text:
|Publication number
|Publication type
|Application number
|Publication date
|Jul 4, 1989
|Filing date
|Sep 28, 1987
|Priority date
|Sep 27, 1986
|Fee status
|Also published as
|Publication number
|Ronald D. Clark
|Original Assignee
|Clark Ronald D
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (3), Referenced by (89), Classifications (5), Legal Events (6)
|External Links: USPTO, USPTO Assignment, Espacenet
This invention relates to surgical diathermy apparatus using indifference plates of the capacitance coupling type.
1. Surgical Diathermy and the Indifference Plate
Surgical diathermy is an electrical method of producing cuts in human tissue. This is done by striking an electric arc by a stylus which completes an electric circuit through the human body. A typical frequency of operation is 600 kHz and the range of power outputs from a diathermy set varies from a few Watts to about 400 W. A diagrammatic view of a typical arrangement is shown in FIG. 1, where:
Z1 represents the contact impedance between the active electrode and the patient, say 400 ohm, mainly resistive;
Z2 represents the impedance of the current path within the patient's body, typically 20 ohm, mainly resistive;
Z3 represents the contact impedance between the plate electrode and the patient, typically 2 ohm, mainly resistive; and
Z4 represents the impedance of the plate cable, typically 30 ohm, mainly inductive.
The return path of the electric current is conveyed from the human body to the diathermy machine by a metal plate of sufficient area that the current density is harmlessly low. This is called the "neutral plate", "indifference plate" or simply "plate electrode".
The indifference plate has undergone some development. Initial types were of lead sheet bound to the patient by dressings and with saline solution added to give good electrical conductivity. Modern types are of foil, the more recent types having an adhesive conductive film which gives good electrical contact with the patient and high security of fixing. The claimed contact impedance of one example of this adhesive type is 15 ohms.
2. The Capacitive Type of Indifference Plate
An alternative approach, which also claims an improved current distribution over former types is the capacitive type. This also is a foil type with an adhesive layer. This adhesive, however, is not conductive but an insulator, so that the return circuit is made through a capacitive reactance. A measured value of the reactance of a commercial plate of this type when attached to a stiff aluminium plate was found to be around 30 ohms. This is twice the contact impedance of the above-mentioned conductive plate and 15 times that of a typical plate electrode of FIG. 1. Some crude measurements of the impedance of a capacitive type plate when on human flesh yielded a value of about 80 ohms.
The advantage of this type of plate is that, when the plate is pulled from the skin, as might accidently but rarely happen, then the reactive impedance is so sized that as the plate area diminishes, the impedance rises quickly to a value that limits the total current and hence the total power and power density delivered. This effect is shown by the curves in FIGS. 2 and 3 where both the total power and the power density (i.e. power dissipated per unit area of skin) are held to moderate levels in the capacitive type of neutral plate for the maximum power settings on a diathermy machine. Although in this particular case `safe` limits as shown arbitrarily in these graphs are exceeded, it is evident that the possibility of a severe burn is very much reduced. Indeed a redesign, by for instance thickening the dielectric of the neutral plate, would easily permit `safe` limits to be held.
Note that a power density of 300 mW/sq cm is about the value at the surface of a 25 cm 60 W incandescent strip light and appears quite bearable to the touch, whereas the power density of 600 mW/sq cm at the surface of a 60 W pearl lamp is distinctly unbearable. Of course there are several complicating factors in the above picture, not least being that the surgeon applies power in an intermittent fashion, which would permit somewhat higher short term power densities. The overall conclusion is, however, that the capacitive type of plate is a fail-safe device in the type of accident where the plate is pulled off.
The other advantage claimed for capacitive type plates is similar to the last-mentioned one, but concerns normal use. In this case it is evident that the plate or adhesive does not actually have to touch the skin to achieve a capacitive value. A much more even distribution of current is therefore likely than from the conductive type of plate, which relies on special preparation to achieve good plate skin contact resistance. Even with the modern self-adhesive conductive types it is possible that current density is higher at the edges than directly under the plate where it might not be so firmly pressed down. In short, hot spots are inherently avoided.
The capacitive type neutral plate has, however, the following disadvantages:
a. Because of the somewhat higher impedance of the capacitive plate as presently manufactured, it does mean that for earth-referenced diathermy machines the patient (FIG. 4) can be at a significant potential, Vf, with reference to earth and any inadvertent contact of the patient to earthed metal could allow current to flow and an RF burn to be sustained (FIG. 4A). Because of the low contact resistance, Z3, of the resistive neutral plate, this accident is much less likely to produce a damaging current. For FIG. 4:
Z5 represents impedance of alternative internal current path;
Z6 represents contact impedance of patient to metal object;
Z7 represents impedance to earth of metal object; and
Z8 represents the overall impedance (mainly capacitive) of the patient to earth.
b. Again because of the high impedance of the capacitive neutral plate any shunt path can carry a current of damaging proportions. Thus if there is a point failure in the insulation, or if an uninsulated tab were to contact the skin then a large proportion of the current could be carried through this failure at very high power density. This path is represented by any impedance in parallel with Z3.
The possibility of this type of event occurring has led to this type of neutral plate not being used except for applications below 50 W.
It is an object of the present invention to provide surgical diathermy apparatus which obviates or mitigates the abovedescribed disadvantages. It is also an object of the invention to provide suitable forms of electrodes and control systems for use with such apparatus.
According to the present invention there is provided surgical diathermy apparatus comprising a power source, an active electrode for operation on a patient, a capacitive neutral plate for attachment to the patient, and a potential compensator for providing a compensating potential in series opposition to the potential drop across said capacitive neutral plate.
Preferably the apparatus includes a compensation potential level adjuster for adjusting the level of the compensation potential provided by the potential compensator.
Preferably the apparatus also includes an open circuit fault condition detector for detecting open circuit fault conditions of the capacitive neutral plate.
Preferably the apparatus also includes a short circuit fault condition detector for detecting short circuit fault conditions of the capacitive neutral plate.
Most preferably the apparatus includes an alarm system associated with said fault condition detectors and including deactivation means for deactivating the diathermy apparatus. Preferably the apparatus also includes two or more compensated capacitive neutral plates or plate elements which are included in the circuit such as to provide an automatic reduction of current in a faulty plate.
Preferably the apparatus also includes an array of neutral plates or a multi-element capacitive neutral plate such that the statistical probability of a significant fault current occurring is minimised.
Preferably the apparatus also includes an automatic control system for automatic control of the apparatus. The automatic control system is preferably based upon a microprocessor.
Embodiments of the present invention will now be described, by way of example, with reference to the accompanying drawings, in which:
FIG. 1 is a schematic diagram illustrating a typical layout of conventional diathermy apparatus;
FIG. 2 is a graph illustraing the power dissipation relative to skin contact area for conventional conductive and capacitive type neutral plates;
FIG. 3 is a graph illustrating the skin power density in relation to skin contact area for conventional conductive and capacitive type neutral plates;
FIG. 4A is a schematic diagram illustrating a potential fault condition of a capacitive neutral plate circuit;
FIG. 4B is an equivalent circuit diagram of the schematic of FIG. 4A;
FIG. 5 is a circuit diagram of one embodiment of surgical diathermy apparatus in accordance with the present invention;
FIG. 6 is a circuit diagram of a further embodiment of surgical diathermy apparatus;
FIG. 7 is a circuit diagram of surgical diathermy apparatus of the present invention incorporating an alarm circuit;
FIGS. 8A and 8B illustrate surgical diathermy apparatus of the present invention incorporating a plate continuity alarm;
FIG. 9 is a circuit diagram of dual capacitive neutral plate diathermy apparatus of the present invention;
FIG. 10 is a circuit diagram of dual capacitive neutral plate diathermy apparatus incorporating a fault detecting circuit;
FIG. 11 is a circuit diagram of multiple element capacitance neutral plate surgical diathermy apparatus;
FIG. 12 illustrates one embodiment of a multiple element capacitance neutral plate;
FIG. 13 illustrates a second embodiment of a multiple element capacitance neutral plate;
FIG. 14 illustrates a third embodiment of a multiple element capacitance neutral plate;
FIG. 15 is a schematic diagram of microprocessor controlled surgical diathermy apparatus in accordance with the present invention;
FIGS. 16, 17 and 18 illustrate three further variations of the invention;
FIG. 19 illustrates the equivalent circuit of a multi-element capacitive neutral plate under fault conditions; and
FIG. 20 is a graph illustrating power dissipation in a partial fault at maximum generator current.
(Note:- FIGS. 1-4 illustrate the background to the invention and have already been described in the "Background of the Invention" section above).
Details (Section 1):
This section describes improvements such that the above disadvantages of the capacitive type indifference plate are overcome. The basic improvement is to arrange that voltage is injected in series with the capacitance neutral plate of such a value that it equals or nearly equals the potential drop across the capacitive reactance. FIG. 5 shows the basic arrangement and the following equations describe the effect: ##EQU1## where w is the angular (radian) frequency of the diathermy set, VAB is the potential drop between points A and B, r residual resistance, C capacitance and p the subscript denoting that these quantities refer to the plate circuit. Also V2 is the injected voltage which is derived from the amplifier (or other electronic power source slaved to the generator) which device not only provides gain G but also provides 90� phase shift with respect to the current, I, and current to voltage conversion. T1 is a current sensing transformer with a step down turns ratio of q1 so that the amplifier delivers a signal V1 as follows: ##EQU2## The current in the secondary of the transformer T1, namely I/q1 is set by q1 so that any reference resistor in the amplifier need only be a low wattage type, but at the same time high enough to give a sufficient signal at all times. Similarly transformer T2, a voltage step down type of ratio q2 in relation to the amplifier output signal, V1, affords protection to the amplified output by reducing backward current injection from the main circuit into the output of the amplifier. The injected voltage V2 is given by: ##EQU3## Since V2 is arranged to be equal or nearly equal in magnitude to I/jwCp, compensation of the neutral plate capacitance, Cp, takes place such that for a given value of I, VAB is minimised.
Disadvantage b. (background section above) is now removed or nearly so since according to equation (3.3) the worst case potential of the patient with respect to earth is Imax. rp where Imax is the maximum diathermy current and since this is usually around 1 Ampere, the worst case potential will be quite small.
If we assume that in practice the injected voltage V2 has a 10% mismatch with a capactive plate potential drop of 80 volts, then the value of VAB is about 8 volts. If the patient makes contact with an earth path with a contact area of 10% of that of the neutral plate, i.e. 20 sq cm and ten times the postulated contact resistance, or 20 ohm, then the current through the patient due to the earth path would be 0.4A and the power density would be about 160 mW/sq cm which appears to be very safe according to the previously adopted limit of 300 mW/sq cm. So, on a conservative estimate the improved circuit renders the earth-referred diathermy safe. A more thorough analysis is presented in the Appendix below.
However there is also concern with the case where the plate incurs either a failure in use or a failure in manufacture such that the insulating material breaks down at a point such as to have a low value resistive path. In effect the capacitance of the neutral plate is shorted out. In the conductive type plate this would mean high current and power densities at the point failure with a high chance of a burn to the patient. With the invented circuit however, it can be arranged so that this does not happen. For if the capacitance is shorted out the cancelling effect is removed and the injected potential, V2, becomes VAB, the potential developed across points A and B as shown in FIG. 5. This effect can be used in different ways as described below.
This arrangement is more elaborate than the inductive compensation circuit (FIG. 6) but it affords easier and more reliable means of automatic control as described below.
The required injection voltage could be derived by different means--e.g. from the output voltage of the diathermy unit--and via different coupling methods--e.g. optical or capacitance--but the general principle of achieving an injection voltage proportional to the current, is the substance of the idea here presented.
Also, a mixture of the two techniques, i.e. the voltage injection method of compensation as described herein, and the inductive compensation method (as shown in FIG. 6) can be employed. By this method the bulk of the neutral plate capacitance can be compensated for by a series inductance and the remainder by voltage injection also in series. Although this method would permit relaxation of the design of the amplifier since the power handling would be reduced, it is not discussed further for simplicity of presentation. The general principles discussed herein would be equally applicable.
The more severe restraint where we have only a partial short is described in the Appendix. Here it is useful to note that the form of equation (3.3) above and the impedance terms therein are similar in form. In particular an equivalent inductance can be defined: ##EQU4## This allows the earlier equations to apply with equal validity in this case. The proviso is made that since, although gain G may be made frequency dependent, it does not in this application need to be made so, since operation is almost invariably at a fixed value of w, say wo. Particular reference is made to the equations in the Appendix below, because it may be taken that all the inductances therein represent compensating injection voltages.
Details (Section 1.1):
The arrangement for this version is shown in FIG. 7. This shows a suggested arrangement of a diathermy unit connected through a patient via an alarm box which is specifically designed to detect a fault in the capacitance plate and disconnect the diathermy from the patient. It is here assumed that the diathermy is of a type which has no arrangement for detecting plate continuity--i.e. for detecting whether the conventional type neutral plate is not sufficiently adhering. In addition to the usual active and neutral leads a reference lead is required in order to establish the reference potential of the alarm box detection circuits at a value near to that of the patient's body. Thus the potential presented to the signal processing section of the alarm box is that of the injected potential in series with the potential drop across Cp.
The reference lead thus has to be attached to the patient by means of a small adhesive and electrically conductive pad. This could be either integral with the neutral plate or be a separate pad similar in type to those used in e.c.g. measurements, but placed close so that it takes up a correct potential.
To operate this diathermy system the surgeon or assistant would first of all take the active lead fitted with a dummy probe and touch it on a test pad. He would operate his foot switch and set the diathermy control to obtain a suitable power level. By operation of push buttos or continuous adjustment the amplifier gain is varied so that the level indicator now shows a maximum. In this way the capacitive reactance, Cp, of the neutral plate is matched. Note that the current path is:- R2, patient, Cp, neutral lead, the secondary of T2 --and is limited to a safe value by R2. Contacts A1 and A2 are normally closed.
The surgeon now proceeds in his usual way. The current then flows--active lead, patient, Cp, neutral lead, secondary of T2. As already described, if Cp experiences a fault and the current is above 0.3A, then a threshold potential of V2 of say 24 v, would be exceeded. The signal processing electronics would detect this event, alarm lights or sounders would be activated, the level indicator would indicate high or danger and the relay would be activated to open contacts A1, A2 and disconnect the supply to the diathermy unit.
To guard against the event of the reference lead being pulled off its conductive pad, a resistor R1, at a value sufficiently higher than the reactance of Cp (say 400 ohm) not to draw significant current in normal operation, but low in relation to the input of the signal processing electronics, is placed across the neutral and reference leads as shown in FIG. 7. If the reference lead or pad becomes disconnected, only the potential V2 would be detected and the alarms and relay would be operated as above. Alarm circuits would latch on and have to be reset after appropriate corrective action.
Neither details of the signal processing nor power supply are shown as these could be conventional. Suffice it that a high level of isolation from mains power is required. The box could be a separate unit or a retrofit to an existing diathermy unit. British Standard BS5724 has to be observed. Values shown are indicative and would be subject to alteration as circumstances require.
Suggested variations of the single neutral plate system are shown in FIG. 8, which shows arrangements suitable for diathermies which have an intrinsic alarm for plate continuity. This type of alarm uses a two element neutral plate and operates when the resistance between the two elements and through the patient rises above a predetermined value. In Eschmann machines this is detected when the current, from an auxiliary oscillator operating at 30 kHz and in a separate neutral plate loop, falls below a preset threshold.
This type of alarm may be adapted as in FIG. 8 to detect an increase in the impedance of the loop. In FIG. 8A the current IA is injected into the above mentioned loop operating at a radian frequency of wo. The compensating amplifier, compensating transformer T2 and sensing transformer T1 are contained in an auxiliary box with its own set of leads. The auxiliary current is given, by analogy with equation (3.3), by:
IA =VA /[rp +j (G/q1 q2 -1/Wo Cp)](3.7)
The amplifier gain could be set by similar means to those described above and in FIG. 7. If perfect balance were to be achieved equation (3.7) would become:
IA =VA /rp (3.8)
This would be a rather large current and would simulate the low resistance condition of the diathermy set's loop integrity circuit. However if the neutral plate were to be flawed--in the extreme shorted--then equation (3.7) would approach:
IA =VA /(jG/q1 q2) (3 9)
Since wo now relates to a frequency of 30 kHz, the neutral plate reactance would now be between 600 and 2000 ohms and therefore so also would be the synthetic reactance in the denominator of equation (3.9). IA would drop to a low value, thus simulating the open plate condition of the diathermy and so the diathermy current I would be shut off. FIG. 8B shows the arrangement of FIG. 7 again, with the exception that the relay acts on the contact A which is in the plate continuity detection circuitry.
These more elaborate circuits could become most attactive for fixed installations which already lack an alarm feature i.e. the plate potential alarm. Retrofitting of one of the above mentioned circuits could be most economically carried out if this alarm feature was also installed at the same time.
Another advantage of the capacitance plate already mentioned is that it can also be designed such that if it is peeled off by some error, then the impedance (capacitive reactance) of the plate rises to limit the current to a safe value. This value probably depends on a threshold of power (from a near-point scource) which can be dissipated harmlessly into the mass of surrounding flesh. Considerations of available point sources such as cautery burners and small lamps seem to suggest a threshold of about 1 Watt before pain ensues, although one make of diathermy specifies a tolerance of 7 W presumably inclusive of zero setting. It is also hard to envisage that the capacitive plate would form a point source. Rather there would be small area (of about say a quarter of a sq cm) below which the plate would fall off completely.
The design of the plate would accord with the following considerations. Assume Z3 in FIG. 4 is made up of a `contact` resistance, rc, and the capacitive reactance Xc. Assume rc is 2 ohm as in FIG. 4, then for a fault condition where only 0.25 sq cm of the plate is adhering and the whole plate has an area of 180 sq cm. Then rfc, the fault contact resistance would be:
(180/0.25)�2=1440 ohm
For a fault current, if we wish:
if 2 rfc <1 Watt (3.10)
if <26.4 mA
If Xcf, the fault capacitive reactance, is designed to be the current limiting impedance from a generator e.m.f. of 800 v, then we obtain: ##EQU5## The design reactance is given by the ratio of the areas already assumed, fringing effects being as elsewhere in this specification ignored: ##EQU6## On this reasoning, the measured reactance of about 80 ohm is more than adequate to limit current to a safe value in the "tear off" situation. If other starting values are shown to be more valid, these can be substituted in the above equations to obtain other design values. Confirmation of these assumptions could be obtained by experimental work.
The current could also be limited for the second type of fault--i.e. a capacitive neutral plate short, since the compensation injection voltage (or inductance) would also have to be high. So, if we were to abandon the present commercial design and were to accept an alteration in the plate physical design then the simple circuits of FIGS. 5 and 6 could form the basis of an intrinsically safe circuit against the two faults discussed. The condition that requires to be satisfied is that, if the plate is shorted out, the current is limited by the compensating voltage V2, which by equation (3.6) is equivalent to IwLp, where Lp is an equivalent inductance.
If we assume that the diathermy excitation voltage is 800 v, that a safe maximum current is 0.25A and that both diathermy source impedance and Z2 are 400 ohm resistive, then the maximum equivalent inductive reactance is given by:
(2�400)2 +(wL)2 =(800/0.25)2 =32002 ##EQU7## The patient is now exposed to a third hazard, since he would now be at a high potential and in danger of drawing curent through the sneak path (see FIG. 4) by chance contact through an earthed metal object while also being attached to an earth referred diathermy. Utilisation of disconnect/alarm circuits such as those of FIGS. 7 and 8 would solve this problem for single plate sets. For the multi-element or multi-plate circuits described below, intrinsic safety is approached for this hazard since the good compensated branches strap the patient to a small potential with respect to the low voltage side of the machine. Nevertheless the high potentials indicated would impose high demands on the specification of equipment and compromises might have to be made.
To resume the design of the plate, if we take 80 ohm as the reactance of the present plate, which has an adhesive film of 0.004 inch (or 0.1 mm) then the design thickness of the new plate would be approximately:
0.1�(3200/80)=4mm or about 0.16 inch
This is a considerably different type of plate since it would be thicker yet still be flexible to fit different body shapes. However possibilities are opened up for different types of plate of strip or spiral type. Not only thickness but area and shape could be altered. FIGS. 12, 13 and 14 show some general concepts for these designs.
An advantage of the capacitive type of plate is that, since good electrical conductivity is not required no shaving or scrubbing of the skin need be required. A good reactive value is obtained so long as good proximity with the skin is obtained.
The main disadvantage of these plate systems is the requirement for the setting up procedure already mentioned above. The surgeon or assistant would adjust this while using a flat probe and at a low setting (see FIG. 7) so that a peak in current was obtained. A faulty plate would show up as a low curent unable to be peaked. It could then be changed. During surgery any fault current across the neutral plate would be limited as mentioned by V2. However as discussed later control techniques could eliminate this disadvantage.
Details (Section 1.2):
The disadvantages of the conventional type neutral plates as described in the last section are:
(a) they require an additional reference lead; and
(b) special circuitry is required to disconnect power in the event of a fault.
A development of the basic circuit, which could permit the utilisation of present commercial plates and also in some measure be intrinsically safe, is shown in FIG. 9. Here a box containing two amplifier/transformers is placed in the neutral plate circuit. The main current is split into two parallel paths (i1 and i2) which flow through the transformer secondaries and neutral plates. In each path the current is sensed as already described above so that the equivalent inductances representing the injected voltages, V2A, V2B (see equations 3.5, 3.6 earlier) balance the negative reactance of the neutral plate.
This condition is described by the following equations: ##EQU8## where r1 and r2 are the values of the winding resistances of each inductor. At maximum diathermy current, if r1 and r2 are about equal, then:
i1 =i2 =0.5A
However a wide tolerance on resistance and thus current is acceptable.
Under the condition, say, that neutral plate 2 shorts out, then jX2 becomes zero and the equation now becomes: ##EQU9## i.e.
i2 =i1 r1 /jwL2 (3.15)
If we suppose r1 is about 4 ohm and jwL2 about 80 ohm, then i2 is about a twentieth of i1, or less then 50 mA for full power where i is 1 Amp. Note that V would be 1�4=4 v and V2B would be 80�0.05 also 4 v and the patient thus be at a very low potential with respect to the generator low terminal. Similar results would be obtained with the high reactance plate of the last section and so the patient would be safe from this hazard. A similar result is found if the other neutral plate shorts. That is, the bulk of the current is forced down the good neutral plate, while the current of the damaged plate is held at a safe level.
This is an important feature of this and the other compensated multi-plate circuits and is the basis of their inherent safety.
Since the expected average incidence of flawed plates is not known a method is suggested to take care of the simultaneous appearance of two flawed neutral plates. The circuitry of FIG. 9 would not now be sufficient since at least one of the fault plates could now be carrying 0.5A or more at full power.
A method of solving this problem is suggested in FIG. 10. A d.c. supply is derived from a transformer TA. LS is of sufficient value that R.F. current does not feed back into the d.c. supply. As the current in the primary of TA, i.e. also in the neutral plate line, is increased, so also is the level of the d.c. supply. In normal operation litle or no current is drawn from this supply, because C1 and C2 block d.c. current in the upper loop of the double plate circuit, while so long as the plates are intact no current is drawn in the lower loop. Even if one of the plates shorts, the other will still block d.c. current.
However, in the rare event that both plates short, a current is drawn, a p.d. is developed across Rs and the disconnect and alarm circuits are triggered. Much detail is omitted for clarity.
Of course to take care of the double fault eventuality it is again necessary to sacrifice simplicity.
Details (Section 1.3):
The advantages of the capacitive neutral plates as seen so far might be listed as follows:
(a) can be designed to be intrinsically safe from the peel-off fault condition;
(b) can be designed to be intrinsically safe against the punch through or short circuit condition;
(c) simpler preparation of the patient's skin, since adhesion not required for good electrical conductance; and
(d) the promise of economies since it may be possible to design a re-usable capacitance neutral plate.
However advantage (b) is conditional on either the feasibility of close tolerance matching of the p d. across the neutral plates and their compensating voltages or the possibility of precise adjustment of that compensating voltage through say the setting of amplifier gain. It has also been assumed that any plate fault is a dead short. In addition there is no data on the incidence of faults in capacitive neutral plates. Where the elaboration of additional circuitry described in the previous sections for either the single or the dual capacitance neutral plate is unnacceptable, the multi-plate capacitance neutral plate promises a simpler approach.
The following calculation illustrates how a multi-element plate reduces the statistical incidence of a fault to negligible proportions. Repeating the format of equation (3.10) in section 3.1, but using a contact resistance rc of say 20 ohm, we might wish to limit the power dissipation to 1 Watt. This gives, where immax is the maximum current in a branch, m: ##EQU10## Since the total delivered maximum current is typically 1 Ampere from a diathermy machine, this implies a minimum of (m=5) good branches to share current; m could be made equal to 1, but this assumes a large area element to achieve a low value of rc, and would leave little safety margin. So, if a failure rate is now postulated at 1 in 20 for the neutral plates (i.e. probability=0.05) then we can stipulate an extra number of sacrificial elements, n, such that the probability of all of them failing at once is extremely low. If n is 5 then this probability is 3 in 10 million. If this were an acceptable risk, then the design number of elements would be 10.
A four-element plate is assumed in the following simple analysis (See FIG. 11), while in the Appendix the case of partial shorts is analysed. This more complex analysis allows for tolerance variation and illustrates the design criteria to be considered.
In the following analysis, a set of power resistors, R1 R2 R3 and R4, all of value 5 ohm, ensure aproximately equal current division. The equations describing normal working are: ##EQU11##
where the equivalent inductive reactances therein are equal in turn to V2A /i1, V2B /i2, V2C /i3, V2D /i4,
and i1 =i2 =i3 =i4 =0.25 A.
Suppose now one plate shorts, say plate 1, then equations (3.17) become: ##EQU12## or to a first approximation, i.e. if jwL1 >>R1 +r1 : ##EQU13##
jwL1 =120 ohm
R2 +r2 =7 ohm
i1 =(7/360)i=20 mA
V˜120�0.02=2.4 v=V2A ˜i2 (R2+r2)
for a diathermy current, i, of 1A, i.e. i1 is very low indeed and shows how most of the current is forced through the good plates. The patient (and also the amplifiers) are also subjected to very low potentials.
We can move up the scale of improbability and postulate that two out of the four plates short, then 3 out of the 4 plates. If we repeat the above analysis we find that the current in the faulty plates is, respectively 30 mA and 60 mA, which are still very low values.
Even if all the plates fail, we get: ##EQU14## which is still moderately safe.
The odds against a combination of types of fault i.e. punch-through and tear-off in such a way that all the current would go down the fault branch must also be very high. Even so, the geometries of the multi-element designs suggested in FIGS. 12, 13 and 14 must increase these odds to astronomical proportions, since they are arranged in such a way that several elements are still in contact with the skin right up to the moment of separation in accidental removal. So it can be concluded that the proposed multi-element design of neutral plate affords an extrememly high measure of patient safety.
Details (Section 2):
In FIG. 12 a disc type plate is illustrated with 60 elements radiating from the centre. One arrangement could be 5 groups of 12 elements. So the lst, 13th, 25th, 37th and 49th would be connected together, the 2nd, 14th, 26th, 38th and 50th together and so on. This design would ensure that several elements are in close contact with the skin in the tear-off type fault. The designs of FIGS. 13 and 14 are multi-element variations to suit other geometric shapes. The design of FIG. 12 would probably be best effected as an adhesive type, but could also be re-usable. The design of FIG. 13 (and of FIG. 14) is such that no adhesive is required. This would permit a re-usable type of plate, such as the cuff type illustrated. The problems would be to design the cuff such that it would be kept tight at all times without impeding the circulation and also to be conformable with all shapes of human body.
The longitudinally segmented plate of FIG. 14 can be realised in at least two forms. The bandage type illustrated is likely to be easily designed to be conformable to the human body by manufacturing it of stretchable material. It would be wrapped round a limb and clipped by means of suitable fasteners. The conformable type would be of extremely resilient material like foam plastic and could even be based on the evacuable bag principle, which allows tailoring of the device to the shape of the particular person. It could be a whole mattress under the patient and as such could include both the safety features of reactive current limitation and multi-element current sharing. These devices would be finished in flexible washable surfaces capable of being disinfected.
Details (Section 3):
The drawback of a pre-operation setting up procedure, mentioned earlier to adjust amplifier gains may be eliminated by an automatically controlled system. For a single plate the arrangement of FIGS. 7 and 8 form the basis of such a control system. To do this the loop is closed such that the signal processing box (which may be integral with the amplifier) adjusts the gain of the amplifier to obtain a minimum at the input of the signal processing box. I.e. V2 becomes automatically adjusted so that the quadrature term in equation (3.3) above becomes zero. V2 remains proportional to the sensed current to keep in step with various settings of the diathermy power. If preset levels are reached within the control unit the diathermy would be disconnected and alarms made. This would occur if the compensation failed because:
(a) the neutral plate shorts, or
(b) the neutral plate peels off, or
(c) the reference lead peels off.
For multi-element or multiple plates there is the extra task of sharing the currents between the various branches. This might be best performed incrementally in an iterative fashion, both to prevent control interactions and interloop resonances. The aim would be to express the equations (3.17) in the real world.
Electronic logic might be used to achieve this overall control, but a more powerful solution would be a mircroprocessor plus an appropriate amount of memory. The microprocessor centre of the control system would accept and deliver signals under program control and by means of algorithms such that the currents in each branch would be incrementaly adjusted to achieve equality and simultaneously to minimise the potential drop between patient and diathermy unit, Vcc, on the neutral plate side. In addition, the presence of some plate punch-through faults or of a peel-of fault would be detected by internal preset limits being reached when the range of adjustment required became excessive. Alarms could be sounded or disconnection made.
The technique would be to sense the currents in each branch by the transformrs Ts1, Ts2, etc up to (in this case) Ts10. Vcc would also be sensed. These signals would be converted to a suitable digital form by the ADCs (analogue to digital convertors) and the digital processing electronics for manipulation by the microprocessor. The reverse process is achieved by further digital processing and DACs (digital to analogue convertors) to drive the transformers T1, T2 etc up to T10 such the reactive balance is achieved in each branch.
The advantages would be an easily operated, ultra-safe and rapidly adapting device giving clear signals to the operating staff. Accommodation of slight movement of the patient would also be an attractive feature.
Details (Section 4):
The background to this section is that many surgical operating theatres have special anti-static floors, the construction of which inhibits the installation of electrical services which come up through the floor. To overcome the consequential difficulties of this inhibition, three further variations of the invention are presented as follows:
(a) the concept of a multi-generator/multi-element neutral plate surgical diathermy apparatus;
(b) that the reference lead can come from a capacitive type pad; and
(c) that the concept of a full body length plate may readily extend the area of application to animals.
(Section 4.1):
FIG. 16 illustrates this arrangement of diathermy machine. In this set-up a separate generator is asociated with each single element of a multi-element capacitance neutral plate. It may be either a diathermy generator in its own right or an amplifier which is slaved from a master oscillator. However it is achieved, the effect would be the same, namely that the total current flow would comprise the sum of all the currents flowing through the neutral plates fromall the generators and would equal in sum the current flowing in the active lead.
One conceptualism of this idea would be to locate these generator elements close to or, by utilising the miniaturisation of electronic components, even to integrate the generator and plate as one element. The remainder of the circuitry, e.g. logic and R.F. (radio frequency) sections could be located elsewhere. This would permit distributed solutions to the problems of implementing surgical diathermy systems in an ergonomic way. A distributed nature of the diathermy system permits installation of the plate alarm/disconnection feature. FIG. 17 illustrates one conception of this idea. Some of the circuitry is located above the operating table, while the remainder is located in a telescopic boom. Power supplies are derived from power units located remotely. In order that the boom is uncluttered by controls, these could be mounted on an infra-red signalling remote control pad similar to these used for TV and video sets. This is an advance on equipments presently employing boom extensions since the R.F. leads (active and neutral) could be kept quite short to minimise stray reactances and hence to permit the inclusion of all alarm circuits including plate voltage. The latter feature permits compaction and simplification of controls and electronic hardware.
Care would be required that the complex waveforms needed for the different diathermy modes were not distorted due to phase or timing differences between component generators. So a master oscillator is required either for the primary generation of the required waveform for subsequent amplification by matching amplifiers or for triggering the waveforms of the equivalent elemental generators.
Summing circuitry would be required to ensure that one generator did not feed directly into the output of another.
Since each generator would be provided with current limitation circuitry, the current through any individual plate element would never reach an unsafe level even with a short in or across a plate. This is an extension of the idea that by means of a large number of current branches the incidence of damaging current can be kept to a low value.
Compensation of the capacitive reactance of the capacitive neutral plate element could be incorporated within each generator if required. However, compensation in this configuration would now only be required if the diathermy machine were earth-referred at the power output. Since modern diathermy machines are without exception non earth-referred, compensation would be unlikely to be required, the above mechanisms of current division and limitation being sufficient.
One can envisage a diathermy machine which can be built up in generator modules of say 40 W output capacity each. Ten such modules would comprise a generator of capacity 400 W. This would allow a production of a basic machine with the provision of, or switching in of, varying numbers of generator/neutral plate elements. Thus the machine could be matched to different applications such as ENT (ear, nose and throat), paediatrics on to orthopaedic surgery. Possibilities open up of building up machines to accommodate all sizes of animals, as well as humans.
Control might well be simplified compared with other previously described configurations. Current summing and limitation as well as alarms would be important aspects. Microprocessor-based control of all functions would allow considerable rationalisation and miniaturisation of the electronics.
Configuration and Orientation of Neutral Plate Elements
In the versions of this invention using compensation of the capacitive plate it was important to keep all the elements in roughly the same proximity to the patient so that no one branch started to draw excessive current. In this version (and indeed in compensated versions similarly controlled) since the current is limited to a safe value in any one branch the physical design of the plates is not so important. Thus the transverse pattern of plate element as depicted in FIG. 17 would be perfectly acceptable and this freedom from this constraint could allow the designer to arrive at the optimum arrangement for the plate.
These are:
(a) Compensation not required for modern machines;
(b) Simplified control requirements;
(c) The frequency problem disappears--no electronic compensation required;
(d) A basic machine can be manufactured with modules added for various applications;
(e) The advantages of the multi-element capacitive plate are retained;
(f) Development and production costs may be significantly lower; and
(g) The option of integrating elements of plate and generator together opens up the possibility of superior designs from the point of view of ergonomics.
These are:
(a) The generators have to be precisely synchronised to achieve the various complex waveforms of surgical diathermy.
(b) In some countries earth-referred diathermies may still exist. These would be more easily retrofitted by means of a compensated version, which would ensure that the patient was kept at safe voltage levels.
(Section 4.2):
The requirement of a reference lead and pad for the purposes of supplying a reference voltage point on the patient for electronic signal processing equipment has been described above (see FIG. 7). The pad for this purpose need not be conductive as indicated, but could just as well be capacitive like the other elements of the multi-element neutral plate (see FIG. 18). The constraint would be that the reactance of the reference pad be small in relation to the input impedance of the associated signal processing equipment. The advantage is that the reference pad could be an element integral with the multi-element plate and thus simplify design and fabrication.
(Section 4.3):
The advantages of the full length capacitive plate (FIGS. 14 and 17) are probably also applicable in animal surgery. Since it may be that no removal of covering of hair, fur or feather would be required for the application of a neutral plate, the beast could be operated on with the minimum trauma. Indeed it might be the only way that electrical diathermy could be used with some animals. This application possibly may open up a completely new market for this invention.
Modifications and variations may be incorporated without departing from the scope of the invention.
There now follows the Appendix referred to in the description above.
FIG. 19 is the equivalent circuit of a diathermy machine connected to a patient through a compensated multi-element neutral plate. The e.m.f. of a generator is Vs with a total current iT. The source resistance, Rs, is composed of the diathermy source resistance and the body impedance of the patient. Body and cable reactance components are assumed negligible for the sake of simplicity.
The neutral plate is assumed to be composed of (m+n) elements with (m+n) compensated branches, m of which are intact and of admittance YB. There are also n elements each with a partial short depicted by identical resistance, Rf, across the capacitive element. The admittance of each of these branches is YBS. The total admittance of the neutral plate is given by:
YT =mYB +nYBS (A 1)
The potential drop across the compensated plate set is given by:
Vcc =ZT.Vs /(Rs +ZT)˜(Zr /Rs)�Vs =(Vs /Rs YT)=Vs /Rs (mYB +nYBs)
If we make the assumption that the maximum value of iT is when the maximum value of Vs is Vsmax, then assuming a practical value of 1 Ampere,
iTmax =Vs /Rs =1
This gives:
Vccmax =1/(mYB +nYBs)
so that the current in a fault branch is given by:
in =Vccmax /ZBs =1/ZBs (mYB +nYBs) (A2)
The current in a partial short, Rn, is given, by current division, by:
iFn =[(X2 -jRn X)/(Rn 2 +X2)].in
The reactance of the capacitive neutral plate is -jX. This gives, using equation (A2):
iFn =(X2 -jRn X)/{(Rn 2 +X2)(mYB +nYBs)ZBs } (A3)
Equation (A3) can be reduced to form expressed in terms of constants a and b and the variable Rn. Of the following equations (A4) is the version giving the magnitude of the fault current:
iFn =1/[(a+bRn)+j(C+dRn)]mod.(iFn)=[(a+bRn)2 +(c+dRn)2 ]-1/2 (A 4)
Values of the fault resistance can then be substituted to yield the current. The power dissipation in the fault resistance is given by:
PF =iFn 2 Rn =Rn /[(a+bRn)2 +(c+dRn)2 ] (A5)
The value of fault resistance which gives maximum power dissipation in the patient's flesh is found by taking the derivitive of expression (A5). So: ##EQU15## The maximum value is then obtained by putting the numerator of this expression equal to zero. This gives:
Rn 2 =(a2 +c2)/(b2 +d2) (A7)
These equations enable the plate parameters to be set to give a worst case power dissipation which is safe for the patient. For one plate element this maximum power value might be in the region of 1 Watt.
Note, however that this analysis is somewhat artificial because all the fault branches have been assumed identical. The case where one of the fault branches is different is examined below. If DX is the amount by which the inductive reactance in a branch exceeds the capacitive reactance of an element of a capacitive neutral plate then equation (A3) can be reduced to:
iFn =1/{[-Rn DX/X+jX(1+DX/X)](mYB +nYBs)}(A8)
(where "D" as used above and below signifies "DELTA").
This simplifies to:
iFn =1/[n+m(1+X/DX)+j(m+n)Rn /X] (A9)
and the maximum power dissipation is, from equation (A7) given by a fault resistance of:
Rn =[n+m(1+X/DX)]X/(m+n) (A10)
So, for a 10 element neutral plate with 5 equal partial shorts (m=n=5), a reactance of X=800 ohm and a reactive tolerance of DX=80 ohm, the current, iFn, giving maximum power dissipation is about 11.79 mA and the power dissipation, Pn, is 0.667 W, which is quite modest. The power dissipation curve versus fault resistance is shown in FIG. 20, where the curve for the wider tolerance of reactive match of 25% (DX=200 ohm) is also shown. This results in a higher predicted maximum power and so leads to the conclusion that the closest matching of inductance and neutral plate capacitance reactance is desirable.
Note that perfect matching (DX=0 ohm) would give a fault current of zero for all values of fault resistance, Rn. As already discussed, this desirable condition may be approached by inserting a trimming capacitor in parallel with the inductor and adjusting for a mximum, current in each branch. However, the disadvantages are:
(a) some circuit complexity,
(b) a setting-up procedure, and
(c) the possibility of over-adjustment so that the branch goes capacitive with the danger that inter-branch resonance results in high loop currents.
We have seen that a fairly wide mismatch can be tolerated and the random values expected for the capacitive neutral plate elements might lead one to expect that at least some would closely match their compensating inductors. It would require just one neutral plate/inductor set to almost match for mYB in equation (A2) to approach infinity and therefore for Vcc to be almost zero and hence also the fault current iFn. It would be prudent for correct current division for a small value power resistor to be placed in each branch. It follows also from this analysis that the division of currents among the branches depends very much on the reactance tolerance. This being so, it may be worthwhile to consider the automatic adjustment of currents in each branch to achieve even distribution. This could be done by coupling each branch into sensing circuits and then sequentially adjusting each branch by the injection of a balancing potential again via transformer. This technique would really only be viable by the use of microprocessor control to give the fast response required.
It is unlikely that multiple faults would exhibit in the coherent equal-valued fashion already described. We can repeat the analysis of the circuit of FIG. 19 inserting the values for the fault resistance, Rn, of 4800 ohm for the 10% reactance tolerance case into four of the fault branches. The fifth branch was allowed to vary and it was discovered that a new value of Rn ' in that branch, namely 7740 ohm, was found to give the maximum power dissipation. This last was found to be 0.742 W, somewhat higher than the previous case of all values being varied together. Other values are illustrated by the dotted curve in FIG. 20.
The above analyses may represent fairly typically what might happen in practice given that in practice the neutral plate element values will be fairly random. Nevertheless we must ask what are the most extreme conditions that could ever occur. Examination of the equation for Vccmax preceding equation A2 show that it is at a maximum when the branch impedances are at a maximum. Two conditions can be conceived as leading to maximum impedance in a branch:
(a) a dead short across the capacitance neutral plate so allowing the compensating inductance to be fully in play;
(b) an open circuit in the branch to given infinite impedance.
If we assume that of the 5 faulty branches 4 are in one of the above conditions then the fifth would take the highest fault current due to the high value of Vccmax. We analyse what happens in this fifth branch.
If YBs ' is the admittance of the fifth faulty branch and dashed symbols refer also to this branch then the equation for Vccmax may be rewritten:
Vccmax =1/(5YB +4YBs +YBs ') (A11)
Equation (A2) may be rewritten
in '=1/(5YB ZBs '+4ZBs '/ZBs +1) (A12)
and (A3)
iFn '=in (X2 -jRn 'X)/[(Rn ')2 +X2 ](A13)
iFn '=1/{[-Rn 'DX/X+jX(1+DX/X)][-5j/DX-j4/(X+DX)]+1+jRn '/X}(A14)
For condition (a):
iFn '=1/(60+j4Rn '/8800+j6Rn '/800) (A15)
where DX=80 ohm i.e. a mismatch of 10%.
Applying equation (A7) we find:
Rn '=(a/d)X=7543 ohm
Also iFn ' is 11.785 mA and the maximum power (Pmax) is 1.048 W. The figures for the case of a 25% mismatch (DX=200 ohm) are:
Rn '=3529 ohm, iFn '=23.57 mA and Pmax =1.96 W
For condition (b):
iFn '=1/(60+j6Rn '/800) (A16)
for a mismatch of 10%.
The figures for this case are:
Rn '=8000 ohm, iFn '=11.785 mA and Pmax =1.111 W
For the 25% mismatch case:
Rn '=4000 ohm, iFn '=23.57 mA and Pmax =2.222 W
This last set of figures could be taken as the worst case figure. It is a little higher than might be comfortable for the patient, but it could be assumed to be rather unlikely or the above equations could be reworked in the light of revised design parameters. Similarly if tolerances are found to be wider than here assumed the design cycle can be repeated.
Again, but with some complexity in the electronics, automatic adjustment of the plate currents could be achieved. Simultaneous control of the plate-inductor voltage, Vcc, would achieve very high levels of patient safety indeed.
|Cited Patent
|Filing date
|Publication date
|Jan 12, 1978
|Feb 19, 1980
|Valleylab, Inc.
|Multiple source electrosurgical generator
|Nov 17, 1977
|Apr 29, 1980
|Valleylab, Inc.
|Contact area measurement apparatus for use in electrosurgery
|Title not available
|Citing Patent
|Filing date
|Publication date
|Jul 10, 1991
|May 17, 1994
|Electroscope, Inc.
|Electrosurgical apparatus for laparoscopic and like procedures
|Mar 30, 1993
|Nov 18, 1997
|Electroscope, Inc.
|Electrosurgical apparatus for laparoscopic and like procedures
|Jun 13, 1995
|Jun 23, 1998
|Electroscope, Inc.
|Electrosurgical apparatus for laparoscopic and like procedures
|Mar 5, 1996
|Aug 11, 1998
|Medicor Corporation
|Electrosurgical device for preventing capacitive coupling and the formation of undesirable current paths
|Aug 12, 2005
|Sep 9, 2008
|Encision, Inc.
|System and method for performing an electrosurgical procedure
|Aug 12, 2005
|Dec 16, 2008
|Encision, Inc.
|System and method for performing an electrosurgical procedure
|Oct 18, 2007
|Jan 6, 2009
|Covidien Ag
|Return pad cable connector
|Nov 6, 2003
|Jul 28, 2009
|Boston Scientific Scimed, Inc.
|Methods and apparatus for dispersing current flow in electrosurgery
|Oct 24, 2007
|May 25, 2010
|Covidien Ag
|Return pad cable connector
|Sep 28, 2006
|May 25, 2010
|Covidien Ag
|Smart return electrode pad
|Jan 12, 2006
|Jun 15, 2010
|Covidien Ag
|RF return pad current detection system
|Feb 26, 2009
|Feb 8, 2011
|Boston Scientific Scimed, Inc.
|Methods and apparatus for dispersing current flow in electrosurgery
|Mar 14, 2005
|Mar 1, 2011
|Vanderbilt University
|System and method for surgical instrument disablement via image-guided position feedback
|May 1, 2006
|Apr 5, 2011
|Stryker Corporation
|Bipolar cannula for use with an electrode assembly having a separate supply electrode
|Sep 28, 2006
|Apr 19, 2011
|Covidien Ag
|Temperature sensing return electrode pad
|Nov 7, 2006
|May 10, 2011
|Covidien Ag
|Multiple RF return pad contact detection system
|Apr 26, 2007
|Aug 30, 2011
|Encision, Inc.
|Device and method to prevent surgical burns
|Apr 3, 2007
|Sep 20, 2011
|Tyco Healthcare Group Lp
|System and method for providing even heat distribution and cooling return pads
|Mar 31, 2010
|Nov 22, 2011
|Covidien Ag
|Smart return electrode pad
|May 7, 2007
|Dec 20, 2011
|Tyco Healthcare Group Lp
|Capacitive electrosurgical return pad with contact quality monitoring
|Dec 28, 2010
|Jan 17, 2012
|Boston Scientific Scimed, Inc.
|Methods and apparatus for dispersing current flow in electrosurgery
|Aug 1, 2007
|Jan 24, 2012
|Tyco Healthcare Group Lp
|System and method for return electrode monitoring
|Feb 3, 2009
|May 29, 2012
|Tyco Healthcare Group Lp
|System and method for return electrode monitoring
|Apr 13, 2011
|Jul 10, 2012
|Covidien Ag
|Temperature sensing return electrode pad
|May 11, 2007
|Jul 31, 2012
|Tyco Healthcare Group Lp
|Temperature monitoring return electrode
|Dec 14, 2011
|Aug 7, 2012
|Tyco Healthcare Group Lp
|Electrosurgical system for measuring contact quality of a return pad
|Jun 13, 2007
|Aug 28, 2012
|Encision, Inc.
|Combined bipolar and monopolar electrosurgical instrument and method
|Jul 20, 2009
|Jan 22, 2013
|Syneron Medical Ltd.
|Method and apparatus for fractional skin treatment
|Jul 17, 2012
|Feb 26, 2013
|Covidien Lp
|Temperature monitoring return electrode
|May 11, 2007
|Mar 5, 2013
|Covidien Lp
|Temperature monitoring return electrode
|Jan 4, 2012
|Apr 30, 2013
|Covidien Lp
|System and method for return electrode monitoring
|Oct 24, 2008
|Jun 11, 2013
|Encision, Inc.
|Multiple parameter fault detection in electrosurgical instrument shields
|Nov 20, 2012
|Jul 30, 2013
|Syneron Medical Ltd.
|Method and apparatus for fractional skin treatment
|Jan 23, 2009
|Sep 3, 2013
|Covidien Lp
|Hybrid contact quality monitoring return electrode
|Feb 14, 2013
|Apr 8, 2014
|Covidien Lp
|Temperature monitoring return electrode
|Oct 13, 2009
|Jun 24, 2014
|Encision, Inc.
|System and method for monitoring electrosurgical systems
|Jan 22, 2009
|Jul 8, 2014
|Syneron Medical Ltd
|Device, apparatus, and method of adipose tissue treatment
|Apr 3, 2007
|Jul 15, 2014
|Covidien Lp
|System and method for providing even heat distribution and cooling return pads
|Aug 20, 2009
|Jul 15, 2014
|Syneron Medical Ltd
|Method and apparatus for personal skin treatment
|May 25, 2012
|Jul 29, 2014
|Covidien Lp
|System and method for return electrode monitoring
|Aug 1, 2007
|Aug 12, 2014
|Covidien Lp
|System and method for return electrode monitoring
|Oct 23, 2003
|Aug 19, 2014
|Covidien Ag
|Redundant temperature monitoring in electrosurgical systems for safety mitigation
|Mar 31, 2006
|Sep 2, 2014
|Covidien Ag
|Temperature regulating patient return electrode and return electrode monitoring system
|Mar 8, 2011
|Oct 7, 2014
|Stryker Corporation
|Electrode assembly with separate bipolar cannula and supply electrode
|Feb 9, 2010
|Dec 2, 2014
|Syneron Medical Ltd
|Method and system for invasive skin treatment
|Feb 15, 2011
|Dec 9, 2014
|Syneron Medical, Ltd
|Method and system for invasive skin treatment
|Sep 30, 2013
|Jan 20, 2015
|Syneron Medical Ltd.
|Device, apparatus, and method of adipose tissue treatment
|Mar 27, 2014
|May 12, 2015
|Syneron Medical Ltd.
|Method and apparatus for fractional skin treatment
|Apr 15, 2014
|Jul 7, 2015
|Syneron Medical Ltd.
|Method and apparatus for fractional skin treatment
|Jun 16, 2014
|Aug 18, 2015
|Syneron Medical Ltd.
|Method and apparatus for fractional skin treatment
|May 15, 2014
|Jan 5, 2016
|Syneron Medical Ltd.
|Method and apparatus for fractional skin treatment
|Mar 6, 2013
|Feb 9, 2016
|Encision, Inc.
|Multiple parameter fault detection in electrosurgical instrument shields
|Sep 13, 2013
|Mar 1, 2016
|Syneron Medical Ltd.
|Method and apparatus for personal skin treatment
|Feb 22, 2010
|Mar 8, 2016
|Syneron Medical Ltd
|Electrical skin rejuvenation
|Jul 15, 2009
|Mar 29, 2016
|Syneron Medical, Ltd
|Applicator for skin treatment with automatic regulation of skin protrusion magnitude
|Nov 4, 2013
|Apr 5, 2016
|Syneron Medical Ltd
|Hair removal apparatus for personal use and the method of using same
|Mar 12, 2009
|Apr 19, 2016
|Syneron Medical Ltd
|RF electrode for aesthetic and body shaping devices and method of using same
|Mar 7, 2013
|Apr 19, 2016
|Encision, Inc.
|Enhanced control systems including flexible shielding and support systems for electrosurgical applications
|Aug 14, 2015
|Aug 16, 2016
|Syneron Medical Ltd.
|Method and apparatus for fractional skin treatment
|Aug 11, 2014
|Jan 10, 2017
|Covidien Lp
|System and method for return electrode monitoring
|Nov 6, 2003
|May 12, 2005
|Scimed Life Systems, Inc.
|Methods and apparatus for dispersing current flow in electrosurgery
|Mar 14, 2005
|Oct 13, 2005
|Vanderbilt University
|System and method for surgical instrument disablement via image-guided position feedback
|Aug 12, 2005
|Feb 23, 2006
|Odell Roger C
|Electrosurgical system and method
|Aug 12, 2005
|Feb 23, 2006
|Odell Roger C
|System and method for monitoring electrosurgical instruments
|Aug 12, 2005
|Feb 23, 2006
|Newton David W
|System and method for performing an electrosurgical procedure
|May 1, 2006
|Jan 18, 2007
|Tullis Philip J
|Medical Bipolar Electrode Assembly With A Cannula Having A Bipolar Active Tip And A Separate Supply Electrode And Medical Monopolar Electrode Assembly With A Cannula Having A Monopolar Active Tip And A Separate Temperature-Transducer Post
|Nov 7, 2006
|Mar 29, 2007
|Sturm Thomas A
|Multiple RF return pad contact detection system
|Jan 11, 2007
|May 17, 2007
|Ehr Chris J
|Return pad cable connector
|Jan 18, 2006
|Jul 19, 2007
|Sherwood Services Ag
|RF return pad current distribution system
|Aug 1, 2007
|Feb 5, 2009
|Gregg William N
|System and method for return electrode monitoring
|Oct 24, 2008
|Apr 30, 2009
|Encision, Inc.
|Multiple Parameter Fault Detection in Electrosurgical Instrument Shields
|Feb 26, 2009
|Jun 25, 2009
|Boston Scientific Scimed, Inc.
|Methods and apparatus for dispersing current flow in electrosurgery
|Jan 23, 2009
|Aug 6, 2009
|Tyco Healthcare Group Lp
|Hybrid Contact Quality Monitoring Return Electrode
|Feb 3, 2009
|Aug 6, 2009
|Behnke Robert J
|System and Method for Return Electrode Monitoring
|Mar 3, 2009
|Sep 17, 2009
|Tyco Healthcare Group Lp
|Variable Capacitive Electrode Pad
|Mar 10, 2009
|Sep 17, 2009
|Tyco Healthcare Group Lp
|System and Method for Detecting a Fault in a Capacitive Return Electrode for Use in Electrosurgery
|Feb 18, 2009
|Aug 19, 2010
|Shimon Eckhouse
|Method for body toning and an integrated data management system for the same
|Jul 20, 2009
|Jan 20, 2011
|Adanny Yossef Ori
|Method and apparatus for fractional skin treatment
|Dec 28, 2010
|Apr 21, 2011
|Boston Scientific Scimed, Inc.
|Methods and apparatus for dispersing current flow in electrosurgery
|Jan 25, 2011
|May 19, 2011
|Vanderbilt University
|System and method for surgical instrument disablement via image-guided position feedback
|Mar 8, 2011
|Jun 30, 2011
|Stryker Corporation
|Bipolar cannula for use with an electrode assembly having a separate supply electrode
|Feb 20, 2012
|Jun 14, 2012
|Adanny Yossef Ori
|Method for fractional skin treatment
|May 20, 2005
|Aug 13, 2008
|Method and measuring device for determining the transient impedance between two partial electrodes of a divided neutral electrode
|Jul 15, 1998
|Dec 29, 1999
|Indiba, S.A.
|Hyperthermia device
|Jan 17, 2007
|Jul 25, 2007
|Sherwood Services AG
|RF return pad current distribution system
|May 6, 2008
|Nov 26, 2008
|Covidien AG
|Capacitive electrosurgical return pad with contact quality monitoring
|Jul 8, 1992
|Jan 21, 1993
|Electroscope, Inc.
|Electrosurgical apparatus for laparoscopic and like procedures
|Jul 8, 1992
|Mar 18, 1993
|Electroscope Inc
|Electrosurgical apparatus for laparoscopic and like procedures
|Mar 30, 1994
|Oct 13, 1994
|Electroscope, Inc.
|Electrosurgical apparatus for laparoscopic and like procedures
|U.S. Classification
|International Classification
|Cooperative Classification
|European Classification
|Feb 3, 1993
|Maintenance fee reminder mailed
|Jun 25, 1993
|Fee payment
Year of fee payment: 4
|Jun 25, 1993
|Surcharge for late payment
|Feb 11, 1997
|Maintenance fee reminder mailed
|Jul 6, 1997
|Lapse for failure to pay maintenance fees
|Sep 16, 1997
|Expired due to failure to pay maintenance fee
Effective date: 19970709


text:
Toggle navigation
Bob Garfield:
Garfield at Large
Barbara Lippert:
MediaPost Week in Review
Online Spin
Current Stories
MediaPost Home
Today's News
Today's Opinions
Agencies &
The Agency Business
MAD: MediaPost's Agency Daily
Accounts in Review
Events Calendar
Out to Launch
People on the Move
Stan Mack's Real Mad
MAD - London
Online Media, Marketing
and Advertising
OMMA Today
Daily Online Examiner
Email Marketing Daily
Email Insider
EU Email Marketing Daily
Email Insider Europe
Mobile Shop Talk
Mobile Insider
Search Marketing Daily
Search Insider
Social Media & Marketing Daily
The Social Graf
Social Insider
Online Media, Marketing
and Advertising
Real-Time Daily
Programmatic Insider
Data & Targeting Insider
Video Daily
Video Critique
Video Insider
Publishing & Publishers
Publishers Daily
The Daily Blog
Publishing Insider
Supply Side Insider
Native Insider
IoT Daily
Connected Thinking
VR/AR Insider
AI Insider
Media Buying/Planning/Research
& The Media
Audience Buying Insider
Metrics Insider
Research Brief
TV Watch
TV Everywhere
Brand and Product Marketing
Thom Forbes: Top of the News
Engage Target Market Newsletters
Industry Sector Newsletters
Marketing Tools & Strategies
Content Marketing,
Events Home
2017 Events Calendar
Upcoming Events
Email Insider Summit
May 31 - June 3, 2017, FL
Publishing Insider Summit
June 4 - 7, 2017, FL
Email Insider Summit Europe
June 25 - 28, 2017, Spain
July 19, 2017, CA
TV + Video Insider Summit
August 13 - 16, 2017, CA
Brand Marketers Insider Summit - Cross-Channel
August 16 - 19, 2017, CA
Digital Out of Home Awards
September 12, 2017, NYC
Publishing Insider Summit
October 15 - 18, 2017, TX
Email Insider Summit
December 3 - 6, 2017, UT
Search Insider Summit
December 6 - 9, 2017, UT
Recently Concluded
IoT Marketing Forum
May 18, 2017, NYC
Appy Awards
May 17, 2017, NY
Marketing AI
May 17, 2017, NYC
Outfront Forum
May 3, 2017, NYC
Search Insider Summit
April 26 - 29, 2017, FL
Data & Programmatic Insider Summit
April 23 - 26, 2017, FL
Marketing Automotive Awards
April 12, 2017, NYC
Marketing: Automotive
April 12, 2017, NYC
TV + Video Insider Summit
March 19 - 22, 2017, AZ
OMMA at SXSW
March 11, 2017, TX
Brand Marketers Insider Summit - Cross-Channel
March 7 - 10, 2017, TX
More Events
Agency of the Year
Content Marketing Insider Summit
Creative Media Awards
Cross-Channel Marketing Insider Summit
Future of Media
IOT: Shopping
Marketing Sports: Soccer in America
Marketing: Health
Marketing: Politics
Masters of Media Selling
Mobile Insider Summit
OMMA at Advertising Week
OMMA at Internet Week
OMMA Atlanta
OMMA Attribution
OMMA Audience Buying
OMMA Awards
OMMA Chicago
OMMA Denver
OMMA Marketing Tech
OMMA mCommerce
OMMA Miami
OMMA Programmatic
OMMA Programmatic in New York
OMMA Programmatic TV
OMMA San Francisco
OMMA Social
OMMA Video
Online All Stars
Social Media Insider Summit
The Reckoning
Video Insider Summit
Awards Home
Digital Out of Home Awards (September 12, 2017, NYC)
Appy Awards (May 17, 2017, NY)
Marketing Automotive Awards (April 12, 2017, NYC)
Creative Media Awards (January 26, 2017, New York City)
Agency of the Year (January 25, 2017, NYC)
Online All Stars (January 19, 2017, NY)
OMMA Awards (September 26, 2016, NYC)
Become A MediaPost Member
Subscribe to your choice of industry specific newsletters, save $100 on conferences, search member directories, comment on stories and more. Free to qualified media, marketing and advertising professionals.
Join Now
Member Resources
All Member Directory
Manage My Account
Manage My Subscriptions
Additional Resources
Feedback Loop (Comments)
Job Postings
Research Briefs
Industry Events
About and Contacts
About MediaPost
Contact Editorial
Media Kit
Advanced Search
Sign in
Remember me
Forgot your password?
5 Brands That Understand Moms
by Holly Pavlika on Sep 28, 1:52 PM
Moms love brands for myriad reasons, but first and foremost, brands have to make great products. If the brand creates the right experience around the product, it will have the beginnings of a great relationship with mom. What is entailed in creating a great brand experience? It can be anything from supporting a cause to offering great tools and information to creating something that touches an emotional chord to being a brand that brings you just plain fun.
21 Shares
Share on Twitter
Share on Facebook
Share on LinkedIn
Share on Google+
0 Comments
New Survey: Moms, Brands And Twitter
by Stephanie Azzarone on Sep 28, 8:37 AM
How are moms using Twitter? Earlier this month, we asked that question of our Social Savvy research panel, comprising moms active in the social media space. We are announcing the results here.
42 Shares
Share on Twitter
Share on Facebook
Share on LinkedIn
Share on Google+
0 Comments
Reaching Moving Moms
by Maryanne Conlin on Sep 21, 10:30 AM
This new "head-down" mom will force brands to rethink advertising strategies. What role will outdoor advertising play in this new world? How effective will traditional advertising be?
10 Shares
Share on Twitter
Share on Facebook
Share on LinkedIn
Share on Google+
0 Comments
The Parent Trap: Marketers Need To Break Old Habits
by Colleen Stroh on Sep 16, 9:06 AM
I'm perplexed by how marketers speak to parents, especially working parents. Some seem to get us, but most seem to use the same words and phrases to remind us of our current plight.
23 Shares
Share on Twitter
Share on Facebook
Share on LinkedIn
Share on Google+
0 Comments
Back-to-School Trends You Might See This Holiday Season
by Patti Minglin on Sep 14, 9:30 AM
In July, the National Retail Federation shared results from its 2011 Back-to-School survey conducted by BIGresearch: combined K-12 and college spending was expected to reach $68.8 billion, serving as the second-biggest consumer-spending event for retailers behind the winter holidays.
7 Shares
Share on Twitter
Share on Facebook
Share on LinkedIn
Share on Google+
0 Comments
Y-ired Moms: New Rules For Digital Engagement With Today's Generation
by Melissa Lavigne-Delville on Sep 14, 8:03 AM
Gen Y moms hold a unique and powerful position in today's marketplace. As the first generation of female consumers who grew up with the Internet enter motherhood, it is more important than ever to understand what makes her "click." Today's young mom is more social, more tech savvy and more connected than ever before. And she is integral to a brand's success both on and offline. Buzz is about what's happening right now, and immediacy and access have become the new norm. Moms with young children are not only at the forefront of this frontier, but are also driving many ...
33 Shares
Share on Twitter
Share on Facebook
Share on LinkedIn
Share on Google+
0 Comments
4 Reasons To Watch Tumblr
by Maryanne Conlin on Sep 7, 7:43 AM
Yes, yet another new social media platform to watch! But Tumblr is one that could change the landscape for those of us who are targeting moms ... which is of course, a large percentage of the marketing community.
9 Shares
Share on Twitter
Share on Facebook
Share on LinkedIn
Share on Google+
1 Comments
Subscribe to
Today's Most Read Stories
The Future Of Social VR Is NOT Facebook
Screenless TV The Next Wave?
Lead Development Quality With Account Based Marketing
Amazon Leads Boom In Virtual Assistants, With Apple Way Behind
Goodbye To All That
May 2017
April 2017
March 2017
February 2017
January 2017
December 2016
All Archives >
November 2016
October 2016
September 2016
August 2016
July 2016
June 2016
May 2016
April 2016
March 2016
February 2016
January 2016
December 2015
November 2015
October 2015
September 2015
August 2015
July 2015
June 2015
May 2015
April 2015
March 2015
February 2015
January 2015
December 2014
November 2014
October 2014
September 2014
August 2014
July 2014
June 2014
May 2014
April 2014
March 2014
February 2014
January 2014
December 2013
November 2013
October 2013
September 2013
August 2013
July 2013
June 2013
May 2013
April 2013
March 2013
February 2013
January 2013
December 2012
November 2012
October 2012
September 2012
August 2012
July 2012
June 2012
May 2012
April 2012
March 2012
February 2012
January 2012
December 2011
November 2011
October 2011
September 2011
August 2011
July 2011
June 2011
May 2011
April 2011
March 2011
February 2011
January 2011
December 2010
November 2010
October 2010
September 2010
August 2010
July 2010
June 2010
May 2010
April 2010
March 2010
February 2010
January 2010
December 2009
November 2009
October 2009
September 2009
August 2009
July 2009
June 2009
May 2009
April 2009
March 2009
February 2009
About MediaPost
Contact Editorial
RSS Feeds
Privacy/Terms & Conditions
©2017 MediaPost Communications. All rights reserved.
1460 Broadway, New York, NY 10036
tel. 212-204-2000, fax 917-591-3261,
All content published by MediaPost is determined by our editors 100% in the interest of our readers ... independent of advertising, sponsorships or other considerations.

subset: commoncrawl


text: The Index of Industrial Production's capital goods index is an extremely volatile index. It is also well known that Indian industry suffers from plenty of spare capacity. Nevertheless, is it really true, as the chart shows, that the capital goods index in June 2016 is below the level it was in July 2011, or nearly five years ago? If true, the state of investment demand is dire indeed.


text:
With First Tutors: Music you can locate a Bala music theory tutor. Music Theory lessons are an excellent method to begin a stimulating new hobby, acquire a new skill and gain confidence in music theory. Use our site to find Bala music theory tutors and determine which best matches your requirements so you can begin your music theory tuition in Bala immediately!
Heulen is 22 years of age and lives in the small village of Parc near Bala, North Wales. She began her vocal studies with Eirian Owen at the age of 7....
This is the only place to discover Bala music theory tutors in the UK, helping you discover Bala music theory tuition close to home and Bala tutors to assist with any instrument like piano, violin or guitar. We can help you find Bala music theory tuition or a Bala music theory teacher to prepare for music theory exams and improve your music theory skills! We order Bala music theory teachers near you according to their hourly charges and proximity to you.
If you are a music theory tutor and would like to offer your music theory lessons in Bala or anywhere else across the UK, please register with First Tutors: Music or learn about us here. Becoming a Bala music theory tutor is a fantastic way to improve your own knowledge through providing music theory tuition while helping someone else, all in Bala!


text:
Whether it be through his books, one of the many workshops he conducts, or his various public speaking topics, Jay's goals are to always incorporate what he calls the "Four E's." Engagement. Entertainment. Education. And Encouragement.
More Than An Autograph
I am extremely proud to have partnered with author Brian Hester on this latest project.
Brian began collecting autographs as a kid for fun but quickly realized that this journey was teaching him more about himself and life in general. After being asked by local broadcaster, Barry Lee, from 92.5 WINC FM a number of years ago to come on his weekly TV show to discuss some of his autographs, Brian quickly realized that each signature in his collection was More Than An Autograph. Within the covers of this book are his stories and subsequent teachings, told with refreshing comedy and acuity that only Brian could bring to life.
List Price: $19.99 Order Now
Read Lucinda Cross's interview with Jay Foreman on The Huffington Post
Brand New Promotional Material … hot off the presses!
More Than A Teacher
I am extremely proud of my daughter for releasing her first book. Pick up your copy today!
More Than A Teacher is a tribute to all teachers who rise above and beyond what is required of them to help ensure that every student they interact with is prepared to be successful in life. Viewed through a student's eyes, Kayla Foreman showcases the fact that teachers care not only about students' grades, but also that the students are growing and maturing into well-rounded individuals. Kayla wanted to write this book as her way of saying Thank You to the wonderful teachers with whom she has been blessed, and to all the teachers around the world that share these same values.
Kayla Foreman is an author and illustrator. This is the first book she has written and the second book for which she has provided illustrations. Kayla is a freshman in high school and hopes to pursue a career in writing. She is thankful for all of the teachers in her life who have encouraged her to follow her dreams.


text:
Matthew K. Franklin
Matthew Keith "Matt" Franklin is an American cryptographer, and a professor of computer science at the University of California, Davis.[1]
Education and employment[edit]
Franklin did his undergraduate studies at Pomona College, graduating in 1983 with a degree in mathematics, and was awarded a master's degree in mathematics in 1985 by the University of California, Berkeley.[2] He earned his Ph.D. in computer science from Columbia University in 1994, under the joint supervision of Zvi Galil and Moti Yung.[3] Prior to joining the UC Davis faculty in 2000, he worked at Xerox PARC, Bell Labs, and AT&T Labs.[2]
From 2009 to 2014, Franklin was editor-in-chief of the Journal of Cryptology.[2][4]
Research contributions[edit]
Franklin is particularly known for the Boneh–Franklin scheme, a cryptography scheme he developed with Dan Boneh that uses the mathematics of elliptic curves to automatically generate public and private key pairs based on the identities of the communicating parties. In 2013, he and Boneh were winners of the Gödel Prize for their work on this system.[5]
Selected publications[edit]
- Franklin, Matthew K.; Reiter, Michael K. (1996), "The Design and Implementation of a Secure Auction Service", IEEE Transactions on Software Engineering, 22 (5): 302–312, doi:10.1109/32.502223.
- Cramer, Ronald; Franklin, Matthew; Schoenmakers, Berry; Yung, Moti (1996), "Multi-authority secret-ballot elections with linear work", Advances in cryptology—EUROCRYPT '96: International Conference on the Theory and Application of Cryptographic Techniques Saragossa, Spain, May 12–16, 1996, Proceedings, Lecture Notes in Computer Science, 1070, Berlin: Springer, pp. 72–83, doi:10.1007/3-540-68339-9_7, MR 1421580.
- Franklin, Matthew K.; Reiter, Michael K. (1997), "Fair exchange with a semi-trusted third party", Proceedings of the 4th ACM Conference on Computer and Communications Security (CCS '97), pp. 1–5, doi:10.1145/266420.266424.
- Boneh, Dan; Franklin, Matthew (1999), "An efficient public key traitor tracing scheme", Advances in Cryptology — CRYPTO' 99: 19th Annual International Cryptology Conference Santa Barbara, California, USA, August 15–19, 1999, Proceedings, Lecture Notes in Computer Science, 1666, Springer, pp. 338–353, doi:10.1007/3-540-48405-1_22.
- Boneh, Dan; Franklin, Matthew (2001), "Efficient generation of shared RSA keys", Journal of the ACM, 48 (4): 702–722, doi:10.1145/502090.502094, MR 2144927.
- Boneh, Dan; Franklin, Matthew (2003), "Identity-based encryption from the Weil pairing", SIAM Journal on Computing, 32 (3): 586–615, doi:10.1137/S0097539701398521, MR 2001745.
- ^ Department faculty, UC Davis Computer Science, retrieved 2013-05-31.
- ^ a b c Curriculum vitae, retrieved 2013-06-01.
- ^ Matthew K. Franklin at the Mathematics Genealogy Project
- ^ Journal of Cryptology Editorial Board, International Association for Cryptologic Research, retrieved 2013-06-01.
- ^ ACM Group Presents Gödel Prize for Advances in Cryptography: Three Computer Scientists Cited for Innovations that Improve Security, Association for Computing Machinery, May 29, 2013.
External links[edit]
- Home page at UC Davis


text:
Want to see less ads, private areas and be able to post? Register Today to receive all these benefits and more!
Dismiss Notice
Want to see less ads, post content and the ability to buy & sell Oakleys?
Madman-Badman Replacement Parts
Discussion in 'Oakley Sunglasses Discussion' started by AP Jester, 1/14/16.


text:
High Yield
Robeco High Yield Bonds Strategy uses a high quality and long-term investment approach and prefers to invest in high-quality issuers within the high yield universe. The highly experienced team performs detailed issuer analysis to enable the portfolio managers to pick the winners and avoid the losers.Highlights:
- A proven strategy results in attractive and stable returns
- Highly experienced and stable team of investment professionals
- ESG is an integrated part of fundamental analysis
Investment Grade CreditsRobeco Investment Grade Credits Strategy is based on a top-down analysis of global credit markets, combined with thorough bottom-up issuer analysis. The process allows the team to identify and assess credit risks to avoid portfolio losses.
- Long history in managing corporate bonds
- ESG research is used across the research-driven investment process
- The proprietary state-of-the-art risk management tools ensure in-depth risk monitoring
Aggregate Fixed IncomeRobeco Aggregate Fixed Income Strategy uses a consistent and disciplined investment approach, mainly based on strategic asset and duration allocations over a wide range of fixed income asset classes. The team emphasizes on in-depth company and country research in the investment process.
- A systematic investment approach focused on strategic allocation
- Freedom to allocate to any fixed income asset class, dependent on its relative attractiveness
- Focus on both absolute and relative risks
Quant Fixed IncomeRobeco is an expert in the field of quantitative investing. Quant duration management is one of the most important alpha generators in fixed income investing. Especially in today's low-interest rate market, active duration strategies offer bond investors opportunities for additional returns.
- Exposure to global interest rate movements and opportunities
- Highly active duration management to generate alpha
- Based on time-proven quantitative duration model
Remortgaging shareholder equity
- Corporate America is raising ever more debt to repurchase shares
- Central banks are trying to extend the recovery with ever cheaper money
- The main negative short-term technical is the Brexit aftermath
Read the full Credits Quarterly Outlook Q3 2016
Yield drought & bond scarcity
- As prospects for further rate cuts are slim, we prefer long-dated bonds
- We are cautious on credit and favor Europe, specifically subordinate financials
- This year's star performer emerging local debt has further to go
Read the full Fixed Income Quarterly Outlook
This web page is published and operated by Robeco Singapore Private Limited ("Robeco"), which holds a capital markets services licence for fund management issued by the Monetary Authority of Singapore ("MAS") and Robeco is subject to certain clientele restrictions under such licence. The MAS has not reviewed the content in this web page and the MAS takes no responsibility for the information contained herein.
The information contained in this web page is provided for informational purposes only and should not be construed as an offer to sell or an invitation to buy any securities or products, nor as investment advice or recommendation. You should seek independent advice before making any investment decision. The information on this web page is not intended for persons located or resident in jurisdictions where the distribution of such information is restricted or unauthorised. Persons in respect of whom such prohibitions apply must not rely on any information herein and should exit this web page immediately.
All material has been obtained from sources believed to be reliable, but its accuracy is not guaranteed. Some of the content on this web page may contain projections or other forward looking statements regarding future events or future financial performance of countries, markets or companies and such statements are not indicative of the future or likely performance. Actual results or developments may differ materially from those projected. You should also be aware that past performance is not indicative of future performance. An investment will involve a high degree of risk, and you should consider carefully whether an investment is suitable for you.
Robeco does not undertake to update this web page regularly. Information posted on this web page is current only as at the date of posting and may no longer be accurate, complete or reliable when viewed by you. All information contained herein may be changed or amended without prior notice. Robeco does not represent or warrant the accuracy, adequacy, reliability or completeness of any information contained in this web page. Robeco and its affiliates, and any of their directors, employees, and officers will not be liable for any claim or loss arising directly or indirectly from any errors or omissions in the information herein.


text:
It is a tragedy that we have wasted 20 years debating whether climate change exists when we should have been planning how we will cope with it and prevent it becoming even more destructive.
We need more funding for flood defences and to reverse cuts to the Environment
Agency. The UK is installing disaster preparedness systems in Bangladesh and
across Africa — we need them here too, to help evacuate people in time.
But there's a limit to how much difference such measures can make. When the ground is already sodden and the Thames Barrier closed, the water still has to go somewhere when the river reaches capacity.
We need to stop building on London's floodplain. There are alternatives such as higher density suburbs, as in Barcelona and Paris. Even better would be a national plan to take the pressure off the South-East and revive struggling ex-industrial communities elsewhere.
Environment Secretary Owen Paterson should accept his failure to act and resign. And we need a sensible government that will work with affected communities and start planning ahead for a climate in which extreme weather events will become more frequent and severe.
Jonathan Essex, Surrey county councillor (Green)
Let's stop blaming the Environment Agency for not dredging rivers and the Treasury for flood defence ratios that favour urban areas, and focus instead on the institutional pickle behind the floods.
Without the funding to defend all properties at risk of flooding, the Environment Agency must prioritise. Advising against developments in flood-risk areas has been a key strategy to avoid building defences later on. But this relies on local councils cooperating. Incentivised by Government grants to encourage development, faced with a diminishing technical workforce and given watered-down flood planning guidance, it's not surprising that councils permit more and more building in flood risk areas each year.
The Environment Agency needs the power and resources to hold councils to account, if necessary passing on the burden for financing future defences to developers. In turn the Government must stop sending out mixed messages that leave councils pulled in two directions at once. Unless developments in flood-risk areas are curbed, or are at least made more resilient, householders will keep paying the price.
Dr James Porter, University of Leeds
denigration of the Environment Agency's flood expertise by Communities
Secretary Eric Pickles on the Andrew Marr show on Sunday was little short of
disgraceful and has caused outrage among UK water and environmental
The Environment Agency employs flood management experts who are highly regarded around the world. Its staff have worked tirelessly to prevent flooding from rainfall events that are the worst on record while the funding needed to enable the building of defences has been cut by £100 million, with front line staff now at risk of losing their jobs.
This country is living on borrowed time in terms of the ability of its flood defences to provide the protection needed for homes and businesses. Our river channels, flood walls and coastal defences were designed to withstand substantial yet infrequent storms. The weather that we are experiencing risks exceeding the capacity of our defences on a far more frequent basis than in the past and we now face the need to completely rethink the way we provide flood protection.
We have the experts and the will. With the right funding and resources they can deliver the flood defence infrastructure so desperately needed at regional, local and individual property level.
Alastair Moseley, honorary vice president, Chartered Institution of Water and Environmental Management
For all the
magnanimity shown by political leaders yesterday, and talk of "getting on
with the job", it is clear that all parties have been guilty of trying to
score points. Of course, it's the job of politicians to react to public
opinion, and to win elections. It is also the job of scientists and independent
experts to give the facts, however unpalatable.
When budgets are tight, it's important that the government spends our cash in the most effective way possible. But no amount of money can make the rain go away. It seems all but certain that this will be the wettest winter on record, and the water has to go somewhere. It is likely that climate change will lead to more extreme rainfall events in future, as well as causing sea levels to rise. In the face of such a challenge, we need to rethink our whole approach to flooding, making use of innovative methods to protect existing communities, and making sure new developments make the problem better, not worse. Most important, we must base policy on evidence – not emotion, knee-jerk reaction, or the limits of human memory.
Dr Hannah Cloke, Associate Professor of Hydrology, University of Reading
In light of the current flooding in the Lower Thames area, serious thought needs to be given as to how expanding Heathrow may exacerbate the situation. The Airports Commission is currently conducting a consultation on the criteria to be applied to their shortlisted options. One of these, on water and flood risk, surely needs a heavier weighting than others.
Andrew McLuskey, Staines


text:
The CCL is looking for a new postdoctoral associate to join our team!
The Child Cognition Lab at Boston University invites applications for a second...
A new study coauthored by Boston University Assistant Professor of Psychology Peter Blake finds that young children endorse fairness norms related to sharing, but often act in contradiction to those norms when given a chance to share. The article, titled "I Should but I Won't: Why Young Children Endorse Norms of Fair Sharing but Do Not Follow Them," was published this week in the journal Plos ONE.
This phenomenon has rarely been explored in the context of a single study. Using a novel approach, the researchers presented clear evidence of this discrepancy and went on to examine possible explanations for its diminution with age. In one part of the study, 3- to 8-year-old children readily stated that they themselves should share equally, asserted that others should as well, and predicted that others had shared equally with them. Nevertheless, children failed to engage in equal sharing until ages 7–8.
Read the full story on the College of Arts & Sciences website.


text:
Thank you Brawley for supporting Measure M!
Investing in our schools is investing in the future generation of students to come.
Heartfelt appreciation from the BESD Board and Staff.
Jaime Silva, Superintendent
Brawley Elementary School District
Mission Statement
We, the staff
and parents of Phil D. Swing Elementary, are committed to making
education relevant to all students in order for them to reach their
maximum potential by creating and maintaining an environment that
ensures that every member of the school community reaches a high level
of academic achievement as determined by local, state, and national
standards. We commit to a comprehensive system of support to assure
this outcome.
Vision Statement
Our vision is to create
and sustain a Family Friendly atmosphere that empowers all the
individuals to develop into lifelong learners ready to meet the
challenges of the 21st century. Phil Swing students will move
beyond the basics and embrace the four "super skills:" Communication,
Collaboration, Critical Thinking, and Creativity.


text:
Displaying the first of 10 photos of Barnston.
Historic maps of Barnston and the local area, hand-drawn by Ordnance Survey cartographers and Samuel Lewis.
Browse 15 photo books on Barnston and the surrounding area.
I am serching for my relations, Dulice Waring, married Gerald Lewis, I know they had a son called Gerald, I think he was an only child. Before Dulcie and Gerald moved to this area they lived in Eastham, Merseyside. I believe the son is still alive, but have no idea of the address ...Read full memory
The Fox and Hounds was a great place to meet friends and have a game of darts and a pint or two. I can remember a warm fire on a winters night and a friendly publican. The place had a typical English country pub feel and my mates and I were always made welcome. I worked on the place as a painter in the 60`s, and was always given a free pint with lunch. Pete
I attended here circa 1969/70 for one year. I hated every minute. I lived about 3 miles away and we had to get a bus there every morning.


text:
All Categories
Bookstore Categories
Item description for Deep River: Spirituals by Robert Fountain...The Robert Fountain Legacy Project presents the University of Wisconsin--Madison Concert Choir, 1971--1994, Robert Fountain, conductor
An album of Fountain's signature concert encores: American spirituals.
The Robert Fountain Legacy Project began in 2000 when the Mills Music Library at the University of Wisconsin--Madison embarked on digitizing the recorded archives of the distinguished choral conductor Robert Fountain. The collection of 350 concert recordings documents the history of choral music at the university.
Promise Angels is dedicated to bringing you great books at great prices. Whether you read for entertainment, to learn, or for literacy - you will find what you want at promiseangels.com!
Item Specifications...
More About Robert Fountain
Product Categories
Reviews - What do customers think about Deep River: Spirituals?
Be the first to write a review about Deep River: Spirituals
Ask A Question or Provide Feedback regarding Deep River: Spirituals
Add This Product Widget To Your Website
Looking to add this information to your own website? Then use our Product Widget to allow you to display product information in a frame that is 120 pixels wide by 240 pixels high.
|Order toll-free weekdays 10am thru 10pm EST by phone: 1-888-395-0572 (Lines are closed on holidays & weekends.)
|Customer Service | My Account | Track My Orders | Return Policy | Request Free Catalog | Email Newsletter


text: 54 5 Reviews Butternut squash recipe using the bounty of squash from the garden that comes with autumn's crisp, cool days. 20 min Prep Time 40 min Total Time 4 servings Ingredients 2 cups water1 (1 1/2-pound) butternut squash, peeled, seeded, cut into 1 1/2-inch chunks1/4 cup Land O Lakes® Butter2 tablespoons chopped fresh parsley1 tablespoon honey1/4 teaspoon salt1/8 teaspoon dried thyme leaves1/8 teaspoon pepper 1 teaspoon Land O Lakes® Butter Chopped thyme leaves, if desired Directions Place 2 cups water into 4-quart saucepan; cook over medium heat until water comes to a full boil. Add squash. Cover; cook 12-15 minutes or until squash is fork tender. Drain. Mash squash; stir in all remaining ingredients. Cover; cook over medium heat, stirring occasionally, 5-7 minutes or until heated through.Top with 1 teaspoon butter at serving time; garnish with chopped thyme, if desired. Recipe Tips Nutrition Facts (1 serving) Calories 200 Cholesterol 30mg Carbohydrates 24g Protein 2g Fat 12g Sodium 250mg Dietary Fiber 4g Your Name: Your Email: Your Friend's Email: Page Name:


text: I found this to answer a question from a friend and thought it might be of use if you travel to remote areas for any length of time, or if the worst happens. You're probably not going to have the conditions or equipment shown in this video, but the principles are important. Abscess Incision and Drainage - Surgery - YouTube