text:
|Publication number
|Publication type
|Application number
|Publication date
|Nov 28, 1995
|Filing date
|Nov 3, 1994
|Priority date
|Mar 26, 1993
|Fee status
|Also published as
|Publication number
|Forrest W. Briese, Charles C. Hyder, John M. Schlarb, Christophe J. A. Fouquet, Coy S. Lowe
|Original Assignee
|Schlumberger Industries, Inc.
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (4), Referenced by (26), Classifications (13), Legal Events (15)
|External Links: USPTO, USPTO Assignment, Espacenet
This application is a division of U.S. patent application Ser. No. 08/037,938, filed on Mar. 26, 1993, pending, entitled "SYSTEM CHECKING AND TROUBLESHOOTING PACKAGE FOR AN ELECTRICTRONIC METERING DEVICE".
The present invention relates to an integral method and apparatus for conducting system installation diagnostics in a solid state electronic meter.
Induction-type watt hour meters typically employ a pulse initiator which generates pulses in proportion to the rate of rotation of a meter disk. These generated pulses are transmitted to electronic registers for deriving current, voltage, power and/or time of use energy consumption.
Various types of solid state polyphase electronic meters are also in common use today. These meters, which monitor electrical energy consumption and record or report such consumption in kilowatt hours, power factor, KVA, and/or reactive volt amperes, typically employ solid state components, and may utilize analog-to-digital converters to provide digital data rather than pulse data from which various demand/consumption indicators can be extracted.
It is also well known to provide solid state electronic meters which may be configurable for installation in any one of a variety of single or multi-phase electricity distribution systems. One example of a solid state electronic watt hour meter is disclosed in U.S. Pat. No. 5,059,896, issued to Germer et al.
An example of a solid state electricity demand recorder which may be used in conjunction with a conventional watt hour meter is disclosed in U.S. Pat. No. 4,697,182, issued to Swanson.
Various ancillary equipment and diagnostic techniques are utilized by service personnel during installation of these meters in attempting to confirm that the installation has been wired correctly. However, many installation checks, such as polarity and cross-phase checks, are derived on site by field personnel and are therefore dependent upon the knowledge and competence of those personnel.
While various diagnostic equipment is available for use by field personnel during installation and periodic maintenance, a need exists for an integral apparatus which automatically and periodically performs a standard series of system and installation diagnostics without interrupting the operation of the meter. In addition, there is a need for periodic self-checks of the meter to determine and record the occurrence of selected pre-defined fatal and non-fatal errors in the meter's operation.
An object of the present invention is therefore to provide an integral system checking and trouble-shooting package for a solid state electronic meter.
It is another object of the present invention to provide a method and apparatus which is integral with a solid state meter and which automatically performs a series of predefined system installation and diagnostic tests on the meter.
It is still another object of the present invention to provide a system checking and troubleshooting package which supports and is integral to an electronic meter, and which includes means for displaying the results of selected self-checks and system diagnostic tests when interrogated by service personnel.
It is yet another object of the present invention to provide an automated system checking apparatus which periodically checks for the existence of certain pre-defined conditions and which, depending upon the nature of the error, takes predefined action in response to the detection of any such errors.
It is another object of the present invention to provide a method and apparatus for determining the phase angles of each voltage and current phasor with respect to a preselected base phasor, for the purpose of verifying that all meter elements are sensing and receiving the correct voltage and current for each phase of a multi-phase electric service.
In accordance with the present invention, an integral electronic meter self-checking and system diagnostics package is provided, including a microprocessor, storage memory, logic for automatically and periodically performing a preselected set of meter self-checks and recording any errors therefrom, logic for automatically performing a preselected series of system diagnostics tests, and recording any results which exceed predefined programmable thresholds, and display means for displaying error and/or diagnostic messages identifying, respectively, one or more self-check errors or selected diagnostic data and/or errors discovered in the meter self-checks during a predefined period.
The device of the present invention is preferably integrated into a solid state meter which utilizes an analog-to-digital converter and associated digital sampling techniques to obtain digital data corresponding to current and voltage for one or more phases of a single phase or multi-phase system to which the meter is connected.
The present invention automatically performs the preselected meter self-checks, preferably once per day, and/or when power is restored to the meter following an outage, and/or when a full meter reconfiguration is performed, to verify the continued functionability of selected meter components. In the preferred embodiment, for example, the device of the present invention checks its own memory, microprocessor, and selected registers in the meter to determined whether the billing data has been corrupted since the last check. Since the corruption of billing data is considered a fatal error of the meter, the device of the present invention would generate and display an error code indicating the nature of the error, lock the display on the error code, and cease all meter functions (except communications functions) until the meter is re-configured.
In addition, the device also periodically checks for other, non-fatal, errors such as for register overflows, clock, time of use, reverse power flow, and low battery errors. The frequency of error checking may vary depending upon the component and/or condition checked, as well as the potential effect of the error on the continued operation of the meter. Once discovered, non-fatal errors may or may not lock out the display depending upon the nature of the error and how the particular meter is configured.
The present invention also periodically performs a series of preselected system diagnostics tests. These tests are at installation of the meter and preferably about once every five seconds during the normal operation of the meter. In the preferred embodiment, the device conducts a polarity, cross-phase and energy flow diagnostic, a phase voltage deviation diagnostic, a current transformer diagnostic, and a per-phase power factor diagnostic utilizing factory-defined parameters as well as user-defined parameters which may be specified by personnel in the field at installation.
In conducting the polarity, cross-phase and energy flow diagnostic, the device of the present invention utilizes accumulated current and voltage information to determine the phase angle of each voltage and current phasor (for example VB, VC, IA, IB, and IC) with respect to a reference phasor (for example VA) in a multi-phase system. The proper position of each phasor for this installation is pre-defined and used as an exemplar for comparison to the calculated phase angle to determine whether each angle falls within a pre-defined envelope. If any one of the calculated phase angles falls outside its corresponding pre-defined envelope, a diagnostic error message may be displayed. This diagnostic is particularly useful at installations since this error may indicate cross-phasing of a voltage or current circuit, incorrect polarity of a voltage or current circuit, reverse energy flow of one or more phases (co-generation), or an internal meter measurement malfunction.
The device of the present invention also preferably includes a "Toolbox" display which, when manually activated by field personnel, causes the display to scroll through a list of preselected values, such as voltage and current for each phase, the angles associated with each voltage and current phasor, and the numbers of occurrences of each diagnostic failure, for review by field personnel.
The system diagnostics and Toolbox display are performed by the device of the present invention without interruption in the operation of the meter except when such operation is purposely suspended as a result of a fatal error.
The above objects and other objects, features, and advantages of the present invention are readily apparent from the following detailed description of the best mode for carrying out the invention when taken in connection with the accompanying drawings.
FIG. 1 is a block diagram of the system;
FIG. 2 is a perspective view of a meter into which the system of the present invention may be integrated;
FIG. 3 is a block diagram of the meter of FIG. 2;
FIG. 4 is a flowchart of the electrical system diagnostics checks of the present invention;
FIG. 5 is a flowchart of a first portion of the polarity, cross phase and energy flow diagnostic implemented by the present invention;
FIG. 6 is a flowchart of the second portion of the polarity, cross phase and energy flow diagnostic implemented by the present invention;
FIG. 7 is a flowchart of a first portion of the phase voltage deviation diagnostic routine implemented by the present invention;
FIG. 8 is a flowchart of a second portion of the phase voltage deviation diagnostic implemented by the present invention;
FIG. 9 is a flowchart of a first portion of the current transformer diagnostic implemented by the present invention;
FIG. 10 is a flowchart of a second portion of the current transformer diagnostic implemented by the present invention;
FIG. 11 is a flowchart of a first portion of the per-phase power factor diagnostic implemented by the present invention;
FIG. 12 is a flowchart of a second portion of the per-phase power factor diagnostic implemented by the present invention;
FIG. 13 is a flowchart of a third portion of the per-phase power factor diagnostic implemented by the present invention;
FIG. 14 is a list of the items displayed in the Toolbox display;
FIG. 15 is a phasor diagram for a typical three-phase meter installation;
FIG. 16 is a graph illustrating the relationship of the wave forms representing two phase quantities tracked by the system;
FIG. 17 is a block schematic of the frontend module 42 of FIG. 3; and
FIG. 18 is a block schematic of the register module 48 of FIG. 3.
Referring to FIG. 1, the system of the present invention, generally designated as 20, includes a central processing unit 22, storage memory 24 adequate for storing digital data corresponding to the periodic samples of the voltage and current data from the voltage A/D converter 26 and current A/D converter 28, respectively, logic 30 for performing the meter self-check and system and installation diagnostics supported by the system, and display means 32 for displaying error and diagnostic information.
Referring to FIG. 2, the system 20 is preferably incorporated into a solid state polyphase Kilowatt/Kilowatt-hours ("KW/Kwh") single function meter 34 (as illustrated in FIGS. 3, 17 and 18 and hereafter described in greater detail) including a generally circular base 36, conventional molded plastic housing (not shown) to which a faceplate 38 is affixed, and a meter cover 40. The meter 34 also includes conventional current sensing elements adapted for connection to existing electrical systems.
Referring now to FIG. 3 in the preferred embodiment, the diagnostics logic 30 for the system 20 of the present invention is incorporated into the front-end module 42 of the meter including a microprocessor 44, an 8 bit A/D converter which serves as the voltage A/D converter 26, random access memory 45, which serves in part as part of the system storage memory 24, and read-only memory and EEPROM, where the system diagnostics logic is located, at 46. The front-end module also preferably supports other meter functions, including meter component self-checks, A/D sampling, energy calculations, present demand, instantaneous values, any optional outputs, and meter communications in addition to the system and installation diagnostics and Toolbox display performed by the system 20 of the present invention. The display in this embodiment is a liquid crystal display 33 preferably including nine seven-segment digits, three decimal points and a plurality of icons useful in displaying electrical system information normally displayed by conventional meters as well as the diagnostic data generated by the system of the present invention, substantially as shown in FIG. 3.
The meter 34 also includes a register module 48 having a microprocessor 50 including: read only memory; random access memory 51, which also serves in part as system storage memory; a 96 segment LCD display driver; and 24 I/O lines. In this embodiment, the read only memory and register CPU 50 include the display logic for generating the Toolbox display as well as the diagnostic error codes generated by the system 20 of the present invention. The register module 48 also supports other meter functions such as maintaining the billing values and billing register related functions, as well as time related functions including self-read, time of use, time of operation, and mass memory.
It should be noted, that in the embodiment of the meter 34 shown in FIG. 3, the system 20 of the present invention utilizes an 8 byte A/D converter 26 for sensing voltage signals, and an external 12 byte A/D converter 28 for sensing current samples. As will be appreciated by those skilled in the art, the current converter 28 requires higher resolution since current varies over a wider range than voltage. It will also be appreciated by those skilled in the art that it is preferable to have separate converters for simultaneously sensing the current and voltage so that the phase error caused by the current transformer may be directly compensated by adjusting the delay between the current sample and the voltage sample. Thus, in the event the current transformer is ideal and imparts no phase delay, then voltage and current can be sampled simultaneously with the independent converters 26 and 28.
The display logic for generating the Toolbox display and diagnostic error message of the system 20 is part of the display logic 52 which is implemented by the register CPU 50 in the particular embodiment of FIG. 3. It will be appreciated by those skilled in the art, however, that the logic and CPU capabilities of the system of the present invention may be implemented in a simpler single processor architecture (such as shown in FIG. 1), as well as the architecture shown in FIG. 3, or other hardware implementations without departing from the spirit of the present invention.
The system 20 of the present invention provides a full range of system diagnostic capabilities and diagnostic display functions through the "Toolbox" display. The system and installation diagnostics are defined in part by the user via the programming software. The Toolbox is a display of a fixed set of diagnostic information contained in a special mode of operation that can be accessed by a user, typically field personnel, preferably by activating a magnetic switch on the meter. Each of the diagnostic capabilities will be discussed in further detail below.
System and Installation Diagnostics
The system 20 of the present invention performs a plurality of system and installation diagnostics which may indicate potential problems with the electrical service, the incorrect installation of the meter, or internal meter malfunctions. Although these diagnostics may vary depending upon the type of electrical service supported by the meter, the below-described diagnostics are typically performed by the system.
Referring to FIG. 4, the system and installation diagnostics are also preferably implemented as a state machine. In the preferred embodiment, the diagnostics consist of four diagnostics that the user may choose for the meter to perform--(1) Polarity, Cross Phase and Energy Flow Check; (2) Phase Voltage Deviation Check; (3) Current Transformer Check; and (4) Per-Phase Power Factor Check. All selected diagnostics are performed by the meter at least once every 5 sample intervals.
When any error condition occurs according to the parameters defined by the user corresponding to the failure of a diagnostic, the meter displays information to indicate the error condition, and optionally triggers an output contact closure, such as a mercury wetted relay or a solid state contact programmed as an "Error Condition Alert." When an optional output is programmed as an Error Condition Alert, this output contact will close whenever any diagnostic error that has been selected by the user is triggered.
Referring again to FIG. 4, the system 20 of the present invention preferably iterates through a series of calculations and diagnostic checks, shown at 54-62. In the preferred embodiment, processing time is divided into sample intervals equal to 60 periods of the power line clock. For example, in a 50 Hz installation, this is 1.2 seconds. In a 60 Hz installation, the sampling interval would be 1 second.
Using a simple counter, the system 20 performs the necessary sampling and calculations to determine the angle of IA (preferably relative to the base phasor VA), as well as performing Diagnostic Check #1 during the first interval, as shown at 54.
In the second interval, at 56, the system 20 accumulates the necessary samples to calculate the angle for IB and performs Diagnostic Check #2.
In the third interval, at 58, the system accumulates the necessary samples to calculate the phase angle for IC and performs Diagnostic Check #3.
In the fourth interval, at 60, the system accumulates the necessary samples to calculate the phase angle for VB and performs Diagnostic Check #4.
In the fifth sample interval, at 62, the system accumulates the necessary samples to calculate the phase angle for VC and sets the counter to zero. The counter is incremented (at 64) at the end of each of these intervals, and the sequence is repeated continuously. Thus, in a 60 Hz system, the phase angle for each of the current and voltage phasors is calculated, and each of the four diagnostic checks are performed, once every 5 seconds. As will be appreciated by those skilled in the art, different time intervals can be implemented and/or the sub-routines of 54-62 can be modified to accommodate more frequent or infrequent checks of one or more of the selected diagnostics as desired.
DIAGNOSTIC #1--Polarity, Cross Phase and Energy Flow Check
Referring to FIGS. 5 and 6, the Polarity, Cross Phase and Energy Flow Check is designed to check for reversed polarity of any phase voltage or current, and to check for voltage from one phase being incorrectly wired to the current from a different phase. This condition may also result from the presence of cogeneration. This check is accomplished by periodically measuring the angle for each voltage and current phasor with respect to a reference phasor (preferably VA). Each angle is compared to its ideal angle, defined as the angle which would result from a balanced, purely resistive load. If any voltage angle is lagging or leading its ideal angle by more than a predefined amount, (preferably 10�), or if any current angle is lagging or leading its ideal angle by more than a second predetermined amount (preferably 90�), the meter indicates a Diagnostic #1 error.
As shown in FIG. 5, the Polarity, Cross Phase and Energy Flow Check diagnostic routine 66 of the system 20 first checks each angle (where applicable for the particular electrical system to which the meter is connected) of each of the current and voltage phasors (at 68-76) to determine whether each is within tolerance of the predetermined ideal for an ABC rotation. If any of the angles are not within tolerance of the ideal, the system sets the abc flag false (at 78) and proceeds (as shown in FIG. 6) to check each of the angles, assuming a CBA rotation.
If all of the angles are determined at 68-76 to be within tolerance of their predetermined ideal, the system 20 sets the abc flag true, at 80, and proceeds to check the angles assuming a CBA rotation.
Referring now to FIG. 6, once the ABC rotation check is performed, the system proceeds at 82-90 to check the angles for each of the current and voltage phasors to determine whether, for a CBA rotation, the phase angles are within tolerance of the predetermined ideal angles. If any one of the phase angles is outside of the range of tolerance for the predetermined ideal angle for that phasor, the system sets the cba flag false, at 92. If all of the phase angles are determined to be within tolerance of the predetermined ideal angles, the system sets the cba flag true, at 94. The system 20 then determines whether either the abc or the cba flag is true. If either is true, this diagnostic check is passed. If neither the abc flag nor the cba flag is true, the diagnostic check has failed for both ABC and CBA rotations, indicating a diagnostic error. It will be appreciated that the system may alternatively provide a prompt which allows the user to specify an ABC or CBA rotation at installation. If this information is supplied, the system would then check the angles only for the appropriate rotation.
When a diagnostic error is determined, the system records the occurrence of the error and displays the error as further described hereinafter. In the preferred embodiment, however, the initial display of this diagnostic error will not occur until the error condition has been present for three consecutive checks.
As will be appreciated by those skilled in the art, this diagnostic may indicate one of several problems, including cross phasing of a potential or current circuit, incorrect polarity of a potential or current circuit, reverse energy flow of one or more phases, or internal meter measurement malfunction.
DIAGNOSTIC #2--Phase Voltage Deviation Check
Referring now to FIGS. 7 and 8, the Phase Voltage Deviation Check is designed to check, at 98, for any phase voltage being outside an envelope defined by the user. This is actually a check of the distribution transformer voltage gap. This check is accomplished by periodically measuring the voltage for each phase and checking it against a predefined voltage envelope referenced by the program software.
The formula used for this check is: ##EQU1##
If any phase voltage is above Vupper or below Vlower, the meter will indicate a Phase Voltage Envelope Diagnostic Error.
It should be noted that in the preferred embodiment, the system 20 checks, at 100, to determine whether the electrical service supported by the meter incorporating the system 20 is a three element, four wire delta service. If so, the system calculates special case upper and lower bounds for the phase C voltage, as shown at 102.
Again, if either of the phase B or phase C voltages exceeds the predetermined bounds, the system indicates the failure of this diagnostic check (at 104 or 106), indicating a diagnostic error, and the error is recorded and the appropriate error message is displayed as hereinafter described. Otherwise, this diagnostic check is passed (at 108) and this check is completed.
It should be noted, however, that in the preferred embodiment, the initial display of this diagnostic error will not occur until the error condition has been present for three consecutive checks.
This diagnostic may indicate a loss of phase potential, incorrect potential transformer ratio, shorted potential transformer windings, incorrect phase voltage, and internal meter measurement malfunction, as well as other potential problems.
DIAGNOSTIG #3--Current Transformer Check
Referring now to FIGS. 9 and 10, in performing the Current Transformer diagnostic, the system 20 will periodically compare the instantaneous RMS current for each phase to the anti-creep value (0.05% of current class). If all three phase currents are above anti-creep, or all three phase currents are below anti-creep, this diagnostic will pass. Any other combination will result in a Diagnostic #3 failure, and a Diagnostic #3 error will be indicated.
Again, however, the recording and display of this diagnostic error will preferably not occur until the error condition has been present for three consecutive checks.
It will be appreciated by those skilled in the art that this diagnostic check can be utilized to indicate any one of several potential problems, such as an open or shorted current transformer circuit.
DIAGNOSTIC #4--Per-Phase Power Factor Check
Referring to FIGS. 11-13, the Per-Phase Power Factor Diagnostic Check is designed to verify that, for each meter phase, the angle between the current phasor and the idealized voltage phasor is within an envelope specified by the user (�1�-90�). Since this tolerance is more restrictive than for Diagnostic #1, the system 20 does not perform this diagnostic check until Diagnostic #1 has passed. This diagnostic may indicate any one of a series of potential problems, including poor load power factor conditions, poor system conditions, or malfunctioning system equipment.
The system 20 first checks the abc and cba rotation flags at 114 and 116. If both of these flags are false, this indicates that Diagnostic #1 has failed. Since the tolerances of this diagnostic are more restrictive than Diagnostic #1, the diagnostic check is aborted.
If either the abc or cba flags are true (indicating that Diagnostic #1 has passed), the system 20 performs the appropriate ABC or CBA rotation checks at 114 and 116, respectively. For an ABC rotation, the system checks the angle between the appropriate current phasor and the idealized voltage phasor, at 118-122, to determine whether the angle is within an envelope specified by the user. If the angle is between the predetermined envelope, the diagnostic is passed at 124. If not, the diagnostic is failed (at 126), indicating a Diagnostic #4 error. In the event of a CBA rotation, the system 20 performs similar envelope checks at 128-132 for the applicable current phasor.
User Definition of Diagnostics
The system preferably allows the user to enable or disable the performance of any one or more of the system diagnostics during installation of the meter. If the diagnostics are implemented, the system also provides for user-defined parameters, preferably as described below.
To activate or deactivate any of the above-described diagnostics checks, the user must respond to the following types of prompts in the programming software for each diagnostic check supported by the system:
For each "Diagnostic N" (where N represents one of the diagnostic numbers 1-4), the user, upon pressing the return key, gets a menu, preferably including the following options:
The Disable option disables the implementation of that diagnostic.
The Ignore option, if implemented, means that the diagnostic will affect the error condition alert (as hereinafter described), but will not be displayed.
The Lock option, if implemented, will cause the meter display to lock on the diagnostic error message in the event a diagnostic error is determined.
The Scroll option, if implemented, will cause a diagnostic error message to be displayed, when discovered, during the "off time" between each normal mode meter display item.
In addition to the above prompt, the user will be prompted to program the electrical service type (e.g., 4-wire WYE) supported by the particular meter installation.
For Diagnostic #2, the user will also be prompted to program the tolerance for all voltages by inserting a number (preferably corresponding to the percent tolerance) in response to the following prompt:
In the preferred embodiment there is no additional input for Diagnostic #3.
Diagnostic #4 preferably also prompts the user to program the allowable angle difference by inserting a number (1�-90�) in response to the following prompt:
If either the Lock or Scroll option was selected, the meter will display the following message as soon as a diagnostic error is detected:
Er DIAG N (where N=the Diagnostic #)
Also, the Number of Occurrences of this Error Counter is incremented by one whenever the error is detected. As previously mentioned, however, in the preferred embodiment the system acknowledgement and initial display of a diagnostic error will not occur until the error condition has been present for three consecutive checks. Likewise, the error will not be cleared from the display until it has been absent for two consecutive checks.
Again, depending on how the system is programmed at installation, the display will either lock on the error message, or scroll the error message by displaying it during the "off time" between each normal mode meter display item. Various other error display regimes may be adopted consistent with the teachings of the present invention.
Meter Self-Checks
The system 20 of the present invention is also preferably suitably programmed to periodically perform a series of meter self-checks and, if any errors are detected, the system will record the existence of an error condition, display an error code corresponding to the type of error detected, and, depending upon the type of error, take other suitable action, depending upon the type of error detected.
The system preferably implements a series of routines which periodically check for fatal errors and non-fatal errors. Errors are classified as fatal where the detected failure may have corrupted billing data or where the detected failure may cause the meter to operate unreliably in the future. The system 20 preferably conducts meter self-checks of the internal RAM of the meter's register module, the ROM of the register module, the EEPROM of the register module, a spurious RESET of the register module, and the internal RAM, ROM and EEPROM of the front-end module. These meter components are preferably checked whenever power is restored to the meter following an outage or otherwise when the meter is reconfigured. If a RAM, ROM, EEPROM, front module processor error, or other fatal error, is detected, the system 20 will display a predetermined error code corresponding to the detected error, lock the display on the error code until the meter is reinitialized, and cease all meter function except communications.
The system 20 checks for a power-down error by determining if the register module processor has encountered a hardware RESET without first going through a predetermined power outage routine. This event may occur, if a transient on the power line asserts the RESET line momentarily. One method of checking for a spurious RESET is to write a special byte to the register EEPROM as the last step in handling an outage. If this special byte is not present on power-up, a spurious RESET has occurred. The system 20 will then display the power-down error code and cease all meter functions except communications.
The system similarly checks for RAM, ROM, EEPROM, and processor failures in the front-end module, as described above. In the embodiment integrated in the meter of FIG. 3, the front-end module will stop communicating with the register module if any front-end module fatal errors are discovered. If the front-end module fails to communicate with the register module for over five seconds, it is presumed that one of these errors has been detected, the front-end processor failure error code is displayed, and the 68HC11 RESET line is asserted until the front-end module resumes normal operation.
The meter self-checks implemented by the system also preferably include a series of non-fatal errors, such as register full scale overflow, system clock, time of use (TOU), mass memory, reverse power flow, and low battery error conditions.
For example, a register full scale overflow error will be reported if the peak Kw register exceeds a pre-programmed register full scale value. If this event is detected, the system displays a register full scale overflow error, which error will be cleared when the meter is reset or when the error is cleared by a predefined programming device.
A clock error will be reported if the minute, hour, date or month data are out of a predefined range. If a clock error occurs, the TOU and mass memory options will be disabled and will cease recording interval data until the meter is reconfigured.
A TOU error will be reported if an internal TOU parameter becomes corrupted and contains a value outside of its predefined accepted range. If a TOU error occurs, the appropriate error code will be displayed and the TOU option will be disabled.
A mass memory error will be reported if an internal mass memory parameter becomes corrupted or is out of its predefined acceptable range. If a mass memory error occurs, the appropriate error code will be displayed and the mass memory option will be disabled.
A reverse power flow error will be reported if the front-end module detects the equivalent of one complete and continuous disk revolution in the reverse direction. This error will be reported regardless of whether energy is detented or undetented.
A low battery error will be reported if the LOBAT signal on the power supply integrated circuit is asserted when its level is checked. If a low battery error is detected, the appropriate error code will be displayed and, as with a clock error, all TOU and mass memory options will be disabled. If the battery is replaced prior to any power outage, the low battery error will be cleared when the battery voltage rises above a predefined threshold value. However, if the battery voltage was below the threshold when a power outage occurred, the meter must be reconfigured to clear this error.
The system also preferably checks for register full scale overflows at the end of each demand interval, and preferably checks for clock, TOU and mass memory errors at power up, 2300 hours, and on any type of meter reconfiguration. The reverse power flow error is preferably checked by the system each second, and the low battery error is checked on power up and once each interval.
In the preferred embodiment of the system 20, the system allows the user to select which of the meter self-checks will be implemented. In the preferred embodiment, if any one of the selected non-fatal errors is detected, the system will display a predetermined error code corresponding to the detected error during the off-time between normal display item. Alternatively, the system may allow for the user to program the system to lock the display on the error code of any non-fatal error, once any such error is detected. In this event, activation of a switch by the user will cause the meter to scroll through the normal display list one time and then lock back on the non-fatal error display.
It should be noted that, in the preferred embodiment, fatal error checks cannot be disabled. If any non-fatal error is not selected, it will not be displayed or flagged.
It will be appreciated by those skilled in the art that various display regimes may be implemented. For example, the system may be programmed to lock the display on the error code corresponding to any non-fatal error detected until a magnetic switch is activated. Upon activation of the magnetic switch, the system may then scroll through its normal display, then lock back on the display of the non-fatal error code. Alternatively, the system could be programmed to continue to scroll through a predefined display list, periodically displaying any and all non-fatal error codes.
Other meter components may similarly be periodically checked using conventional means and assigned error codes which may be displayed when appropriate to alert the user of possible data corruption or unreliable operation of the meter.
Toolbox Mode
The Diagnostics Toolbox is a fixed selected set of display items preferably in the format illustrated in FIG. 14. In the preferred embodiment, the Toolbox display is accessed via a magnetic reed switch which is located at the 12 O'clock position on the meter board, and is activated by keeping a magnet next to the reed switch for at least 5 seconds. This may be accomplished by the user by placing a magnet on top of the meter.
When accessed, the Toolbox display items are each displayed individually as shown and in the sequence indicated in FIG. 14. Once the meter is in Toolbox display mode, it will scroll through all of the Toolbox display items at least one time. When the magnet is removed, the meter will finish scrolling to the end of the Toolbox display list and then revert to Normal mode operation. The TEST annunciator will flash two times per second during the entire time the meter is in Toolbox mode.
All of the # DIAG Error counters are preferably cleared by an external device, such as by a hand-held personal computer, or through normal communications. In the preferred embodiment, the maximum value of each counter is 255.
While the meter is in Toolbox mode, it continues to perform meter operations as usual. This assures that meter operation is not affected even if the magnet is left on top of the meter for an extended period of time. The system continually updates the displayed Toolbox quantities as they change in value during the entire time the meter is in Toolbox mode.
While in Toolbox mode, the Watt Disk Emulator scrolls at the rate of one revolution per 1.33 seconds in the direction of power flow of the phase for which information is being displayed at that point in time. For example, while A phase voltage, current, voltage angle and current angle are being displayed, the Watt Disk Emulator scrolls once per second in the direction of power flow of phase A. As soon as the phase B values (if present) are displayed, the Watt Disk Emulator reverses direction if the power flow in B phase is opposite that of A phase. The Watt Disk Emulator is turned off while the four diagnostic error counters are displayed.
Because continuous potential indication is required by the customer, three potential indicators, preferably labelled VA, VB and VC, are present on the display. These indicators are "ON" as long as the corresponding voltage is above the predefined threshold. The threshold is preferably defined as 75% of the lowest voltage the meter is rated to operate at. If any voltage drops below the threshold, its indicator will flash, preferably at a rate of two times per second.
When more than one error exists at the same time, the information relating to only one of the errors is displayed, based upon a predefined priority. The following priorities are established in the preferred embodiment of the system:
1. Meter Self-check errors take priority over System and Installation Diagnostic errors.
2. Since only one System and Installation Diagnostic error can be displayed at a time, the highest priority error will be the one that is displayed using a pre-defined priority list.
If two or more System and Installation Diagnostic errors exist, the highest priority error will be the one that is displayed and the one that triggers the output contact closure. If this error is then remedied, the next highest priority error that still exists will then be displayed and will again trigger the output contact closure. The output contact closure (Error Condition Alert) thus remains asserted as long as one or more of the diagnostic errors have been triggered.
As described above and illustrated in FIG. 14, the Toolbox display also preferably displays the instantaneous value of the current and voltage for each phase, and their phase relationship to the voltage on phase A. With this information, the user can construct a phasor diagram which assists in determining the correct installation and operation of the meter. This display also shows the number of diagnostic errors accumulated for each diagnostic since the last time the system was cleared.
An example of the desired relationship between a phasor diagram for a three phase meter installation and a Toolbox display is shown in FIGS. 14 and 15, respectively. With the phase current, voltage and angle information given in the Toolbox display, the user should be able to construct a phasor diagram as shown in FIG. 15. This will allow the user to get a snapshot of the power system status, and to identify any peculiarities or errors. As mentioned before, the Toolbox display will also give the status of the four diagnostic counters which will provide the user with more detailed status information for the system.
Calculation of Phase Angles
In the preferred embodiment, the angle information for phase current and voltages utilized in system Diagnostics #1 and #4, and required for display in the Toolbox display, are determined from accumulated current and voltage values for each phase, as well as the accumulated products, Q and Y (as hereinafter defined). The voltage on phase A is preferably used as the reference (or base phasor) for the other angles. The phase A voltage angle will thus appear as 0.0� in the display. The five other angle values for (IA, IB, IC, VB, VC) will be reported with respect to the voltage on phase A, and will always be given with respect to a lagging reference.
1. The Phase Angle Between VA and IA
If the Power and Apparent Power are known, the Power Factor can be derived. The relationship is as follows: ##EQU2## The phase angle (θ) between voltage and current can then be calculated as follows:
θ=arccos(Power Factor)
The device of the present invention can also determine whether the current is leading or lagging the voltage by examining the sign of the reactive power. If the reactive power is positive, then the current is lagging the voltage, and if the reactive power is negative, then the current is leading the voltage.
In the preferred embodiment, the power, RMS voltage, and RMS current are calculated every 60 line cycles for each phase on the meter. This is accomplished by taking 481 samples of the voltage and current over a 60 cycle period. The necessary multiplications and accumulations are done, and then these values are averaged to yield the power, RMS voltage, and RMS current for a given 60 line cycles. These quantities are then used at the end of each 60 line cycle to calculate a power factor for each phase.
The reactive power can be calculated much the same way as the power, except that a 90 degree phase shift must be induced between the current and voltage measurements. This phase shift can be achieved by taking the present current sample and multiplying it by a delayed voltage sample (stored in memory) corresponding to a 90 degree phase shift.
2. Derivation of a Generalized Phase Angle Calculation Method
As demonstrated below, the method of calculating the phase angle of VA to IA can be generalized to calculate the angle between any reference phasor (such as VA) and any other phasors (such as VB, IB, VC, or IC).
Referring now to FIG. 16, consider two sinusoidal waves of the same frequency, different magnitude, and phase shifted one from another as follows:
a(t)=A cos (ωt)
b(t)=B cos (ωt-θ).
By representing the cosine argument as (ωt-θ), the implicit assumption is that θ represents a lagging phase shift from reference a(t) to b(t). The respective position refers to whether b(t) reaches its maximum value before or after a(t) with respect to time. If b(t) reaches a maximum after a(t), then it is said to lag a(t). If b(t) reaches a maximum before a(t), then it is said to lead a(t).
In order to isolate the phase angle θ, the average value of the product of the two sine waves will be evaluated. This average value will be denoted by Q. The equation for the average value is as follows: ##EQU3## where A and B represent the amplitudes of sinusoidal waves a(t) and b(t) respectively. The amplitude, XMAX, of a sinusoidal wave is related to the RMS value, XRMS, by the following relationship ##EQU4## Therefore, ##EQU5## Substituting these relationships into the equation for Q, the equation becomes:
Q=ARMS BRMS cos θ
or, ##EQU6## and finally, ##EQU7##
Therefore, if the average value of the product of two sine waves and the RMS values of the two individual waves is known, then the angle between the two waves can be calculated. This information alone will not allow us to determine whether b(t) is lagging or leading a(t). However, if the sine of the angle θ were known, then whether the angle was a leading or lagging angle could be determined.
In order to determine the sine of the angle, consider the average value of the products of two sinusoidal waves, where a(t) is shifted by 90 degrees or π/2 radians. An expression for the shifted version of a(t) is as follows: ##EQU8## The average value of the product of a(t) and b(t) will be referred to as quantity Y. The equation is as follows: ##EQU9## Solving the integral yields the following relationship: ##EQU10##
Therefore, if the average value of the product of the two sine waves (Q) is known, the average value of the product of the sine waves with the reference wave delay shifted by 90 degrees (Y) is known, and the RMS value for each of the waves is known, then the phase angle can be calculated and a determination made whether the unreferenced wave is lagging or leading the reference wave. The two equations which can be used to determine the magnitude of the phase angles are as follows: ##EQU11## Whether the angle is leading or lagging can be evaluated by examining the signs of the arccosine and arcsine arguments. Since a positive angle corresponds to a lagging angle, then the following is true for determining whether the angle is leading or lagging:
Arccosine argument(+), arcsine argument (+)-Lagging between 0 and 90 degrees;
Arccosine argument (-), arcsine argument (+)-Lagging between 90 and 180 degrees;
Arccosine argument (-), arcsine argument (-)-Leading between 90 and 180 degrees; and
Arccosine argument (+), arcsine argument (-)-Leading between 0 and 90 degrees.
Therefore, if Q, Y, and RMS values for a(t) and b(t) are available then the phase angle between these sinusoidal waves can be determined.
The above-described technique for finding the phase angle will thus apply to any pair of voltages or currents. For instance, to determine the angle between VB and VA, the two required quantities that will have to be calculated are the average value of the product of two waves (QVAB), and the average value of the product of the two waves with VA shifted by 90 degrees (YVAB).
As previously mentioned, the meter incorporating the preferred embodiment of the system 20 samples VA and VB 481 times every 60 line cycles. If the product of VA and VB is calculated for each of the 481 samples and accumulated over a sample interval, then at the end of the sample interval the average value of the product of the two waves, QVAB, can be calculated. The equation for QVAB is as follows: ##EQU12## where C is a calibration scaling factor used to compensate for the reduction of the phase voltages to a measurable level.
YVAB can be found in a similar fashion from: ##EQU13## where the C for the YVAB calculation is the same as the C for the QVAB calculation and VA(n-2) is the voltage VA two samples previous to the sample, VA(n).
The sampling is designed so that two consecutive samples of a signal are 44.91 degrees apart. Therefore, if the voltage sample from the two samples ago is taken, this will result in a phase shift of 89.82 degrees which is approximately 90 degrees.
It should be noted that instead of using shifting samples of VA, the other quantities could be shifted by 90� to calculate the phase angle. This will result in the same results for the magnitude of the Y value. However, this will change the sign information because the phase angle is shifted by 180�. With this implementation, the following sign relationships between the arcsine and arccosine arguments exists:
Arccos (+), arcsine (-)-Lagging angle between 0 and 90 degrees;
Arccos (-), arcsine (-)-Lagging angle between 90 and 180 degrees;
Arccos (-), arcsine (+)-Leading angle between 90 and 180 degrees;
Arccos (+), arcsine (+)-Leading angle between 0 and 90 degrees.
If the new values were to be calculated every sample interval for the phase angles needed for the Toolbox display, then the ten product and accumulation terms shown above would have to be calculated every sample interval. Due to the excessive use of processor time and RAM required to accumulate all ten terms every sample interval, only one pair of terms is preferably considered for each sample interval. This limits the use of processor time and RAM, and it makes new phase angle values available for the Toolbox display every five sample intervals.
In the preferred embodiment, the product terms are calculated and accumulated in the following order:
1. First sample interval-VA *IA and VA(-90�) *IA for phase angle IA ;
2. Second sample interval-VA *IB and VA(-90�) *IB for phase angle IB ;
3. Third sample interval-VA *IC and VA(-90�) *IC for phase angle IC ;
4. Fourth sample interval-VA *VB and VA(-90�) *VB for phase angle VB ; and
5. Fifth sample interval-VA *VC and VA(-90�) *VC for phase angle VC.
After the fifth sample interval, the sequence begins again, accumulating the necessary Q and Y values for phase angle IA. The samples for VA are stored during each sample interval. This thus requires that two additional values be stored for VA at each interval, the two previous VA.
In the preferred embodiment, these functions are implemented in 68HC11 assembly code. The multiplication and accumulation of these product terms occurs in the front-end sampling interrupt routine. The voltage values are 8-bit values and the current values are 12-bit values. Since VA is always involved in any of the multiplications, this will mean some of the multiplies will be 8�8 bit and some will be 8�12 bit. Since it is desirable to use the same algorithm to do all the multiplications, the 8-bit values are extended to 12-bit values such that an 8�12 bit multiplication algorithm is used exclusively in the preferred embodiment.
The 8-bit voltage values for VB and VC are sign extended to 12-bit values so that all the multiplication and accumulation of product terms for finding the phase angles are handled by two algorithms, one for the accumulation of product terms for the Y value and one for the accumulation of product terms of the Q value. The sign extension of voltage values VB and VC are performed during every sample period. This makes special checks unnecessary for identifying the sample intervals in which these quantities are needed, because they are available during every sample interval.
All 12-bit values for current and voltages are preferably stored in 16-bit registers in the memory, because the memory is segmented into byte boundaries.
The front-end sampling routine must have a way of identifying which product term is to be calculated at each sample interval. A counter identifier is preferably utilized as an index to access the correct value for the multiplications necessary in the accumulation of the Q and Y values.
In order to accumulate the two product terms, two accumulators are set aside in the memory map. The size of each of these accumulators is the same, since both are doing 8�12 bit multiplies. The largest possible accumulated value is as follows:
Largest 8 bit value=128
Largest 12 bit value=2048
Largest accumulated result=481*128*2048=07 84 00 00(hex)
Therefore, each accumulator is four bytes long to accommodate the worst case result. Two four-byte accumulators are therefore set aside to accumulate each pair of product terms for each sample interval.
At the end of each sample interval, the results in the two four-byte accumulators are stored in two four-byte holding areas to await processing by the background routines necessary in completing the angle calculation during the next interval.
Once the accumulated pairs have been transferred to the holding registers at the end of a sample interval, then the remaining calculations needed to determine the phase angle take place during the next sampling interval in the background, while the accumulation for the next pair is taking place in the foreground. These background routines must also have a way of determining on which pair of accumulated product terms they are working. A separate counter identifier is used for these background routines which operates in a similar fashion to the counter identifier for the front-end sampling interrupt. However, it is possible to use the same counter, since this identifier will always be one count behind the counter identifier for the front-end module sampling interrupt routine.
The meter 34 illustrated in FIGS. 2, 3, 17 and 18, into which the system 20 of the present invention is preferably integrated, is a solid state single function KW/Kwh meter utilizing digital sampling techniques to provide conventional Kw/Kwh demand, time of use, and other conventional real time billing information in addition to the diagnostic information generated by the system 20 of the present invention. The meter 34 is preferably programmed using software that runs on an IBM compatible personal computer using the MS-DOS operating system. This software includes the logic for prompting the user to provide meter configuration parameters and preferably includes the installation prompts which provide for user-defined parameters for the diagnostics supported by the system 20 of the present invention, so that a hand-held personal computer can be plugged into a communications port on the meter to program the meter at installation.
FIG. 17 illustrates the front-end module 44 of the meter 34 into which the system 20 of the present invention is preferably incorporated. The front-end module 44 preferably includes a Motorola MC68HC11KA4 microprocessor 140 running in single chip mode, an integral 8-bit A/D converter 142, which serves as the voltage converter 26 in the system 20 of the present invention, 24K bytes read only memory (ROM), 640 bytes electrically erasable programmable read only memory (EEPROM), and 768 bytes random access memory (RAM), all shown at 144. The ROM and EEPROM include the diagnostic logic, and the RAM serves as storage memory for the present invention. An external 12 bit A/D converter, shown at 146 serves as the current A/D converter 28 for the system 20 of the present invention.
An additional error condition alert function may be implemented as an option on the front-end module 44. This function utilizes a line out to, for example, an external communication device, which can be activated whenever an error condition is determined. This optional function may be utilized by the system 20 of the present invention for activating and communicating the existence of error conditions for any one of the diagnostics performed by the system 20 of the present invention.
An option board 146 may be incorporated into the front-end module 44 to provide various signals to the outside world. For example, the error condition alert may be assigned to a low current solid-state or Mercury-wetted relay to indicate when one or more diagnostic errors have been determined. Other known ancillary functions, such as automated meter reading or real time billing, may be implemented on option board 146, or on a similarly configured option board utilized with the front-end module 44.
Referring now to FIG. 18, the register module 48 of the meter 34 into which the system 20 of the present invention is preferably incorporated, includes a NECuPD75316GF single chip microprocessor 148, including 16K bytes of ROM, shown at 150, 512�4 bits of RAM, shown at 152, and a 96 segment LCD display driver 154, suitable for driving an LCD display 156 such as the particular type of display 33 shown in FIG. 3 and utilized in the preferred embodiment of the meter 34.
Serial data will be transferred between the front-end module 44 and the register module 48 via a four wire synchronous serial data link shown respectively at 158 in FIG. 17 and 160 in FIG. 18. The front-end module will monitor and update the status of all of the diagnostics performed by the system 20 of the present invention and, periodically (preferably once per second) communicate these statuses to the register module 48 via the above-described serial communications link for display, as well as for storage of volatile data in the event of a power outage. In addition, any instantaneous quantity required for display in the Toolbox display of the present invention, will be communicated by the front-end module as needed to the register module. The front-end module 44 also communicates various other conventional meter information to the register module 48, such as the amount of energy (in Kwh) registered for the past 60 line cycles, as well as its direction (delivered or received), present demand and end-of-interval information.
Information which may be communicated from the register module 48 to the front-end module 44, typically includes periodic meter register status information.
Referring again to FIG. 17, the front-end module 44 enables the measurement of per-phase voltage, current and watts for one sample interval (60 line cycles). As previously described, the front-end module preferably performs 481 samples per 60 line cycles, which corresponds to 481 Hz when the line frequency is 60 Hz, and approximately 401 Hz when the line frequency is 50 Hz. The sampling frequency is recalculated every 60 cycles, based on the measured line frequency. As previously described, the diagnostic functions of the present invention, including determination of instantaneous per-phase current, voltage, watts and phase angle, are preferably performed by the front-end module 44 when the system is incorporated in a meter of the type shown in FIG. 3.
Referring again to FIGS. 3 and 18, the register module 48 preferably performs the function of driving the LCD display 33 in the meter 34. As previously described, the Toolbox display of the present invention may be implemented by activating an alternate display switch (not shown) for a predefined period. When activated, the Toolbox display mode is activated and the display will scroll through the Toolbox display list as previously described herein. During a Toolbox display, the "TEST" icon preferably flashes continuously, and the watt disk emulator, shown as the five rectangular icons at the bottom of the display 33, will scroll at a rate of about one revolution per 1.33 seconds. The direction of the watt disk emulator will be the same as the direction of power flow for the phase being displayed (left to right if received, right to left if delivered). The meter will leave the Toolbox display mode when the end of the display is reached and the alternate display switch is no longer activated. It should be noted, as previously described, the meter will continue to perform all normal mode meter operations while the Toolbox display sequence is active.
When the alternate display switch is not activated, the meter display 33 operates in normal display mode for the meter 34.
Communication to or from the meter may also be accomplished through the front-end module 44 via connection to the optical port 162.
Thus, the integral electronic meter system diagnostic package of the present invention provides the capability for continuous self-checking of the internal components of the meter, as well as alert field personnel to any discovered error, without interruption of the meter's operation. The system also provides the capability for constant system diagnostic checks, and display of those diagnostic results, to provide pertinent diagnostic data to system personnel during or following installation of the meter.
The system provides the flexibility of allowing the user to program the system to select and define the functions and parameters suitable to the particular service supported by the meter installation.
Finally, the Toolbox display capability of the present invention allows for periodic display of valuable information respecting the internal functioning of the meter as well as the character of the service supported by the meter, again without interruption of normal service and meter operation.
While the best mode for carrying out the invention has been described in detail, those familiar with the art to which this invention relates will recognize various alternative designs and embodiments for practicing the invention as defined by the following claims.
|Cited Patent
|Filing date
|Publication date
|Mar 2, 1983
|Sep 16, 1986
|Siemens-Allis, Inc.
|Method and apparatus for monitoring instantaneous electrical parameters of a power distribution system
|Feb 25, 1992
|May 18, 1993
|Basic Measuring Instruments, Inc.
|Harmonic-adjusted power factor meter
|Oct 6, 1992
|Apr 12, 1994
|Basic Measuring Instruments
|Harmonic-adjusted power factor meter
|Jan 21, 1993
|May 24, 1994
|General Electric Company
|Electrical energy meter with programmable register component
|Citing Patent
|Filing date
|Publication date
|Jun 26, 1996
|May 5, 1998
|Korea Electric Power Corporation
|Optical-electronic demand meter using optical metering outfit
|Aug 7, 1997
|May 25, 1999
|Abb Power T&D Company Inc.
|Current measurement methods and apparatus employing second harmonic scaling
|Nov 30, 1998
|Jul 25, 2000
|Abb Power T&D Company Inc.
|System and method for automatically determining the electrical energy service type to which an energy meter is connected
|Jun 24, 1997
|Aug 29, 2000
|Siemens Power Transmission & Distribution, Llc
|Robust electrical utility meter
|Nov 30, 1998
|Jul 31, 2001
|Abb Power T&D Company Inc.
|System and method for detecting flicker in an electrical energy supply
|Nov 30, 1998
|Apr 16, 2002
|Abb Automation Inc.
|Energy meter with instrumentation systems and methods
|Mar 26, 1998
|Nov 19, 2002
|Abb Automation Inc.
|Apparatus for metering electrical power that determines energy usage data based on downloaded information
|Nov 30, 1998
|Jan 14, 2003
|Abb Automation Inc.
|Energy meter with power quality monitoring systems and methods
|Apr 13, 2001
|Aug 26, 2003
|Badger Meter, Inc.
|Meter register with programming and data port and meter input resolution factor
|Apr 24, 2000
|Dec 2, 2003
|Siemens Power Transmission & Distribution, Inc.
|DC load detection in an electric utility meter
|Jul 27, 1999
|Dec 16, 2003
|Siemens Transmission & Distribution, Llc
|Utility meter having primary and secondary communication circuits
|Jul 21, 2000
|Feb 8, 2005
|Utility Consult Hinzmann & Koenig Ohg
|Supply meter for liquid and gaseous mediums
|Mar 10, 2003
|Jun 14, 2005
|Itron Electricity Metering, Inc.
|Electronic revenue meter with automatic service sensing
|Sep 25, 2002
|Feb 7, 2006
|Landis+Gyr, Inc.
|Utility meter power arrangements and methods
|Sep 12, 2003
|Mar 7, 2006
|Landis & Gyr, Inc.
|Electricity meter with power supply load management
|Feb 13, 2006
|Nov 14, 2006
|Landis+Gyr, Inc.
|Electricity meter with power supply load management
|Mar 1, 2006
|Mar 25, 2008
|Landis+Gyr, Inc.
|Electricity meter with power supply load management
|May 16, 2006
|May 5, 2009
|Avery Long
|Method for detecting and correcting wiring errors in power monitoring applications
|Nov 12, 2009
|Aug 16, 2011
|Itron, Inc.
|Apparatus for electronically measuring or distributing electrical energy
|Mar 10, 2003
|Sep 25, 2003
|Briese Forrest Wayne
|Electronic revenue meter with automatic service sensing
|Sep 12, 2003
|Jul 1, 2004
|Ramirez Anibal Diego
|Electricity meter with power supply load management
|Feb 10, 2005
|Sep 29, 2005
|Landis+Gyr, Inc.
|Utility meter power arrangements and methods
|Mar 1, 2006
|Jul 6, 2006
|Landis+Gyr, Inc.
|Electricity meter with power supply load management
|Feb 13, 2006
|Jul 20, 2006
|Landis+Gyr, Inc..
|Electricity meter with power supply load management
|May 16, 2006
|Sep 14, 2006
|Avery Long
|Method for detecting and correcting wiring errors in power monitoring applications
|Nov 12, 2009
|May 6, 2010
|Itron, Inc.
|Apparatus for electronically measuring or distributing electrical energy
|U.S. Classification
|324/750.3, 708/426, 324/623, 702/79, 324/76.11, 708/422
|International Classification
|Cooperative Classification
|European Classification
|Apr 21, 1997
Owner name: GREYROCK BUSINESS CREDIT, A DIVISION OF NATIONSCRE
Free format text: SECURITY INTEREST;ASSIGNOR:FUTURE LABS, INC.;REEL/FRAME:008454/0760
Effective date: 19970401
|Nov 6, 1998
Owner name: SCHLUMBERGER RESOURCE MANAGEMENT SERVICES, INC., G
Free format text: CHANGE OF ADDRESS;ASSIGNOR:SCHLUMBERGER INDUSTRIES, INC.;REEL/FRAME:009547/0919
Effective date: 19980113
|Mar 22, 1999
|Fee payment
Year of fee payment: 4
|Oct 8, 2002
Owner name: SCHLUMBERGER SEMA INC., GEORGIA
Free format text: CHANGE OF NAME;ASSIGNOR:SCHLUMBERGER RESOURCE MANAGEMENT SERVICES, INC.;REEL/FRAME:013718/0449
Effective date: 20011231
|May 6, 2003
|Fee payment
Year of fee payment: 8
|Jan 19, 2004
Owner name: SCHLUMBERGER ELECTRICITY, INC., SOUTH CAROLINA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNOR:SCHLUMBERGERSEMA INC.;REEL/FRAME:014892/0800
Effective date: 20040116
|Jul 7, 2004
Owner name: WELLS FARGO BANK, NATIONAL ASSOCIATION, AS ADMINIS
Free format text: SECURITY AGREEMENT;ASSIGNOR:SCHLUMBERGER ELECTRICITY, INC.;REEL/FRAME:014822/0327
Effective date: 20040701
|Jul 19, 2004
Owner name: ITRON ELECTRICITY METERING, INC., SOUTH CAROLINA
Free format text: CHANGE OF NAME;ASSIGNOR:SCHLUMBERGER ELECTRICITY, INC.;REEL/FRAME:014863/0313
Effective date: 20040716
|Oct 30, 2004
Owner name: WELLS FARGO BANK, NATIONAL ASSOCIATION, AS ADMINIS
Free format text: AMENDMENT TO GRANTOR NAME CHANGE UNDER SECURITY AGREEMENT;ASSIGNOR:ITRON ELECTRICITY METERING, INC.;REEL/FRAME:015312/0692
Effective date: 20041005
|Jan 18, 2006
Owner name: ITRON, INC., WASHINGTON
Free format text: MERGER;ASSIGNOR:ITRON ELECTRICITY METERING, INC.;REEL/FRAME:017025/0439
Effective date: 20060101
|Apr 24, 2007
|Fee payment
Year of fee payment: 12
|Apr 25, 2007
Owner name: WELLS FARGO BANK, NATIONAL ASSOCIATION, WASHINGTON
Free format text: SECURITY AGREEMENT;ASSIGNOR:ITRON, INC.;REEL/FRAME:019204/0544
Effective date: 20070418
Owner name: WELLS FARGO BANK, NATIONAL ASSOCIATION,WASHINGTON
Free format text: SECURITY AGREEMENT;ASSIGNOR:ITRON, INC.;REEL/FRAME:019204/0544
Effective date: 20070418
|Jun 25, 2007
Owner name: ITRON, INC., CALIFORNIA
Free format text: TERMINATION AND RELEASE OF SECURITY INTEREST IN INTELLECTUAL PROPERTY;ASSIGNOR:WELLS FARGO BANK, NATIONAL ASSOCIATION;REEL/FRAME:019466/0486
Effective date: 20070418
|Aug 15, 2011
Owner name: ITRON, INC., WASHINGTON
Free format text: RELEASE BY SECURED PARTY;ASSIGNOR:WELLS FARGO BANK, NATIONAL ASSOCIATION;REEL/FRAME:026749/0263
Effective date: 20110805
|Aug 16, 2011
Owner name: WELLS FARGO BANK, NATIONAL ASSOCIATION, WASHINGTON
Free format text: SECURITY AGREEMENT;ASSIGNOR:ITRON, INC.;REEL/FRAME:026761/0069
Effective date: 20110805


text:
.0915 -- (House of Representatives - May 19, 2006)
Mr. LARSON of Connecticut. Mr. Speaker, I regret that I could not be present today because of a family medical emergency and I would like to submit this statement for the Record in opposition to the previous question and H. Res. 821, the rule providing for consideration of the FY2007 Military Quality of Life appropriations bill (H.R. 5385).
The Veterans' Administration has treated more than 144,000 returning veterans from Operations Iraqi Freedom and Enduring Freedom, and nearly 30,000 veterans are waiting in line for their first appointment--nearly double the number last year. However, funding for veterans and military retiree health care has barely kept pace with the increasing demand. As a result our young men and women will return home from Iraq and Afghanistan to a health care system that is struggling to take care of current veterans--let alone new ones.
While the bill before us today is certainly an improvement over last year's bill, it still under-funds critical programs and services that our veterans and military retirees rely on. For instance, even as the Defense Authorization Bill (H.R. 5122) we passed last week rightfully rejected the President's plan to increase fees for military retirees, this bill still falls $735 million short of the level needed to ensure that military retirees do not face having their TRICARE fees doubled or tripled. In addition, this bill uses a budgeting gimmick to designate $507 million for 20 military construction projects as emergency spending so that the committee could keep the overall total under the bill's allocation level--jeopardizing this critical funding by leaving it vulnerable to procedural points of order that could strip it from this bill. Finally, this bill provides $25.4 billion for veteran's medical services--$2.6 billion more than last year, but still $400 million below the recommendation of the Independent Budget and $2.8 billion below the level recommended by the House Veterans Affairs Committee Democrats.
Unfortunately, Democratic amendments to address these shortcomings were rejected by Republicans on the Appropriations Committee and are blocked from being considered here today by this rule. These pragmatic measures would have made this a stronger bill that fulfils our promise to our military retirees and veterans. I urge the defeat of the previous question and this rule so that we can debate critical amendments to ensure that veterans and military retirees get the timely, quality, and affordable health care they deserve.

subset: commoncrawl


text:
[new] how to design a classroom download [MnUr] file now
About how to design a classroom
FileName: how to design a classroom
Languages: ENGLISH
Release date:
03/31/2015 20:21:12
Visit Website | More Programs
Last month's downloads::
Windows XP/Vista/7/8 (32-Bit/64-Bit)
Download URLs
Related software:
03/31/2015 20:21:12
Eventually most marketers are presented with the task of creating a positioning statement for their business, brand, product or service. While some may dismiss this ...By clicking "Create Account" you are indicating that you have read and agree to the Terms of service.Learning How to Learn: Powerful mental tools to help you master tough subjects from University of California, San Diego . This course gives you easy access to the ...We talk about current trends in the web design and web development community.Understanding By Design [Grant Wiggins, Jay McTighe] on Amazon.com. *FREE* shipping on qualifying offers. Drawing on feedback from thousands of educators around …An entrepreneurial, non-profit organisation that fosters the efficient and restorative use of resources to create a more secure, prosperous, and life-sustaining world.Big News on Design. Includes blogs, news, and community conversations about Design.By Rain Noe - Mar 27 Who Knows More About Working Wood, Furniture Designers or Shipwrights? In this YouTube channel, a Master Shipwright shares ...12/19/2008 · Learn how to lose weight FAST in Photoshop! Use Adobe Photoshop to virtually put someone on a diet! The same method can be used to add weight to a …New Ways to Push Your Practice Bringing you SIX new workshops and courses in customer experience, brand strategy, leadership, product definition and design, …
how to design a classroom
Most downloaded this week:
dance martha munizzi mp3 download
sony sound forge full crack
tiesto mp3 break my fall
unkle war games
shrek 2 the game walkthrough
mom love son
moroff 3d game
podzemnuu perexod serial
how to get over rejection
saturday night fever soundtrack torrent
sobipro joomla google map download
futurama tyng theme download
RuntimePack 14.4.12 Full (32 and 64 bit)
microsoft live lab
paintball sound
building layout programs
full family guy
clymer bmw r1200c manual
sorting and searching algorithms a cookbook
nuvaring starter pack
mp3 o great in de
dreamfall the longest
pre paid game card
samsung le46a556p
hotel rwanda 1080p


text:
Chappell builds support for cancer program at Children's Hospital Central California
November 22, 2013
FRESNO, Calif.- PGA TOUR player Kevin Chappell will return to his native Fresno on Tuesday, Nov. 26 to host a golf clinic for Kevin Chappell's "Champion for Children's", the program that he launched last year in support of the oncology program at Children's Hospital Central California.
The event will take place at San Joaquin Country Club from 4:00 - 6:30 p.m. Chappell will lead the golf clinic followed by a reception for those attending. Chappell also plans a visit to the hospital to greet patients and staff. Chappell's support is a part of the "Together, Anything's Possible" program, a PGA TOUR initiative to build a community of tournaments, players, charities, sponsors, partners, volunteers and fans who are committed to making a social investment by helping others.
"Children's Hospital Central California has been a part of my life since I was born," said Chappell. "Being about to provide meaningful support and shine a light on Children's while encouraging others to do the same is one of the best things about my association with the PGA. Each time we visit children in the oncology unit, we learn how far a little gift can go. A simple, good deed can go a really, really long way. I feel if I can give a little of my time and money to help children with life threatening diseases, it can, at the very least, change a person's day for the better."
As a newborn infant, Chappell was treated in the hospital's Neonatal Intensive Care Unit. He went on to have a healthy childhood and discovered his talent and passion for golf. The Buchanan High School (Clovis, CA) graduate led the UCLA Bruins golf team to a national championship and turned professional in 2008.
Chappell visited the hospital's oncology unit in 2011 and decided it was where he wanted to focus his resources and persuade his fans to join him in this support.
"Patients being treated for cancer at Children's Hospital Central California receive world-class care and the most up-to-date treatment possible," said Todd Suntrapak, Children's president and CEO. "We participate in the Children's Oncology Group, partnering with research scientists from around the world in our efforts to understand the causes of cancer and find more effective treatments, and research done here contributes to this effort. Thanks to supporters like Kevin Chappell, our children and families have an incredible network of care and support."
For more information on Kevin Chappell's Champion for Children's, visit Kevin's website at kevinchappell.com or go to the Children's Hospital Central California giving website at waystogive.childrenscentralcal.org.
Related News
- Winners announced for 2017 David Hearn Foundation Kia grant 07/21/2017
- Despite injury, Van Pelt gives back 07/21/2017
- John Deere Classic's impact extends beyond tournament week 07/17/2017
- Creighton Farms Invitational hosted by Jack Nicklaus raises $1.4 million for children's health charities 07/17/2017
- Scott Stallings: Working with wounded soldiers is 'life changing' 07/15/2017
- VIDEO: Notah Begay helps raise $218K for charities 07/13/2017
- Shriners Hospitals for Children launches Birdies for Charity 07/11/2017
- Valero Texas Open's 2017 Birdies for Charity campaign nets $2.35 million for community 07/11/2017
- Rickie Fowler Endows NAE Grand Challenges at OSU 07/11/2017
- WATCH: FedEx Junior Business Challenge at the Travelers Championship 07/11/2017


text:
This property is no longer available.Search for Available Property.
220 acres in Blanco County, Texas
Beautiful land near Johnson City once owned by LBJ himself. Creek flowing through property, barn, stock tanks, fenced and cross fenced, vistas, 30 mile views - perfect for horses or fun recreational land. Deer, hogs, dove, quail, turkey and other native hill country wildlife. Call for more information and personal showing. Updated survey conveys. Property adjoins 2,000 acre conservation ranch known as "The Preserve at Walnut Springs." Owner will sell all/part with possible owner financing.
Flat Creek Rd 5.5 miles to Shiloh Rd
Financing Options
Mortgage Options
Owner Finance
Tax Data (Show Tax Data)
Loading Tax Data
Other Land for Sale in Blanco County
- Blanco County Properties by City
- Blanco, TX Land for Sale
- Johnson City, TX Land for Sale
- Round Mountain, TX Land for Sale
- Hye, TX Land for Sale
- Cypress Mill, TX Land for Sale
NOTE: The information contained herein is provided as general information only and has been obtained from sources deemed reliable. It is provided without any guaranty, warranty or representation, expressed or implied, made by Lands of America, or any related entity, as to the accuracy or completeness of the information. The information is presented subject to errors, omissions, change of price or conditions, prior sale or withdrawal without notice. Prospective purchasers should make their own investigations, projections and conclusions concerning the information.


text:
Top 10 Problems and Issues Teenagers Face
Are Teenagers the Problem?
I was pushed into writing the ' Top ten problems teenagers face' after I attended a school program, where I had the opportunity to interact with other parents of teenagers. The normal flow of conversation was in the form of complaints about raising teenagers. I could see that each parent was only adding one more item to the list of complaints. I searched the net - the result wasn't much different... it was mostly about how to… get your teen to clean their rooms, handle behavior that is rebellious or aggressive, problematic teens,defiant teenagers, out of control teenagers, make them more responsible etc… This got me thinking, and I concluded that what we need today is to look at the problem in a different light..
Kids go where there is excitement. They stay where there is love - Zig Ziglar
I took a good look at the teenager I raise, followed him around, spoke to him awhile and got in touch with more teenagers and this is what I found...
The health of the teenager is most important
Problems that Teenagers Face Today
Teenagers and Body image
At the very outset they are struggling to come to terms with their body image. The cherubic appearance of childhood gone, they are looking at strangers in the mirror, gawky, disproportionate limbs, facial features, facial hair and acne to mention a few. They don't know what to do with their hands and legs and are often termed 'fidgety'
Search for Identity - the troubled teenager
I see teenagers as those struggling to find a place in society, be recognized by their peers and be accepted for what they are. This is the most important task of their lives at this juncture and they are faced with a lot of anxiety and insecurity as they work around this. I take my son as an example. My son is brought up to be respectful, helpful, kind, gentle, tolerant, follow moral and biblical principles of honesty, integrity etc. He has chosen to live by what he is taught. The struggle that I see each day is food for thought. Looking at the world around, these values are passé- this teen sticks around like a sore thumb because he is not like the rest of his peers - aggressive, pushy, vulgar, defiant and lack moral values, to name a few. His friends don't accept him though they know that when they are in trouble they can find help here. He dares to be a loner because of his choice.
All around him, the media is not so subtly preaching that it is okay to do anything that makes you feel good. You are cool only if you have 'things' is what is shouted from housetops. The shows on television for this age group say all is fair in teenage - be it bullying, premarital sex, aggressive / violent behaviour, breaking rules. What is the teenager learning? What is a teen to choose? Who is to blame? Is it their fault if they become are confused teenagers , defiant teenagers or out of control teenagers?
Problems Faced by Teenagers
Scheduling work and play
Teenage stress increases with school work and as they move higher, a number of projects to be done, tests to cope up with, professional choices to be focused on, drama, literary forums, sport and other extra curricular activities from the school front add to the pressure. On the social front, a new and tantalizing horizon opens up- dating, partying, hanging out with friends, vying for attention with the distraction of the media, sports, music, internet relationships etc. The teenager is under much duress to schedule time, prioritize work and achieve goals. Effective learning skills could be of great help to these teenagers.
High parental expectation Parenting teenagers is like walking on thin ice. Parents need to strike the right balance. Often times parents live their dream through their children this is when the pressure builds. Expecting the teenager to excel in academics, get good grades, be the child of their expectations – well behaved, responsible for themselves and sometimes for their younger siblings and bring in accolades from extra curricular activities - is in itself enough pressure.
Added to this is peer pressure. Teens, to be accepted among peers, feel the need to become more popular through other channels like music, sport and if need be by bulling and getting into school gangs… and are often under great stress to become acceptable and popular. Amid such hectic activity, the seeds of restlessness, anxiety, fear of failure, unbearable stress is all planted. The most immature and easy way out could lead them to other problem areas like formation of unacceptable habits, teenage depression, substance abuse, getting into trouble with the law etc.
Social and parental pressure
Every significant other adult around the teen is attempting to mould him / her into some acceptable shape. Parents, teachers, elders in the family, social groups are all party to this. The teen is hardly ever able to do what he / she wants to do with his / her life. Even though people do not push their views down their throats, they are still under a lot of strain because of the variety of suggestions given and the fear of failure. Frequently, teens arrive at decisions which conflict with their own skills or fields of interest.
The teen is suddenly expected to act like an adult, with good social and self- help skills in areas like managing work independently, decision making, managing their finances well, being responsible, etc. Though every parent complains about filthy bedrooms, untidy shelves and cabinets, smelly socks, missing stuff, I wonder where is the time! Their inabilty to handle pressure make them defiant in the face of expectations, more prone to anxiety, depression etc.
Problems of teenagers - Health and wellness
Teen health The eating habits of teens are poor and unmonitored. The girls are forced towards an anorexic look (to be popular) and the boys stuff themselves with unhealthy, fatty, high calorie junk food. Teen health is at high risk and its no wonder that this generation now is the victim of every health problem in the book. Wonder of wonders, they survive it all!!
Relaxation and rest. The most important and yet the most ignored item on the teen's agenda. In an effort to do it all and be it all, the teen sacrifices highly on rest. It is a known fact that teens require about nine hours of sleep but at an average, teens get roughly about seven plus hours of sleep. This is one of the main reasons for performance being negatively affected.
Emotional health. Hormonal changes in the body add to the chaos of strain of scheduling, prioritizing, achieving, fulfilling parental expectations and conforming to peers' expectation. The teen is unable to effectively handle all this is under considerable emotional pressure. This is reflected in mood swings, aggression, depression and sometimes even a complete break down of the individual. Here is where the skills of parent in parenting teenagers comes into play.
Searching for role models - who are their heroes?
In our schools we find the biggest bully, the richest spoilt brat lording it over all, the ones who break all rules (considered daring) are the most popular.
The media glorifies the people, especially in filmdom, sports, music etc and focuses on the negative aspects of their lives. Our children are fed on a diet that television programs and internet churn out, and are imbibing the values of their heroes as they see them portrayed in the media.
Helping teenagers deal with the problems they face
What a paradox the life of a teenager is today!! The teens today are forced to live on the edge, at a very superficial level with no acceptance and very little positive affirmation from parents, peer pressure, learning to live from the internet and television with no emphasis on moral value or excellence.
Parenting teenagers
What we need is a paradigm shift and a clearer view of our parenting skills.
We need to look at ways to help teens, our parenting skills, stop complaining and take responsibility for what we are exposing our teens to right now.
We cannot afford to ignore the parental duties of bringing them up well from the time they are little. Televisions, computers and other gadgets that they spend most of the time with, cannot give them the nurturing that only parents can.
Provide a healthy and complete meal whenever you pack it or put it on the table. Make vegetables and fruits interesting.
Build lines of communication that are so strong that your children always look at you as allies and not as enemies. Communicate positively and avoid commands and 'I told you so' lines. Establish rules for teen driving, discuss and set limits on teen datingetc. and ensure that they conform to those rules. This will tackle the problem of the out of control teenager.
Let them know that you don't always have all the answers and you too are not always right. Take their opinion and help when ever needed. I always take my son's input on what concerns him.
Be graceful enough to thank and appreciate them at every possible occasion.
Punishing teenagers does not really work. Disciplining teenagers does.
Be willing to listen to their point of view. Give it a good thought before you trash it. An encouraging and supporting attitude goes a long way.
Fear of failure is one of the greatest reason for stress, help them build self esteem, techniques to program the mind, self affirmation etc could be taught to overcome this.
Teaching techniques of goal achievement, helps them break down goals into smaller, achievable modules and will take much of the stress out of their life .
Look out for sign of stress, anxiety, lack of concentration, poor food intake, poor oral and personal hygiene, disturbance in sleep, plummeting of interest in social activities, address them immediately. It could be teenage depression, get professional help.
Most importantly, let them know that you care!
Wishing you all success!!
I would love to hear from you ....
The biggest problem i face is my shortness, any ways to get taller???
Very good hub on this topic. You nailed it with this hub. Good for you. Thumbs up..
Wow. This is good! As a teen I face this all the time!
Thank you so much! I would love to know about anything you have based on axiety and fears in teenagers. I am one facing this, and was just wondering if you have anything that goes along with that.
This article is spot on. Good job.
adolescence is the most complex stage of life of a person. so we really need guidance and love at this stage, and the opportunity to build our own identities because we are made to be unique from each other. we do not need to imitate our idols we just need to know who we are and just be confident with ourselves. :)
hey yall
Voting this up.
Great topic and great information, glad I found you on here, I'm looking froward to reading more:]
Great and informative hub.Great work for teenagers.
Having Asian parents but living in North America is one of the toughest things you could ever face. Asian on the outside, Western on the inside. But still expected to meet those Asian standards your parents put up for you, and not being accepted into society because of your looks.


text:
login with
feeling guilty
ok so i had my wisdom tooth pulled yesterday and my sister came over to help me with the girls and sense i was right there i felt real bad not taking care of my own kids. my other sister is always saying to just let her know when i need a break cause she would love to have the girls over and man do i need a break sometimes but i feel like the worst mom ever by asking her to take my kids cause i need time away from them. i mean i dont have like separation anxiety when they do go away im fine but the whole asking someone to take them makes me feel horrible.
anyone else ever feel that way?
I have three kids and my cousin and my friend are always saying the same thing to call them if I need help. I feel bad because I know that my kids are a lot for me sometimes and these ladies have kids of their own also. Don't take for granted that you have a kick ass support system right there.
Moms also like
26How often should you bath your newborn
7When to start a routine with baby
14Diaper rash creams that really work
About Sarah Weems
Current: Smyrna, Delaware
Birth: August 31
On Moms.com since: May 18, 2014
I am a 23 year old stay at home mommy. I love to cook, bake, arts and craft projects, and any outdoor projects. I'm in the process of trying to get back into college and would love to own my own bakery one day.


text: Just when we thought Japanese technology couldn't get any freakier (Love Pillows, anyone?) — now comes a new wacky perversion. Atami, a resort town located just outside of Tokyo, now caters to men with virtual girlfriends — guys who play a Sims-like virtual relationship game called LovePlus. The game features a teenage boy who works… READ MORE »


text:
Cable Operators
PAY TV IN 2020: $200 PER MONTH?4/16/2012 12:01 AM Eastern
Millions of Americans may
get priced out of cable TV
within a decade, if fees for
sports networks and other
programming keep rising
at 6% annually. By 2020,
the average subscription-
TV bill in the U.S. will
jump to $196 — more
than double the $86 average
last year, according to
research firm NPD Group.
A big caveat: NPD's totals
include premium services.
But even $110 a month
for baseline pay TV would
likely drive many consumers
to cheaper Internet
video services.
Want to read more stories like this?
Get our Free Newsletter Here!
Get our Free Newsletter Here!


text:
Hugo Boss is one of the market leaders in the premium and luxury fashion world offering an attractive collection of wrist watches for men and women. Watches from Hugo Boss are defined by their stylish casings, well-designed faces and comfortable bands.
Hugo Boss HB1512631
Wholesale Price
Do you want to know the wholesale price? Registered Users benefit from discounts up to 90% off, register now.
Retail price: €. 299
- quartzchronograph with date
- Leather band , stainless steel case, mineral glass
- Measures: case 44x11mm, band 20cm x 22mm
Prices VAT excluded
Out of stock


text:
Liberty Safe - Centurion 22
2 Attachment(s)
Today I became a "responsible" gun owner. They delivered the Centurion 22 I bought from Gander Mountain. I've given that store too much money over the past month.
It was on sale for $550. I'm very pleased with that deal. Access point at bottom right corner for an in-safe heater/dehumidifier. 30min fire rating. Lot's of room for my current four long-guns, the one or two more long-guns I plan to buy in the near future, and plenty more handguns than what I have right now.
I recently got a gun safe with the fire rating and all from Gander Mountain. It looks similar to yours. I noticed the other night, that there are four holes in the bottom of the safe, visible if you lift up on the interior panel with the pull tab. It is not going to be easy to fix that, but i think i will have to do something about it or just bolt it to the concrete under the tile floor.
Is you safe holey too?
No plans on doing anything with the holes. I think you'll find them on just about every safe out there.
I saw them on sale and was thinking about getting one. What are your impressions of it?
I definately need a safe and I think this one would work for me for the next few years. It has to fire resistant though.
I think you would want to bolt it to the ground whether you have kids that could knock it over or not.
Bolting the safe to the ground makes a safe harder to break in to.
Not sure the fire rating would be accurate without it being bolted down.
Ya think the engineer that designed the safe expects the owner to bolt it down? The safes floor is its achilles heel. If its designed to be bolted down they will cut costs by using a thinner floor.
Any firefighter peeps with input on the holey issue?
edit* What cane said, but with a question mark.
Scott, did you get the dehumidifier rod?
As for bolting it down. It was enough to get my wife to agree to this in the house. I'm 95% in a better situation with the safe than without, I'm not going to push her to mess up her carpet and floor for the other 5%. That's for the next house with a dedicated work room.
Scott, did you consider the door storage option? The one i ended up with has some pistol holsters/pockets and other mag pockets and such on a thing that mounts to the inside of the door. It saves space on the handgun storage.
|All times are GMT. The time now is 08:47 PM.
Copyright ©2000 - 2017, Jelsoft Enterprises Ltd.


text:
Welcome to the ineedhits Search Engine Marketing blog, where we share the latest search engine and online marketing news, releases, industry trends and great DIY tips and advice.
Last week Rene posted about Google's new homepage design, along with some of the other changes that didn't make the final cut. With millions of people worldwide using the new design since last week, the revamp appears to have been a success with little or no complaints from users so far.
With such a drastic change to the worlds most popular search engine, I was expecting to see a bit more controversy over the design. As I pointed out in a comment on last weeks post, there are a couple of elements that still bug me about the new design.
They aren't major complaints, and I certainly won't be switching search engines over it but I'd love to hear your thoughts. Let us know in the comments below what you think of the new design.
P.S. If you want the classic Google look back, bookmark the following URL http://www.google.com/webhp?hl=all You can also switch back to the new look here.
Agree with all your comments.
It is just so cluttered now, and actually makes it harder to focus on the actual results.
I very much doubt whether 98% of searches will use these additional features. It would have been better to place the new options on the "advanced search" page IMO.
By Clay Cook - May 12, 2010
Thanks for your perspective. I can tolerate most of the changes except for the new styled input box. To me it doesn't have proper input box "feel" and it is REALLY irritating. I've switched my browser home page to Bing.
By Ben S - May 18, 2010
Just want to say what a great blog you got here!
I've been around for quite a lot of time, but finally decided to show my appreciation of your work!
Thumbs up, and keep it going!
Christian, iwspo.net
By ONENSELOX - May 22, 2010
Just want to say what a great blog you got here!
I've been around for quite a lot of time, but finally decided to show my appreciation of your work!
Thumbs up, and keep it going!
Christian,Diet Guide!
By ONENSELOX - May 31, 2010
I do like the various designs when I log into google, my favorite so far was the packman!!!!!!!!! Keep up the great creativity! What is the meaning of the glowing green cube…looks like Krypton…lol! Fun to see your handy work when I log in. All the best with your blog also.
By Tanya Gonzalez - June 5, 2010
Thank you. Not bad website you got going on here. Got some extra sites to direct to which have more info?
By do it - November 24, 2010
Google totally nailed it. Love it.
By Dave - July 1, 2011
Keep up to date with the latest from our blogs.
New Yahoo Ad Campaign Takes Swipe at Google's Home Page
Yahoo is about to launch an $85m advertising campaign which they hope will increase traffic to their home page by ...


text:
HAMMOND — The Southeastern Louisiana University men's basketball team plays four of its next five games at home, beginning Saturday when it hosts arch-rival Nicholls State in a Southland Conference contest at the University Center.
Tipoff is at 6 p.m, and the game can be heard in the Hammond area on KSLU-FM (90.9) and on the internet at www.LionSports.net.
Southeastern (7-9, 1-3 SLC), which is 7-0 at home this season and will host league rivals Central Arkansas (Jan. 25), Lamar (Feb. 1) and Sam Houston State (Feb. 4), had its two-game winning streak snapped on Wednesday in a 61-47 loss at McNeese State. The Lions fell behind by 21 points in the first half.
Jeremy Campbell scored 11 points to lead Southeastern in the contest with Daron Populist adding 10. Roosevelt Johnson grabbed a game-high 10 rebounds for the Lions, who held McNeese State to 8-for-29 (28 percent) from the floor in the second half.
Johnson leads the Lions in scoring (10.1), rebounds (7.6) and blocks (0.9) while Elgin Bailey (8.1 ppg.), Populist (8.0), DeShawn Patterson (6.5) and Campbell (5.6) are next in the scoring column.
Nicholls State (4-13, 0-5 SLC) has dropped its last six games, beginning on Dec. 19 at North Carolina. The Colonels, whose nonconference schedule also included LSU, Tulane, Auburn and Texas, dropped a 99-76 decision at UCA on Wednesday.
Trevon Lewis leads the Colonels in scoring (14.1) and rebounds (6.1) with Chris Talkington averaging 10.4 points. Bryan Hammond (9.4), Shane Rillieux (7.5), Dantrell Thomas (7.4) and Sam McBeath (6.9) are part of an offense that averages 66.2 points and 6.9 3-pointers per game.
Southeastern leads the all-time series between the schools, 47-39, including a 50-43 victory in last season's regular-season finale at the University Center.