text:
|Publication number
|Publication type
|Application number
|Publication date
|Aug 8, 2006
|Filing date
|Jun 24, 2002
|Priority date
|Jun 22, 2001
|Fee status
|Also published as
|Publication number
|Juan Taboas, Rachel D Maddox, Paul H Krebsbach, Scott J Hollister, Tien-Min Gabriel Chu
|Original Assignee
|The Regents Of The University Of Michigan
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (13), Non-Patent Citations (2), Referenced by (59), Classifications (83), Legal Events (2)
|External Links: USPTO, USPTO Assignment, Espacenet
This application claims the benefit of U.S. Provisional Application No. 60/300,353, filed Jun. 22, 2001 now abandoned.
The present invention relates to methods for producing three dimensional structures with designed and controlled micro-porous and macro-porous locally and globally pored internal architectures using solid free form fabrication techniques.
Tissue reconstruction, regeneration of damaged tissues and organs, artificial organ production, and fabrication of living tissue constructs for proteomic, drug interaction, drug delivery, and pathogen analyses often requires the fabrication of tissue templates (scaffolds). Control over structure porosity in such scaffolds is critical. Precise control of scaffold porosity and internal pore architecture parameters (e.g. pore geometry, size, interconnectivity, orientation, and branching) is necessary to maximize nutrient diffusion, interstitial fluid and blood flow, to control cell growth and function, to manipulate tissue differentiation, and to optimize scaffold physical function and regenerated tissue physical properties.
The ability to manufacture with a range of materials and the ability to control the surface properties of the materials is also critical. The material employed allows control of scaffold degradation and mechanical integrity, cell interaction with the scaffold, and cell function. Material surface roughness, charge, hydrophobicity, and stability/dissolution have also been shown to regulate cell function.
Composites of polymer and ceramic are noteworthy because the mixed material composition can confer favorable mechanical properties, including strength via the ceramic phase, toughness and plasticity via the polymer phase, and graded mechanical stiffness. Biologic advantages include enhanced control over cell differentiation and the potential to deliver multiple biofactors, including growth factors, gene therapy vectors, and cells. Composite scaffolds may prove necessary for reconstruction of multi-tissue organs, tissue interfaces, and structural tissues including bone, cartilage, tendon, ligament and muscle.
However, the mechanical property requirements for hard tissue repair are difficult to satisfy using polymer/ceramic blend composites in which the ceramic and polymer components are mixed and occupy the same space. This is because large amounts of ceramic must be incorporated, making fabrication difficult. Therefore, composites with discrete regions of fully sintered HA are desirable for bone/soft tissue interfaces.
Composites scaffolds with discrete regions of materials are desirable for controlling structure mechanical properties, degradation kinetics, and biological interaction, as well as for producing structures that mimic natural tissue properties, structures that deliver several biofactors (drug/gene) vehicles, and living structures that contain multiple tissue components.
Although several manufacturing techniques exist to create scaffolds, precise control of pore geometry and interconnectivity is difficult. Conventional sponge scaffold manufacturing methods are capable of producing structures with locally-porous internal architectures from a diverse array of materials. The local pores in such scaffolds are voids characteristically defined by small struts and plates or spherical and tubular cavities generally less than 300 μm in diameter. These local pores are interconnected within local regions of the scaffold microstructure.
While these local pored scaffolds may include interconnected micro (less than 100 μm wide) pores that may comprise a continuous conduit throughout a scaffold, the pore connectivity is not an intentional result of an a priori global design. Rather, the connectivity is a random product of variable, local void interconnections that are affected by polymer processing parameters. Such random connections may not provide optimal scaffold permeability for tissue ingrowth nor optimal connectivity to maximize regenerated tissue mechanical properties.
Unlike sponge methods, micro-milling, textile, and direct solid free form (SFF) fabrication can be used to create scaffolds with good global pore control. The global pores take the shape of channels with non-random interconnectivity that are designed a priori to interconnect on a global scale across the scaffold. Unlike micro-milling and textile fabrication, full control over global pore architecture in 3D space is available with SFF. Unfortunately, the number of materials that can be used for each direct SFF manufacturing method is limited by the manufacturing process. Additionally, the incorporation of local pores in a direct SFF fabricated global pored scaffold is difficult.
Global and local pores are differentiated based on the extent of non-random designed interconnectivity. These pores may be sub-classified by size as either microporous (width less than 100 μm) or macroporous (width greater than 100 μm). Current direct SFF manufacturing is limited to manufacturing global pores with widths greater than 100 μm because of inherent limitations on the minimum size feature.
Direct SFF fabrication, also known as rapid prototyping, refers to a set of manufacturing processes that consists of building a 3D part in a layered fashion based on a computer representation. The part is often post processed (e.g. cleaning, curing, finishing) to yield the final product.
SFF has been used to mimic the 3D shape of macroscopic structures. However, a versatile method to manufacture biomimetic structures in the microscopic range, for example structures that mimic human trabecular bone or 3D vascular branching, has not been available. Such biomimetic strategies are desirable to replicate native tissue structure and function.
The need for 3D scaffolds with controlled local/global and micro/macro pores is great. However, current 3D local-pore scaffold fabrication methods do not allow fabrication of pores with designed geometry, orientation, branching, interconnectivity, and size. Global-pore manufacturing methods, including current SFF fabrication techniques, are not suitable for varied micro-pore structure fabrication and provide limited incorporation of local-pores within a global-porous structure. Additionally, the need for 3D scaffolds made from a variety of materials is great. Further, control over the discrete spatial location and surface properties of varied materials within the same scaffold is desired.
A method of fabricating tissue scaffolds and other similar structures is provided. The method includes computationally designing the desired structure, fabricating a mold for the desired structure, casting the desired structure in the mold, and removing the mold from the desired structure. Methods for post modification of the desired structure are presented. The step of computationally designing the desired structure preferably includes at least one computer aided design technique and/or an image based design technique. The step of fabricating the mold for the desired structure preferably includes solid free form techniques such as 3D printing, fused deposition, stereolithography, selective laser sintering, direct material deposition, or layered object manufacturing. The step of post modification of the structure preferably includes stabilized plasma functionalization.
In accordance with the foregoing, the present invention provides three dimensional structures with designed and controlled local-porous and global-porous internal architectures. Enhanced control of global-pore orientation, branching, interconnectivity, size, and geometry is provided compared to current SFF techniques. Additionally, a larger variety of local-pore architectures are possible. The disclosed methods allow for fabrication of structures from more materials than currently possible with existing SFF fabrication and for post fabrication surface modification of these materials. Moreover, methods to fabricate mechanically robust porous composite structures containing a number of plastics or combinations of plastics and ceramic (and/or cements) are provided. These methods provide for 3D structure fabrication from more material blends and combinations of discrete material regions than currently available. Additionally, the mechanical strength of structures fabricated using these methods both in the build direction and at the composite material interface regions is improved. Novel designs and constructs created from these structures are presented for use as tissue regeneration scaffolds, bioreactor culture substrates, perfusion (advection) assisted diffusion exchangers, 3D templates for screening material/cell interaction with identical architectures, and living tissue constructs for screening the effects of pathogens and drugs derived from proteomic data and pharmacology science on human tissues in vitro. The disclosed methods enable the fabrication of engineered tissues that are mechanically and biologically functional. Constructs fabricated with this technology constitute a large reduction in drug screening costs and an advancement in "lab on chip", biosensor, and cell manufactured drug technologies.
Further areas of applicability of the present invention will become apparent from the detailed description provided hereinafter. It should be understood that the detailed description and specific examples, while indicating the preferred embodiment of the invention, are intended for purposes of illustration only and are not intended to limit the scope of the invention.
The present invention will become more fully understood from the detailed description and the accompanying drawings, wherein:
The following description of the preferred embodiments is merely exemplary in nature and is in no way intended to limit the invention, its application, or uses.
To circumvent the limitations of the prior art and combine the benefits of local pore manufacturing and direct SFF fabrication, an indirect SFF scaffold manufacturing technique is provided. More particularly, the present invention provides a set of molds, casting methods, mold removal, and structure post fabrication modification techniques that are compatible with image-based design (IBD) methods and with solvent, melt, and slurry casting of polymers and ceramics.
The indirect SFF fabrication method of the present invention combines image based design with SFF fabrication to precisely control scaffold architecture and material composition. This method provides simultaneous control of scaffold external shape to match implant site geometry, internal architecture to yield biomimetic structures, and global pore structure in 3D space (interconnectivity, branching, geometry, diameter, and orientation). Additionally, the indirect SFF method is compatible with most local pore polymer scaffold processing techniques, thereby providing inclusion of local pores within a scaffold global pore framework.
The molds and methods of the present invention allow 1) fabrication of discrete composite scaffolds made from materials that have incompatible processing parameters such as sintered ceramics and polymers, 2) fabrication of scaffolds from the largest pool of materials possible, 3) production of local and global pores within one scaffold, and 4) fabrication of structurally biomimetic scaffolds. These multi-pore-architecture scaffolds are ideal for several tissue engineering applications. For example, the global pores may serve as channels to perfuse culture media or augment diffusion of nutrients to cells seeded within local pores. In turn, the local pores provide a potential space for tissue growth. Global pores may also serve as conduits for blood vessel ingrowth or to anchor the scaffold once implanted in a host.
Referring to
From block 12, the methodology advances to either block 14 or block 16. In block 14, the ceramic or cement mold casting is performed in the premold. Thereafter, the final ceramic global pore mold is provided. From block 14, the methodology advances to block 18, 20, or 22.
In block 18, polymer melt casting is performed. In block 20, polymer solvent casting is performed. In block 22, local pore fabrication is performed by way of PLA casting.
Referring again to block 16, in this step of the methodology PSA removal is performed. Thereafter, the final wax global pore mold is provided. From block 16, the methodology continues to block 22. As recited above, in block 22, local pore fabrication is performed by way of PLA casting.
Referring again to blocks 18 and 20, from either of these steps, the methodology advances to either block 24 or 26. In block 24, mold removal is performed by way of an acid. In block 26, partial mold removal for HA/PLA composites is performed.
Referring again to block 22, after this step, the methodology advances to either block 28, 30, or 32. In block 28, emulsion-solvent diffusion is performed. In block 30, emulsion-freeze drying is performed. In block 32, solvent-porogen leaching is performed.
From any of blocks 28, 30, and 32. The methodology advances to either block 34 or 36. In block 34, ceramic mold removal is performed via an acid. In block 36, wax mold removal is performed by way of cyclohexane.
As noted, the material type from which scaffolds are manufactured influences cell function as well as protein absorption and confirmation. One aspect through which materials can regulate cell function is via material surface properties that may include hydrophilicity, zeta potential, charge, and roughness. Additional control is afforded by post modification of these surfaces. Cell culture surfaces may be coated with ligands that modulate cell functions including cell attachment, spreading, proliferation, differentiation, and synthesis of secreted cell products including extracellular matrix. Post modification of surfaces may also be used to tether desirable compounds, including viral particles and drugs for delivery to cells, and enzymes and/or catalysts for chemical reactions. These modifications are referred to as functionalization of the material surface. Several functionalization methods are applicable to the materials manufactured using the disclosed fabrication methods. The functionalization methods of the present invention are more stable than current state of the are because they inhibit thermodynamically driven detachment from and internalization into the material of the tethered substances.
The method of the present invention will now be described in greater detail. In the first step, the computational design of the desired structure is performed. The external scaffold design and global pores (macro-porous architecture) can be created using either traditional computer aided design (CAD) techniques that are computational geometry based or a newly developed image-based design (IBD) technique whose details are fully described in co-pending patent application Ser. No. 10/178,419 entitled DESIGN METHODOLOGY FOR TISSUE ENGINEERING SCAFFOLDS AND BIOMATERIAL IMPLANTS to Hollister et al. which is expressly incorporated by reference herein. The IBD methods allow design of complex internal porosity (including pore geometry, orientation, interconnectivity, and branching) and of scaffold external shape to match implant site requirements using a patient specific image.
Scaffold designs can also be generated using commercial CAD software, but integration of scaffold design with imaging data and creation of complex porosity is not optimized.
IBD or CAD created mold designs are saved either as stereolithography triangular facet data (*.stl), stereolithography contour data (*.slc) or SolidScape contour data (*.slf) and are then converted to appropriate input files for computer aided manufacturing (CAM) using commercial software for 3D printing (e.g., ModelWorks, Solidscape, Inc) and stereolithography (e.g., 3D Lightyear, 3D Systems).
In an experimental run of the present invention, the external scaffold shape and global-porous architecture were designed using an image based design (IBD) method. Orthogonally pored scaffold designs were 8 mm diameter by 8 mm height cylinders with either square or circular equi-spaced, orthogonally branching and interconnected global pores channels (global pore volume fraction =50%,
The second step of the method of the present invention involves mold fabrication. The casting methods allow fabrication of complex global porosity (including pore geometry, orientation, interconnectivity, and branching), of varied local pores, and of scaffold external shape to match implant site requirements. Molds for casting the designed scaffold are fabricated using direct deposition of the mold material, preferably by using 3D printing. Several other SFF technologies can be used to fabricate molds, including fused deposition (FDM), selective laser sintering (SLS), direct material deposition (DMD), stereolithography (STL), layered object manufacturing (LOM), and 3D printing of binders or solvents on powdered material beds (3DP).
Molds can be fabricated from waxes (any of various natural, oily or greasy heat-sensitive substances, consisting of hydrocarbons or esters of fatty acids that are insoluble in water but soluble in several organic solvents), plastics (any of various organic compounds produced by polymerization), sugars (any of a class of water-soluble crystalline carbohydrates), ceramics (any of various hard, brittle, heat-resistant and corrosion-resistant materials made by shaping and then firing a mineral composition at a high temperature), and cements (any of building materials that form a paste when mixed with water and harden into a solid).
The useful properties of these materials in casting are as follows. Waxes (e.g. paraffin and beeswax) and plastics (thermosetting and thermoplastic, e.g. polysulphonamide, polystyrene, polyesters, silicones) are combustible and/or meltable, and soluble in organic solvents. As such, wax and plastic molds can be removed by using solvents and heating temperatures exclusive for the cast material. Sugars (e.g. starch, cellulose, and saccharides) are water-soluble. As such, sugar molds can be removed by soaking in water and other hydrated solvents. Cements and ceramics (e.g. plaster of Paris, hydroxyapatite) are dissolvable and breakable but can withstand high temperatures and pressures. As such, cement and ceramic molds can be used for injection molding and melt processing.
In the preferred method, wax or polysulphonamide molds are created using a SolidScape, Inc. (Merrimack, N.H.) Model Maker II 3D printer (
Cement molds are created by casting cement in an inverse mold made of wax or polysulphonamide fabricated on a Model Maker II printer. A final cement mold is made by removing the inverse mold via solvation or heating. Ceramic molds are created by casting a ceramic/binder (HA and acrylic) slurry in an inverse mold made of wax or polysulphonamide or by casting in an inverse mold made of epoxy resin (or other stereolithographic resin) (SL 5170, Vantico, Inc. Basel, Switzerland) fabricated using a 3D Systems (Valencia, Calif.) SLA 250 System.
In a preferred method, ceramic molds or scaffolds are created by casting a low-viscosity ceramic slurry made of tricalcium phosphate (TCP) powder, acrylates, dispersants, and initiators. The particle size of the TCP powder should be below 100 microns to allow final sintering, preferably lower than 30 microns. The viscosity of the acrylates should be lower than 100 mPa�S to ensure a low viscosity of the final slurry. The dispersants can be either anionic or cationic dispersant, preferably a mixture of the two. The initiator should allow the curing process to take place below the melting point of the wax mold, preferably the peroxide derivatives or similar. In particular, the a preferred formulation is comprised of 40 volume percent of tricalcium phosphate powder in the size of 1 micron in diameter. The balancing reactive medium is a 50:50 mixture of isobornyl acrylate and propoxylated neopentyl diacylate. The ceramic powder is dispersed with 2.5% of a cationic dispersant (a polypropopoxy quaternium ammonium acetate) and 2.5% of an antionic dispersant (an aromatic phosphate ester). The particular formulation exhibits a Newtonian viscosity of 23 Pa�S at a shear rate of 10 sec−1, a particular flow behavior appropriate for casting in the molds. A thermal initiator, benzoyl peroxide is added to the slurry just prior to casting. The thermal initiation mechanism employed in this disclosure for this ceramic slurry is different from prior art initiation mechanisms. An accelerator, N,N-dimethyl p-toludine, is added to the slurry to allow the initiation of the curing reaction at room temperature, below the melting point of the wax mold. The addition of the N,N-dimethyl p-toludine eliminates the need to heat the mold to 80� C. to initiate the reaction. After curing. the cured ceramic slurry is heat-treated to 1150� C. to remove the cured acrylates and to sinter the TCP powder.
Preferably, cast ceramic molds are subject to a burn out cycle that removes organic binders and the inverse mold. The ceramic pre-mold is then sintered (at about 1300� C. maximum) yielding the final ceramic mold.
In the third step of the method of the present invention, the material selected for the structure to be formed is cast in the mold. The method of the present invention enables novel 3D scaffold fabrication from diverse materials with controlled local/global-porous architectures because the structures are created by casting in IBD or CAD/CAM designed and manufactured molds. These casting techniques allow for the fabrication of scaffolds from more materials than possible with direct SFF techniques. Furthermore, by employing mold materials that shrink during curing or burnout, scaffolds with global pores smaller than the minimum feature size of current direct SFF manufacturing methods can be fabricated.
Advantageously, the material chemistry need not be altered for manufacturing compatibility. Additionally, both solution and molten processing of the material is possible with these casting methods. Most conventional local-pore polymer processing techniques are also compatible with these methods. Local-pore manufacturing methods may be enhanced with an annealing step after polymer casting in order to fuse polymer particles together and increase scaffold mechanical integrity. Coupled with the high-resolution control of global-pore structure dictated by the mold design, controlled local/global-pore internal architecture is possible within one structure. As with other SFF techniques, external scaffold geometry can also be controlled.
The following several paragraphs describe a few casting processes suitable for use in the method of the present invention. The steps for some processes are illustrated through reference to their respective block numbers from one example of the embodied methodology in
First, a process for solvent casting in wax and plastic will be described. Particular plastics are compatible with casting in select waxes and plastics. For example, polymers of α-hydroxyacids and urethane can be solubilized in organic solvents and cast in wax molds. Micro-porosity is created by using any conventional technique. Polylactide (PLLA or PLA) and copolymers of polylactide and polyglycolide (PLGA) are dissolved in methylene chloride. Polyglycolide (PGA) is dissolved in 1,1,1,3,3,3-hexafluoro-2-propanol.
Second, a process for solvent casting in ceramic and cement will be described. Plastics solubilized in solvents that dissolve wax and plastic molds can be cast in ceramic and cement molds (
An example of solvent casting in ceramic will now be made with reference to
Scaffolds containing local pores were made using several methods. For porogen leaching, 104–124 μm sieved NaCI grains were packed into ceramic molds and cast with 7.5% w/v PLA in chloroform. Solvent was evaporated under 15″Hg vacuum overnight (
Third, a process for casting uncured material in wax and plastic will be described. Materials that cure without heat generation or that are modestly exothermic upon curing may be cast in wax and plastic molds. Silicone elastomer (Sylgard 184, Dow Corning Corp, Midland, Mich.) may be cast and cured in wax and polysulphonamide molds. After casting, the elastomer is preferably placed in a 37� C. oven to speed curing over 12 hrs.
Fourth, a process for casting uncured material in ceramic and cement will be described. Materials that are exothermic upon curing may be cast in ceramic and cement, including poly(methylmethacrylate) and commercial bone cements.
Fifth, a process for melt casting in ceramics will be described. Ceramic and cement molds enable melt processing of plastic scaffolds. PLLA, PLGA, PGA, and polyurethane (PU) scaffolds have been manufactured using this method. Ceramic molds are heated in an oven set to 10–20� C. above the plastic melting point (Tm) (PLA Tm=120� C., PGA Tm=150� C.). Molten polymer is allowed to flow into the mold. The molten polymer/molds are then subject to a vacuum while heated in order to remove any air from the mold and melt (
An example of a process for melt casting in ceramics will now be described with reference to
Sixth, a process for injection molding in ceramics will be described. Ceramic and cement molds of designed macro-porosity may be used for injection molding of plastics in melt or uncured forms. Injection molding facilitates material infiltration into the mold, as the material can be processed under high temperatures and pressures. Molds should be places in dies that mount within conventional injection molding devices. These dies may be manufactured to fit external mold shape or the molds may be manufactured with extra material so they fit within standard dies.
Since local injection molding utilizes pressure above atmosphere and no material solvents, most conventional micro-pore inclusion methods should not be used. However, suitable porogens can be included as in melt casting. The porogens are then leached out of the final material.
Seventh, a process for composite material casting will be described. Composite materials may be manufactured using several methods. As used herein, a composite means a structure composed of two or more materials that have different chemical structures or a structure composed of the same material but with differing spatial density or molecular weight distributions. The materials composing a composite may occupy the same volume as a blend, occupy different volumes of one structure as in a discrete composite, or may be combinations thereof.
Blend (ceramic or cement)/plastic composites are manufactured by melt or solution casting of plastics containing ceramic particles of desired size and geometry. HA/PLLA blend composites with designed macro-porosity have been manufactured using melt processing.
Non-blend composites with discrete regions of differing plastics can also be manufactured using solvent and melt processing. In this case, the molds are designed to contain two or more exclusive pore or lattice networks. One plastic is cast per network, interdigitation of the differing plastics is controlled by the mold structure and by using plastics with exclusive solvents, by casting plastics to controlled depths within the mold, or by selectively removing part of the mold before casting subsequent plastics. Exclusive solvents are those that will dissolve one set of the desired materials but not the other sets.
Further, non-blend composites with discrete plastic and ceramic/cement regions that are mechanically interdigitated can be manufactured using solvent and melt processing. The methods for fabricating these structure may also be used to fabricate discrete composites from metals, plastic, polymers, ceramics, and cements. Three methods are preferred to create non-blend (ceramic or cement)/plastic composites.
In the first method, the ceramic or cement scaffold is created using slurry casting. Then a wax or plastic mold is built upon the ceramic/cement base structure. This mold is used to cast the plastic or to cast a second ceramic/cement mold. This second ceramic/cement mold is of different solubility than the scaffold on which it is constructed. For example, a plaster of Paris mold may be built on a HA scaffold. The desired plastic is melt or solvent cast into the secondary mold and into regions of the base ceramic/cement scaffold that have been designed with pores for plastic infiltration and mechanical interdigitation of the plastic region with the ceramic/cement region. The secondary mold is then melted or dissolved away. This method uses two or more molds, one mold to create the base ceramic/cement scaffold and the other secondary molds build upon the ceramic/cement scaffold for plastic casting. This method provides for fine interdigitation control.
In the second non-blend composite manufacturing method, only one ceramic/cement scaffold is used. This structure is created with additional porous or lattice networks that serve as the mold for the plastics to be cast. The additional networks may not be continuous with other pore networks within the ceramic/cement scaffold. This facilitates plastic melt or solution casting of the desired plastic structure without blockage of other pore networks within the ceramic/cement region. Ceramic/cement free regions of the scaffold are then created by selective application of ceramic/cement solvents. An HA/PLLA composite scaffold has been created with the second method. An example of such a composite scaffold is shown in
In the third method, polymer ceramic composites are made by assembly of the polymer and ceramic regions. The ceramic is heated and the polymer pressed onto the ceramic. The polymer melts into the anchor pores in the ceramic, locking the two phases together. The polymer also may be fused to the ceramic via the use of polymer solvents. This procedure provided course control of interdigitation. The benefits of this method is that it avoids exposure of the polymer phase to ceramic solvents
Two examples of processes for discrete composite manufacturing will now be detailed with references to
Polymer discrete composite PLA/PGA scaffolds were made with a two part casting process (
Ceramic/polymer discrete composites were made with a casting and etching process (
The fourth step of the method of the present invention involves mold removal. There are three preferred techniques to separate molds from the desired cast structure. The following lists compatible methods for varied casting techniques.
First, a process for fracturing the mold could be used. This method is adequate for scaffolds of limited internal architectures.
Second, a process for melting wax and plastic molds could be used. Molds can be melted away from cast material if the melting point of the mold material is lower than that of the cast material. For example, the preferred polysulphonamide melts between 95–107� C. while the preferred wax melts between 50–75� C. Wax molds may be removed by melting in an oven at 85� C. for one hour (PSA Tm=95–107� C., Wax Tm=50–75� C.). Residual wax may be removed by washing in cyclohexane for one hour and air drying.
Third, a process for dissolving wax, plastic, ceramic, and cement molds could be used. Wax, plastic, ceramic, and cement molds can be dissolved away from the cast material. The choice of solvent depends on the mold and cast material. Unsuitably buffered acids will increase hydrolysis of plastics. Partial removal of the mold via dissolution of select regions can be achieved using selective spatial application of mold solvents. Table 1 lists suitable solvents for some processing methods.
Mold Material
For ceramic molds, only solvent dissolution was possible. Molds were placed in RDO (APEX Engineering Products Corp, Plainfield, Ill.) under fluid agitation for 1–6 hours. Complete mold removal was verified by x-ray. For HA/PLA composites, the ceramic was selectively removed using a surface etching technique. The cast structures were embedded in wax and the wax selectively removed from surfaces where ceramic erosion was to occur. The depth and uniformity of erosion was regulated by exposing different regions of the surface over time to RDO and adjusting the time in RDO. The eroding ceramic struts that defined polymer pores were cleaned of ceramic particles with pressurized air several times during etching. The scaffolds were then washed with distilled H2O and dehydrated in graded ethanol washes.
Fourth, a process for porogen removal may be used. Hydrated solvents can be used to remove salt and sugar porogens. During the process of ceramic leaching, salt porogen particles are removed as well. Hydrated solvents were used to remove salt porogens. For ceramic mold, salt particles were removed by RDO soaking. For wax molds, scaffolds were hydrated using 15 minute washes in the following order: 1:1 (cyclohexane:ETOH), ETOH, 19:1, 17:1, 14:1, and 6:1 (ETOH:H2O), and H2O. Soaking the molds in H2O for 48 hours removed the salt. Scaffolds were then dehydrated in ethanol washes.
According to another aspect of the method of the present invention, defined chemistry for molecule adhesion on 3D scaffolds can be implemented. That is, scaffolds can be modified with specialized molecular species in order to facilitate substrate dependent chemistry, ligand attachment, cell culture in vitro, and engineered tissue and organ implant fabrication and host integration via control of seeded or migrating cell function.
Three examples of processes for functionalization of scaffolds surfaces are now detailed. Some of the presented methods are only applicable to Sylgard (a silicone) and other plastics that cure using similar addition chemistry.
According to the first technique, functional activation is performed via adsorption. For example, a buffered solution is used to adsorb solubilized ligand onto the scaffold surface. The adsorption is regulated by the hydrophobicity and surface charge of the scaffold material. This technique does not yield a stable surface.
According to the second technique, functional activation is performed via addition reaction. For example, a novel Sylgard functional activation technique may be used. This technique utilizes amide chemistry to bind cell ligands or other chemical species to the silicone surface that has been derivitized with carboxylate and amine moieties. This functional activation can be performed under several conditions, two of which are outlined below.
In the first functional activation condition of the second functionalization technique, the mold used to cast the silicon is coated with a vinylated carboxyl or amine derivitizing compounds (e.g., acrylic acid, vinylacetic acid, 4-pentenoic acid, allylamine, acrylamide). The Sylgard is then poured into the mold. The derivitizing agent becomes covalently bonded to the Sylgard during curing. This occurs because the vinyl groups of the derivitizing compounds react with electrophilic Si groups in Sylgard via addition under a platinum (Pt) catalyst. The desired ligand is covalently coupled to the derivitizing agent. An example amide chemistry employed to achieve covalent coupling utilizes 1-[3-(dimethylamino)propyl]-3-ethylcarbodiimide hydrocloride (EDC). The benefit of this derivitization method is that it utilizes the cross-linking chemistry of Sylgard. Sylgard cures via addition cross-linking between nucleophilic dimethylvinyl-terminated polydimethylsiloxane and electrophilic methylhydrogen siloxane groups. The Sylgard pre-polymer contains the Pt catalyst.
In the second functional activation condition of the second technique, the vinylated derivitizing compounds are mixed and allowed to cross-link with the Sylgard curing agent. Since the derivitizing compounds contain only one vinyl group, no cross-linking occurs between curing agent molecules and the mixture remains liquid. The stochiometry of the reaction is regulated such that free methylhydrogen siloxane groups remain available for cross-linking with the Sylgard base material. The curing agent/derivitizing molecule mix is then added to the Sylgard base material and allowed to cure. The desired ligand is covalently coupled to the derivitizing agent. The second method is better suited to create functionally activated surfaces using conventional porogen leaching micro-pore fabrication techniques.
According to the third functionalization technique, surface modification is performed via plasma activation, derivitization, and subsequent functionalization via covalent modification. For example, the silicone surface is first activated via plasma discharge. Plasma discharge is also referred to as glow discharge or radio-frequency (RF) discharge. Plasma activation is applicable to more materials than the Sylgard chemistry outlined above (e.g. poly(α-hydroxyacids), poly(anhydrides), poly(propalenefumarates), polycaprolactone, polyurethanes, and silicones). Therefore, this technique, as defined by the amide chemistry functional groups used, can be used to functionally activate several materials for cell attachment, including silicone substrates currently used in cell culture (e.g., Flexcell International Corp cell culture plates, Hillsborough, N.C.).
The plasma derivitization is preferably performed under two conditions, direct deposition of the derivitizing group within the plasma chamber or plasma ionization of the surface within the plasma chamber and subsequent addition of the derivitizing group external to the chamber. In the first plasma derivitization technique, the vinylated functional compound or compounds is placed along with the silicone in the plasma chamber. The functional compound is vaporized under plasma. The vinyl groups then react with the ionized silicone surface produced under plasma. The plasma methods and gases to be used can be varied (e.g. microwave or glow discharge using argon or xenon).
In the second plasma derivatization technique, the silicone is subject to plasma such that the surface becomes oxidized. This can be achieved with a small amount of oxygen gas in the plasma chamber or by exposing the silicone to an oxygen atmosphere after plasma modification. The vinylated derivitizing molecule is then covalently bonded to the silicone surface using EDC and/or tosylation chemistry. The functionalized surface may also be stabilized by coupling functionalizing molecules to each other using a similar technique as above.
According to yet another aspect of the present invention, a process for stabilization of modified surfaces yielded using the second and third functionalization techniques is provided. Derivitizing molecules are subject to material surface rearrangements when they are covalently coupled to the material. The derivitized surface is stable under low surface energy interfaces like water and cell culture media. When exposed to air, however, the surface energy increases. The derivitizing groups may then be internalized into the material. This internalization process is driven by the thermodynamically favorable decrease in enthalpy and increase in entropy.
In order to stabilize the derivitized surface, derivitizing groups can be covalently coupled to each other via a primary cross-linker. Consequently, the derivitizing groups cannot be internalized because the cross-linker inhibits bond rotation of the polymer backbone. In order to maintain versatility to attach any desired cell ligand or chemical species using amide chemistry, the primary crosslinker should also contain active carboxyl and/or amine sites. Ideal molecules include, but are not limited to, polylysine, polyarginine, polyapartate, and polyglutamate. These molecules may be linked to the active carboxyl and amine groups of the derivitizing molecule directly using EDC to activate carboxyl groups or indirectly using secondary cross-linkers such as formaldehyde and glutaraldehyde. Cell ligands and other chemical species can be covalently bonded to the primary crosslinkers via the same EDC and secondary cross-linker chemistry.
Control over molecular orientation on the material surface is also desirable. For example, the orientation of cell attachment ligands has been shown to regulate cell binding. Additionally, immobilized enzymes function best if the active site is not occluded by the material surface. Control over orientation of immobilized molecules is possible using the covalent functional activation techniques presented herein. The adsorption method does not afford this control. By altering the use of crosslinkers and EDC, the steric orientation of cell attachment ligands can be inverted. For example, the cell attachment molecule GRGDS (glysine-arginine -glysine-aspartate-serine) can be attached to a surface with the amine or carboxyl end facing away from the surface.
Except for the functionalizing method where the derivitizing molecule is added to the curing agent, the above functionalizing methods only alter surface properties of the desired material. Therefore, the bulk properties a material such as elastic modulus and degradation rate are maintained. This makes these methods more favorable than bulk material modification methods.
Several simulations and working models of the present invention will now be described.
First, a 3D silicone and/or hydroxyapatite (HA) ceramic cell culture substrate with cell adhesion ligands will be described with reference to
The macro-porous network is designed to facilitate nutrient diffusion and waste clearance and provide a network for perfusion assisted mass transport. This in turn enhances cell viability and function. The surfaces of these scaffolds are then derivatized with cell adhesion ligands. Together with the scaffold material, select adhesion ligands can be used to control cell proliferation and biosynthesis rates, including extracellular matrix and soluble protein production. Sylgard and HA scaffolds functionally activated with fibronectin (
Advantageously, the cell culture scaffolds presented may be reused, depending on the fabrication material selected. However, several cell attachment ligands may be damaged by cell culture passaging methods. In order to maintain predefined surface chemistries, these structures would preferably be recoated. Since most of the surface derivitizing molecules are stable under common digestions used during cell passage (e.g. trypsin, ethylenediaminetetraacetic acid, and collagenase), the materials only need be recoated with attachment ligands.
Second, it is predicted that the present invention will be suitable for fabrication of an immobilized cell/enzyme biosensor.
Third, the present invention may be used to fabricate structural prosthetics: plastic/(ceramic/cement) composites for bone and soft tissue. As an example, a ceramic/polymer composite for tendon and ligament replacement will be described. Composite ceramic/plastic polymer scaffolds have been fabricated for replacement of damaged ligaments and tendons in accordance with the teachings of the present invention. Several designs have been created that consist of three parts: the rigid base, the interface region, and the flexible plastic polymer block.
The materials used to fabricate the scaffold may be HA and α-hydroxyacids, all FDA approved materials for human implant use. The scaffold long axis traverses from the base, through the interface region, and along the polymer block.
The ceramic base is designed to be implanted into bone at the tendon insertion site. It contains a macro-porous architecture designed to promote angiogenesis, bone ingrowth, and anchoring of the implant into bone. The ceramic base can be fabricated with a screw shaped external architecture so that initial bone anchoring is provided via surgical screwing into place. The base can also be fabricated with holes for initial bone anchoring with screws or pins. After bone ingrowth and tissue anchoring of the base occurs, the screws or pins may be removed or resorbed.
The polymer block consists of longitudinal polymer fibers that are interconnected through an off axis lattice network. The longitudinal fibers carry the tendon and ligament tensile loads. The interconnecting lattice provides some load transfer between fibers and block structural rigidity in the off axes. The lattice network can be orthogonal to the scaffold long axis. Several lattice networks are suitable. The interspace between fibers is composed of locally porous polymer that serves as a carrier for cells and a template for cell migration and tissue ingrowth. Alternatively, the interspace may be filled with a polymer gel such as collagen type I, fibrin, or alginate.
A preferred lattice consists of individual struts interconnecting longitudinal fibers at an acute angle, akin to trusswork. As longitudinal fibers are stretched during loading, the interconnecting lattice deforms and orients along the scaffold long axis, thereby bearing some longitudinal loading. The lattice network carries an increasing percent of the overall scaffold load as the scaffold load increases. This endows the scaffold with nonlinear stiffening behavior. This is beneficial in that it provides a flexible scaffold that behaves similarly to native tissue and delivers substrate loads to cells adherent to the polymer surface, while becoming very stiff at high loads to minimize the risk of scaffold rupturing. Additional designs for the ligament portion that include wavy fibers to match the nonlinear mechanical behavior of ligaments or other soft tissues can be fabricated using this technique.
The polymer block is anchored to the muscle or bone attachment site using surgical sutures, staples, or screws. However, the ligament scaffold can be fabricated with two ceramic bases at both ends, avoiding the use of staples or screws during surgical implantation. This is possible because ligaments have two bone insertion sites at each end. These ligament scaffold ends are anchored into the two bone insertion sites occupied by the damaged ligament.
The interface region is composed of the polymer fibers anchored in the ceramic base through an interconnected network. A preferred interface design consists of ellipsoidal longitudinal channels interconnected with orthogonal links. The ellipsoidal structure and orthogonal links increase the pullout strength of the fibers from the ceramic base.
The above scaffolds have been manufactured from HA/PLLA and the design of the polymer ceramic interface shown in
Fourth, the present invention can be used to create a ceramic/plastic composite for joint resurfacing. Diarthroidal joint resurfacing and construction of living artificial joints is hampered by the lack of tissue integration with preexisting host tissue, inadequate mechanical properties of the regenerate cartilage, and failure at the bone/cartilage interface. Methods similar to the ligament/tendon engineering methods have been developed in accordance with the teachings of the present invention to fabricate cartilage implants with a ceramic/cement base. This method allows for the entire resurfacing of joints as the implant becomes anchored to the subchondral bone bed via bone ingrowth into the ceramic base.
Again, the implant consists of three parts: the rigid base, the interface region, and the flexible plastic polymer bloc. The ceramic base is similar to the ligament/tendon base and may be manufactured from HA, an FDA approved material. It can be fabricated to encompass an entire joint surface. Referring again to
On the other hand, in the case of cartilage engineering, the polymer block is a sponge with defined local/global-porosity that provides mechanical force support and load transfer to the ceramic block. The sponge may be used to carry cells and/or contain hydrogels. In the preferred embodiment, the scaffold is loaded with chondrogenic cells during implantation. Several sources for these cells are well known.
The cells can be delivered without carriers or in secondary gels. The cells produce a cartilagenous matrix in vivo providing full load bearing capability and mechanical function by the implant. The hydrogels, including polyvinyl alchohols, polyethylene glycols, alginates, and pluronics (PEO-PPO-PEO, i.e. PEO=poly(ethylene oxide), PPO=poly(propylene oxide)), can be used to enhance maintenance of chondrogenic cell phenotype and mechanical load bearing. The polymer bloc sponge can be made of α-hydroxyacids that are FDA approved. The preferred polymer bloc material is PLGA, as the increased hydrolytic rate of the polymer enhances chondrogenic phenotype.
The interface region consists of a pore network within the ceramic base into which the polymer sponge interdigitates and is anchored. Pores may be fabricated that allow osteogenic cells from the base and chondrogenic cells from the polymer block to access the same volume and produce a natural cartilage/bone tissue interface, further enhancing the integration of cartilage and bone tissues.
Fifth, the present invention may be used to create a living ceramic/plastic composite for ligament replacement. Autologous ligament/tendon transplant is the gold standard for current ligament/tendon replacement. However, there are two major limitations of this method. First, the donor tissue site suffers from decreased mechanical strength and function while the risk of site morbidity increases. Second, the implanted tissue's mechanical property degrades post-implantation due to cellular degradation of the implant matrix secondary to decreased functional loading. The loading of the implant must be limited post operatively in order to assure implant integration into the host tissue. However, if the ligament/tendon transplant can be engineered to provide immediate mechanical functionality and integration with host tissue, the risk of implant degradation is minimized.
The ligament/tendon scaffolds presented above provide immediate mechanical integration with the host tissue. Additionally, these scaffolds can be pre-seeded with cells in vitro and pre-conditioned to produce tissue with similar mechanical properties to natural ligaments and tendons, thereby providing immediate natural functionality. These scaffolds can be created from HA/PLGA, currently FDA approved implant materials. The PLGA lactide/glycolide ratio used is chosen such that the polymer has been degraded prior to implantation. Three methods to create living ligament/tendon constructs are presented as examples.
In the first construction method for the living ceramic/plastic composite ligament replacement, the polymer block is seeded with fibroblasts. These cells can be derived from the patient or from universal cell lines. The cell are seeded within a carrier matrix that fills the space between polymer fibers. Several carriers can be used, including but not limited to collagen gels, fibrin clots and alginate. A preferred design utilizes a collagen type I carrier. The scaffold is then placed in a bioreactor and subject to tensile mechanical load conditioning and perfusion of culture media. The mechanical loading influences the extracellular matrix production of the cells and yields a stiffer tendon structure than otherwise possible in culture. The perfusion is initially used to maintain cell viability.
As the amount of extracellular matrix increases in the scaffold, the number of cells decreases. The decreased metabolic demand of the tissue coupled with then forced perfusion induced by mechanical loading of the scaffold eliminates the need for perfusion of culture media over time.
In the second construction method, the polymer scaffold is seeded with a gel carrier containing stromal cells derived from the patient bone marrow or from a universal stromal cell line. These cells have been shown to produce connective tissue in vitro. The cells are induced to produce fibrous tissue using two methods. First, the culture media may be supplemented with basic fibroblast growth factor and/or connective tissue derived growth factor (CTGF) that induces increased stromal cell collagen I secretion and fibrous matrix production. Second, the cells are infected to produce CTGF and thereby signal increased fibrous tissue production via paracrine/autocrine pathways. The cells are infected with an adenovirus created with Cre-lox recombination according to a known method carrying an expression vector for CTGF coupled to a cytomegalovirus (CMV) promoter (AdCMVCTGF). The transfection is transient, minimizing the risk of immunological rejection post implantation in the host.
The cells may be transfected under three different protocols. First, the cells can be transfected in vitro before seeding onto the scaffold. Second, the scaffold may be coated with virus and the cells seeded onto the scaffold. The virus may be covalently liked to the scaffold surfaces using the chemistries outlined above. Third, the virus and cells can be mixed into the gel carrier and then seeded onto the scaffold. In the latter methods, cell infection occurs either in vitro or in vivo after scaffold implantation. The scaffold is then placed in the bioreactor and subject to tensile mechanical load conditioning and perfusion of culture media.
The third construction method is similar to the second, except that non-stromal cells are used. These cells are induced to produce connective tissue using the AdCMVCTGF virus as outlined above. The adenovirus is replication deficient and the expression of CTGF is transient. Therefore at implantation time, the implant tissue does not produce any exogenous drug and therefore may not considered a pharmaceutical but rather a biological. The implant may also be devitalized with freeze/thaw processing rendering it a medical device. Devitalizing the construct would render a structure similar to allogenic tendon/ligament transplants albeit with enhanced mechanical integration into the host site.
The scaffold is ready for implantation after mechanical load conditioning during in vitro culture. The bioreactor can monitor tissue stiffness increase over culture time and this information may be used to discern when adequate tissue stiffness has been achieved. Prior to implantation, the ceramic base of the scaffold may be seeded with bone-derived osteoblasts and/or bone marrow stromal cells to enhance the rate of bone formation and tissue anchoring of the implant. These cells may be seeded with or without gel carriers.
Sixth, the present invention may be used to provide a living ceramic/plastic composite for joint resurfacing. The scaffold fabricated for joint resurfacing may be pre-seeded and mechanically conditioned in vitro in order to produce tissue with similar mechanical properties to natural joint cartilage, thereby providing immediate natural functionality. Cells may be seeded directly onto the polymer sponge or in secondary gel carriers. The preferred gel for cell seeding is collagen type I & II/chondroitin gel that resembles the natural cartilage tissue constituents. The gel is manufactured by solubilizing collagen types I and II in acetic acid, adding chondroitin sulfate, diluting the mix to an appropriate concentration in cell culture media supplemented with sodium bicarbonate, neutralizing with a biocompatible base, suspending cells, and allowing gellation to occur. The chondroitin sulfate concentrations are low enough to prevent complete precipitation of the collagen from solution.
Alternative gels include but are not limited to collagen I (�II)/hyluronan, fibrin/hyluronan, and fibrin/chondroitin. Collagen II sponges have been shown to induce chondrogenesis by bone marrow stromal cells when compared to collagen I sponges. Multiple cell sources may be used, including differentiated chondrocytes, stromal cells driven to a chondrogenic phenotype via chondrogenic media supplements and culture conditions (including insulin, transforming growth factor beta, basic fibroblast growth factor, dexamethosone, dextran sulfate, micromass culture, and spherical cell culture), and cells transfected to progress along the chondrogenic phenotype, including stromal cells and fibroblasts. The cells may be derived from the patient or from universal cell lines. After seeding, the scaffold is placed in the bioreactor and subject to compressive mechanical load conditioning and perfusion of culture media. The mechanical loading influences the extracellular matrix production of the cells and yields a stiffer cartilage tissue with higher glycosaminoglycan content than otherwise possible in culture. As the amount of extracellular matrix increases in the scaffold, the number of cells decreases.
The decreased metabolic demand of the tissue coupled with then forced perfusion induced by mechanical loading of the scaffold eliminates the need for perfusion of culture media over time. The bioreactor can monitor tissue stiffness increase over culture time and this information used to discern when adequate tissue stiffness has been achieved. Prior to implantation, the ceramic base of the scaffold may be seeded with bone-derived osteoblasts and/or bone marrow stromal cells to enhance the rate of bone formation and tissue anchoring of the implant. These cells may be seeded with or without gel carriers.
Seventh, the present invention may be used to provide scaffolds with designed branching channels for guided blood vessel ingrowth. In this technique, seeding of angiogenic cells is unnecessary. The scaffold global-pores are designed to facilitate nutrient diffusion and blood vessel ingrowth by providing a less tortuous path for cell migration. The local-pores are spaced so that ingrown blood vessels penetrate the entire scaffold. Angiogenic cells may be preseeded into the global pores to facilitate blood vessel formation.
A paramount requirement for survival of implanted living tissue constructs is adequate nutrient supply to the cells. Most tissue-engineered constructs rely on diffusion for transport of nutrients and waste. However, diffusion is unsuitable to large tissue construct fabrication. Molecule diffusion rates depend on diffusing media and the molecular species. Dense tissue have lower diffusion rates compared with less dense tissues. Additionally, the cells have different nutrient demands, depending on the cell type and metabolic state. The combination of diffusion rate limitations and cell metabolic demand forces living tissue constructs that rely on diffusion to be small in size. For example, liver cells are highly metabolic. In order for the cells to remain viable, diffusion rates limit thickness of constructs to less than 1 mm. Bone cell constructs have similar limitations, in particular due to the density of the tissue. Contrarily, cartilage tissue can be made much thicker, for example 1 cm, because the chondrocytes have a much lower metabolism. In order to circumvent this limitation, the tissue construct has to be designed and fabricated with pores for force perfusion or with blood vessels that will carry nutrients and waste.
Eighth, similar scaffold design and fabrication principles as used in the seventh embodiment can be used to fabricate living branching blood vessel template onto which any cells can be seeded. Such a device will facilitate in vitro production of living organ systems. Additionally, such a device will facilitate survival of the living construct once implanted in the host.
The manufacturing methods outlined in this application allow for the fabrication of scaffolds with global-pores that may be seeded with angiogenic cells. Endothelial cells can be seeded in these macro-pores in order to form blood vessels via angiogenesis. Pericytes and smooth muscle cells can be used to line the vessels at later time. Conversely, angioblasts can be seeded in these pores to form blood vessels via vasculogenesis. Angioblasts will form a vascular plexus. Upon implantation, the vessel network can be anastomosed with pre-existing vasculature. In order to enhance vessel production, angiogenic biofactors can be incorporated in the macro-pores or bulk scaffold material. VEGF-1 (vascular endothelial cell growth factor) can be used as an angiogenic factor specific to endothelial cells. Additional factors include ANG-1, PDGF-β, and TGFβ. The later have additional effects on cells other than endothelial cells and angioblasts. After blood vessel formation in the scaffold in vitro, the scaffolds are then seeded with cells that will produce the desired tissue type. Nutrient and waste clearance functions are carried out by media perfusion through the engineered blood vessel network that permeates the scaffold with the use of a bioreactor.
The presented methods provide several benefits for the production of living branching blood vessel devices. First, the blood vessel network can be made such that all arterioles branch from one synthetic artery. The same is true for the venial path. This is possible due to the control of pore branching structure afforded by SFF methods. Consequently, the scaffold contains one artery and vein that are anastomosed with a pre-existing artery and vein in the implant site and thereby supply nutrient and waste transport to the entire scaffold. Second, the macroscopic blood vessel structure can be fabricated to mimic natural blood vessel constructs using the presented global pore design and manufacturing methods. Third, living tissue constructs can be fabricated in vitro on scaffolds where the scaffold material makes up less than 10% of the scaffold volume. This is possible because the disclosed casting manufacturing technique allows for the fabrication of local-pores. Unlike blood vessel construct fabricated using two dimensional templates, these constructs can be seeded with large amounts of cells in the remaining local-pores. Therefore, the living tissue is not restricted to the global-pores, but the entire scaffold volume. Fourth, the use of angioblasts simplifies the vessel maintenance in vitro because the vascular plexus that is formed can be maintained under static culture media conditions. During in vivo vasculogenesis, the forming vascular plexus is not connected to the systemic circulatory system, unlike the blood vessels formed during angiogenesis.
Ninth, the present invention may be used to provide scaffolds mimicking existing tissue structure. Any tissue structure that can be 3D imaged, including blood vessels via computed tomography (CT) angiography, trabecular bone via micro-CT, and ligaments and tendons via magnetic resonance imaging (MRI) can be replicated from ceramic, cement, and plastic and polymeric materials. Structures created from these image sets can be used to mimic natural tissue function, create tissue replacements, and investigate the contribution of tissue structure on tissue function and adherent cell behavior. In addition, inverse molds of actual tissue structure can be made and tissue grown into the pores of the mold to generate multiple repetitions of actual tissue. For example, a scaffold has been manufactured in accordance with the teachings of the present invention from PLA that replicate trabecular bone architecture. See
Tenth, the present invention may be used for branching polyurethane vascular stents. These stents are designed to replace branching vessel structures. The present invention may be used to produce a pre-made living branching blood vessel template onto which any cells can be seeded. The present invention may also be used to provide pre-vascularized implantable tissue constructs.
Eleventh, the present invention may provide living tissue constructs for proteomic data, pathogen, and drug screening. For example, one could use living cartilage tissue created with osteoarthritic tissue for screening of therapies. Using the method of embodiment ten, inverse molds of actual trabecular bone structure can be made and tissue grown into the pores of the mold to generate multiple repetitions of actual trabecular tissue for in vitro drug testing. This approach could be used to test osteoporosis drugs on in vitro generated bone tissue.
Twelfth, the present invention may also be used to provide living tissues for drug production and drug discovery using the methods outlined in embodiment eight. That is, living tissue structures can be fabricated that produce therapeutic biofactors. In this application, cells that produce the desired biofactor are seeded onto the vascularized scaffold in vitro. For biofactor production, the scaffold is maintained in vitro and the biofactor harvested. A vascularized scaffold is not required for biofactor production in vitro. In this case, the global channels will serve for nutrient perfusion and drug harvest. For therapeutic delivery in vivo, the tissue construct is implanted and anastomosed with preexisting vasculature. An example of this is the delivery of Factor IX to treat hemophilia B.
Hemophilia is a sex-linked hereditary bleeding disorder transmitted on a gene of the X chromosome. Persons with hemophilia have prolonged bleeding when injured due to reduced functional amounts of one or more of the plasma proteins needed to form a clot. Hemophilia B, which is sometimes called Christmas disease, is a condition in which the body does not make functional Factor IX. Factor IX is an antihemorrhagic protein that participates in the blood clot cascade. Factor IX isolated from donor plasma and recombinant Factor IX are used to treat Hemophilia B, and in some cases Hemophilia A, via systemic injections. These treatments are designed to stop blood loss after it has begun. However, a more desirable treatment should inhibit the spontaneous bleeding that often occurs in hemophilia. This can be achieved by maintaining normal levels of Factor IX in the system circulation. Delivery via implantable pumps or doped materials is impractical because Factor IX has a short storage life once reconstituted. Therefore, delivery of Factor IX via an implantable tissue construct is ideal. In this therapy, bone marrow stromal cells are isolated from the patient and expanded in culture. The cells are transduced to produce Factor IX. These cells are then seeded on a prevascularized tissue template to create a mass of tissue that produces Factor IX. This tissue consists of fibrous tissue and bone surrounding the predefined vessel structure. The living tissue construct is implanted into the patient and anastomosed, whereupon systemic delivery of Factor IX commences. The implant is immunocompatible as the patient's endogenous cells are used. Additionally, the stromal cell transfection occurs ex vivo. The architecture of the tissue construct, both external shape and internal pore structure, regulates the tissue mass and vascular access, thereby regulating the amount and rate of Factor IX released. Depending on the patient size, the scaffold template is adjusted accordingly. The construct is radiographically identifiable since it contains bone tissue. Therefore, the construct can be surgically retrieved should necessity arise.
The disclosed indirect SFF scaffold manufacturing methods provide versatility because assorted scaffold designs, materials, and casting methods can be used. Such versatility is not possible with direct SFF techniques. This is essential for tailoring designs to meet specific tissue engineering applications. Compatible materials include ceramics, cements and polymers such as silica, hydroxyapatite, plastics and polymers (e.g. α-hydroxyacids, polycaprolactone, polyanhydrides, polymethylmethacrylate, polypropylenefumarate), rubbers (e.g. polyurethane, polydimethylsiloxane), biological materials (e.g. collagen, alginate, fibrin and dextran gels), and composites thereof. Compatible polymer processing techniques include melt processing, emulsion freeze drying, emulsion solvent diffusion, phase separation and freeze drying, solvent casting and particulate leaching, gas foaming, and combined gas foaming and particulate leaching. Each processing method has advantages that suit different tissue engineering applications.
The outlined methods of the invention contain several other inherently useful properties that contribute to the utility of the presented embodiments above.
For example, melt processing yields dense and tough polymer structures ideal for load bearing tissues but precludes the incorporation of biofactors within the polymer phase as is possible with gas foaming. In these methods, leaching of porogens was facilitated by the presence of the interconnected global pores, allowing the removal of salt from relatively thick scaffolds. Porogen leaching from thick scaffolds is often difficult because solvent permeability is low when the porogen is present. Additionally, 3D molds containing global pore templates provide for casting of larger local pore scaffolds than possible when casting in hollow containers.
Most local-pore fabrication technology is not compatible with direct SFF fabrication methods. For instance, the use of emulsion, phase separation, solvent diffusion, and gas foaming local-pore fabrication techniques is not possible with direct 3DP, SLS, STL, DMD, LOM, and FDM. In contrast, porogen leaching can be used with these methods. For example, 75% and 90% void fraction PLA scaffolds have been made with controlled local pore size ranging from <38–150 μm using TheriForm™ (a 3DP system, Therics, Princeton, N.J.) and salt leaching. However, high local porosity, which is a function of void fraction and pore interconnectivity, is difficult to achieve in direct SFF. In particular, the distance between porogens increases during extrusion in FDM. In STL and SLS, the light beam used may diffract through large porogens, adversely affecting curing and melting. Finally, the densely packed porogen bed necessary in 3DP for high porosity may interfere with powdered build material salvation and binding. As noted, high pore interconnectivity is required to promote cell and tissue growth. In 3DP manufacturing, local pore interconnectivity may be improved by using volatile porogens that sublimate upon contact with solvents or direct polymer deposition upon an interconnected porogen bed. Such methods are also compatible with indirect 3D scaffold fabrication via casting. Nevertheless, it is the ability to employ varied polymer and porogen processing techniques using indirect SFF fabrication that facilitates construction of high porosity scaffolds.
Solvent casting and injection molding of blend composites are also feasible with indirect SFF casting methods. Blend composites offer greater resistance to fatigue, increased ceramic resorption rates, and facilitated biofactor adhesion compared to conventional 100% ceramic scaffolds. Direct SFF methods may also be used to manufacture blends. For example, local pore 80:20 PLGA:TCP scaffolds have been made using TheriForm™ and salt leaching. Additionally, using SLS and 3DP, graded composites with gradients in ceramic concentration may be manufactured by altering the polymer to ceramic ratio between bed material layers. However, composite manufacturing with 3DP, STL, and FDM is limited because these methods require incorporation of binders within the entire scaffold ceramic region, use a low ceramic volume ratio, or have inadequate bonding of the polymer and ceramic regions. These direct SFF methods preclude the use of sintered ceramic and the fabrication of discrete polymer/ceramic scaffolds.
The use of sintered dense ceramic is desirable for several reasons. The polymers in blends may interfere with the osteoconductivity of HA. In fact, calcium phosphates have been nucleated onto polymers in the hope of increasing osteoconductivity. Unlike polymers, HA exhibits direct collagen and bone apposition with a strong mechanical interface between ceramic and bone. This concept has been used to improve bone/polymer bonding strength via incorporation of HA in PLA. Finally, although sintered HA is much stiffer than cortical bone, porous HA scaffolds with similar porosity and compressive modulus of trabecular bone have been made. Blends have commonly been manufactured as bone fillers or as non-porous bone fixation devices with stiffness similar to cortical bone and bending strengths of similar order of magnitude. However, the diminished load bearing capability of porous blends may make them more suitable for non-load bearing applications.
50–65% porous ceramic scaffolds made from replamine form coralline have successfully been used for hard tissue repair (ProOsteon and Interpore, Interpore Cross International, Inc. Irvine, Calif., for review). Coralline compressive modulus is on the order of trabecular bone modulus, making these scaffolds ideal for hard tissue repair. However, ceramic scaffolds are exceedingly stiff for soft tissue repair, precluding their use for interface tissue engineering. Discrete composites containing dense ceramic for bone growth and polymer for soft tissue growth have been created by adhering PLA fleece to a Coralline™ or calcite substrates. However, their function under physiologic skeletal loads might be inadequate as no mechanical interdigitation or strong chemical adhesion of the composite phases was provided. As shown in
The utility of biomimetic approaches for scaffold design and fabrication is under investigation. IBD has been used to design temporomandibular condylar prosthesis that replicates the natural condylar shape. Nano-fibrous α-hydroxyacid matrices have been created to mimic natural matrix collagen size and structure. Bio-inspired mineral deposition of carbonated apatite onto α-hydroxyacids has been used to render them osteoconductive. Several groups have developed tubular tissue engineered constructs for blood vessel replacement. These studies have replicated macroscopic (vessels and condyles) and nanoscopic (fibers and crystals) features of naturally occurring tissues. The disclosed methods provide for bio-mimicry on the microscopic and macroscopic scale.
Referring now to
In view of the foregoing, it can be appreciated that the synthesis of indirect SFF technologies, current scaffold fabrication methods, and image based design provides the means to precisely control scaffold material composition and architecture, including local and global porosity. This control allows for the manufacture of scaffolds for tissue engineering that are optimized for mechanical and biological function through precise manipulation of scaffold shape, porosity, and internal pore architecture (pore size, geometry, orientation, branching, and interconnectivity). This control facilitates engineering of biomimetic scaffolds and scaffolds for complex biomechanical applications, including perfused tissues (concurrent local/global porous scaffolds) and interfaces between bone and soft tissues (ceramic/polymer discrete composite scaffolds). Future scaffolds constructed with these integrated manufacturing techniques should aid tissue reconstruction, regeneration of damaged tissues and organs, artificial organ production and living tissue construct fabrication.
In conclusion, the present invention provides novel casting methods to manufacture three dimensional (3D) structures from diverse materials with designed and controlled local-porous and global-porous internal architectures using solid free from fabrication (SFF) techniques. These methods allow fabrication of mechanically robust porous composite structures containing a number of plastics, polymers, ceramics, cements, and metals, . These structures can be used as tissue regeneration scaffolds, bioreactor culture substrates, perfusion assisted mass diffusion exchangers, biosensors, "lab-on-chips", and living tissue constructs for proteomic, drug interaction, and pathogen analyses. The ability to manufacture scaffolds from diverse materials is essential for tailoring designs for specific applications. The methods presented allow fabrication of scaffolds from more materials than currently possible, including ceramics, cements and polymers like silica, hydroxyapatite, plastics, rubber, biological materials, and composites thereof. Composite structures consist of two formulations: blend structures made from materials embedded within a second material(s) that is used as a binder or structures with discrete regions of differing material composition that are mechanically interdigitated and/or chemically bonded. These manufacturing methods also provide simultaneous control over structure local and global pore architecture, including scaffold pore and strut geometry, orientation, branching, and interconnectivity. Consequently, these manufacturing methods enable the advancement of tissue engineering, biochip, and bioreactor technologies. In particular, the flexibility of manufacturing scaffolds from varied materials with designed macro/micro-porosity provides enhanced control over:
And allow the fabrication of:
The description of the invention is merely exemplary in nature and, thus, variations that do not depart from the gist of the invention are intended to be within the scope of the invention. Such variations are not to be regarded as a departure from the spirit and scope of the invention.
|Cited Patent
|Filing date
|Publication date
|Oct 12, 1990
|Jul 13, 1993
|Hitachi Metals, Ltd.
|Method of designing cavity shape of mold
|May 17, 1995
|Oct 21, 1997
|Mitsubishi Denki Kabushiki Kaisha
|CAD device for metal mold
|Jun 1, 1995
|Feb 3, 1998
|Johnson & Johnson Professional, Inc.
|Bone prostheses with direct cast macrotextured surface regions and method for manufacturing the same
|Sep 12, 1994
|Apr 21, 1998
|The University Of Queensland
|Stereolithographic anatomical modelling process
|Sep 30, 1996
|Jun 9, 1998
|Johnson & Johnson Professional, Inc.
|Custom bioimplantable article
|Apr 30, 1997
|Nov 9, 1999
|Lawton; John A.
|Solid imaging process using component homogenization
|Feb 15, 1994
|Jan 11, 2000
|The University Of Texas System
|Continuous release polymeric implant carrier
|Mar 26, 1998
|Nov 7, 2000
|Carnegie Mellon
|Assembled scaffolds for three dimensional cell culturing and tissue generation
|Jul 6, 1998
|Dec 26, 2000
|Microvention, Inc.
|Vascular embolization with an expansible implant
|May 27, 1998
|Feb 20, 2001
|Fujitsu Limited
|Injection mold design system and injection mold design method
|May 14, 1999
|Apr 3, 2001
|Align Technology, Inc.
|Creating a positive mold of a patient's dentition for use in forming an orthodontic appliance
|Jun 16, 1998
|May 15, 2001
|Sanyo Electric Co., Ltd.
|Method and apparatus for designing molds using 3-dimensional CAD system
|Dec 3, 1999
|Oct 30, 2001
|Align Technology Inc.
|Attachment devices and method for a dental applicance
|International Search Report mailed Feb. 26, 2003.
|International Search Report mailed Oct. 2, 2002.
|Citing Patent
|Filing date
|Publication date
|Mar 4, 2005
|Aug 3, 2010
|The Trustees Of Columbia University In The City Of New York
|Multi-phased, biodegradable and osteointegrative composite scaffold for biological fixation of musculoskeletal soft tissue to bone
|Jul 21, 2006
|Aug 10, 2010
|Tomotherapy Incorporated
|Method and system for evaluating quality assurance criteria in delivery of a treatment plan
|May 12, 2005
|Oct 19, 2010
|Massachusetts Institute Of Technology
|Manufacturing process, such as three-dimensional printing, including solvent vapor filming and the like
|Aug 10, 2005
|Nov 23, 2010
|California Institute Of Technology
|High aspect ratio template and method for producing same
|Jul 21, 2006
|Nov 23, 2010
|Tomotherapy Incorporated
|System and method of evaluating dose delivered by a radiation therapy system
|Feb 24, 2006
|Jun 7, 2011
|Cadman Patrick F
|Method and apparatus for modulating a radiation beam
|Sep 2, 2008
|Dec 13, 2011
|California Institute Of Technology
|High aspect ratio template and method for producing same for central and peripheral nerve repair
|Sep 30, 2004
|Dec 27, 2011
|Medical Modeling Inc.
|Method for design and production of a custom-fit prosthesis
|Jul 21, 2006
|Jul 24, 2012
|Tomotherapy Incorporated
|System and method of detecting a breathing phase of a patient receiving radiation therapy
|May 10, 2006
|Jul 31, 2012
|Tomotherapy Incorporated
|System and method of treating a patient with radiation therapy
|Oct 29, 2007
|Sep 25, 2012
|The Regents Of The University Of Michigan
|Engineered scaffolds for intervertebral disc repair and regeneration and for articulating joint repair and regeneration
|Aug 10, 2010
|May 14, 2013
|Tomotherapy Incorporated
|Method and system for evaluating quality assurance criteria in delivery of a treatment plan
|Aug 13, 2009
|Jul 2, 2013
|Smed-Ta/Td, Llc
|Orthopaedic screws
|Jun 23, 2010
|Aug 20, 2013
|Postech Academy-Industry Foundation
|Manufacturing method for 3D structure of biomaterials using stereolithography technology and products by the same
|May 16, 2013
|Apr 22, 2014
|Smed-Ta/Td, Llc
|Orthopaedic Screws
|Aug 12, 2009
|Jun 17, 2014
|The Trustees Of Columbia University In The City Of New York
|Fully synthetic implantable multi-phased scaffold
|Jul 21, 2006
|Jul 1, 2014
|Tomotherapy Incorpoated
|System and method of delivering radiation therapy to a moving region of interest
|Dec 22, 2009
|Aug 12, 2014
|The Trustees Of Columbia University In The City Of New York
|Multi-phased, biodegradable and osteointegrative composite scaffold for biological fixation of musculoskeletal soft tissue of bone
|Jan 9, 2013
|Oct 21, 2014
|Vivex Biomedical, Inc.
|Bone implants and method of manufacture
|Aug 24, 2010
|Oct 21, 2014
|The Trustees Of Columbia University In The City Of New York
|Biomimmetic nanofiber scaffold for soft tissue and soft tissue-to-bone repair, augmentation and replacement
|Feb 18, 2011
|Oct 28, 2014
|Simpleware Limited
|Image processing method and method of three-dimensional printing incorporating the same
|Jul 1, 2014
|Aug 11, 2015
|MetaMason, Inc.
|Methods for additive manufacturing processes incorporating active deposition
|Aug 2, 2012
|Sep 22, 2015
|Apollo Endosurgery, Inc.
|Mucosal tissue adhesion via textured surface
|Dec 14, 2012
|Nov 10, 2015
|The Regents Of The University Of Michigan
|Porous bidirectional bellowed tracheal reconstruction device
|Aug 28, 2009
|Jun 7, 2016
|Smed-Ta/Td, Llc
|Orthopaedic implant
|Sep 8, 2008
|Jun 21, 2016
|Renishaw Plc
|Relating to joints and/or implants
|Mar 11, 2014
|Aug 9, 2016
|Smed-Ta/Td, Llc
|Removable augment for medical implant
|Aug 4, 2014
|Aug 30, 2016
|The Trustees Of Columbia University In The City Of New York
|Multi-phased, biodegradable and oesteointegrative composite scaffold for biological fixation of musculoskeletal soft tissue to bone
|Oct 29, 2007
|Sep 13, 2016
|The Regents Of The University Of Michigan
|Degradable cage coated with mineral layers for spinal interbody fusion
|Feb 19, 2014
|Sep 13, 2016
|Accuray Incorporated
|Electromagnetically actuated multi-leaf collimator
|Jul 11, 2014
|Dec 6, 2016
|Tundra Composites, LLC
|Surface modified particulate and sintered or injection molded products
|Oct 2, 2014
|Feb 7, 2017
|Smed-Ta/Td, Llc
|Drug delivery implants
|Aug 28, 2009
|Apr 11, 2017
|Smed-Ta/Td, Llc
|Drug delivery implants
|Jan 17, 2014
|Apr 25, 2017
|Ehsan Toyserkani
|Systems and methods for additive manufacturing of heterogeneous porous structures and structures made therefrom
|Mar 11, 2014
|Jun 20, 2017
|Smed-Ta/Td, Llc
|Method of manufacturing a tubular medical implant
|Mar 3, 2015
|Jul 11, 2017
|Smed-Ta/Td, Llc
|Orthopaedic implant with porous structural member
|Aug 8, 2016
|Jul 18, 2017
|Smed-Ta/Td, Llc
|Removable augment for medical implant
|Mar 13, 2014
|Aug 8, 2017
|Smed-Ta/Td, Llc
|Porous tissue ingrowth structure
|Sep 4, 2002
|Dec 9, 2004
|Rebecca Petersen
|Filtration membrane
|Sep 30, 2004
|Jun 23, 2005
|Medical Modeling Llc
|Method for design and production of a custom-fit prosthesis
|Mar 4, 2005
|Feb 16, 2006
|Lu Helen H
|Polymer-ceramic-hydrogel composite scaffold for osteochondral repair
|Mar 4, 2005
|Mar 30, 2006
|Lu Helen H
|Multi-phased, biodegradable and osteointegrative composite scaffold for biological fixation of musculoskeletal soft tissue to bone
|Oct 1, 2003
|Jun 8, 2006
|E-Tec Co., Ltd.
|Cold-curing binder and process ror producing molding with the same
|Oct 29, 2007
|Aug 14, 2008
|Chia-Ying Lin
|Engineered Scaffolds for Intervertebral Disc Repair and Regeneration and for Articulating Joint Repair and Regeneration
|Jun 20, 2008
|Dec 25, 2008
|Philadelphia Health & Education Corporation D/B/A Drexel University College Of Medicine
|Method of local therapy using magnetizable thermoplastic implant
|Mar 16, 2009
|Jul 9, 2009
|Microfabrica Inc.
|Medical Devices and EFAB Methods and Apparatus for Producing Them
|Sep 2, 2008
|Aug 13, 2009
|California Institute Of Technology
|High aspect ratio template and method for producing same for central and peripheral nerve repair
|Jun 6, 2009
|Feb 25, 2010
|Lu Helen H
|Graft collar and scaffold apparatuses for musculoskeletal tissue engineering and related methods
|Aug 10, 2005
|Mar 4, 2010
|California Institute Of Technology
|High aspect ratio template and method for producing same
|Aug 12, 2009
|Nov 18, 2010
|Lu Helen H
|Fully synthetic implantable multi-phased scaffold
|Sep 8, 2008
|May 26, 2011
|University Of Bath
|Improvements in or Relating to Joints and/or Implants
|Feb 18, 2011
|Aug 25, 2011
|Simpleware Limited
|Image processing method and method of three-dimensional printing incorporating the same
|Jan 9, 2013
|May 23, 2013
|Amendia Inc.
|Bone implants and method of manufacture
|Aug 9, 2010
|Jun 13, 2012
|Rolls-Royce Corporation
|Method for forming a cast article
|Aug 9, 2010
|Apr 9, 2014
|Rolls Royce Corp
|Method for forming a cast article
|May 10, 2011
|Nov 17, 2011
|Allergan, Inc.
|Porogen compositions, methods of making and uses
|Aug 1, 2013
|Feb 6, 2014
|Allergan, Inc.
|Mucosal tissue adhesion via textured surface
|Sep 27, 2013
|Apr 3, 2014
|Allergan, Inc.
|Porogen compositions, methods of making and uses
|Mar 14, 2014
|Sep 18, 2014
|The George Washington University, A Congressionally Chartered Not-For-Profit Corporation
|3d biomimetic, bi-phasic key featured scaffold for osteochondral repair
|U.S. Classification
|264/49, 264/317, 264/51, 264/219, 264/299, 264/328.1, 264/232, 264/212, 264/53, 623/11.11, 164/47, 264/50, 264/255
|International Classification
|Cooperative Classification
|European Classification
|Feb 8, 2010
|Fee payment
Year of fee payment: 4
|Feb 10, 2014
|Fee payment
Year of fee payment: 8


text:
Removal & Installation
At this time the manufacturer does not provide removal and installation procedures for this component. The following procedure is a guideline and may differ from the vehicle you are servicing.
Before servicing the vehicle, refer to the Precautions Section.
NOTEIf working near and/or around the SRS system and components, be sure to disable the SRS system. After disabling the system wait three minutes or more before servicing the vehicle. Disconnect the air bag module connector.
- Disconnect the negative battery cable.
- Raise and safely support the vehicle.
- Remove the tire and wheel assemblies.
- Carefully siphon brake fluid from the master cylinder to an acceptable level.
- Compress the piston(s) in the brake caliper. Remove the caliper slide pin bolt.
- Swivel the caliper up and away from the torque plate. Tie the caliper to a suspension member so it's out of the way. Do not disconnect the brake line. Do not allow the caliper to hang by the brake hose.
- Remove the grease cap from the hub. Remove the cotter pin and the castellated nut.
- Remove the wheel hub with the brake disc attached.
Remove the rotor.
To install:
Be sure to use new fasteners, as required.
- Installation is the reverse of the removal procedure.
- Be sure to refill the master cylinder with the proper grade and type fluid.
- Bleed the brakes, as required.
- Using the CONSULT-III diagnostic tool, or equivalent, perform the SRS self diagnostic test and any reconnect/relearn procedures, as required.

subset: commoncrawl


text:
Author | Rummana Ahmed | Wednesday, Dec 15,2004 7:56 AM
A peak into the glamorous lives of our celebrities, the biggest gossips and the smallest rumours, their well-kept secrets, we want to know it all. 'Koffee with Karan' does just that, a chat over a cup of coffee on Friday nights at 10 PM. Star World claims that the treatment of the show is very different, as the guests are not being profiled. Instead, it's like a chat among friends, between host Karan Johar and his friends in Bollywood. At times Karan treads almost on forbidden ground, asking his celebrity guests from tinsel town questions that most interviewers wouldn't dream of asking.
Ajay Vidyasagar, Senior Vice-President, Marketing, Star Network, explains, "The unique programming of the show is its selling point. In no other show do super celebrities chat in such a casual and free manner, cutting jokes about people and about each other." He added that "people love trivia and they are getting it on this show".
The show was launched on November 19 and is just a few a few episodes old. The first show featuring Shahrukh Khan and Kajol recorded a TVR of 0.37 in the four metros for the target audience Sec A,B,C in the age group of 24+. The second episode featuring Kareena Kapoor and Rani Mukherjee recorded a TVR of 0.58. The third show with Saif and Preiti Zinta showed a dip with a TVR of 0.19. Ratings for the fourth episode with Aishwarya Rai and Sanjay Leela Bhansali are yet to come out.
Radha Barua, Associate Business Director, Carat India, said, "There might be several reasons for people liking the show; the kind of chemistry Karan Johar manages to build works for the show. It also has two guests, and a three-way conversation always works better. It is also interesting to see how all of them, the host and the guests, interact with each other."
The show also works because there is no stiffness between the interviewer and interviewees. For now Karan has managed to brew coffee and conversation well. Time will tell how 'Koffee with Karan' manages to score over the other run-of-the-mill chat shows on a sustained basis.
Programming HeadRadio Mango


text:
Alessi Diva + Anna Gong
Sneak peek in to Alessi's Fall/Winter 2011 product line ~ two grabbed my attention in particular. Eero Aarnio's Diva watering can, is both elegantly simple and sassy all at once ~ in black, green, and a creamy white. And the Anna Gong is a nod at Mendini's famous "Anna G" corkscrew and legendary folding cake stand "art.898". Take a peek at closeups and more details on the next page.
TO PAGE 2 of "Alessi Diva + Anna Gong"! ----->
Tags: food - gifts - home - kitchen - plants - shopping - water


text:
100 Creatives: Chance McClain
What he does: Chance McClain is probably best known for having co-written the "Yao Ming Song," a rousing rally cry for the Houston Rockets fans in the earlier part of this millennium. The song was made famous outside Our Fair City via Sports Illustrated and was even featured on the Chinese program "Breakfast With Yao," in which the previous night's Rockets game was broadcast.
Sports are definitely in his blood, and McClain was a producer for years on AM 1560 The Game. But recently, he's started turning heads in an unexpected way: As the writer of Kissless, a new musical that will premiere at the end of the month at the New York Musical Theatre Festival. It's a big step for McClain, who was into musical theater when he was younger but traded it in for sports and music.
It was only recently, when he took his young son to a community theater audition, that he got back into musicals.
"As soon as I was back in the theater, I fell in love with it all over again," he says.
Austin Jacobs, David Allen III, Kristin Rice and Raegan Roberts rehearse Kissless, The Musical.
Upcoming Events
Mike Epps
TicketsSat., Oct. 21, 7:00pm
TicketsSat., Oct. 21, 7:00pm
Love On The Floor
TicketsFri., Oct. 27, 8:00pm
Jeanne Robertson
TicketsSat., Nov. 4, 7:00pm
Nick Cannon Presents: Wild 'N Out Live!
TicketsSat., Nov. 4, 8:00pm
Why he likes it: McClain says he likes theater because of the live experience. "There's real people working right in front of you," he says. "It's just so much more legit when it's real people acting things out live."
McClain agrees that people sometimes have the concept of a "sports guy" and a "musicals guy" and never the twain shall meet, but he doesn't think the two are mutually exclusive.
"It's all entertainment to me," he says. "You know, part of writing the book is crafting jokes, and that's not so different from crafting jokes for the morning show. They're only different in context."
If not here, then where? McClain was born here, and, except for a brief stint in the army, has lived in Texas all his life, so it's hard for him to imagine living anywhere else, he says. Still, the best resources for what he wants to do are in New York. "The way this community has embraced me and the play has been remarkable, but sometimes I can't find anyone to answer some of the questions I have here," he says. "Ideally, I guess I'd split my time between here and New York. But Houston's always home."
What's next? Houston previews for Kissless, The Musical are set for September 11, September 17 and September 18 at The Berry Center, 8877 Barker Cypress Road in Cypress. The NYMF starts September 26.
More Creatives (In order of most recently published; click here for the full page).
Jodi Bobrovsky, properties master Brittany Bentine, photographer Paul Hope, actor Marilu Harman, dancer Van G. Garrett, poet Dominic Walsh, dancer and choreographer Ibis Fernandez, animator Alex "PR!MO" Luster, filmmaker Chris Nguyen, designer Sophia Vassilakidis, animator Sandra Lord, tour guide Scarlett St. Vitus, model Wayne Stevens, actor Bill Davenport, sculpture and found art Julie Zarate, painter Margo Toombs, actor and writer Shelby Hohl, graphic designer Timothy Dorsey, writer and illustrator Lucas Gorham, musician Tracy Manford Carlson, photographer Lauren Rottet, architect and designer John Robertson, visual artist John Adelman, visual artist Chandos Dodson, interior designer Cliff Franks, painter Kim Hartz, photographer Katy Heinlein, visual artist Robert Shimko, dramaturg Galina Kurlat, photographer Wayne Slaten, filmmaker Jane Weiner, dancer and choreographer El Franco Lee II, visual artist Chris McKay, photographer Jason Ransom, visual artist Mr. SINched, fashion desiger "Uncle" Charlie Hardwick, poster designer Avital Stolar, playwright and educator Katherine Houston, visual artist Christopher Olivier, visual artist Dennis Lee Harper, sculptor David A. Brown, photographer Rachel Harmeyer, visual artist Kia Neill, installation artist Stacy Davidson, filmmaker Jennifer Wood, choreographer GONZO247 Kevin DeVil, filmmaker Kerry Beyer, photographer and filmmaker Robert Ellis, musician Davie Graves, musician and visual artist Robert Hodge, multimedia Mary Magsamen, photo and video artist John Harvey, theater Bret Harmeyer, visual artist Joel Orr, puppet master Rodney Waters, photographer and pianist Jeremy Choate, lighting designer Chuck Ivy, visual artist Tra'Slaughter, visual artist Jen Chen - visual art, designer Howard Sherman - Painter Nancy Hendrick - Founder of Dance Salad Misha Penton - Opera Singer and Theater Artist Ben Tecumseh DeSoto - Photojournalist Tracy Robertson aka Batty - Goth Fashion Designer Tierney Malone - Creative Type Dolan Smith - Painter Jenny Schlief - Mixed-Media Artist David Eagleman - Writer Anna Sprage - Painter Philip Lehl - Actor Andy Noble - Choreographer David McGee - Painter
Follow Art Attack on Facebook and on twitter ArtAttackHP
Get the ICYMI: Today's Top Stories Newsletter Our daily newsletter delivers quick clicks to keep you in the know
Catch up on the day's news and stay informed with our daily digest of the most popular news, music, food and arts stories in Houston, delivered to your inbox Monday through Friday.
Recommended for You
Powered by SailThru
Upcoming Events
104.1 KRBE's Roula & Ryan Road Show
TicketsFri., Sep. 22, 7:30pm
The Comedy Get Down
TicketsSat., Sep. 23, 8:00pm
John Mulaney: Kid Gorgeous
TicketsSat., Oct. 14, 7:00pm
TicketsWed., Sep. 6, 7:30pmpowered by goldstar
Sponsor Content


text:
Company Updates
Pedigree Technologies Introduces the Cab-Mate Connect
FARGO, N.D. — Pedigree Technologies, a leading provider of fleet management and supply and service-chain solutions, has announced the release of …
Pedigree Technologies among 20 Most Promising Logistics Tech Solution Providers
Pedigree Technologies made the annual list put out by CIO Review Magazine FARGO, N.D. — Pedigree Technologies has been included on …
We're Proudly Sponsoring Farm Rescue
Pedigree Technologies is now a proud sponsor of Farm Rescue, a one-of-a-kind nonprofit organization that provides planting, harvesting and haying assistance …
Alex Warner's Interview on CONEXPO-CON/AGG Radio
Pedigree Technologies Founder Alex Warner was recently interviewed by Peggy Smedley on CONEXPO-CON/AGG Radio. Peggy Smedley is an internationally known speaker …
See all News posts.
On the Web
Media Contact
Want to write about us?
Email firstname.lastname@example.com