3.
A basic operation that must be accomplished to make the above-mentioned amplitude demodulation technique work in the present invention is the separation of the modulation signal from the carrier signal in the frequency domain by use of a low pass filter. For example, in an AM radio, the low pass filtering is typically accomplished by using a simple RC network. However, the filtering of the fringe intensity data becomes more complicated, due to the fact that the intensity data through focus for each pixel is not available as a continuous analog signal, as in the case of an AM radio, but instead is sampled and digitized once each video frame.
Therefore, a digital technique must be used to perform the low pass filtering function. One possibility to accomplish the low pass filtering would be to perform a Fourier transform on the data collected for each pixel for an entire measurement and then perform "computational filtering" in the Fourier domain. This approach would require storing a large number of frames of video data, thereby imposing massive memory requirements on the system. Another possibility for performing the low pass filtering would be to use a digital filter algorithm to calculate a new filter output at each step (i.e., at each OPD) of the profiling process. The amount of memory required then would be reduced to one frame of video data for each "order" of the filter. The higher the order of the filter, the steeper its cutoff characteristics are, indicating better separation of the modulation envelope and the carrier signal. The details of the design are common knowledge, and are selected from pages 218 to 223 of "Digital Signal Processing" by J. V. Oppenhiem and R. W. Schafer, Prentice-Hall, New Jersey, 1975.)
The flow chart of FIG. 5 shows the operation of "extracting" or detecting the "envelope" 25 of waveform 23 in FIG. 2. The peak of the envelope 25 then corresponds to the point of best focus, and therefore correlates to the relative height of the present pixel of rough surface 3.
The intensity data obtained from the phase shifting operation is shown in FIG. 5 as an input to block 80, in which the sampled intensity data I(n) is stripped of the DC component I0, by use of either a digital high pass filter or by subtracting an average DC value from the intensity signal produced by each pixel (by digital low pass filtering or averaging techniques).
As shown in block 82, the resulting signal x(n), then is "rectified" to perform the needed frequency separation. The rectification process in an AM radio typically is implemented using a simple solid state diode circuit. However, to rectify the signal in a digital or time-sampled domain, a non-linear operation such as taking an absolute value or performing a mathematical squaring or the like must be applied to the discrete data. The result is a signal r(n) that now has the frequency components of the modulation signal moved to a lower frequency, while the frequency components of the fringes have been multiplied by a factor of 2 in the frequency domain.
The modulation signal M(n-N), where N is the "order" of the filter, then can be separated by the digital low pass filtering operation, as indicated in block 84 of FIG. 5. This modulation signal M then is input to a correlator that detects the peak (and/or other features) of the modulation signal M and correlates the feature(s) to the relative surface height z of the present pixel, as indicated in block 86.
For the initial implementation of a so-called IIR (infinite impulse response) digital filter (which is preferred because it generally will lead to a simple implementation and require far less memory space for storage of the delay terms than an FIR (finite impulse response filter), a fourth order Chebyshev design with a cutoff frequency of approximately 0.2π was chosen.
The algorithm executed to implement the above digital filter is as follows. First, the intensity signal from the detector is averaged for eight frames at the beginning of a measurement, during which time the data points all remained above "focus", i.e., outside the coherence length of the light source. The PZT stage 5 (FIG. 1) or other OPD translating technique in effect moves rough surface 3 through "focus" at a fixed rate, and the data is taken at fixed time intervals such that the distance moved between frames is approximately 50 nanometers.
After the eight frames of intensity data are taken for each pixel of solid-state imaging array 18, the average of the eight frames is compuoted and subtracted from each intensity value obtained throughout the measurement. At this point the delay terms (4 frames) are cleared, and the signal can begin to be input to the digital filter. Meanwhile, the PZT continues to ramp at the same rate of 50 to 100 nanometers per step.
The above technique involving averaging the first eight frames is performed for the purpose of obtaining the "DC" value I0 referred to previously. That I0 then is subtracted from each intensity value obtained during the measurement process in order to obtain a signal with no DC component. Alternatively, the same objective can be achieved by filtering the intensity signal using a digital high pass filter with a form similar to the low pass filter described above.
The DC-subtracted signal then is rectified by one of two methods, by taking the absolute value of the data or by squaring the data. The method to be used is selected by the user at the start of each measurement. The rectified signal then is input to the fourth order IIR low pass digital filter described above. At each step after the fourth, a new value of modulation is output for each pixel.
The modulation calculated by the filter then is compared with the maximum modulation computed up until that step, and if the current value is greater than the previous value, the current value is stored along with values computed at the previous steps. This requires the storing of the last three frames of modulation output for all pixels. The step where the maximum modulation occurred also is stored.
After the measurement is complete, interpolation by quadratic fitting is performed on three points saved for each pixel, to determine the actual surface height at that point. If the three modulation values taken in the vicinity of the peak of the modulation function are Y1, Y2, and Y3, with Y2 being the highest value of the three, three equations can be written relating each point to a point on a quadratic equation:
y=a.sub.0 +a.sub.1 z+a.sub.2 z.sup.2 (8)
where z1, z2, and z3, are relative positions or offsets from the position of the peak value, with z1 =-1, z2 =0, and z3 =1.
The three equations can be solved for the three unknowns a0 a1 and a2, and the expression for the interpolated offset from z2 to the position of the peak modulation value is given by the expression ##EQU4##
The result then is multiplied by the step size to obtain the relative height of the present pixel of the rough surface 3.
The above digital filter algorithm was used to test a wide variety of rough surfaces, and much better results were obtained than for the first two approaches described above. No phase ambiguity offset errors were observed, and no errors due to phase shifts (for example due to presence of thin films or phase shift on reflection), were observed. Measurement resolution of less than 10 nanometers was achieved. The technique worked well over a wide range of PZT ramp signal slopes, source illumination levels, source filtering, etc. No recalibration of PZT ramp signal slope was necessary for different sample types.
It therefore was concluded that the use of digital filtering algorithms is a very robust technique, the performance of which can be further improved by better filter design, better DC tracking (which can be accomplished by the above-mentioned high pass filtering), and more "intelligent" peak detectors.
In another embodiment of the invention, relative height values for the various pixels of rough surface 3 that correspond to the peak of modulation envelope 25 can be used to eliminate the phase ambiguity errors from the height measurement for each pixel. By determining the peak of the modulation envelope, the described digital filtering techniques, in combination with the described phase measurement techniques, it is possible to obtain better resolution of the surface profile. The peak of modulation envelope 25 indicates a "coarse" measurement resolution, and the phase measurement technique provides additional more accurate measurement values which, in accordance with the second term of equation (6), are added to the initial measurement to provide increased resolution. Thus, relatively smooth surfaces can be very accurately profiled using the disclosed phase measurement techniques, and very rough surfaces can be very accurately profiled using the described modulation techniques. The immediately foregoing peak modulation "interpolation" techniques allow measurement of surfaces of "intermediate" roughness with approximately the same resolution as can be obtained in profiling very rough surfaces by combining both techniques sequentially in accordance with equation (6). That is, the modulation envelope peak measurement technique is first used to get a rough measurement, and then the phase measurement technique is used to further refine that measurement.
The present invention thus provides a means of using a combination of interferometric techniques to make fast area measurements very accurately of surfaces of any degree of smoothness or roughness.
While the invention has been described with reference to several particular embodiments thereof, those skilled in the art will be able to make the various modifications to the described embodiments of the invention without departing from the true spirit and scope of the invention. It is intended that all combinations of elements and steps which perform substantially the same function in substantially the same way to achieve the same result are within the scope of the invention.
|Cited Patent
|Filing date
|Publication date
|Mar 19, 1987
|Mar 22, 1988
|Zygo Corporation
|Interferometric surface profiler
|Oct 15, 1986
|Dec 13, 1988
|Eastman Kodak Company
|Sub-nyquist interferometry
|Feb 27, 1987
|Mar 21, 1989
|Alps Electric Co., Ltd.
|Method and apparatus for measuring the floating amount of the magnetic head
|May 6, 1986
|Apr 4, 1989
|Kla Instruments Corporation
|Method and apparatus of using a two beam interference microscope for inspection of integrated circuits and the like
|Apr 4, 1989
|Jun 5, 1990
|Wyko Corporation
|Apparatus and method for automatically focusing an interference microscope
|"An Application of Interference Microscopy to Integrated Circuit Inspection and Metrology", by M. Davidson, K. Kaufman, I. Mazor, and F. Cohen, SPIE vol. 775, Integrated Circuit Metrology, Inspection & Process Control (1987), pp. 233-247.
|"First Results of a Product Utilizing Coherence Probe Imaging For Wafer Inspection", and Process Control II (1988); pp. 100-114.
|"Profilometry With a Coherence Scanning Microscope", by B. S. Lee, and T. C. Strand, Applied Optics, vol. 29, No. 26, Sep. 10, 1990, pp. 3784-3788.
|An Application of Interference Microscopy to Integrated Circuit Inspection and Metrology , by M. Davidson, K. Kaufman, I. Mazor, and F. Cohen, SPIE vol. 775, Integrated Circuit Metrology, Inspection & Process Control (1987), pp. 233 247.
|First Results of a Product Utilizing Coherence Probe Imaging For Wafer Inspection , and Process Control II (1988); pp. 100 114.
|Profilometry With a Coherence Scanning Microscope , by B. S. Lee, and T. C. Strand, Applied Optics, vol. 29, No. 26, Sep. 10, 1990, pp. 3784 3788.
|Citing Patent
|Filing date
|Publication date
|Jan 29, 1996
|Jul 8, 1997
|Medar, Inc.
|Scanning phase measuring method and system for an object at a vision station
|May 30, 1996
|May 19, 1998
|Photon Dynamics, Inc.
|Substrate inspection apparatus and method
|Jun 26, 1996
|Sep 15, 1998
|The United States Of America As Represented By The Secretary Of The Navy
|Method for remotely determining sea surface roughness and wind speed at a water surface
|Feb 21, 1997
|Nov 17, 1998
|Micron Technology, Inc.
|Method and apparatus for determining surface roughness
|May 13, 1996
|Mar 2, 1999
|Boxer Cross, Inc.
|System and method for measuring the microroughness of a surface of a substrate
|Oct 21, 1997
|Jun 1, 1999
|Beaty; Elwin M.
|Three dimensional inspection system
|Jan 26, 1998
|Dec 7, 1999
|Zygo Corporation
|Method and apparatus for performing interferometric measurements with reduced sensitivity to vibration
|Apr 24, 1998
|Dec 21, 1999
|Heui Jae Pahk
|Non contact measuring method for three dimensional micro pattern in measuring object
|May 5, 1997
|Apr 25, 2000
|Beaty; Elwin M.
|Three dimensional inspection system
|May 28, 1999
|May 16, 2000
|Elwin M. Beaty
|Apparatus for three dimensional inspection of electronic components
|May 28, 1999
|May 16, 2000
|Elwin M. Beaty
|Process for three dimensional inspection of electronic components
|Dec 2, 1997
|May 30, 2000
|Hitachi, Ltd.
|Method and apparatus for measuring the height of an object
|Jan 16, 1998
|Jun 6, 2000
|Beaty; Elwin M.
|Method and apparatus for three dimensional inspection of electronic components
|Aug 1, 1996
|Aug 8, 2000
|Nanopro Luftlager-Produktions-Und Messtechnik Gmbh
|Device and process for measuring two opposite surfaces of a body
|Mar 8, 2000
|Aug 7, 2001
|Nanopro Luftlager-Produktions-Und Messtechnik Gmbh
|Apparatus and method for measuring two opposite surfaces of a body
|May 18, 2001
|Oct 29, 2002
|Mitutoyo Corporation
|Scanning wide-area surface shape analyzer
|Jun 20, 2000
|Nov 26, 2002
|Ppt Vision, Inc.
|Precision 3D scanner base and method for measuring manufactured parts
|Oct 16, 2001
|Dec 3, 2002
|Ppt Vision, Inc.
|Color-adjusted camera light and method
|Jun 20, 2000
|Dec 31, 2002
|Ppt Vision, Inc.
|3D scanner and method for measuring heights and angles of manufactured parts
|Jun 20, 2000
|Jan 21, 2003
|Ppt Vision, Inc.
|Binary optical grating and method for generating a moire pattern for 3D imaging
|Jul 8, 1999
|Feb 18, 2003
|Ppt Vision, Inc.
|Combined 3D- and 2D-scanning machine-vision system and method
|Jul 8, 1999
|Aug 5, 2003
|Ppt Vision, Inc.
|Circuit for machine-vision system
|Jul 10, 2000
|Sep 7, 2004
|Ppt Vision, Inc.
|Method and apparatus for adjusting illumination angle
|Nov 18, 2002
|Oct 26, 2004
|Ppt Vision, Inc.
|Method and apparatus for a pulsed L.E.D. illumination source
|Jun 12, 2001
|Nov 9, 2004
|Dana Corporation
|Method and apparatus for measuring irregularities on an outer surface of a rotatable cylindrical shaft
|Jul 13, 1999
|Mar 1, 2005
|Elwin Beaty & Elaine Beaty
|Method and apparatus for three dimensional inspection of electronic components
|Oct 24, 2002
|Apr 26, 2005
|Nanopro Luftlager-Produktions-Und Messtechnik Gmbh
|Apparatus and method for measuring two opposite surfaces of a body
|Apr 27, 2001
|Jul 5, 2005
|Elwin M. Beaty
|Method and apparatus for three dimensional inspection of electronic components
|Jan 8, 2001
|Oct 18, 2005
|Ismeca Europe Semiconductor Sa
|Imaging for a machine-vision system
|Feb 28, 2005
|Jul 18, 2006
|Scanner Technologies Corporation
|Electronic component products made according to a process that includes a method for three dimensional inspection
|Feb 28, 2005
|Aug 1, 2006
|Scanner Technologies Corporation
|Method of manufacturing electronic components including a method for three dimensional inspection
|Jul 30, 2004
|Nov 28, 2006
|Ppt Vision
|Method and apparatus for adjusting illumination angle
|Sep 20, 2004
|Mar 6, 2007
|Lebens Gary A
|Method and apparatus for a variable intensity pulsed L.E.D. light
|Jul 8, 1999
|Apr 8, 2008
|Charles A. Lemaire
|Tray flipper and method for parts inspection
|Mar 13, 2003
|Jun 10, 2008
|Taylor Hobson Limited
|Surface profiling apparatus
|Feb 12, 2007
|Jul 1, 2008
|Charles A. Lemaire
|Method and apparatus for constant light output pulsed L.E.D. illumination
|Jun 29, 2004
|Jul 8, 2008
|Charles A. Lemaire
|Parts manipulation and inspection system and method
|Jan 19, 2006
|Sep 23, 2008
|Zygo Corporation
|Interferometer for determining characteristics of an object surface, including processing and calibration
|Jan 19, 2006
|Nov 4, 2008
|Zygo Corporation
|Interferometer for determining characteristics of an object surface
|Nov 19, 2007
|Nov 25, 2008
|Zygo Corporation
|Methods and systems for interferometric analysis of surfaces and related applications
|Jun 26, 2007
|Dec 16, 2008
|Zygo Corporation
|Profiling complex surface structures using scanning interferometry
|Feb 28, 2005
|Mar 24, 2009
|Scanner Technologies Corporation
|Electronic component products and method of manufacturing electronic component products
|Nov 21, 2003
|Apr 14, 2009
|Taylor Hobson Limited
|Surface profiling apparatus
|Jul 19, 2007
|Apr 21, 2009
|Zygo Corporation
|Compensation of systematic effects in low coherence interferometry
|Nov 25, 2006
|Jul 7, 2009
|Lebens Gary A
|Method and apparatus for auto-adjusting illumination
|Oct 31, 2008
|Sep 8, 2009
|Zygo Corporation
|Methods and systems for interferometric analysis of surfaces and related applications
|Jul 9, 2005
|Sep 29, 2009
|Kla-Tencor Corporation
|Measurement of the top surface of an object with/without transparent thin films in white light interferometry
|Jan 19, 2006
|Nov 10, 2009
|Zygo Corporation
|Interferometer with multiple modes of operation for determining characteristics of an object surface
|Jul 19, 2007
|Nov 17, 2009
|Zygo Corporation
|Generating model signals for interferometry
|Oct 11, 2006
|Dec 22, 2009
|Zygo Corporation
|Interferometry method and system including spectral decomposition
|Jan 22, 2008
|Mar 23, 2010
|Zygo Corporation
|Interferometer and method for measuring characteristics of optically unresolved surface features
|Jul 8, 2008
|May 18, 2010
|Charles A. Lemaire
|Parts manipulation, inspection, and replacement system and method
|Jul 1, 2008
|Jun 22, 2010
|Charles A. Lemaire
|Method and apparatus for pulsed L.E.D. illumination for a camera
|Mar 3, 2009
|Aug 10, 2010
|Charles A. Lemaire
|Method and apparatus for parts manipulation, inspection, and replacement
|Jul 16, 2008
|Oct 12, 2010
|Zygo Corporation
|Interferometry method for ellipsometry, reflectometry, and scatterometry measurements, including characterization of thin film structures
|Jul 20, 2007
|Jan 11, 2011
|Zygo Corporation
|Multiple-angle multiple-wavelength interferometer using high-NA imaging and spectral analysis
|Jun 5, 2007
|Feb 8, 2011
|Zygo Corporation
|Interferometry for determining characteristics of an object surface, with spatially coherent illumination
|Jun 4, 2007
|Feb 15, 2011
|Zygo Corporation
|Interferometry for lateral metrology
|Dec 21, 2007
|Apr 12, 2011
|Zygo Corporation
|Apparatus and method for measuring characteristics of surface features
|May 6, 2008
|May 24, 2011
|Taylor Hobson Limited
|Surface profiling apparatus
|Feb 8, 2010
|May 24, 2011
|Zygo Corporation
|Interferometer and method for measuring characteristics of optically unresolved surface features
|Oct 15, 2009
|May 31, 2011
|Zygo Corporation
|Interferometer with multiple modes of operation for determining characteristics of an object surface
|Nov 7, 2008
|Jul 12, 2011
|Zygo Corporation
|Interferometer utilizing polarization scanning
|Aug 14, 2009
|Jul 12, 2011
|Zygo Corporation
|Compound reference interferometer
|Jul 24, 2009
|Aug 23, 2011
|Zygo Corporation
|Scan error correction in low coherence scanning interferometry
|Apr 8, 2008
|Nov 15, 2011
|Charles A. Lemaire
|Tray flipper, tray, and method for parts inspection
|Oct 9, 2008
|Dec 6, 2011
|Zygo Corporation
|Interferometric analysis of under-resolved features
|Sep 2, 2009
|Jan 31, 2012
|Zygo Corporation
|Methods and systems for interferometric analysis of surfaces and related applications
|Jul 24, 2009
|Feb 21, 2012
|Zygo Corporation
|Interferometric systems and methods featuring spectral analysis of unevenly sampled data
|Dec 11, 2008
|Feb 28, 2012
|Zygo Corporation
|Analyzing surface structure using scanning interferometry
|Jun 22, 2010
|Apr 17, 2012
|Lemaire Illumination Technologies, Llc
|Apparatus and method for pulsed L.E.D. illumination
|Nov 15, 2011
|Oct 16, 2012
|Charles A. Lemaire
|Parts manipulation, inspection, and replacement system and method
|Sep 25, 2011
|Jan 29, 2013
|Led Tech Development, Llc
|Apparatus and method for L.E.D. illumination
|Aug 31, 2009
|Feb 19, 2013
|Zygo Corporation
|Fiber-based interferometer system for monitoring an imaging interferometer
|Oct 4, 2012
|Apr 2, 2013
|Charles A. Lemaire
|Parts manipulation, inspection, and replacement
|Jan 4, 2005
|Jun 18, 2013
|Entropic Communications, Inc.
|Distributing candidate vectors based on local motion complexity
|Jan 8, 2013
|Feb 4, 2014
|Lemaire Illumination Technologies, Llc
|Apparatus for L.E.D. illumination
|Oct 8, 2013
|Sep 9, 2014
|Led Tech Development, Llc
|Apparatus and method for pulsed L.E.D. illumination
|Feb 13, 2013
|Dec 2, 2014
|Zygo Corporation
|Low coherence interferometry with scan error correction
|Aug 13, 2009
|May 5, 2015
|Zygo Corporation
|Interferometry for lateral metrology
|Dec 21, 2012
|Jul 28, 2015
|L-3 Communications Security And Detection Systems, Inc.
|3D mapping with two orthogonal imaging views
|Oct 24, 2002
|Mar 27, 2003
|Dieter Muller
|Apparatus and method for measuring two opposite surfaces of a body
|Nov 18, 2002
|May 22, 2003
|Ppt Vision, Inc.
|Method and apparatus for a pulsed L.E.D. illumination source
|Sep 20, 2004
|Feb 24, 2005
|Ppt Vision, Inc.
|Method and apparatus for a variable intensity pulsed L.E.D. light
|Jun 29, 2004
|May 26, 2005
|Hackney Joshua J.
|Parts manipulation and inspection system and method
|Feb 28, 2005
|Sep 1, 2005
|Beaty Elwin M.
|Electronic component products and method of manufacturing electronic component products
|Feb 28, 2005
|Sep 1, 2005
|Beaty Elwin M.
|Electronic component products made according to a process that includes a method for three dimensional inspection
|Feb 28, 2005
|Sep 1, 2005
|Beaty Elwin M.
|Method of manufacturing electronic components including a method for three dimensional inspection
|Mar 13, 2003
|Oct 13, 2005
|Bankhead Andrew D
|Surface profiling apparatus
|Nov 21, 2003
|Jan 26, 2006
|Bankhead Andrew D
|Surface profiling apparatus
|Jan 19, 2006
|Jul 20, 2006
|Xavier Colonna De Lega
|Interferometer with multiple modes of operation for determining characteristics of an object surface
|Jan 19, 2006
|Jul 20, 2006
|Xavier Colonna De Lega
|Interferometer for determining characteristics of an object surface
|Jul 9, 2005
|Jan 11, 2007
|Phase Shift Technology, Inc.
|Measurement of the top surface of an object with/without transparent thin films in white light interferometry
|Feb 12, 2007
|Jun 14, 2007
|Charles Lemaire
|Method and apparatus for a pulsed l.e.d. illumination
|May 6, 2008
|Sep 4, 2008
|Taylor Hobson Limited
|Surface profiling apparatus
|Dec 4, 2007
|Nov 13, 2008
|Zygo Corporation
|Methods and systems for determining optical propertis using low coherence interference signals
|Jul 8, 2008
|Mar 19, 2009
|Charles A. Lemaire
|Parts manipulation, inspection, and replacement system and method
|Apr 8, 2008
|Mar 26, 2009
|Charles A. Lemaire
|Tray flipper, tray, and method for parts inspection
|Mar 3, 2009
|Jul 16, 2009
|Charles A. Lemaire
|Method and apparatus for parts manipulation, inspection, and replacement
|Dec 11, 2008
|Jul 16, 2009
|De Groot Peter
|Analyzing surface structure using scanning interferometry
|Jan 4, 2005
|Aug 20, 2009
|Erwin Bellers
|Distributing candidate vectors based on local motion complexity
|Aug 13, 2009
|Dec 10, 2009
|Zygo Corporation
|Interferometry for lateral metrology
|Jul 8, 1998
|Jan 1, 2008
|Ismeca Europe Semiconductor Sa
|Scanning phase measuring method and system for an object at a vision station
|May 22, 2001
|Apr 26, 2012
|Mitutoyo Corp.
|Abtastender Grossbereich-Oberfl�chenformanalysator
|Sep 27, 2001
|Apr 17, 2002
|Tokyo Seimitsu Co.,Ltd.
|Roughness measuring method and apparatus
|Sep 27, 2001
|Jan 2, 2008
|Tokyo Seimitsu Co.,Ltd.
|Roughness measuring method and apparatus
|Nov 6, 2001
|May 8, 2002
|Tokyo Seimitsu Co.,Ltd.
|Roughness measuring method and apparatus
|Nov 6, 2001
|Mar 31, 2004
|Tokyo Seimitsu Co.,Ltd.
|Roughness measuring method and apparatus
|Jan 13, 1997
|Jul 24, 1997
|Photon Dynamics, Inc.
|Substrate inspection apparatus and method
|U.S. Classification
|356/497, 382/108, 250/559.05, 250/559.06, 250/559.29, 356/512
|International Classification
|Cooperative Classification
|European Classification
|Apr 10, 1998
|Fee payment
Year of fee payment: 4
|Apr 10, 2002
|Fee payment
Year of fee payment: 8
|Apr 30, 2002
|Maintenance fee reminder mailed
|Apr 26, 2006
|Maintenance fee reminder mailed
|Jul 17, 2006
|Surcharge for late payment
Year of fee payment: 11
|Jul 17, 2006
|Fee payment
Year of fee payment: 12
|Sep 7, 2010
Effective date: 20030603
Free format text: CHANGE OF NAME;ASSIGNOR:WYKO CORPORATION;REEL/FRAME:024946/0669
Owner name: VEECO TUCSON INC., CALIFORNIA
|Sep 9, 2010
Owner name: VEECO METROLOGY INC., CALIFORNIA
Free format text: CHANGE OF NAME;ASSIGNOR:VEECO TUCSON INC.;REEL/FRAME:024953/0612
Effective date: 20060407
|Jun 11, 2012
Owner name: BRUKER NANO, INC., CALIFORNIA
Free format text: CHANGE OF NAME;ASSIGNOR:VEECO METROLOGY INC.;REEL/FRAME:028350/0511
Effective date: 20101007


text:
Make and receive calls from anywhere in the world with just one click.
You don't even need to pick up the phone and dial — just press one button and you're already on the line. The call is automatically synced inside your lead. That's a great plus!
Sales Manager at Case3D
Spend less time dialing numbers and more time closing deals.
Stop switching windows or dialing numbers, and focus on making more calls in less time.
Our software saves you time from dial numbers and makes it easy to quickly go through a list of leads.
Close.io keeps track of all your calling activity so you can focus on selling, not organizing.
Calls made and received are automatically logged in Close.io, including the duration of the call.
Call went to voicemail? Simply click the "Leave voicemail" button and Close.io will leave your pre-recorded voicemail.
Should your prospect speak with another member of your team? Click the "Transfer Call" button, select your team member and your call will get transferred.
Away from your desk? Forward your calls to another number to never miss an important call.
Enable call recording and listen to calls directly in your timeline. Use your recordings to review your call activity, train new employees or monitor progress.
Download recordings in one click and share them with your team.
Get an overview of the calling activities across your entire team in just a few clicks.
Want to see who's ahead and who's falling behind? Our powerful reporting gives you an overview in seconds.
See the total number of calls made, calls by user, call duration and more.
All calls, including missed and snoozed calls and voicemails, will show up in your inbox.
Your team will never again let a lead fall through the cracks.
No fees. No minutes. Make and receive as many calls as you want.
Don't want to switch numbers? Keep using your existing number.
Add a main sales line that goes to any available salesperson.
Our engineers work hard to give you the crystal clear audio you need.

subset: commoncrawl


text: So I bought this Firewire card used online the other day, and I'm trying to install it, but my system doesn't seem to be recognizing it (no Install New Hardware flag). Not only that, but when I try to manually install 1394 IEEE drivers, my computer says that it doesn't have any (I'm running WinXP Pro). No 1394 section shows up in my device manager, either. The website of the manufacturer says that in WinXP the drivers should already be there. So, I either need to know where to find 'em on my computer, or where to download 'em... or what ever the heck else I'm doing wrong. I know the firewire card is at least getting power, because I have a light up firewire cable that glows when I connect it.


text:
mike has yet to fill out a bio.
Rob Williamson posted 3352 days ago
Thanks bro...I try to shoot straight from the hip.
Lisa Horne posted 3359 days ago
It was only up from 9-5 PST on MSN.com. It's on FOXSports.com tomorrow.
mike does not have any awards.
mike is not a member of any communities.
mike does not have any favorite writers.


text:
What we already know. NoteSheet 10.1. The Rise of Islam (pg. 263). 1. What is the Story of Muhammad?. Orphaned at the age of six , Muhammad was raised by his grandfather and uncle At the age of 25 he married a wealthy businesswoman of about 40 named Khadijah
Download Policy: Content on the Website is provided to you AS IS for your information and personal use and may not be sold / licensed / shared on other websites without getting consent from its author.While downloading, if for some reason you are not able to download a presentation, the publisher may have deleted the file from their server.
The Rise of Islam
(pg. 263)


text:
Oliver, Stoned. Trailer (2014)
The world's biggest stoner, Oliver, loses a high profile car, forcing him to steal an ice cream truck and enlist his wacky friends to help track down the thief before it's too late.
Watch the official Oliver, Stoned. 2014 trailer in HD below or find more Oliver, Stoned. videos on Vidimovie.


text:
Terminator T-1000 killer robots may soon be reality after major breakthrough
KILLER shape-shifting droids like Terminator T-1000 may soon be possible after a major scientific breakthrough.
Boffins have managed to create liquid metal circuits – just like those used in the terrifying baddie in Terminator 2: Judgement Day.
In a terrifying leap forward that will send chills down most people's spines, the molten metal switches and pumps were able to operate themselves – independently.
The prof behind the development boasted: "It may be possible to build a 3D liquid metal humanoid on demand – like the T-1000 Terminator."
"It may be possible to build a 3D liquid metal humanoid"Prof Kourosh Kalantar-zadeh
As any 1990s film fan knowns, T-1000 could change shape – sliding under doors or through the bars of a prison cell – and repair any damage to itself in seconds.
Its shape-shifting ability made it virtually indestructible.
But engineers from RMIT University in Melbourne, Australia, are trying to make the apocalyptic robots a reality.
Professor Kourosh Kalantar-zadeh said: "We adjusted the concentrations of acid, base and salt components in the water and investigated the effect.
"Simply tweaking the water's chemistry made the liquid metal droplets move and change shape, without any need for external mechanical, electronic or optical stimulants.
"Using this discovery, we were able to create moving objects, switches and pumps that could operate autonomously – self-propelling liquid metals driven by the composition of the surrounding fluid.
"Eventually, using the fundamentals of this discovery, it may be possible to build a 3D liquid metal humanoid on demand – like the T-1000 Terminator."
British super-brain Stephen Hawking has already warned mankind could be wiped out by Terminator-style killer robots.
Now inevitably we will need another team to create a T-800 to travel back in time and destroy the scientists at RMIT before they can finish their evil project.


text:
|your cart is empty
In Stock
Mind Awake[CD]
~ For Your Eyes Only
Product Notes
BIOGRAPHY: I need to make this clear right now: This ISN'T your average Bio. I'm not going to make delusional comparisons to the band and the current music 'scene' or create some false sense of fame by listing some of the national bands we've played with. You may have grown to know For Your Eyes Only as the screamo / fast-rock band we were or if you're listening to For your Eyes Only for the first time, then read on. For Your Eyes Only has gone through a few tiny construction stages to produce the ultimate friendship based, musicianship based, and dedicated band. Everyone in this band has an unprecedented passion just to push our music forward. Not to be the next revolutionaries like Led Zeppelin or The Beatles (I'm not going to lie, that'd be amazing) but just to step it up to that next level in our career. That is were you come in. You are a very important part of For Your Eyes Only; you are the fuel to our fire. Without YOU we are NOBODY. You are the ONLY reason we can play our music out of state, or go on long tours coast to coast. You put gas in our tank, feed us, give us the motivation to make new music, and keep us going. That is why I ask; If you like us then tell EVERYONE that you think would like us too. Bring all of your friends to our shows, and come talk to us. We love talking to you guys. You might need to get used to our weird sense of humor though. :) Love us or hate us we're going to keep pushing on, harder each and every time. You have NOT heard the last from us. Expect new music soon. This is 100% the truth.-Eric, Jarred, Randy, Jim, Maxx.
You May Also Like
Page 1 of


text:
£18.36 (£22.03 with VAT) On Hold
WEI EAST WHITE LOTUS Defining Moment Serum 1.69oz Retails for $145!
Only 1 available
Shipping: United Kingdom: free (more destinations)
Condition: Brand new
*The store has not been updated recently. You may want to contact the merchant to confirm the availability of the product.


text:
May 4, 2015
Why Job-Hopping May Not Be So Bad
The Pros and Cons of Job-Hopping What Baby Boomers Have to Say Baby Boomers and Millennials have different views when it comes to
The Pros and Cons of Job-Hopping What Baby Boomers Have to Say Baby Boomers and Millennials have different views when it comes to
It's no secret that advances in technology are revolutionizing the university and college experience. There are many apps in existence (and in development) which
We're always on the lookout for tips and tricks that will help us get things done more efficiently, because… well… startup life! As we were
The CEO Experience in the Differences Between Corporate and Startup Culture By Zula CEO: Raz Yalov Have you ever thought about how a
Students are typically asked to participate in group projects to explore unique viewpoints and strategies as they complete their coursework. This is, of course,
What does it mean to work smarter, but not harder? We hear this when on the verge of being burned out upon realizing that
"Teamwork" has a positive connotation to it, but oftentimes brings discomfort to participants. From work, to school, to the organization of events- once 'teamwork'
It feels like it was only yesterday when we featured Triberr on our blog for their team communication and collaboration practices. Well it was
In remote team management, it gets complicated to keep everyone aligned. It's hard to juggle remote employees with their varying mindsets, in addition to different tools
Startup nation is on a roll! In January 2015 alone, startups based in Israel have raked in over $1.3B in funding and acquisitions. But


text: Say, if I stored a save on my SD card from say.... Wario Ware... And I brought it over to my friend's house, with my copy of Wario Ware. And i put the SD card in his wii, and copied the save state to his wii.. Would I be able to play MY saved games on HIS wii? PS: I know, it HAS HAD to have been answered before... But that search function, the only things you get are "I GIT FRII WII GAMEZ YA?" or "HOEMBROO 4 MII WIT SD CURDZ RITE?" PLEASEDON'TKILLME.


text:
Quick Look
Property brokered by
Get more listing details on brokerage site
Presented By:
Betsy Egan
(413) 687-6830
Email Agent
Betsy Egan
(413) 687-6830
Email Agent
Property DetailsSpacious colonial in North Amherst on the bus route within walking distance to UMass. Also easy walk to North Amherst conservation park with in-ground pool .Large yard ideal for play area or great garden space. Many improvements. New Furnace and water heater 2009. New roof (metal) on garage in 2015. Great location, very central to everything. Solidly built home! A wonderful opportunity to live on the 1st floor and rent the second floor apartment (with a separate kitchen) and pay all expenses with money left over. A GREAT INVESTMENT! There is potential for 2 more bedrooms on the first floor! Perfect for the student with four or more years at Umass to live and rent out rooms to other students and bring in an income! Price reduction to $329,900.00!
Area Recent Sales
Area Detailed Information
Area Schools
Area Weather
Area Cost Index
Mortgage Calculator
Additional Property InformationThis single family listing located at 1117 North Pleasant St, Amherst, MA has four bedrooms and two bathrooms. The property is currently selling for $329,900.
Features include amenities description: public transportation, shopping, park, walk/jog trails, laundromat, bike path, conservation area, highway access, house of worship, public school, university; appliances description: dishwasher, disposal, refrigerator, washer, dryer; basement description: full, interior access, bulkhead, concrete floor; bathroom 1 description: bathroom - full, bathroom - with tub & shower, flooring - vinyl; bathroom 2 description: bathroom - full, bathroom - with shower stall, bathroom - with tub, flooring - vinyl; bedroom 2 description: flooring - vinyl; bedroom 3 description: flooring - hardwood; bedroom 4 description: flooring - hardwood; bedroom 5 description: skylight, flooring - wall to wall carpet; construction description: frame; dining room description: closet/cabinets - custom built, flooring - hardwood; directions: north pleasant north until 1117 on right @sign; electrical features: 110 volts, circuit breakers, 100 amps; energy features: storm windows, insulated doors, storm doors; exterior features: deck - roof, screens, garden area; flooring description: vinyl, wall to wall carpet, hardwood; foundation description: concrete block; frontage (feet): 88.00; garage/parking description: detached, garage door opener, storage; hot water description: electric, tank; insulation features: unknown; interior features: cable available; kitchen description: flooring - vinyl, exterior access; living room description: fireplace, flooring - hardwood, cable hookup; lot description: corner, paved drive, cleared, level; master bedroom description: flooring - hardwood; number of rooms: 11; parking description: off-street; parking spaces: 5; pricing description: $172.72 (square foot); road description: public, publicly maint., sidewalk; utility connections: for electric range, for electric oven, for electric dryer, washer hookup.
This property is listed by Betsy Egan of Jones Group REALTORS located in Massachusetts.


text:
Tunisia vs Cape Verde Islands 01/18/2015 African Cup of Nations Preview,Odds and Prediction
Cape Verde Islands vs Tunisia
Sunday 18th January 2015 19:00(GMT)
The Line: Tunisia 11/10 Draw 21/10 Cape Verde Islands 11/4 See the Latest Odds
TV: Eurosport; Bein Sports
Tunisia are a heavy favourite to progress in this group and they are unbeaten in their last seven in all competitions. Tunisia are a physical strong team with Younes and Ben Youssef up front this Tunisia team look likely to win this group and could be a competitior in the knockout stages but first they must negotiate this group which is tougher than it looks.
They face a tough game against a Cape Verde team that have some young talent in their rranks. Gary Rodriguez and Heldon will be the key men for the islanders and they will take stopping as both are talented players plying their trade in a the Portugese and Spanish leagues. Cape Verde actually beat Tunisia 2-0 in a World Cup qualifier but the game got awarded to Tunisia as Cape Verde fielded an ineligible player. There is definite revenge angle here for Cape Verde if they can pull it together.
Tough game to handicap here as although Tunisia have a strong physical team I have a feeling Cape Verde could turn over the favourites here. I am going to take both teams to score though as the value I see in the market is there.


text:
GraphicRiver Mythic Ribbons 5147401
Category: Web Element | Comment: ( 0 )
GraphicRiver Mythic Ribbons 5147401
Web Element \ Miscellaneous
Ribbons \ Interfaces \ Miscellaneous \ Misc \ Game \ Gaming \ Green \ Brown
Web Element - Mythic Ribbons 5147401 by KEvil @ GraphicRiver
Web Element \ Miscellaneous
Graphics Files Included: Photoshop PSD, Transparent PNG, JPG Image; High Resolution: No; Layered: Yes; Minimum Adobe CS Version: CS; Pixel Dimensions: 1000x555.
Mythic Ribbons pack contains ribbons that can be used in all kind of web site, mobile application and game development.
- Fully Layered
- Layer Styles
- Grouped layers
- 10 Elements
- 3 Color Variants (Red, Green, Brown)
- Used Font : Lao UI
Grunge User Interface; Galaxy User Interface.
Keywords: brown, game, gaming, green, gui, misc, miscellaneous, red, ribbons, ui, web interfaces.
Download Link


text:
We recommend upgrading to the latest Internet Explorer, Google Chrome, or Firefox.
If you are using IE 9 or later, make sure you turn off "Compatibility View".
Handpicked brands for you
Himalaya iPad sling, high in the Himalayas. This bag is made from a vintage Tibetan apron, worn 50-150 years ago and woven on a backstrap loom from wool collected in the high plains.


text:
Sharing a victory...
View Single Post
June 14th, 2011, 06:56 PM
Platinum Supermommy
Join Date: Mar 2007
Location: Cape May, NJ
Posts: 7,825
Sounds like your in a good grove. Don't worry about the two "gated off" little ones. My girls spent a lot of time "gated off" in their playroom (and still sometimes are) while I did housework etc... They whined and cried from time to time, but I found that they soon realized that playing with their toys was a lot more fun (and they learned that Mommy would pop in with kisses and hugs periodically during this work time). Now that they are older I get comments all the time about how imaginative they are and how "unusual" they are for being able to entertain themselves. I attribute this to them being given time to just explore and play without me interrupting them.
View Public Profile
Find all posts by Jill0924


text:
Mammootty The Great Father Wiki Detail Star Cast News Release Date Poster
Mammootty next movie "The Great Father" first look poster out now. After a very long time Mammootty has updated about his next project first look poster on his official account. Earlier shadow of his first look got release, now new poster got release which show a industrial workplace and his back on the poster.
The latest upcoming Malayalam movie The Great Father is expected to release next years. Shoot of the film stared on 25 August, and makers decided the places to shoot the movie in Kochi, Wagon and Thrissur.
Haneef Adeni is going the direct the movie. This is Haneef's debut directorial movie and he has narrated the story as well. Film starring Mammootty and Sneha, she is going to play opposite to lead actor, Baby Sara Arjun will be seen as daughter of Mammootty. Prithviraj Sukumaran will be seen in cameo role.
The film is produced by Santosh Sivan, Prithviraj Sukumaran, Shaji Nadesan and Arya. The movie is brought under the banner of August Cinemas. Last movie by the banner was a huge success.
This is a family drama thriller genre movie, who's music source is done by Gopi Sunder. The movie is expected to release at the end of this year or begging of next year. With such a strong star cast makers cannot extent the release date far.


text: I can't seem to fingur it out and I've checked the help file and looked through the preferences & menus. I want the bookmarks to be listed in a sidebar and a website to the right of it. You can achieve this simply in Firefox by hitting Apple-Tab. I noticed you can drag bookmark folders up top, and it's nice, but you can't open them into a new tab (from what I figured out). Reason why I want to use Safari more is because it's just less buggy on the Mac than Firefox. I have a lot of hang ups when copying images or viewing some sites. Sometimes just out of the blue. Safari does seem a little snappier and the scroll-click works.


text:
2 Casual Dining Restaurants in Swellendam
Showing 1 to 2 of 2 results
Roosje van de Kaap Restaurant
Where? Swellendam
If you are taking a journey along the Garden Route make sure you have an overnight stay in Swellendam to enjoy not only the pictureques scenery but fine dining at the Roosje van de Kaap restaurant. We ...
The Old Gaol Restaurant
Where? Swellendam
Dine alfresco with stunning views of the Langeberg mountains, while at The Old Gaol restaurant. This coffee shop and restaurant serves traditional home made foods such as good old bobotie, old fashioned ...


text:
Systemacy of Managing Small Public Catering Enterprises
The article studies a general notion of public catering as a specific branch of economy. It analyses economic essence of the "management" notion. It reveals structural elements, which carry out management of small public catering enterprises. It identifies typification and types of public catering enterprises. It identifies structural and branch specific features of public catering. It identifies influence of the branch specific features upon the state of management in public catering. It studies the state of the function of public catering enterprises. It identifies systemacy of management of small public catering enterprises. It underlines features of public catering enterprises. It analyses main elements of the mechanism of management of a public catering enterprise: economic (commercial) accounting, internal planning, pricing, financing and crediting.
If you experience problems downloading a file, check if you have the proper application to view it first. In case of further problems read the IDEAS help page. Note that these files are not on the IDEAS site. Please be patient as the files may be large.
Volume (Year): (2013)
Issue (Month): 9 ()
Pages: 383-387
|Contact details of provider:
| Web page: http://www.business-inform.net
No references listed on IDEAS
You can help add them by filling out this form.
This item is not listed on Wikipedia, on a reading list or among the top items on IDEAS.
When requesting a correction, please mention this item's handle: RePEc:idp:bizinf:y:2013:i:9:p:383_387. See general information about how to correct material in RePEc.
For technical questions regarding this item, or to correct its authors, title, abstract, bibliographic or download information, contact: (Alexey Rystenko)
If you have authored this item and are not yet registered with RePEc, we encourage you to do it here. This allows to link your profile to this item. It also allows you to accept potential citations to this item that we are uncertain about.
If references are entirely missing, you can add them using this form.
If the full references list an item that is present in RePEc, but the system did not link to it, you can help with this form.
If you know of missing items citing this one, you can help us creating those links by adding the relevant references in the same way as above, for each refering item. If you are a registered author of this item, you may also want to check the "citations" tab in your profile, as there may be some citations waiting for confirmation.
Please note that corrections may take a couple of weeks to filter through the various RePEc services.


text:
Spreckles Theater, San Diego
Friday, September 10, 1999
There was some insane action on the big screen at Spreckels Theater in downtown San Diego at the double-feature premiere of the new Standard Films and Mack Dawg movies on Friday night, but by far the best turns were performed by the SDFD fire engine that barreled around the corner onto Broadway"in full lights-and-sirens code-three regalia"fishtailing through oncoming traffic right by the cheering throng of moviegoers that'd just exited the theater.
Standard Films' latest offering, TB8: Infinity, wasn't too shabby on the action quotient, either. Although the movie confirmed suspicions that Standard wasn't holding as much of the jaw-dropping, breath-taking AK footage as they usually have, it actually made for a good balance between huge hucks off kickers and the gnarly big-mountain footage the Totally Board series is famous for.
Word filtered through the grapevine this past spring that Standard was forced to nix its annual pilgrimage to Alaska due to shitty and dangerous conditions. But these guys are pros, and they improvised, scoring killer freeriding footage in other locales.
As Infinity started, I feared that Standard had abandoned the big-mountain stuff altogether as I saw huck after huck flash across the screen, but it wasn't long before I got what I wanted"steep pow, trees, cliffs, etc.
So without further adieu, here are some random thoughts from my first glance of Infinity: Jeremy Jones is the hot big-mountain rider right now. Temple Cummins only has one run in the movie, but it's a sweet one. North Cascades Heli looks to have some insane terrain. Jim Rippey knows how to ride a sled (his snowboarding part is good, too). Rippey's also a glamour glutton, as evidenced by how he was lapping up the attention of admirers on a tour of Japan. A segment filmed in Whistler/Blackcomb really captures the place"lots of snow, grim weather, badass pillow drops. Brian Savard rules the it (Whistler, that is), as does Shin Campos. There's a scene in which Savard pays the piper, getting caught in a slough slide"heavy stuff mere mortals rarely get to see. Victoria Jealouse breaks off a well-balanced segment, showing off her serious freeriding skills. It's extremely obvious that Jimmy Halopoff, Dave Downing, and Kevin Jones are super solid all-around riders who know how to boost & spin like crazy AND how to rule the natural terrain. Last but not least, Kevin Jones' dachshund stole the show, full stop.
There was plenty of supertech backcountry freestyle riding in Infinity"probably more than in past TB flicks"but I must admit that throttling back the big-mountain stuff (whether on purpose or out of necessity) and making it a roughly 50/50 blend of freeride/freestyle resulted in a movie that was always entertaining and never monotonous (which was a complaint with some of the other premieres). Throw in a well-balanced soundtrack and stylie interstitial segments, and once again Standard Films has delivered a winner.


text:
"On this lackluster Boxing Day dominated by illiquid moves in every asset class, we thought a few succinct minutes spent comprehending the US and European government policies of social welfare and their outcomes was time well spent. Canadian MP Pierre Poilievre delivers a rather epic speech destroying the myths of US and European 'wealth' noting that "Once the US citizen is in debt, the US government encourages them to stay in debt," noting that "the US government encouraged millions of Americans to spend money they did not have on homes they could not afford using loans they could never repay and then gave them a tax incentive never to repay it." His message, delivered seamlessly, notes the inordinate rise in the cost of all this borrowing, adding that "through debt interest alone, soon the US taxpayer will be funding 100% of the Chinese Military complex." From Dependence to Debt to the Welfare State and back to Dependence, this presentation puts incredible context on the false hope so many believe in the US and Europe. Must watch."
- Source: Zerohedge, view the full article here:


text:
Last Saturday I re-entered the arena with my very first author appearance at the library here in Hillsboro, Oregon. (Arena? Yeah, it was a real Thunderdome) To say it went swimmingly would be both an under and over statement. That's me alright. Under and over, around and through.
Over-wise, it was a small (nay, minuscule) affair with only a handful in attendance, most of which I filled myself (totally grateful for all family and friends that showed).The library events director, the person who helped put this together, said that author readings tend to have very small audiences and that I shouldn't take it personally, which I didn't. She seemed to be pleased that anyone showed at all, which I also didn't take to heart.
From the under-side, I have riding on a cloud ever since. Maybe it's not Cloud 9, but a cumulus not the less, billowy enough for a comfortable trip around the ego. My nerves were pretty much shot going into this day. There has been the inherent fear that I just don't have it in me anymore because I've waited too long to get my own show on the road. I seemed to have forgotten who I was, almost as if I was running away from the real Scott Cherney. Maybe that's why I turned my talk into a near retrospective, a dip into Lake Me and hopefully not delivering my own eulogy. What's closer to the truth was what that day represented to me in general. I told the "crowd" that I was proud to be a part of the library not only that day but to have my works, my books on a library shelf. It's all about validation and for the first time in awhile, I felt comfortable being in my own skin. The road to find myself had led to the Hillsboro Library on Saturday, February 27, 2010. Nervous? You bet your ass I was nervous. I had no one other than my wife to express my trepidation and even that I kept to a minimum. When Grant-Lee Phillips visited Portland in January for a concert, I expressed my anxiety over lunch and felt rather embarrassed that I used my friend as a therapeutic backboard. I mentioned it on the dreaded Facebook and fortunately I know a lot of great people who gave me some much needed encouragement even though I felt so insipidly needy about mentioning my stage fright at all. But hey, I had a lot riding on it. The stakes were my very existence. Fine. I made a u-turn into overstatement again. So sue me. It's my life. I know what it all meant. Part of me actually wanted to fail so i could go back to my non-existence, but fortunately, the real me prevailed and won the day.
I done good.
To paraphrase from my own book PLEASE HOLD THUMBS, since Saturday afternoon, I realized I wasn't dead to the world after all. I am alive.
Now it's time to move forward.
Best Picture: Avatar (Better pictures: The Hurt Locker, District 9 and Up, but Avatar is such an all-around cinematic achievement it could pull out the win, despite the fact that Hollywood really doesn't like James Cameron)
Best Actor: Jeff Bridges (No contest. Kudos to former Palace Showboat Player Jeremy Renner for his nomination)
Best Actress: Meryl Streep (Could be Sandra Bullock, for some unknown reason)
Best Supporting Actor: Christoph Waltz (That's a bing0!)
Best Supporting Actress: M'onique (That's a Yahtzee!)
Best Director: Katheryn Bigelow (Fingers crossed on this one, but could be a slam-dunk, perfect for this tall drink o' water)
Best Adapted Screenplay: Up in the Air (Booby prize for the former front-runner)
Best Original Screenplay: The Hurt Locker (Sorry, Quentin)
Best Animated Film: Up (Where was Ponyo in all this?)
See ya!


text:
Fort Oglethorpe
The 6th Cavalry occupied Fort Oglethorpe during World War I and it was the home of some 4,000 German Prisoners of War. It served as an induction and processing center during World War I and II Also it was a training center for the Women's Army Corps during World War II.
The General Locomotive Chase Monument
The Andrews Raid of April 12, 1862 delivered the first Union soldiers into north Georgia and led to an exciting locomotive chase which lasted seven hours and included about two dozen men. It actually began during the spring of 1862 as the Northern forces advanced on Huntsville, Alabama, heading for Chattanooga, Tennessee when a civilian spy by the name of James J. Andrews led a Union raiding party behind Confederate lines to Atlanta. The force stole a locomotive and racing northward, destroyed track, telegraphy lines and bridges toward Chattanooga, Tennessee. The exciting event become known as the Andrews Raid. They intended to knock out the Western and Atlantic Railroad which supplied Confederate forces at Chattanooga. The twenty-two volunteers were taken from three Ohio infantry regiments and wore plain clothes as they slipped through the lines to Chattanooga and entrained to Marietta. Two of the men overslept on the morning of April 12th as the Andrews party boarded the northbound train and traveled eight miles to Big Shanty (Kennesaw). While crew and passengers ate breakfast, the raiders uncoupled most of the cars. At about 6 a.m. they steamed out of Big Shanty aboard the locomotive General, a tender, and three empty boxcars. Three railroad men noted the action and began their pursuit, eventually overtaking a platform car. Meanwhile, the Andrews raiders steamed out of the Big Shanty depot aboard the locomotive the General. For the next seven hours and 88 miles, Anthony Murphy and William Fuller persisted in their chase, first suspecting the train thieves to be Confederate deserters. Andrew's men cut the telegraph lines and pried up rails. Murphy and Fuller switched locomotives and picked up more men to keep up the chase. When the train theirves reached the Oostanaula River near Resac, that attempted to burn the bridge, but the pursuers were too close behind and could only take on only a little water and wood. At about 1 p.m. it ran out of steam two miles north of Ringgold, as the Southerners aboard the Texas, caught up. All of the raiders were rounded up, but only eight (Andrews included) were tried as spies and executed in Atlanta. The rest either escaped or were exchanged. Nonetheless, the Union train thieves in the Andrews Raid were hailed in the North as heroes, receiving the Medal of Honor. Today, a monument marks the site of the final stop of the Locomotive Chase, approximately one mile north of the Ringgold Depot on the former Western and Atlantic Rail Line.
Catoosa County Wills, Estates, MarriagesCatoosa County Records Available Records to Members of Georgia Pioneers
- Index to Catoosa County Will Bk 1874 to 1961
- Images of Catoosa County Wills 1874 to 1880
Testators: Alexander, Sophronia
Anderson, David
Anderson, Nathan
Anderson, Rachel
Bolton, Peter
Bothar, Daniel
Brotherton, Obedience
Camp, Wilburn
Colhouse, Alexander
Farmer?, Samuel
Fowler, Adolphos
Fowler, Edward
Gray, William
McClure, George
Payne, Thomas
Yates, Presly
- Catoosa County Marriage Index 1858-1887
See how easy it is to view Wills, Estates, Inventories, Returns, Sales online
Genealogy History
Jeannette Holland Austin Profile


text:
District 27-4A is arguably one of the toughest 4A districts in the state of Texas, with all top five teams appearing to be worthy playoff competitors. Steele by far seems to have the most athletic team, but everybody in this district always seems to have a way of beating each other, as evidenced by last year's three-way tie for first place. This should definitely be one of the most interesting districts to watch in the area.
1. Cibolo Steele Knights
• As the third largest school in the district, and the second largest out of the top five it seems unlikely that the Knights will avoid the draw of a Lake Travis team early in the playoffs again. This year's team could have the talent to end the Cavs' season and their win streak if it doesn't happen earlier. Powerful running back Malcolm Brown gained 1,352 yards, and scored seventeen times in a short seven games. Brown isn't the Knights' only weapon though; OU commit wide receiver Sheldon McClain also will add excitement. The offensive line will be anchored by a four-year starter in Austin Landin, along with Drew Phillips. Defensively, Steele will rely on several returning starters, including defensive tackle Corey Hill, and defensive backs DJ Jones and Shane Huhn, each of whom had four interceptions last year.
Article Continues Below
2. Alamo Heights Mules
• Alamo Heights once again will need to replace a Division I quarterback, but QB High School will soon be grooming Cullen Mills to fill in the vacancy. The Mules return six to each side of the ball, including wide receivers John Cabell and Trevor Lovelady, who combined for over 1,000 yards last season. Particularly strong points for Heights will be their O-line and secondary. Cullen Wernette, Nick Morton and Matt Cohen return to the line, while
Lucas Maloy, Michell Dan and Phillip Massari should shore up the secondary.
3. Kerrville Tivy Antlers
• For Tivy, they must figure out how to replace two Division I caliber players in Colton Palmer and Logan Vick. However, Johnny Manziel proved to be a gem of a dual-threat when asked to step up to the plate, gaining over 1,000 yards and 13 touchdowns passing in limited action, while still gaining 806 yards rushing and another 13 touchdowns. Andy Kinnison should be his primary target on offense. Most of the Antler's starting experience comes back on defense, where five starters return including nose guard Brian Sherman, linebacker Tyler Meyer, and safety Michael McKenzie's 152 tackles. So far the offensive line has been Tivy's key trouble area, but a tough pre-district should toughen up the four that are new to the starting role.
4. Schertz Clemens Buffaloes
• Clemens has three solid seniors in running back Stanley Harris, and linebackers Eric Hunsaker and future Red Raider Zac Winbush. Sophomore quarterback Brice McCoy has been given the keys to the offense, but they should be in good hands, as McCoy has only a bright future ahead of him and his big arm. Harris could be the Buffs' best all around player, as he not only will run the ball, he is also all district in the secondary and returned two kickoffs for touchdowns.
5. Boerne Champion Chargers
• A lack of experience could hurt the Chargers, but with the successful history such as Champion (formerly Boerne) has, it shouldn't be so hard to fill in the holes. The Chargers return three burners at wide receiver; James Hurd, Miles Albrect and Blake Allemand, which could make quarterback Taylor Davis' transition into the starting role much smoother than otherwise. Defensive tackle David Roberts leads all returners on defense with 58 tackles, so somebody must step up at linebacker or in the secondary to fill the tackling void that has been left by graduation. The best candidates to come along are linebacker Josh Free and free safety Connor Embry
6. New Braunfels Canyon Cougars
• Don't be surprised if the Cougars finish in the top four either. While they only return three to each side of the ball, those who are there were within twelve points of finishing 5-2 in the district last year, but this was denied in losses to Steele, Clemens and Champion. While returning only three to each side of the ball, a key defensive and offensive player returns in defensive back/wide receiver Tyler Figol. Figol brings his 75 tackles with three interceptions to the table, along with the potential to play more of a role in the passing game this year, after showing the he can catch passes that are not going his way. Throwing his way will be quarterback Lane Fife, who hopes to also utilize receiver Alex Lawrence.
7. Seguin Matadors
• The Matadors should be in a key phase of building their program back. This year's team will rely on their juniors, which could make Seguin's 2010 team a dark horse. Big Brig Gerlich starts at quarterback, and standing 6'4" he won't be afraid to take a hit. Protecting him will be Joseph Cheek and Anthony Razo, and the best defensive line prospect is also a junior; Chris Carpenter. It should be interesting to see how Seguin grows over this coming season.
8. San Marcos Rattlers
• San Marcos' slot-T offense will have several legitimate options in their repertoire with Andre Ray, Tyler Walker and Michael Wilburn. The Rattlers hope to have a déjà vu season like '06, when they went four rounds deep into the playoffs, after an average 5-5 season. Defensively, Walker and Wilburn will also have to get it done in the secondary, while Jeremy Gonzalez will provide the big hits again this year at the linebacker position.