text:
|Publication number
|Publication type
|Application number
|Publication date
|Dec 28, 2004
|Filing date
|Sep 25, 2001
|Priority date
|Sep 25, 2001
|Fee status
|Also published as
|Publication number
|R. Matthew Mosley, Douglas J. Paul, John N. Pike
|Original Assignee
|Tennessee Scientific, Inc.
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (4), Non-Patent Citations (8), Referenced by (34), Classifications (13), Legal Events (8)
|External Links: USPTO, USPTO Assignment, Espacenet
1. Field of the Invention
The subject invention relates to instruments for measuring the characteristics of fluids, and more particularly, to an improved instrument and method for testing the color, turbidity and/or fluorescence of fluids such as water mixed with a reactant.
2. Background of the Related Art
Water quality monitoring at all levels of production and usage is rapidly becoming a global necessity as sources of fresh potable water become taxed by increasing populations. Even low levels of foreign matter or contaminants can pose significant health and safety risks when undetected. As a result, fast and accurate water testing results are required of an ever-expanding source of test samples. Consequently, there is a growing need for rugged portable instruments and methods for monitoring water quality as the treatment and usage of water expands.
Typically, the testing of water quality has involved the addition of a specified reagent to a fluid sample. Conventionally, a reactant of known concentration is mixed with a water sample which contains a reactant of an unknown concentration. Alternatively, the known reactant is continually added until a sample property change indicates the endpoint of the reaction. In either case, the reagent reacts with the contaminant to create a reaction proportionate to the concentration of the contaminant. Often, a color indicator is included so that a color change occurs, or the color change is inherent to the chemical reaction. Thus, subsequent to the addition of the reagent, visual inspection against a printed color chart can determine the absence or level of an associated contaminant. Such purely visual comparisons are inherently subjective, and therefore unreliable for sensitive measurements. Generally, a skilled technician is required to determine the degree of the reaction and interpret the results. Alternatively, a colorimeter or photometer can consistently measure the degree of reaction (e.g., the depth of color or the spectral transmission) and, hence, the concentration of the contaminant. However, traditional calorimeters and photometers are not practical for field use and provide only color related data.
When the reaction product is a fine-particle precipitate, the sample can be measured by optical turbidimetric methods, i.e. scattering. A high concentration of precipitate as a result of a high concentration of the contaminant creates increased scattering. Therefore, the level of turbidity corresponds to the level of contaminant concentration. Alternatively, the presence or absence of a level of turbidity from a source in a natural fluid sample may be a critical indicator of the quality of such a fluid sample.
An additional technique is to add a reagent including a fluorescent marker. The reaction with the contaminant may either enable or quench the fluorescent moiety. For example, the marker rhodamine will fluoresce in the red when excited by blue light. Thus, transmitting blue light through the sample will generate a red fluorescence proportionate to the level of contaminant. Accordingly, determining the change in the level of fluorescence will indicate the concentration of the contaminant. Additionally, if the blue exciting light is repetitively pulsed and the fluorescence intensity is measured at a particular time after each pulse, the time-decay rate of the fluorescence can provide further information on the chemical nature of the contaminant.
In view of the above, several systems have been developed to ascertain the color, turbidity or fluorescence of a liquid sample. A traditional calorimeter includes a broadband light source. Filters are moved in and out of the optical path to provide different wavelengths. The filters may be moved manually or by motors. A lightpipe or lens system may collect and direct the light to a point on the object to be tested. At such points, the light reflects off opaque objects and passes through translucent objects to receivers. The receivers, usually photodiodes, convert the light signal into an electrical signal for processing. To prevent erroneous readings, the receiver must be isolated from ambient light. To control the environment, the conventional colorimeter is usually utilized exclusively in a laboratory.
For example, U.S. Pat. No. 5,137,364 to McCarthy discloses an optical spectral analysis device having light emitting diodes (hereinafter "LEDs") and receivers mounted on the same substrate. Thus, only reflected light is analyzed. U.S. Pat. No. 5,229,841 to Taranowski et al. shows using a plurality of different colored LEDs which are run according to timing pulses. In synchronism with the LED timing pulses, the outputs of the photodiodes are sampled, and thus each output is indicative of an individual colored LED's signal. U.S. Pat. No. 6,094,272 to Okamoto discloses receiving a sum total of reflected light and comparing the summed value to a value associated with a reference target. The resulting comparison value is displayed in numerical form to indicate a match degree between the tested item and the reference item. U.S. Pat. No. 6,157,454 to Wagner et al. discloses a miniature calorimeter. The miniature colorimeter includes a body having a light pipe for transmitting reflected light to a light sensor, three different primary colored light sources, a display panel and a measure button. In operation, the miniature colorimeter generates three color data points representing the reflectance of the target measured at the wavelengths of the three primary colors. A microprocessor analyzes the three data points and displays the results in various commonly known formats. Further, several patents are directed specifically to water testing methods and devices. U.S. Pat. No. 5,618,495 to Mount et al. automates the process of determining when the endpoint of the reagent reaction is reached with the use of a computer in communication with a colorimeter and other devices. U.S. Pat. No. 5,691,701 to Wohlstein et al. uses the voltages generated by photosensors to produce a ratio which indicates the condition of engine oil. If the test fluid is outside a preset acceptable limit, an alarm indicating the same is triggered.
A multitude of patents are directed to particular aspects of photoelectrically sensing the color of an object. For example, U.S. Pat. No. 5,303,037 to Taranowski discloses a color sensor illumination source which generates a white light evenly composed of red light, green light and blue light directed at the same angle. The importance of a balanced source is to yield relatively balanced color output readings. U.S. Pat. No. 5,471,052 to Ryczek shows a secondary photosensitive element which receives the light directly from the light source. As a result, the signal from the secondary photosensitive element is used to create a closed loop feedback signal to regulate the light source power output.
Additional patents have recognized that certain materials display different colors depending upon the angle of observation. In particular, U.S. Pat. No. 5,592,294 to Ota et al. recognizes the need to accurately determine the angle of observation in order to render reproducible results. To solve this problem, Ota et al. incorporated an angle detector which controls an adjustment mechanism in order to set the desired angle of observation repeatably.
U.S. Pat. No. 5,083,868 to Anderson discloses the need for a portable colorimeter. The colorimeter is enclosed in a housing for receiving a sample. When a vial is placed in the sample compartment, a cap member is positioned in grooves to prevent interference from external light. U.S. Pat. No. 5,872,361 to Paoli et al. discloses a portable turbidimeter having a non-imaging optical concentrator between a sample cell and an optical detector. A cover is utilized to reduce the effect of ambient light on the readings. U.S. Pat. No. 5,604,590 to Cooper et al. discloses a nephelometer instrument for measuring very high water turbidities, such as 10,000 NTUs. The nephelometer instrument has one light source and four detectors. The detectors receive back scatter, forward scatter, 90° scatter and transmitted light.
Still further, several patents are directed to devices for only determining the light penetrability of liquids. U.S. Pat. No. 5,696,592 to Kuan teaches immersing a light guide and a liquid-tight photosensor in a liquid to be measured. When a light source illuminates the light guide, the photosensor generates a signal indicative of the penetrability of light for the test liquid. U.S. Pat. No. 6,055,052 to Lilienfeld is directed to a system for monitoring airborne particulates. Lilienfeld appreciates the need for a portable instrument which can determine ambient air quality in real time at remote locations. Each of the U.S. patents described above are incorporated herein by reference.
Despite the teachings of the above-mentioned patents, there are various problems associated with the systems and methods of the prior art as known in the field of water testing. Many systems require the time-consuming process of acquiring a sample and sending the sample to a laboratory for analysis. Alternatively, on-site analysis is often subjected to inconsistent results due to human error. Further, devices of the prior art, particularly those designed for portable field use, yield only limited results. In the past, providing a device that could consistently and accurately indicate contaminant levels was far less than economic regardless of the location. There is a need, therefore, for an improved water testing instrument and method which permits efficient and accurate readings of at least one, and preferably a plurality of parameters for a variety of applications and operating conditions.
The present invention is directed to a method and device for optically measuring qualities of a substance in ambient light, wherein the device includes a translucent wall defining a sample chamber and an axis. The translucent wall contains the substance to be measured. A radiation source, such as an LED or other light source, is mounted adjacent to the sample chamber and emits a modulated beam of radiation distinguishable from the ambient light because the radiation is modulated. After passage through the sample chamber, the radiation is received by a detector angularly spaced about the axis of the sample chamber relative to the radiation source. The detector generates a modulated output signal indicative of the intensity of the radiation of the beam impinging thereon. A controller activates the radiation source and the detector and processes the modulated output signal for show on a display.
The present invention is also directed to an instrument and method for measuring characteristics of a substance, wherein the instrument includes a sample chamber for receiving therein a sample of the substance. A signal generator has a radiation source mounted adjacent to the sample chamber. The radiation source emits a beam of radiation through the sample chamber onto a detector of the signal generator angularly spaced about the axis of the sample chamber relative to the radiation source. The detector receives the beam of radiation after passage through the sample chamber and substance to be measured therein. Upon receiving the radiation, the detector generates an output signal indicative of the intensity of the radiation impinging thereon. A memory stores reference measurement data based upon a plurality of different reference samples, wherein each reference sample has a different concentration of an impurity. A controller in communication with the memory is operative to receive a signal from the signal generator based upon a sample within the sample chamber having an unknown concentration of an impurity. The controller automatically compares the signal to the reference measurements to determine a concentration of the impurity in the sample and generate an output signal indicative of the concentration.
The present invention is also directed to a method and device for analyzing color and scattering of an elongated sample, such as a water sample contained in a vial, wherein the elongated sample defines an axis. The device includes a first channel defining a first meridional plane having a first radiation source mounted adjacent to the sample. The first radiation source emits a first beam of radiation through the sample onto a first sensor angularly spaced about the axis of the sample relative to the first radiation source. The first sensor generates a first output signal indicative of the intensity of radiation impinging thereon. A second channel defines a second meridional plane and includes thereon a second radiation source mounted adjacent to the sample. The second radiation source also emits a beam of radiation through the sample onto a second sensor angularly spaced about the axis of the sample relative to the second radiation source for generating a second output signal. Electronics activate each of the channels and process the signals generated thereby. Preferably, the channels are about 45° apart to allow for color, transmission and 45° turbidity measurements.
One advantage of the instrument and method of the present invention, is that they employ light sources which are modulated, and therefore the signals generated thereby can be isolated. As a result, the instrument and method are capable of functioning in ambient light. Further, multivariate analysis can be conducted simultaneously to gather data faster than with traditional mechanisms.
Still another advantage of the subject invention is the simplified arrangement of stationary components which perform a multitude of measurements. As a result, the reliability is improved and production costs are reduced.
These and other unique features and advantages of the instrument and method disclosed herein will become more readily apparent from the following description, the accompanying drawings and the appended claims.
So that those having ordinary skill in the art to which the disclosed instrument and method appertains will more readily understand how to make and use the same, reference may be had to the drawings wherein:
FIG. 1 illustrates a perspective view of an instrument for measuring optical transmission, turbidity and fluorescence of a fluid sample constructed in accordance with subject disclosure;
FIG. 2a is a partial schematic, perspective view of the optics of an instrument constructed in accordance with a preferred embodiment of the subject disclosure, wherein only two of three envisioned meridional planes are shown for simplicity;
FIG. 2b is a partial schematic, side view of one meridional plane showing the principle rays of three LEDs of an instrument constructed in accordance with a preferred embodiment of the subject disclosure;
FIG. 2c is a partial schematic, top view of one meridional plane showing the principle ray and outermost rays of one LED of an instrument constructed in accordance with a preferred embodiment of the subject disclosure;
FIG. 3 is a schematic view of the optics and electronics of an instrument constructed in accordance with a preferred embodiment of the subject disclosure;
FIG. 4 is a partial schematic, top view of the optics of an instrument constructed in accordance with a preferred embodiment of the subject disclosure;
FIG. 5 is a partial schematic, top view of the optics used for a scattering measurement in accordance with a preferred embodiment of the subject disclosure;
FIG. 6a illustrates a perspective view of another instrument for measuring optical transmission, turbidity and fluorescence of a fluid sample flowing in a transparent conduit in accordance with the subject disclosure;
FIG. 6b is a cross-sectional view taken along line A—A of FIG. 6a;
FIG. 7a is a perspective view of a sample vial in accordance with an additional preferred embodiment of the subject disclosure;
FIG. 7b is a cross-sectional view of the base of the sample vial of FIG. 7a; and
FIG. 7c is a bottom plan view of the base of the sample vial of FIG. 7a.
The present invention overcomes many of the prior art problems associated with water testing devices and methods. The advantages, and other features of the instrument and method disclosed herein, will become more readily apparent to those having ordinary skill in the art from the following detailed description of certain preferred embodiments taken in conjunction with the drawings which set forth representative embodiments of the present invention, and wherein like reference numerals identify similar structural elements.
Referring to FIG. 1, a multi-axis photometric sensing instrument 100 includes a housing 110 for protecting and aligning optical components mounted therein, a display 112 for indicating the results of a plurality of measurements to a user, and a keypad 114 for receiving input from a user. The housing 110 defines a recess 116 for receiving therein an elongated sample vial 130 defining a chamber therein for receiving a sample to be tested. As described further below, the recess 116 properly aligns the sample vial 130 in the optical path for analysis. A port 118 for communicating with a computer (not shown) and a socket 120 for charging an internal power cell 121 (as seen in FIG. 3) are also integrated within the housing 110. Preferably, the port 118 is a serial, USB, IEEE1394 port or the like as is known to those of ordinary skill in the pertinent art.
Preferably, the instrument 100 is watertight to increase its ruggedness. The optics and electronics, which in the currently preferred embodiment are LEDs and photovoltaic detectors (hereinafter "PVDs"), are fixed within the housing 110. It is envisioned that a multitude of optical and electrical components may be used to fulfill the performance requirements of the instrument. For example, without limitation, semiconductor lasers may replace the LEDs and photoFETs (photosensitive field-effect transistors) or avalanche photodiodes may replace the PVDs. It is envisioned that any useful combination of analog and digital signal-processing electronic components may be used as the electrical components. Reducing the number of moving parts provides for increased reliability. As described in further detail below, the multi-axis, multi-source and multi-detector instrument 100 is selectively operated by electrical switching to perform a plurality of tests.
Preferably, the sample vial 130 is a thin-walled, visually transparent or translucent, non-scattering scintillation tube in the form of a wide-mouth, capped bottle. In a preferred embodiment, the sample vial 130 is in the form of a clear glass or plastic cylinder having a height of about 2 inches and a diameter of about 1 inch. As a result, the sample vial 130 will refract light passing therethrough. The medium within the sample vial 130 will also interact with light passing through the sample vial. It is also envisioned that the sample vial 130 can be square, rectangular, oblong or a multitude of other shapes as will be appreciated by those of ordinary skill in the pertinent art upon review of the teachings herein.
The sample vial 130 has a tabbed, screw-on plastic cap 132. The cap 132 is preferably provided with a tab portion 133 for labeling, tightening leverage when wet, and ease of handling. In one embodiment, the sample vial 130 is disposable to avoid cross-contamination from previous tests. Preferably, the sample vial 130 has a fill line 131 for indicating the required amount of sample. As may be recognized by those of ordinary skill in the pertinent art based on the teachings herein, the dimensions, materials of construction, and shape of the sample vial 130 or other structures defining the sample chamber disclosed herein are only exemplary, and may be changed as desired depending upon the particular application of the instrument or otherwise as desired.
In operation, the sample vial 130 is filled with the test fluid and placed within the recess 116 of the instrument 100. For example, the test fluid may be a water sample taken from the output of a drinking water supply. In a preferred embodiment, the sample vial 130 acts both as a sample holder and an optically refractive element necessary to the instrument's functioning. In the illustrated embodiment, all other hardware and software resides permanently in the remainder of the housing 110, and specific instructions for any chemical/turbidimetric tests are entered through the keypad 114 of the housing 110. The user selects via the keypad 114 an appropriate test such as, without limitation, color, turbidity and/or fluorescence, as described in further detail below. The instrument 100 analyzes the test fluid and the display 112 indicates the results of the test.
Now referring to FIG. 2a, only meridional planes 154 b and 154 c are shown for simplicity. The sample vial 130 is located between multiple LEDs 150 and PVDs 152 when placed within the recess 116. In one embodiment, nine LEDs 150 are arranged so that the principle rays 151 emitting therefrom define three meridional planes 154 passing through a central axis "z" of the sample vial 130 and converging at the central points of their corresponding PVDs 152. The selection and arrangement of the LEDs 150 and PVDs 152 creates a plurality of defined meridional planes 154 a-c about the sample chamber's axis z. The LEDs 150 emit in a forward direction, but there is no need to further control the cone angle of the emission, other than the refractive focussing performed by the sample vial 130. Preferably, the LEDs 150 are relatively small view-angle LEDs. As shown, each of the LEDs 150 and PVDs 152 lie on defined, fixed axes, and the meridional planes 154 a-c pass through axis z of the preferably cylindrical sample vial 130. Any angle effects are automatically taken into account by zeroing with a pure water sample, as described further below. In a currently preferred embodiment of the present invention, the three defined meridional planes 154 a-c (or "channels") provide a total of nine different wavelength bands distributed over the near ultraviolet, visible and near infra-red (hereinafter "NIR") range for liquid optical absorption. The resulting sensitivity and precision of photometric transmission measurement exceeds that of the human eye, and extends the usefulness of the chemical tests beyond that of classical visual color determination. In one embodiment, the LEDs are selected with approximately 5 mm diameters and high brightness. The color of the light may be, without limitation, blue (430 nm peak emission wavelength), green (565 nm), red (660 nm), near IR (880 or 940 nm) and white. Preferably, the white LED is part number CMD333UWC available from Chicago Miniature Lamp. It is envisioned that other selections of LEDs, PVDs, and other light sources and detectors, including the wavelengths and positions of such light sources and detectors, would be obvious to one skilled in the art upon review of the subject disclosure.
Preferably, each LED's emission cone, the refractive power of the sample vial 130, and the width of each PVD 152, are selected in combination to capture the full cone of light within the corresponding PVD. The circular cone of light emitted by each LED 150 is distorted into an oblong shape by passing through the sample vial 130. Hence, the rectangular shapes of the PVDs are needed to catch all of the light. In a preferred embodiment, each LED 150 is a high-brightness T-1ľ LED located approximately 5 mm from the sample vial 130. Each LED emits into a forward cone which preferably has a view-angle of less than about 30°, with emission roughly uniformly distributed within this cone of light. Each PVD 152 has an approximately 20×40 mm rectangular active area. A suitable PVD is available from Radio Shack under part number #276-124A. Each PVD is placed approximately 19 mm from the sample vial 130 to collect the forward cone of light emitted by the corresponding LEDs. In another embodiment, a PVD of approximately 20×20 mm may be used with a small cylindrical lens (axis horizontal, not shown) placed in front of each LED 150 in the 5 mm space between the LED and the sample vial 130. The cylindrical lens condenses the vertical beam spread to fit on a 20 mm high PVD or other sensor. The PVDs 152 are used in the photovoltaic mode to assure linearity of response to input signals, no dark current, and minimal sensitivity to the operating temperature of the instrument 100. As may be recognized by those of ordinary skill in the pertinent art based upon the teachings herein, the dimensions and angles herein are only exemplary and may be changed as desired or otherwise required.
Preferably, the optical system is simple to construct, and requires no rigid tolerancing, as shown. To facilitate lightweight construction and automatic alignment of the LEDs 150 in each meridional plane 154, a piece of metal or plastic (not shown) may be machined or molded to secure the LEDs in place. In another embodiment, clear glass or like transparent or translucent covers (not shown) are epoxied or otherwise secured over the LEDs 150 and/or PVDs 152 to prevent water and dirt buildup. In still another embodiment, a Wratten blue-pass filter (not shown) is cemented over the blue-emitting LEDs in one of the meridional planes to not only aid cleanliness, but to increase the signal-to-noise ratio during measurements. As described further herein, the instrument 100 includes a microcontroller 174 (as best seen in FIG. 3) programmed in a manner known to those of ordinary skill in the pertinent art to control the various switching regimes. Consequently, the only moving part in the instrument 100 is preferably the hand-inserted sample vial 130.
In another embodiment, four PVDs 152 can be arranged in an octagonal arrangement (not shown). In the octagonal arrangement, four LED channels and four PVDs 152 allow twelve different LEDs 150 to be used simultaneously for photometric, transmission and turbidimetric or colorimetric purposes. It is envisioned that small interference filters (not shown) in front of some of the LEDs can be used to sharpen the spectroscopy, if desired. Preferably, each of the four PVDs 152 is approximately 20×40 mm; however, as may be recognized by those of ordinary skill in the pertinent art based upon the teachings herein, these dimensions are only exemplary and may be changed as desired or otherwise as required.
Referring now to FIGS. 2b and 2 c, only one meridional plane 154 of the sensor is shown for simplicity. As shown, the meridional plane 154 lies in the plane defined by the axes x and z of a rectangular coordinate system. It will be appreciated by those skilled in the art that the meridional plane 154 shown represents the configuration of any of meridional planes 154 a-c and that axis z may be in any spatial direction. For the preferred embodiment's capped sample vial 130, the most sensible direction to define axis z is approximately vertically upwards. For other embodiments, such as the sample vial or chamber defined by a quartz pipe or other transparent or translucent conduit for receiving flowing fluids therethrough and described further below, there are no practical restrictions on the length and orientation of axis z. In another embodiment shown in FIG. 2c, a plain-glass beam-splitter 157 and small reference photocell 159, are positioned so as to provide continuous monitoring of LED output power and feedback to correct for LED variations, if such referencing is needed or desired.
As shown in FIG. 2b, the LEDs 150 are located at a distance "s1" in the x axis from the left glass wall of the sample vial 130 in the drawing, and are spaced some minimal distance "h1" in the z axis from each other. The principal ray 151 of each LED in the meridional plane 154 intersects the central portion of the respective PVD 152 along the z axis. The PVDs 152 are oriented approximately perpendicular to the central axis of the meridional plane 154, and are spaced a distance "s3" in the x axis from the sample vial 130 in the drawing. As shown in FIGS. 2b and 2 c, the exemplary PVD 152 defines a height "H" and width "W" and receives light from any or all of the respective meridional plane's LEDs, sequentially or simultaneously. If two or more of the LEDs are "on" simultaneously, each source is preferably frequency modulated, or otherwise modulated as desired (e.g., by phase or amplitude modulation) so that software of the microcontroller 174 can analyze each PVD's output signal and separate the components arising from each LED's emission, as described further below. For example, in a currently preferred embodiment, the frequency of each LED or other light source is modulated approximately as follows: f1=1613 Hz, f2=1099 Hz, and f3=676 Hz.
With reference to FIG. 2b, the height H of the exemplary PVD 152 is determined by the vertical beam spreads of the cones emitted by the corresponding LEDs 150, as modified by refraction in the meridional plane 154 by the water or other liquid-filled sample vial 130. As shown in FIG. 2c, the width W of the exemplary PVD 152 is determined by the refraction in the x-y plane of the extreme rays 153 emitted in the LED's forward cone of half-angle η (i.e., half the conventional view-angle of the LED 150).
Preferably, three meridional planes 154 are spaced octagonally (i.e., about 45° apart) around the central axis z of the preferably cylindrical sample vial 130. For simplicity, FIG. 2a shows only two meridional planes 154 b and 154 c. Preferably, each of the meridional planes 154 contains a plurality of LEDs 150 of different peak wavelengths, and a single PVD 152. Thus, the instrument 100 of the preferred embodiment has nine distinct waveband LEDs 150 and three independent broad-area PVDs 152. As may be recognized by those of ordinary skill in the pertinent art based upon the teachings herein, depending on the measurement application, a greater or lesser number of meridional planes may be used, with a greater or lesser number of LEDs per meridional plane. If the sample vial 130 is scaled up or down in diameter, then the optical design will require s1 and s3 to vary (see FIG. 2b), particularly so that the PVD widths W (see FIG. 2c) can be constrained to avoid the physical overlapping of adjacent PVDs. In theoretical design and tests of the preferred embodiment, where the diameter s2 of the sample vial 130 is about 25.4 mm, it has been found that for LEDs with view angles where 2η is less than or equal to about 24°, the useful dimensions in the system are an "s1" of greater than about 5 mm but less than about 10 mm, an "s3" of about 15 mm, and "W"דH" equal to about 20×40 mm, which are commonly available PVD dimensions. However, as may be recognized by those of ordinary skill in the pertinent art based upon the teachings herein, any of these dimensions and angles may be changed as desired or otherwise required.
Referring now to FIG. 3, when three meridional planes 154 a-c are used in an octagonal configuration, it is useful to use only a triplet of orange, red and NIR LEDs 150 a in meridional plane 154 a, and only deep green and blue LEDs 150 c in meridional plane 154 c, so that complementary long-pass and short-pass filters 178 and 176, respectively, are deployed in front of PVDs 152 a and 152 c, respectively, to reduce stray light in the instrument. Wratten gelatin filters #25 and #47 have been found to be satisfactory in this regard for the long and short filters, respectively. In the preferred embodiment, a doublet of yellow and green LEDs 150 b are used in meridional plane 154 b.
In the embodiment of FIG. 3, a plurality of oscillators 170 a-c are switchably connected to the seven LEDs 150 a-c, respectively. Upon activation, a power cell 121 provides current to the oscillators. Preferably, each oscillator 170 generates a square wave of a unique frequency. Switching circuitry within a switching circuitry area 172 receives the oscillator outputs and determines which LEDs are illuminated depending upon the specific analysis requested by the user, as will be described in more detail below. For exemplary purposes, a simplified switching scheme is illustrated within the switching circuitry area 172. It will be appreciated by those skilled in the electronic switching art that the specific switching circuitry is not limited to the simplified version illustrated, and that a multitude of combinations beyond that shown are contemplated hereby. The use of electronic switching as opposed to manual allows for rapid sequential readings which are only limited by the settling times of the associated preamplifiers 180 a-c, amplifiers 188 a-c and display 112. A microcontroller 174 provides the instructions to actuate the oscillators and associated switching circuitry. The microcontroller 174 has a microprocessor 184 and memory 186 operatively connected thereto. It will be appreciated by those skilled in the pertinent art that the microprocessor 174 can directly generate the modulating frequencies for the LEDs 150. Thus, the analog oscillators 170 may be unnecessary.
Still referring to FIG. 3, the seven LEDs 150 are of various colors. Preferably, the seven LEDs consist of light blue (B+) and dark blue (B−) LEDs 150 c within the meridional plane 154 c, yellow (Y) and green (G) LEDs 150 b within the meridional plane 154 b, and red (R), orange(O) and infrared (IR) LEDs 150 a within the meridional plane 154 a. The light emitting from the LEDs 150 passes through the sample vial 130. Within the sample vial 130, the light is refracted and scattered, if turbidity exists, as described above with respect to FIG. 2a, 2 b and 2 c. It is envisioned that two of the PVDs 152 will receive filtered light. In a preferred embodiment, and as described above, PVD 152 a in meridional plane 154 c has a red light blocking filter 176 associated therewith (e.g., Wratten #47). and PVD 152 a has a blue-green blocking filter 178 associated therewith (e.g., Wratten #25). The use of optical filters aids in reliably separating the average fluorescence emission intensity from scattered intensities. In the case of sensing fluorescence by pulsed excitation or time-delay gated methods, the physical filters may be omitted.
When the PVDs 152 a-c convert the light into an electrical signal, the resulting electrical signals are modulated at the same frequencies as that of the oscillators 170 a-c which supplied power to the corresponding LEDs 150 a-c. The modulation not only allows sorting out the signal from ambient light, which automatically eliminates sun and room lighting from having any effect on the instrument's performance, but also sorts out the signal as opposed to that generated by the other LEDs if more than one LED is activated at any one time.
In a preferred embodiment, one of a plurality of preamplifiers 180 a-c boosts each signal generated by the corresponding PVDs 152 a-c. The microcontroller 174 determines the timing and duration of the activation of the preamplifiers 180 a-c by controlling additional switching circuitry within switching area 182 of a type known to those of ordinary skill in the pertinent art. The boosted signals are variably routed by switching circuitry 182 to a summing amplifier 184 according to the specific analysis requested by the user, as will be described in more detail below. Whichever preamplifiers 180 a-c are turned on, the corresponding signal is sent to the summing amplifier 184. The plurality of amplifiers 188 a-c are each tuned to a respective frequency of modulation to receive the boosted signals. The tuned amplifiers 188 a-c further boost the portions of the signals which are at the modulation frequencies of the oscillators 170 a-c while rejecting or dampening other frequencies. The outputs of the tuned amplifiers 188 a-c are processed by the microcontroller 174 and input to the display 112 for review by the user. It is envisioned that the display 112 may be a LCD screen, a printout, a digital meter, an analog needle gauge, or the like as is known to those skilled in the art. The outputs of the tuned amplifiers 188 a-c also are stored in memory 186 as data. Software stored in the memory 186 provides instructions to the microprocessor 184 to process the data of the particular test. Preferably, the data and/or results are transmitted to an external computer (not shown) via the port 118 for further analysis and long-term storage on a periodic basis. In another embodiment, the above sequence of preamplification, switching, summing and frequency-selective tuned amplification, relevant to the analog representation of the instrument 100 can be replaced at any point after preamplification by a number of alternative digital methods of signal processing. For example, a switched capacitor filter, a digital software filter, synchronous demodulation and the like can be used to extract the signal directly from the PVD preamplifier outputs 180 a-c. The flexibility of microprocessor-controlled switching allows measurements of any PVD output caused by any LED's modulated emission.
Referring now to FIG. 4, for simplicity, an overhead plan view of the optical components in another preferred embodiment of the instrument 100 includes three LEDs 150 a-c and three PVDs 152 a-c in six of the eight positions allowed by an octagonal configuration. The PVDs 152 measure sample transmission in their respective meridional planes 154 a-c and, by suitable switching, the light scattered by a turbid liquid, i.e., turbidity. An optional fourth PVD 152 d is positioned approximately 22.5 degrees from PVD 152 a. The fourth PVD 152 d allows for additional collection of light which has been scattered within the sample vial 130. In this embodiment, the protective color filters in meridional planes 154 a and 154 c have been dispensed with so as to allow scattering measurements from any of the plural LEDs in any meridional plane 154 a-c to any of the four PVDs 150 a-d. It is envisioned that a multitude of configurations would accomplish the measurements contemplated herein, as would be readily recognized by one skilled in the art upon review of the subject disclosure.
To increase the accuracy of the instrument 100 when post-preamplification and modulated signal measurements are carried out by any desired combination of digitized or analog components, stray light is preferably minimized. Undesirable stray light originates from weak radiation directed from an LED 150 outside the view-angle cone, surface reflection upon entering the sample vial 130, optical scattering by bubbles and/or particulates in the liquid or fluid in the sample vial 130, internal reflection upon exiting the sample vial 130, and/or reflection and scattering at the PVD 152 surface. One advantage of using LEDs in three meridional planes, as shown in the arrangement of FIG. 3, is the reduction of stray light effects. The stray light can be further reduced by the addition of black absorbers 190 a and 190 b, as shown by broken lines in FIG. 4, within the empty spaces of the unused meridional plane between the LED and PVD sides of the optical system. As shown, the black absorbers between adjacent LEDs can extend inwards substantially adjacent to the sample vial 130.
To further increase the accuracy of the instrument 100, periodic calibration is appropriate. For periodic calibration, the sample vial 130 is sealed and contains pure bubble-free water ("the zeroing bottle"). The zeroing bottle is inserted in the instrument 100 to check and reset in memory 186 the instrument's clean-water transmission and any direct or scattered stray light levels. Resetting the stored base-line levels at the time and location of use corrects for instrument ambient operating temperature fluctuations, component aging, and will signal the user if some mis-function or major departure from factory-set levels has occurred. The clean-water scattered stray light values stored in memory 186 are mathematically subtracted from sample turbidity measurements to determine actual turbidity. Similarly, the clean-water transmission signals stored in the memory 186 for each LED are used as the divisors in computing the transmissivity of the test fluid in the sample vial 130.
Generally, in operation, the instrument 100 is loaded with standard calibration data. In another embodiment, the instrument 100 performs a series of laboratory readings to generate unique calibration data for storage. It is also envisioned that the instrument may not require stored calibration data, and that field generated reference data will be sufficient to perform the desired analysis. During use of the instrument 100, a user inserts a sample vial 130 into recess 116. As appropriate for a particular analysis, one sample vial 130 may contain clear bubble-free water for zeroing during periodic calibration and other sample vials 130 may contain an unmodified field sample, or a field sample with a reagent added thereto, wherein several readings based upon different substances may be required to complete a particular analysis. The user selects the desired analysis and initiates the process via keypad 114. The microcontroller 174 activates the required LEDs 150 a-c and PVDs 152 a-c, and provides instruction to the user via display 112 as required for the selected analysis. If necessary, the user manually inserts the appropriately filled sample vial 130 at the appropriate time as prompted by the display 112. The electronics, including without limitation the amplifiers and microcontroller, process the signals generated by the readings, display the results to the user on display 112, and store the results in memory 186 for subsequent download to an external apparatus via port 118. If the instrument 100 becomes low on power, the user may recharge the internal battery via receptacle 120 as would be known to those skilled in the pertinent art.
Referring again to FIGS. 3 and 4, optical absorption of light passing straight-through the sample vial 130 provides color-change data. It is recognized that color is a psychophysical perception, and therefore an opto-electronic device cannot obtain such a measurement. However, for simplicity, the term color is used to refer to the multi-wavelength photometry disclosed herein. Thus, there is no true concern for the visual color of the fluid target, just the photometric transmission at particular wavelengths.
To obtain a baseline for the instrument 100, precision spectroscopy is conducted in a laboratory. The results are stored in the memory 186 to be used in combination with digitized data from the output of the tuned amplifiers 180 a-c (see FIG. 3). Preferably, if a reagent is required to effect the color change, the instrument 100 is carried with a kit containing the various necessary chemicals and reagents.
If a reagent needs to be added to the sample in order to effect a color change, it is envisioned that two samples will be taken. One of the samples will provide a baseline "natural color" reference and the second sample will have the reagent added thereto. Each sample should be measured following substantially the exact same process, but for the addition of the reagent. It is also envisioned that a reagent may need to be added to create a fluorescent effect.
For many color measurements, only red, green and blue+LEDs 150 need to be utilized. In one embodiment, the red, green and blue+LEDs are LEDs 150 a-c, respectively, as shown in FIG. 4. The three LEDs are each modulated at a different frequency. As a result of the three beams of light passing through the sample vial 130, red, blue+and green readings are output to the display 112 indicating a color reading for the sample. For less sophisticated color tests, the redundancy of three different color channels may not be necessary. For example, a bluish sample may only require color analysis using the red channel. For more sophisticated color tests, any subset of up to twelve different color LEDs, e.g., as in an octagonal configuration, operated simultaneously at different modulation frequencies, or sequentially at a single modulation frequency, may be used for a color analysis.
In another embodiment, the instrument 100 indicates the concentration of a predetermined impurity. The concentration is based upon comparison of data from a field sample to data points stored in the memory 186. In the preferred embodiment, the stored data is representative of a log-linear Beer's Law plot. A Beer's Law plot is made by measuring the transmitted light signal (T) of solutions of varying concentration without varying the path length of the light or the wavelength, and dividing T by the transmitted light signal of a pure water sample (To) with the same path length and wavelength. A plot of the logarithm of the ratio against concentration is a Beer's Law plot. A linear Beer's Law plot indicates that the Beer-Lambert relationship holds for the solution at the particular wavelength and the Beer's Law plot is used in determining the concentration of unknown solutions. Symbolically, the Beer-Lambert relationship is "T/To=10−A", in which case "A=εbc", where A is the absorption coefficient, ε is the molar absorbtivity, b is the path length, and c is the concentration of the compound in the solution. Preferably, the stored data points are generated under controlled conditions in a laboratory by measuring several samples of known impurity concentrations with the instrument 100. Upon acquisition of an absorption or transmission reading from a field sample, the stored data is compared to the reading from the field sample to find the corresponding impurity concentration in a manner known to those of ordinary skill in the pertinent art based upon review of the teachings disclosed herein. For example, the microcontroller 174 selects first and second data points above and below, respectively, the reading from the field sample. Based upon the field sample reading, and the first and second data points, the microcontroller mathematically interpolates to arrive at an impurity level for the field sample.
It also will be appreciated by those skilled in the art that if the natural color of the field sample has undesirable turbidity or coloration, a natural color sample may be used as a zeroing reference point to further enhance the accuracy of the resulting concentration analysis. In another embodiment, to further increase the reliability of the concentration analysis, multiple colors may be utilized with corresponding sets of stored data points. Thus, the results for analysis with one color would verify the results of another color.
FIG. 5 depicts a currently preferred configuration of the components for conducting turbidity measurements, and is seen to contain a portion of the same components in the same relative positions as shown in FIGS. 3 and 4. The switching circuitry changes according to microprocessor control to accomplish the selected turbidity measurement in a manner known to those of ordinary skill in the pertinent art based upon the teachings herein. Turbidity data is preferably based upon 45° light scattering at about 45° or thereabouts.
In one embodiment, blue turbidity is measured by activating the blue+beam's LED 150 c. It will be appreciated by those skilled in the art that such an instrument 100 may contain additional components which are not shown for simplicity. For example, such components would process the signals and facilitate performing additional measurements as described hereinbelow. As light from the blue+LED 150 c passes through the sample vial 130, a portion of the light is scattered and impinges upon PVD 152 b of the adjacent channel. PVD 152 b has no color filter so a signal modulated at the frequency of blue light is generated by the scattering. The amplifier 188 (see FIG. 3) is tuned to the blue light frequency. After first measuring the sample transmission by amplifying the signal from PVD 152 a, the amplifier 188 is then switched to amplify the signal of PVD 150 b only, and the strength of the signal is proportionate to the 45° turbidity. As a result, the instrument 100 displays a value on the display 112 indicative of the turbidity of the sample.
Still referring to FIG. 5, NIR emission turbidity can also be measured. Preferably, LED 150 a is a modulated source providing NIR, PVD 152 a is the transmission monitor, and unfiltered PVD 152 b is the sensor for NIR scattering approximately 45° off-axis. Since international practices in monitoring water turbidity currently specify the use of 860 nm NIR light in some instances, and long-wave blue light in others, the ability to measure both turbidities in the same instrument 100, with no moving parts, is easy for the user and avoids potential operator error. In still another embodiment, as shown in FIG. 4, placement of a fourth and/or a fifth PVD (not shown) can receive 22.5° light from meridional planes 154 a and c, respectively, for a corresponding turbidity measurement.
For measurements, and particularly for turbidity, zeroing technique and cleanliness are important because air bubbles, lint, smudges and the like may cause undue scattering which may in turn affect the results. In a preferred embodiment, a technician follows the method below in order to efficiently and accurately conduct a turbidity measurement:
1. Select turbidity from a list of measurements shown on the display 112 via the keypad 114.
2. Perform a periodic zeroing to acquire and store clean-water transmissions and stray light signals for all components to be used in the specific turbidity test. As necessary, cleaning and wiping of sample vial 130 is performed. Additionally, inspect for potential sources of undesirable turbidity such as, without limitation, minute air bubbles clinging to the inner glass and/or water interface of the chamber.
3. Fill a sample vial 130 with a sample of the water to be measured. Inspect the filled sample vial 130 for potential sources of undesirable turbidity similar to the previous step.
4. Insert the sample vial 130 into the instrument 100.
5. Initiate the measurement by depressing an appropriate button on the keypad 114 and view the desired result on the display 112.
If addition of a reagent is required to create the scattering of particles within the sample, additional steps may be required. An additional sample vial 130 should be filled with a sample and sealed. Even though no reagent is added, the additional sample vial 130 should be treated, i.e., shaken, substantially exactly as the sample vial 130 with the reagent, and a baseline reading taken to account for any natural turbidity present in the original sample. Thus, by using such a reference point with a similar shaking history, the effects of settling and preparation within the sample are mathematically subtracted by the microcontroller 174, in a manner known to those of ordinary skill in the pertinent art, to yield a true turbidity result.
Still referring to FIG. 5, fluorescence can be measured using direct and approximately 90° emission. In one embodiment, excitation occurs along the blue channel by LED 150 c. The meridional PVD 152 c indicates blue LED intensity when the sample vial 130 contains clear water, natural liquid transmittance when the sample chamber contains a "natural color" water sample, and reacted liquid transmittance when the sample vial 130 contains the natural water after chemical reaction with a fluorogenic agent. When a fluorescent reaction product emits in the red, the signal generated by PVD 152 c is solely that of the blue exciting light transmitted through the sample vial 130 because the red light is blocked by the filter 176 located in front of PVD 152 c. The red fluorescence is detected by PVD 152 a at about 90° to the exciting blue light beam because the red fluorescence passes through the filter 178 located in front of PVD 152 a. Preferably, during this measurement, all LEDs in the instrument except the blue+LED 150 c are switched off to reduce the signal-to-noise ratio. The red fluorescence signal arising from PVD 152 a has the same modulation frequency as that of the exciting beam, and therefore to still further reduce electronic noise, the signal from direct PVD 152 c is switched off when that from PVD 152 a is measured. It will be recognized by those of ordinary skill in the art that the instrument 100 for fluorescence measurement is physically no different from the embodiments discussed above for transmittance and turbidity measurements, and that only electrical switching changes controlled by the software of the microprocessor 184 in a manner known to those of ordinary skill in the pertinent art are required.
In another embodiment, multiple blue LEDs on the meridional plane 154 c increase excitation intensity, and in turn increase the generated signal strength. To further increase the strength of the fluorescence signal, receiving PVD 152 a may be placed closer to the sample vial 130 to increase the collected fluorescence signal. The fluorescence sensitivity can be increased still further if the receiving PVD 152 a is an appropriately molded Winston cone and avalanche photodiode detector, as would be appreciated by those of ordinary skill in the pertinent art.
Multivariate analysis is analyzing two or more different sets of data simultaneously using matrix mathematics in a manner known to those of ordinary skill in the pertinent art to separate out the results. For example, turbidity and fluorescence measurements can be collected simultaneously with the instrument 100. Simultaneous transmission and scattering measurements can be taken with no cross-talk because the LEDs 150 are modulated at different frequencies. Simultaneous transmission measurements at multiple wavelengths can be made on a liquid sample undergoing multiple, non-interacting chemical tests which produce known optical absorption changes at different wavelengths. Various algorithms of a type known to those of ordinary skill in the pertinent art are carried out by the microcontroller 174 to process such multivariate data. Examples of multivariate analysis are given in "Applications and Limitations of Genetic Algorithms for the Optimization of Multivariate Calibration Techniques" by Matthew Mosley (Clemson University, 1998), which is incorporated herein by reference in its entirety.
A plurality of readings can also be conducted sequentially. The intervals between readings are limited by the settling time of the amplifiers and the display 112. For example, photometric and fluorometric, or photometric and turbidimetric measurements can be conducted within seconds by multiple light beams directed along multiple meridional planes 154 of the sample vial 130. While such sets of sequential measurements require only conventional calculational means, the intentional redundancy of data supplied is advantageously treated by multivariate methods known to those of ordinary skill in the pertinent art to greatly increase the user's confidence in, and awareness of, the statistical accuracy of the instrument's output.
Referring to FIGS. 6a and 6 b, the instrument 200 is adapted to take measurements of a fluid or other material passing through the sample chamber 230 in the form of a transparent or translucent conduit 232. To achieve this, the instrument 200 contains multiple LEDs 250 and multiple PVDs 252 on multiple meridional planes (only one shown for simplicity). A microcontroller 274 is also housed within the instrument 200. Preferably, the LEDs 250 are secured on one side of the conduit 230 and the PVDs 252 on the other side. For simplicity, only one set of LEDs 250 and PVD 252 are shown in FIG. 6a. The instrument 200 is halved along a center line passing parallel to the sample chamber axis z′. Preferably, the two halves of the instrument 200 are secured together by a hinge 240 and a closing latch 242. The transparent conduit 230 defines the sample chamber therein. In one embodiment, the outer diameter of the conduit 230 is about 1 inch to allow a configuration of meridional planes 254 as described above. Such an arrangement permits continuous transmission, turbidity and fluorescence measurements of a material flowing through the conduit 230. It is envisioned that this arrangement is applicable in industrial photochemical and biochemical reactors, municipal water and sewerage plants, nuclear power plant fuel rod storage facilities, and the like.
In monitoring flowing fluids, it may be advantageous to scale down the optical sensing instrument 200 to accept sample pipe outer diameters of 0.5 inches or less. Accordingly, in the scaled-down embodiment, the LEDs should be 3 mm instead of 5 mm with small view angles of approximately 20°. Alternatively, semiconductor lasers could be used as the radiation source where appropriate in lieu of the LEDs 250, with PVDs 252 of even smaller surface area than 20×20 mm as described above. Applications utilizing 0.5 inch diameter quartz pipe sample chambers include everything mentioned above, plus common relatively smaller tube piping in chemical, pharmaceutical and food processing plants and the like. As may be recognized by those of ordinary skill in the pertinent art based on the teachings herein, the optical components of the instrument 200 may be scaled up or down to meet the requirements of any of numerous different applications or uses that are currently or later become known.
Only minor design changes in the positioning of the PVDs 252 along the axes of their meridional planes 254 are required to accommodate monitoring of petrochemical fluids having refractive indices higher than water, e.g., in the 1.4-1.5 range. Transmission and particulate concentration monitoring can also be carried out on low refractive index fluids, such as flowing industrial and engine exhaust gases, and low-temperature super-critical and critical fluids, such as condensed CO2. In order to collect substantially all of the light energy, these measurements require the use of large-area PVDs and very low view-angle LEDs, such as 10° light cones and/or semiconductor lasers, because no appreciable refractive light-beam shaping takes place in passage through the sample. Preferably, for high temperature and low temperature flowing fluids, the sample-containing tubes 230 are vacuum sealed inside somewhat larger outer quartz tubes to thermally insulate the samples from the optical sensing instrument 200. In such an arrangement, the PVDs 252 are preferably designed to fit closely around the outer insulating tube.
Referring now to FIGS. 7a through 7 c, another preferred embodiment makes use of a sample chamber 300 with an integrated circuit 320. The sample chamber 300 has a cap 332 and a plastic base 310. The base 310 is secured by a press fit, epoxy or other method known to those of ordinary skill in the pertinent art. As shown in FIG. 7b, the base 310 contains an embedded integrated circuit or printed circuit board 320 which carries electronic devices such as non-volatile, erasable, writable memory. The electronic devices store specific test software instructions, test-specific calibration data, user interface configuration data and the like as may be desired. The printed circuit board 320 of the sample vial 300 also contains an identifying number which positively relates the sample vial 300 to prepackaged chemicals, such as reagents, that may be inside or supplied with the sample vial 300. Upon insertion in the instrument 100, the instrument 100 recognizes the sample chamber 300 by the identifying number, and receives the necessary instructions to perform the appropriate measurements and to automatically start the selected test. Thus, the user only needs to fill the sample chamber within the sample vial 300 and insert the sample vial 300 in the instrument 100. If desired, the sample chamber 300 may be reusable depending upon the chemical test or series of tests conducted therewith.
Referring to FIG. 7c, the sample vial 300 communicates with the instrument 100 through the base 310, when placed in the recess 116 of the instrument 100. Preferably, the weight of the filled sample vial 300, regardless of its rotational orientation in the recess 116, causes contacts in the base 310 to come into electrical contact with spring-loaded contact buttons embedded in the recess 116 of the instrument 100. In one embodiment, an outer contact ring 312 mounted within the recess 116 of the instrument establishes the common electrical ground, and a central contact 316, also mounted on the base of the recess 116 and electrically coupled to the microprocessor 184, communicates power and multiplexed electrical signals back and forth. It is also envisioned that the base 310 and instrument 100 may be electronically connected by a capacitive coupling, by low-power "RFID"-type technology, by magnetic coupling, by any ordinary opto-electronic remote control, and the like.
It will be appreciated by those of ordinary skill in the pertinent art that the components and dimensions described above are only exemplary and can be changed as desired or required by new applications. Accordingly, the detailed description of preferred embodiments is to be taken in an illustrative, as opposed to a limiting sense. While the invention has been described with respect to preferred embodiments, those skilled in the art will readily appreciate that various changes and/or modifications can be made to the invention without departing from the spirit or scope of the invention as defined by the appended claims.
|Cited Patent
|Filing date
|Publication date
|Aug 11, 1993
|Feb 6, 1996
|Texaco Inc.
|Photomeric means for monitoring solids and fluorescent material in waste water using a falling stream water sampler
|Jul 21, 1999
|Sep 12, 2000
|Hertel; Martin
|Method for identifying particles in a gaseous or liquid carrier medium
|Nov 19, 2001
|Feb 11, 2003
|Point Source Technologies, Llc
|Identification of particles in fluid
|Jan 29, 2002
|Jul 8, 2003
|Pointsource Technologies, Inc.
|Flow through light scattering device
|"Light Scattering Technique Detects Pathogenic Aerosols", Jun. 2003, Photonics Spectra p21, reprinted from Optics letters Apr. 15, 2003, pp 589-591.
|Peter C. Hauser and David W. Chiang, "A Photometric Detector Based on a Blue Light-Emitting Diode," Jan. 26, 1993, Talanta, vol. 40 No. 8, pp. 1193-1200, Copyright (C) 1993 Pergamon Press Ltd.
|Peter C. Hauser and David W. Chiang, "A Photometric Detector Based on a Blue Light-Emitting Diode," Jan. 26, 1993, Talanta, vol. 40 No. 8, pp. 1193-1200, Copyright © 1993 Pergamon Press Ltd.
|Peter C. Hauser, Thusitha W. T. Rupasinghe, and Norman E. Cates, "A Multi-Wavelength Photometer Based on Light Emitting Diodes," Oct. 27, 1994, Talanta, vol. 42 No. 4, pp. 605-612, Copyright (C) 1995 Elsevier Science Ltd.
|Peter C. Hauser, Thusitha W. T. Rupasinghe, and Norman E. Cates, "A Multi-Wavelength Photometer Based on Light Emitting Diodes," Oct. 27, 1994, Talanta, vol. 42 No. 4, pp. 605-612, Copyright © 1995 Elsevier Science Ltd.
|R. James Berry, Jay E. Harris, and Ronald R. Williams, "Light-Emitting Diodes as Sensors for Colorimetric Analyses," Apr. 17, 1997, Applied Spectroscopy, vol. 51, No. 10, pp. 1521-1524, Copyright (C) 1997 Society for Applied Spectroscopy.
|R. James Berry, Jay E. Harris, and Ronald R. Williams, "Light-Emitting Diodes as Sensors for Colorimetric Analyses," Apr. 17, 1997, Applied Spectroscopy, vol. 51, No. 10, pp. 1521-1524, Copyright © 1997 Society for Applied Spectroscopy.
|Raimund Barden and Rainer Peters, "CPM Enables Sizing of SubMicron Particles," Mar. 2002, Laser Focus World, p. 99 et seq.
|Citing Patent
|Filing date
|Publication date
|Jan 6, 2003
|Jan 3, 2006
|Optiscan Biomedical Corporation
|Layered spectroscopic sample element with microporous membrane
|Aug 14, 2002
|Jan 24, 2006
|Optiscan Biomedical Corporation
|Device and method for in vitro determination of analyte concentrations within body fluids
|Jul 19, 2002
|May 23, 2006
|Optiscan Biomedical Corp.
|Reagent-less whole-blood glucose meter
|Dec 22, 2004
|Jun 24, 2008
|Avago Technologies General Ip Pte. Ltd.
|Integrated optoelectronic system for measuring fluorescence or luminescence emission decay
|Aug 2, 2007
|Jul 8, 2008
|Nanodrop Technologies, Llc
|Apparatus and method for measuring the signal from a fluorescing nanodrop contained by surface tension
|Mar 3, 2005
|Feb 17, 2009
|Ecolab Inc.
|Portable multi-channel device for optically testing a liquid sample
|Oct 10, 2006
|Mar 2, 2010
|American Ecotech Llc
|Detection of particles
|Dec 22, 2005
|Apr 6, 2010
|General Electric Company
|Method and apparatus for controlling a laundering process
|Dec 23, 2008
|Jun 15, 2010
|Optiscan Biomedical Corporation
|Device and method for in vitro determination of analyte concentration within body fluids
|Jun 14, 2010
|Jan 18, 2011
|Optiscan Biomedical Corporation
|In vitro determination of analyte levels within body fluids
|Jan 14, 2011
|Aug 16, 2011
|Optiscan Biomedical Corporation
|In vitro determination of analyte levels within body fluids
|Jan 26, 2009
|Jun 12, 2012
|Optiscan Biomedical Corporation
|Fluid handling cassette having a spectroscopic sample cell
|Mar 15, 2013
|Oct 14, 2014
|Innovative Science Tools, Inc.
|Optical analyzer for identification of materials using reflectance spectroscopy
|Oct 29, 2014
|Jul 28, 2015
|The United States Of America As Represented By The Secretary Of The Army
|Method of sensing nitroaromatic electron accepting compounds using a photovoltaic sensor
|Mar 15, 2013
|Nov 10, 2015
|Hach Company
|Mobile water analysis
|Jul 21, 2011
|Nov 10, 2015
|Hach Company
|Lab-on-a-chip for alkalinity analysis
|Mar 15, 2013
|Mar 29, 2016
|Innovative Science Tools, Inc.
|Optical analyzer for identification of materials using transmission spectroscopy
|Jun 7, 2012
|Apr 26, 2016
|Measurement Specialties, Inc.
|Optical sensing device for fluid sensing and methods therefor
|Jul 19, 2002
|May 8, 2003
|Braig James R.
|Reagent-less whole-blood glucose meter
|Aug 14, 2002
|May 8, 2003
|Braig James R.
|Device and method for in vitro determination of analyte concentrations within body fluids
|Jan 6, 2003
|Jul 8, 2004
|Peter Rule
|Layered spectroscopic sample element with microporous membrane
|Dec 22, 2004
|Jun 22, 2006
|Fouquet Julie E
|Integrated optoelectronic system for measuring fluorescence or luminescence emission decay
|Mar 3, 2005
|Sep 7, 2006
|Eugene Tokhtuev
|Portable multi-channel device for optically testing a liquid sample
|Jan 23, 2006
|Nov 30, 2006
|Braig James R
|Device and method for in vitro determination of analyte concentrations within body fluids
|Oct 10, 2006
|Jun 7, 2007
|Ecotech Pty Ltd
|Detection of particles
|Dec 22, 2005
|Jun 28, 2007
|Potyrailo Radislav A
|Method and apparatus for controlling a laundering process
|Aug 2, 2007
|Jan 3, 2008
|Nanodrop Technologies, Inc.
|Apparatus and method for measuring the signal from a fluorescing nanodrop contained by surface tension
|Feb 5, 2007
|Dec 31, 2009
|Industrial Municipal Equipment, Inc.
|Light Collar
|Jan 26, 2009
|Dec 31, 2009
|Optiscan Biomedical Corporation
|Fluid handling cassette having a spectroscopic sample cell
|Feb 9, 2006
|May 27, 2010
|Nanodrop Technologies, Inc.
|Apparatus and method for measuring the signal from a fluorescing nanodrop contained by surface tension
|Mar 30, 2011
|Oct 20, 2011
|Sick Engineering Gmbh
|Method for determining a particle concentration and measuring apparatus
|Oct 29, 2014
|Feb 26, 2015
|United States Of America, As Represented By The Secretary Of The Army
|Method of sensing nitroaromatic electron accepting compounds using a photovoltaic sensor
|Jul 28, 2014
|Oct 11, 2016
|Hach Company
|Cuvette for a water analysis instrument
|Feb 9, 2006
|Jan 25, 2007
|Joel Bruce Hansen
|Fluorescence measurement method utilizing surface tension apparatus
|U.S. Classification
|356/436, 356/73
|International Classification
|Cooperative Classification
|European Classification
|Sep 25, 2001
Owner name: TENNESSE SCIENTIFIC INC., SOUTH CAROLINA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:MOSELY, R. MATTHEW;PAUL, DOUGLAS J.;PIKE, JOHN N.;REEL/FRAME:012221/0107;SIGNING DATES FROM 20010920 TO 20010921
|Jul 7, 2008
|Maintenance fee reminder mailed
|Dec 10, 2008
|Fee payment
Year of fee payment: 4
|Dec 10, 2008
|Surcharge for late payment
|Jun 18, 2012
|Fee payment
Year of fee payment: 8
|Aug 5, 2016
|Maintenance fee reminder mailed
|Dec 28, 2016
|Lapse for failure to pay maintenance fees
|Feb 14, 2017
|Expired due to failure to pay maintenance fee
Effective date: 20161228


text:
There are two NS's in hierarchy, parent and child. When replicated items are deleted from parent, they are not deleted from the child.
Customer has this issue with policies, reports, tasks and filters.
For example:
1. Clone a client task and server task
2. Run complete schedule for the Replication.. these tasks are replicated to child NS
3. Delete the client and server task and the parent NS
4. Run differential schedule
5. These tasks are not deleted in child NS
Since replicated items are read-only on child, these can not be deleted manually either.
Note: The expected behavior for items replicated can be found on HOWTO9696 "Are hierarchy replicated resources which are deleted at the parent NS also deleted at the child NS on subsequent replications?"
Known Issue.
Symantec Development is aware of the issue and it is working on providing a resolution on this. A defect entry has been created and it is under investigation. There is not a workaround at this point due to the complexity of the processes involved on Replication.
Applies To
Symantec Management Platform 7.0 SP1 & SP2
Logged in Etrack (Symantec) database
Terms of use for this information are found in Legal Notices.
Terms of use for this information are found in Legal Notices.
Login to Subscribe
Please login to set up your subscription.
Get support for your product, with downloads, knowledge base articles, documentation, and more.
Maximize your product competency and validate technical knowledge to gain the most benefit from your IT investments.
Submit a suspected infected file to Symantec.
Report a suspected erroneous detection (false positive).
Set default language
Do you wish to save this as your future site?

subset: commoncrawl


text:
– Titus O'Neil and Darren Young are expected to win the WWE Tag Team Titles soon. Many people in WWE thought the title change would happen at SummerSlam and apparently that was the plan at...Read More →
– Earlier this week, developmental diva Caylee Turner's profile page was removed from the FCW/NXT website, leading to speculation that she had been released. PWInsider.com is now confirming that Turner, real name Christina Crawford, was...Read More →
Tough Enough contestant Christina Crawford hopes to one day win the WWE Divas Championship. But the person she would like to take the belt from is the same person who inspired her to be a...Read More →
The USA Network has revealed the entire cast of the upcoming season of Tough Enough, which premieres April 4 following Raw. The cast features nine men and five women, many of whom possessing independent wrestling...Read More →
With the debut of WWE Tough Enough just a few weeks away (the night after WrestleMania), we've got some new details on the men and women who will be competing on the show. WWE held...Read More →
— Former WWE developmental talent Christina Crawford (a/k/a Caylee Turner) could be seen in the Tough Enough trailer that aired during tonight's Raw. Crawford, who is the younger sister of WWE Diva Alicia Fox, was...Read More →


text:
- Description
- Shipping+Return
Retro pattern stitching, three button, v neck, long sleeves this pretty cardigan is a must have! Please purchase according to the following size data.
The cardigan featuring a retro pattern stitching, three button, v neck. Highly Recommend as a pretty outwear! For cute, comfy and casual, you can not go wrong with the cardigan.
Season: autumn
Length: 68cm
Shoulder Width: 74CM
Bust: 130cm
Sleeve: 61cm
Length: 70cm
Shoulder Width: 78CM
Bust: 138cm
Sleeve: 63cm
Length: 72cm
Shoulder Width: 82CM
Bust: 146cm
Sleeve: 65cm
FREE SHIPPING over $100(exclude shoes, bags and sale item)
Receiving time = Processing time + Shipping time
Return within 30 days from the delivered date.
Return within 30 days from the delivered date.
1. Items received within 30 days from the delivered date.
2. Items received unused, undamaged and in original package.
3. Return shipping fee is paid by buyer.


text:
jbaranello Review
Tuscan Bamboo Roman Shades
I've been eying them for quite awhile. Glad I purchased them. They complement my family room perfect.
- How did the color on site compare with the actual product? identical
- Please tell us about the quality of the product. good quality, except the string pull has a slight pull in it, but I don't plan on opening and closing it, so it should be okay
- How accurate was the on site description of the product? perfect
Was This Review Helpful?


text:
My career has been a mixture of mathematics and computing. I started out as a math professor, then for several years I was a software developer and software project manager. Later I brought these together, working in mathematical computing.
I've used maybe a dozen programming languages professionally. When I was a full-time developer I mostly wrote C++. These days I mostly write Python and occasionally C# or R. When these aren't fast enough, I drop back to C++.
I add the most value to projects that require a mix of math and computing. On large projects I can help the software developers and math/stat people communicate, making sure that nothing is lost in translation between the two groups.
Here are some areas of computation where I have experience and may be able to help your company.
- Network analysis
- Signal processing
- Formal methods of validation
- Numerical integration
- Interpolation and extrapolation
- Functional programming
- Statistical computing
- Applied linear algebra
- Numerical analysis


text:
The difficulty of designing such tools in EOO modeling environments is linked to the difficulty of mapping simulation form to the model sources. Furthermore; debugging of complex models consisting of over thousand equations by traversing each equation may be very ineffective; especially when the fault has multiple and not very evident causes.
A model reduction methods is proposed and discussed as a method of verification. With model reduction methods it is possible to identify the most important parts of the model which have contributed to the specific model behavior. Because model reduction can be performed on original model representation; the difficulty of mapping simulation form back to model source is avoided.
Keywords: verification; debugging of EOO models; verification by model reduction


text:
As the year drew on, and summer approached, we began to look forward to the good weather. Well, better weather. In the summer Douitchurch is plagued with less rain, although the thermometer does not rise much above ten degress on the average day.
I spent my days emptying my head of all evil thoughts, and all good thoughts for that matter, and drawing whatever was in front of me. By the end of June I had captured most of my house down on paper once at least.
School finished at the end of July, and my year had a big party out on the field. I wasn't invited, so I didn't go. When school ends it is not a time to rejoice. It means that there is nothing to set one's mind on until work starts again in September. It means that another year has gone by and one has not done anything profitable. It means that one is still wasting his life brooding about a murder that happened nearly six years ago, and there's nothing one feels inclined to do about it. By 'one', as you may have gathered, I mean me.
But then in August something else came about. I was not present at the actual happening of it. But I know someone who was. So I tell her tale as she told it to me in my room a few days after it happened. I must confess it is one of the biggest surprises to me what a sleuth she is. Indeed, I am still unsure as to how and why she came to be there. I believe that someone warned her. Of something. Someone...


text:
Hansel Aquino; One In A Million
In the world of mixed marital arts we often see warriors inside the cage, but rarely do we see the everyday champions that claim triumphs over personal tragedy outside of the cage.
Eighteen year old, Los Angelos, CA, resident Hansel Aquino's life completely changed when a medical mystery turned life threatening. Aquino had just gotten into training when he noticed a boil on his leg – or so he thought it was. Along with his assumptions, the doctors also believed it was a treatable condition; something similar to a rash. They gave him all kinds of medicines and creams until they finally started doing tests after nothing was appearing to heal it. They discovered what was much worse than a then expected. Aquino was diagnosed with a rare form of cancer, Epthiliod Sarcoma.
"At first I laughed and then it became real to me. I was in complete shock. I didn't drink or smoke, I was an athlete; a normal 18 year old kid. It was my senior year."
Epthiliod Sarcoma is extremely rare; Aquino was the only one in California diagnosed with it at the time. He was a true 'one in a million'. Doctors were telling Aquino that the only way he would survive this was if they amputated his leg.
"I told them to take every muscle they can so that I can keep my left leg, and they kept telling me the amount of muscle they had to take would make my leg physically unusable. I really didn't want to lose my leg. I was determined and said right up until the surgery that I would end up walking on it again. The doctors thought I was dreaming big, but my mind is powerful – determined."
After the surgery Aquino was laying in his hospital bed when he got up and decided to try out this unusable leg. Aquino was able to walk on it. He was once again 'one in a million'.
"Walking was a memorable moment for me. I cried for my first and last time. I cried out of happiness – they were tears of joy and thankfulness."
Today Aquino is cancer free and spreads his positivity and enthusiasm to his 'little cancer warriors' at local hospitals.
"I call them my cancer warriors because all they know is to keep going. These kids are 4 and 5 and 6 years old. They don't know why cancer chose them. They don't understand the disease. I want to give them the power to keep pushing, and be there for support."
Aquino is fairly new and also very active in MMA and his training. His record is 1-0 .He trains daily in L.A and Las Vegas and rotates between two gyms. He trains at Danger and Defiant in L.A and flies out to Las Vegas to train with Ryan Couture at Xtreme Couture.
"I love all of my training and its exciting training with Ryan Couture, but I plan on moving out to Las Vegas because it's such a hassle flying back and forth regularly. "
Aquino's next fight is on October 21st, 2012 at the Nokia Theater in Los Angelos, CA. against an unknown opponent at this time
"I make sure to train all areas and be well rounded so that I can be competitive against whoever gets in the cage with me."
From hardship, to rehab, to finalization Aquino has achieved success through his internal conflicts along with his physical and mental abilities. Aquino hopes to continue working towards success with his cage fighting, cancer warriors and newest cancer foundation, IAC (It Ain't Chemo!).
"I'm glad it happened to me, because I can be the one to push others and say that nothing is impossible. This sport has kept me alive and pushing forward, it's all I ever think about, and I hope to reach my full potential in the near future."
Sponsors Hansel Aquino would like to thank:
– Gun Store
– RYU (Respect Your Universe)
– Faction Mouth Guards
– IAC (It Ain't Chemo!)
– Xtreme Couture
To reach Hansel Aquino go to his Facebook at:
Caitlin Dieni MMAMadhouse Sports Writer
To reach me on Facebook:
To reach me on Twitter:


text:
Dharamsala: Almost a year to the day he announced that he had played his last game of cricket, former Australia wicketkeeper turned captain of Kings XI Punjab has confirmed that he will step away from the game at the end of IPL 6. Gilchrist revealed his decision to quit after playing a role in Punjab's seven-run win over Delhi Daredevils on Thursday, which kept their flickering hopes of making the IPL 6 play-offs alive.
On May 20, 2012 after Punjab were beaten by Delhi in Dharamsala, Gilchrist said at the post-match conference that he was stepping away from cricket. However, he made a u-turn and rejoined the Mohali-based IPL franchise as captain ahead of the sixth season.
"Last year, I sat here and said I am finished, I probably won't come back, but I came back. I am not looking for any grandstand finish, but in two days, it will be [the] last time I play on this ground," said Gilchrist at the HPCA Stadium on Thursday. "Little things like that are really exciting and lift your enthusiasm levels. I just love this place, this region, it's been one of the greatest experiences of my cricketing life, to be able to come up here and play cricket."
Gilchrist, 42, has so far totalled 2064 runs at 27.52 in 79 IPL matches - the most by a wicketkeeper in the league. After a lean run in IPL 6, he burst to life with an unbeaten 85 off 54 balls against Royal Challengers Bangalore and 42 off 26 balls to help set up a match-winning total against Delhi.
Punjab play Mumbai Indians on Saturday in Dharamsala needing a win to keep them alive in the tournament. Before that, however, they need Sunrisers Hyderabad to lose on Friday.


text:
Latest gameson glulo.com
Easter Egg Hunt- Played: 734 times
Easter is almost here, and to celebrate, we are having an Easter Egg Hunt! Try to find all of the eggs hidden within each of the 4 scenes.
Use your mouse to click one of the big eggs on the intro screen to pick your scene. Then, click around the scene to try and find the hidden Easter eggs!


text:
ED Magedson – Founder
Ripoff Report
VO Financial2900 Fire Rd suite 101 Egg Harbor Township, New Jersey United States of America
VO Financial VO Group VO Financial Scam All Top Officers Arrested by FBI for Fraud Egg Harbor Township, New Jersey
Does your business have a bad reputation?
Fix it the right way.
Corporate Advocacy Program™
VO Financial is a Scam Run by People who Have Been Arrested for Fraud and are currently being prosecuted by the FBI.All of the top financial officers of VO Financial have been arrested for fraud. They were doing business as VO Group (Vacation Ownership Group). When all the top officers were arrested and news hit the internet, they just changed their name to VO Financial and are continuing to do the same exact illegal fraud that they were doing under the old name. (See news article below) You are just not supposed to notice the similarity in name and the fact that they are working from the exact same addresses that VO Group used with the same exact people and using the same exact fraudulent tactics. (Notice below that the address of both businesses are the same. Do not give them any money.
I have alerted U.S. Attorney Paul J. Fishman who is the lead prosecutor in the case against VO Group about the most recent attempt to fool the public.Most of the people who work for VO Financial are former timeshare salesmen for Wyndham. They were trained by Wyndham to use fraudulent sales practices. They use the same exact fraudulent sales practices that Wyndham uses and that they berate Wyndham for using on a daily basis. They will tell you how horrible the sales practices of Wyndham are and ironically, they are using the exact same techniques to get you to buy their services. They will tell you, for example, that they are calling you because they are investigating a complaint that you or a family member made. This is a lie. They do not have any information that you made a complaint. They just want you to start telling them all the things that you are unhappy with Wyndham about. (or other timeshare companys). They will tell you that they do this service for free. This is a lie. Just ask them if you have to pay any money upfront. Of course, they will not tell you the truth but you will find that you have to send them a check for many thousands of dollars or they will not do anything for you. You can do what they do for free. You do not need to pay them tens of thousands of dollars to help you. They just lead you to think that they are doing something for you that you cannot do yourself. All they do is tell you to stop paying your money to Wyndham. If you do not pay Wyndham for 90 days, they will issue a deed in lieu automatically. This happens as a matter of law. It is simply a quick foreclosure process that is provided by law and Wyndham uses. They make you think they have done some work for you but all they do is just tell you not to pay. That is Step one of the fraud. They get you out of your existing contract with your timeshare company.
Step 2 of the fraud is to buy points cheap from a secondary market for pennies on the dollars. They then sell you the points for astronomical amounts compared to what they paid for the points and for what you can get the points for on the internet from 3rd parties. The scam is complete and you will never know what they did. They fleeced you and they made an obscene profit doing it. So step 2 of the fraud is to sell you back less points than you currently own so you do not have to pay as much and you have rolled back the clock to a point where you have the amount of points that you like instead of the extra points that you do not need. There are so many lies that they tell you that I cannot begin to list them. One you should know about though is that they claim that a committee will decide if they will help you. They will tell you that so many people want their help that they can only assist the people who have the strongest case. What a joke!!! They will take anybody who pays them. They will also tell you that you have to make a decision in a hurry otherwise the committee will drop your case and they will help someone else. Does this sound a lot like the tactics of Wyndham. (You must buy today or you will not get this price again.) They will tell you that this practice of Wyndham is misrepresentation and against the law. They will not tell you that they use the same trick on you.
These people are professional liars and they use every illegal sales trick in the book to get you to purchase their product. These people need to be stopped. Until the FTC and FBI put them out of business for good, I will keep the internet up to date when they try to hide their identity again and again. Dont let them make you their next victim.News Article that was published April 25, 2002 by the newspaper Inside the Gate.Adam Lacerda, VO Group Staff Arrested for Timeshare FraudCAMDEN, N.J. (April 18, 2012) U.S. Attorney Paul J. Fishman announced yesterday that nine people from southern New Jersey were arrested for their alleged roles in a $2.4 million mail fraud conspiracy involving timeshare mortgages.Seven others were also charged in the conspiracy and will have initial court appearances in Florida, South Carolina, Virginia, and the U.S. Virgin Islands before going to New Jersey to answer the charges in Complaints unsealed yesterday. The defendants arrested in New Jersey were expected to appear yesterday afternoon before U.S. Magistrate Ann Marie Donio in Camden federal court.
All sixteen of them are affiliated with The Vacation Ownership Group, (a/k/a VO Group LLC).The nine defendants arrested in New Jersey are:Adam Lacerda, 28, Egg Harbor TownshipAshley R. Lacerda, 32, Egg Harbor TownshipSteven Cox, 48, Ventnor CityIan Resnick, 37, AbseconRyan E. Bird, 34, ClementonFrancis Santore, 52, NorthfieldBrian Corley 27 Egg HarborCatherine Bannigan, 57, Egg Harbor TownshipJoseph Diventi, 32, Somers PointAlso charged are:Alfred Giordano, 32, Horry County, S.C.Vincent Giordano, 27, MargateJoseph Saxon, 38, St. Thomas, Virgin IslandsAimee Allen, 26, Little River, S.C.Genevieve Manzoni, 46, Lake Worth, Fla.Eric Reilly, 33, GallowayEric K. Reiff, 40, Williamsburg, Va.
According to the complaint: In July 2010, law enforcement officers began investigating The Vacation Ownership Group, (a/k/a VO Group LLC). The investigation revealed that from at least March 2009 and to Sept. 1, 2011, the defendants, through the VO Group, participated in a fraudulent scheme in which representatives of the VO Group, often using false identities, telephoned owners of timeshare vacation properties purchased from Flagship Resort Development, Wyndham Vacation Resorts Inc., and other timeshare developers. They convinced the owners in some cases to submit money to the VO Group, purportedly to pay off the owners mortgages on their timeshares.The VO Group claimed that the timeshare owner could pay off the mortgage balance at a substantially reduced amount often by as much as 50 percent the amount of the owners original mortgage by mailing payment to the VO Group at a Post Office box in Pleasantville, N.J. The VO Group representatives also got timeshare owners to send the VO Group money purportedly to have timeshares cancelled or sold. After receipt of payments for the VO Groups service, the conspirators caused those payments to be deposited into a bank account in the name of the VO Group. Rather than paying off the timeshare owners mortgage, cancelling the owners timeshare, or selling the timeshare, the conspirators used the timeshare owners money for their personal use.
The investigation also revealed that in an attempt to cover up the scheme, the conspirators in most cases engaged in a bait and switch tactic by purchasing an additional timeshare in the victims name without the victims knowledge. The victim purportedly had assented to the purchase based on documents the VO Group previously emailed to the victim for signature even though the victim had been led to believe he or she was simply paying off the original timeshare mortgage. During the course of the investigation, law enforcement officers interviewed approximately 225 victims of the conspirators scheme identified to date. Many of the victims are elderly. Law enforcement has determined that the conspirators defrauded the victims of more than $2.4 million.All of the defendants except Alfred Giordano, Aimee Allen and Eric K. Reif were arrested Tuesday and are in custody.Another company associated with Adam and Ashley Lacerda is Timeshare Protection, LLC, but that company was not included in this complaint.
P.O. Box 112
Pleasantville, NJ 08232
Toll Free: +1 800 381 9469
E-mail: firstname.lastname@example.com Vacation Ownership Group
50 Rt 134
Dennis, MA 02638
Toll Free: +1 800 381 9469
E-mail: firstname.lastname@example.com The Vacation Ownership Group
2525 Perimeter Place Drive Ste 118
Nashville, TN 37214
Toll Free: +1 800 381 9469
E-mail: firstname.lastname@example.com
This report was posted on Ripoff Report on 05/24/2012 02:05 PM and is a permanent record located here: http://www.ripoffreport.com/reports/vo-financial/egg-harbor-township-new-jersey-08234/vo-financial-vo-groupnbsp-vo-financial-scamnbsp-all-top-officers-arrested-by-fbi-for-f-887692. The posting time indicated is Arizona local time. Arizona does not observe daylight savings so the post time may be Mountain or Pacific depending on the time of year.
Ripoff Report has an exclusive license to this report. It may not be copied without the written permission of Ripoff Report. READ: Foreign websites steal our content
Click Here to read other Ripoff Reports on VO Financial
If you would like to see more Rip-off Reports on this company/individual, search here:Search Tips
In order to assure the best results in your search:
#1 General Comment
SUBMITTED: Friday, October 12, 2012POSTED: Friday, October 12, 2012
Are you an owner, employee or ex-employee with either negative or positive information about the company or individual, or can you provide "insider information" on this company?
#2 UPDATE Employee
SUBMITTED: Friday, September 21, 2012POSTED: Friday, September 21, 2012
We have verified this posting was a competing business and will have Rip Off Report exposed this within the next few days.
Are you an owner, employee or ex-employee with either negative or positive information about the company or individual, or can you provide "insider information" on this company?
#3 General Comment
SUBMITTED: Thursday, July 26, 2012POSTED: Thursday, July 26, 2012
Are you an owner, employee or ex-employee with either negative or positive information about the company or individual, or can you provide "insider information" on this company?
#4 UPDATE Employee
SUBMITTED: Wednesday, July 04, 2012POSTED: Wednesday, July 04, 2012
Are you an owner, employee or ex-employee with either negative or positive information about the company or individual, or can you provide "insider information" on this company?
Advertisers above have met our
strict standards for business conduct.


text:
Yes we have it in Version 3. 🙂 No it's not in the Implementation Guide. The instructions are available online now.
You can install Microsoft Dynamics CRM 3.0 Server into a clustered Microsoft Exchange or SQL Server environment. A cluster of servers running Microsoft Exchange or SQL can reduce system downtime because if the active server in the cluster fails there will be an automatic fail-over to another server.
Ohh yeah, this only works with the professional edition. (I don't think Small Business Server allows for clustering… 🙂 )
Click Here for the instructions for Exchange
Click Here for the instructions for SQL Server
Click Here for instructions on "clustering" Microsoft CRM
And in case you wanted to know how to setup an Exchange Cluster… Here you go.
And in case you wanted to know how to setup a SQL Server Cluster… Here you go.


text:
Post homework questions online and get free homework help from tutors.
Learn More
Improve Grades
Help Others
Save Time
Accessible 24/7
There are currently 101 guests browsing and 2 members online. So far today, 224 students have registered.
Answer These
Questions in this board that still need answers
Find more unanswered questions in Environmental and Conservation Biology
Which is the WORST study break activity?
Snacking on junk food
Taking a nap
Media-related activities (TV, video games, browsing the web)
Excessive caffeine
Huge meals
Previous poll results: Which is the WORST study habit?


text:
Call 0333 200 2024 lines open Mon to Fri - 9am to 6pm
0 items /
Be the first to review this product
Availability: In stock
The Mark Harris California Brown Faux Leather Dining Chair from Mark Harris is a lovely piece that would add elegance to any dining room. This incredible dining chair set is a fine example of designer furniture that uses high quality and sustainably sourced materials. The Mark Harris California Brown Faux Leather Dining Chair features a superb brown faux leather upholstery with a stylish tufted design. It is completed by the shiny chrome legs. It is supplied as a set of two chairs.
Check items to add to cart or select all
Mark Harris Salento 1500 Clear Glass & Chrome Dining Table
Model Number:
Mark Harris California 1400 Clear Glass & Chrome Extending Dining Table
View the whole Mark Harris Furniture range.
For more information please call us on 0333 200 2024. Lines are open from Monday to Friday - 9am to 6pm. Alternatively please email us at email@example.com.


text:
Need a Homocysteine Test in Kalispell, MT? Simply order online 24/7 and get a fast, affordable Homocysteine Test at a lab near you (Find More Locations):
Why would I get a Homocysteine Test?
A Homocysteine Test is a lab test that checks the homocysteine level. People in Kalispell, MT order the Homocysteine Test for general screening or tracking.
How much does a Homocysteine Test cost?
Our Homocysteine Test price includes the doctor's order that you need to get tested in Kalispell, MT, all lab fees and taxes, and a pdf copy of your results. For current prices, CLICK HERE.
What does a Homocysteine Test report look like?
For a Homocysteine Test sample report, please CLICK HERE.
We are delighted to offer the Homocysteine Test in Kalispell, MT.
Homocysteine Test in other cities near Kalispell, MT:Big Arm | Bigfork | Columbia Falls | Coram | Dayton | Hungry Horse | Kila | Lake Mc Donald | Lakeside | Martin City | Proctor | Rollins | Somers | West Glacier | Whitefish


text:
Palm Beach Hotel
No prices found through our partners. Please contact the business directly or check some of our recommended alternatives.
Travelers also recommend:
TripAdvisor Reviews Palm Beach Hotel Colombo
Travel Blogs from Colombo
Two planes, four airports, sixteen hours.
... in, I even had to wait for security to open before I get could get through to departures. By the time that was all done, the clock was ticking and I only had an hour or so before we started boarding. My seat was actually pretty decent, as I had no one in front of me, except that meant that I was the one next to the exit. The door. So, if anything happened, I was opening that bad boy and saving a plane full of lives. No pressure. Of course, the plane ...
A Traditional Wedding at Ape Gama
... the auspicious time to step off the Poruwa is, and there have been incidents of brides fainting! Thankfully this probably lasted about half an hour.
We went indoors into the blissfully airconditioned dining area. Here's a table showcasing what I believe to be ...
Hello Sri Lanka
... Hotel, when we proceed to slow and travel down a bumpy dirt track. Emily, sitting in the front had the immediate view of what looked like a swamp land, getto....her life flashed before her eyes! She envisioned a group of burley men waiting.... Our driver apologised and quickly realised he had made a wrong turn, reversing quickly and finding the right turn only meters away. The Sanmali Beach hotel looks inviting. The gates are opened by an ...
Please explain to me how Sod's law works?
... and then two officers turned up and I was taken back to the police station I had just been from!!! I then had to wait another while and the young officer (who was actually pretty smart and was an inspector in the police force) eventually took me to the scene of the crime and had me reacting out what happened in broad daylight which the locals found rather humorous.
All joking aside, the guy who was assigned to me was sharp and I liked him a lot, his English was ...
D is for Departure, Delays & my next Destination!
... coconut milk, coconut treacle, coconut shells, coconut oil, coconut palm fronds and thatching. That reminds me of another food favoured by the Sri Lankans.. Buffalo curd (a kind of a yogurt), topped with coconut treacle. It's more mild than yoghurt and I'm now partial to a small bowl after dinner or lunch. Funny story, driving back from Worlds End Trail, Dan spotted a buffalo and referred to it as a "curd machine"... It got awkward when I pointed out ...