text:
|Publication number
|Publication type
|Application number
|Publication date
|Jul 23, 1985
|Filing date
|Apr 20, 1983
|Priority date
|Apr 20, 1983
|Fee status
|Also published as
|Publication number
|Michael J. Labowsky, John B. Fenn, Masamichi Yamashita
|Original Assignee
|Yale University
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (3), Non-Patent Citations (14), Referenced by (172), Classifications (19), Legal Events (7)
|External Links: USPTO, USPTO Assignment, Espacenet
This invention relates to a method for the mass spectrometric analysis of chemical compounds in solution, especially when the solution is the effluent of a liquid chromatograph, which is particularly suitable for compounds which are either thermally unstable or involatile.
The direct mass spectrometric analysis of solutions, especially those in which the solutes are thermally unstable or involatile, has long presented difficulties. There have been a number of different approaches, as reviewed, for example, by P. J. Arpino and G. Guiochon in Analytical Chemistry, June 1979, vol. 51, p. 683A, and as discussed at the first workshop on liquid chromatography--mass spectrometry, held at Montreux in October 1981, the proceedings of which were published in Journal of Chromatography, 1982, volume 251 pps. 91-225. Amongst the many different approaches that have been used, two related techniques, known as electrohydrodynamic ionisation and electrospray ionisation, respectively, will be discussed here in greater detail because of their relevance to the present invention.
In the technique of electrohydrodynamic ionization, which is fully described by B. A. Stimpson and C. A. Evans Jnr. in Journal of Electrostatics, 1978, volume 5 p. 411, and Journal of Physical Chemistry, 1978, volume 82, p. 660, the solution is introduced into the vacuum system of the mass spectrometer through a capillary tube which is charged at high voltage relative to an extractor electrode situated immediately in front of it. This electrode is usually a thin disc with a hole in the centre, and the capillary tube is positioned concentric with the hole and so that its end is situated within the thickness of the disc. The solution to be analysed is ejected into the vacuum system through the capillary by means of a syringe, which is preferably motor driven. A high positive voltage (if positive ions are to be formed) is applied to the capillary, and the syringe plunger compressed to eject liquid into the vacuum system. If the correct conditions are employed (described below), electrohydrodynamic ionisation of the liquid will take place, and a beam of ions characteristic of the solute will be formed, which can be focussed into a conventional mass analyser. In general, it is necessary to use a solvent which has a low volatility (to ensure that the pressure in the vacuum system does not rise too high), and one which is strongly polar and has a reasonably high electrical conductivity. Glycerol with sodium iodide dissolved in it is frequently employed. These requirements are thought to be due to the fact that in electrohydrodynamic ionisation the electrical field does not actually ionise the solute molecules, but merely distorts the forces present at the surface of the liquid to such an extent that ions already present in the solution are directly emitted into the gas phase. These ions are then focussed into the mass spectrometer. Consequently it is necessary for the ions to be present in solution before it encounters the electrical field, and the process works best with with polar sample molecules dissolved in strongly polar solvents, or with a liquid metal sample. Another characteristic is that the flow rate of the solution is best kept at a very low level, so that no droplets of liquid emerge from the capillary. The sample ions are then emitted from sites round the tip of the capillary, and the shape of the capillary and its position relative to the extractor electrode have a profound effect on the efficiency of the ionization.
The electrohydrodynamic ionisation mass spectra of organic samples, obtained from glycerol and sodium iodide solvents, consist in general of peaks due to the molecular ion of the solute clustered with a variable number (between 0 and 10) of glycerol molecules and sometimes sodium ions, or in the case of negative ions, iodide ions. There is little fragmentation of the molecular ion, but the spectra are often difficult to interpret because of the formation of the complex clusters containing an unknown number of glycerol molecules. Further, the use of electrohydrodynamic ionisation sources with solvents other than glycerol and sodium iodide, although possible, is not always satisfactory because the degree of ionisation of the sample in the solution is usually lower, and more volatile solvents can give rise to problems of excessive pressure in the vacuum system due to evaporating solvent vapour. This problem can be reduced by using a nozzle skimmer system and an additional pumping stage in a similar way to that described below, but the ionisation process remains of low efficiency and for organic molecules the only really satisfactory results are obtained with glycerol solvents. Consequently the use of electrohydrodynamic ionisation for liquid chromatograpy-- mass spectrometery is restricted.
In contrast with electrohydrodynamic mass spectrometery, electrospray mass spectrometery does not require glycerol and sodium iodide solvents. It is based on work by M. Dole et al, (described, for example, in Journal of Chemical Physics, 1968, volume 49, p. 2240). A solution containing the sample to be ionised is sprayed from a capillary tube into a region containing gas at approximately atmospheric pressure, towards a small orifice in a plate which leads into the vacuum system of the spectrometer. A high electrical potential is applied between the spraying capillary and the walls of the chamber containing the gas (including the plate with the small orifice). A separation device, usually a nozzle skimmer system like that described by Kantrowitz and Gray in the Review of Scientific Instruments, 1951, volume 22, p. 328, is placed between the region of atmospheric pressure and the vacuum system in order to reduce the quantity of gas flowing into the vacuum system, and to produce a better collimated molecular beam.
The principle of operation of the electrospray source is as follows. The sample to be ionised is dissolved in a solvent, preferably a fairly polar one, and the resultant solution is slowly displaced through the capillary into a region of high gas pressure and electrical field, as explained. As the jet of liquid emerges it becomes charged by the strong field, the solvent begins to evaporate and the jet breaks up into a series of small charged droplets. It was originally thought that these droplets would continue to evaporate until a point known as the Rayleigh limit was reached, where the drop would become unstable because of its increasing charge to volume ratio and break up into smaller drops, at least one of which would carry the charge. This process was thought to continue until all the solvent evaporated, leaving only neutral solvent molecules in the gas phase and ions of the solute, usually clustered with a few solvent molecules. However, the present inventors believe that it is unlikely that a droplet could evaporate sufficiently to reach the Rayleigh limit before an ion, usually solvated, would be lost from the charged drop by a process similar to electrohydrodynamic ionisation. Whatever the principle involved, it is clear from the original work of Dole that the electrospray technique produces ions from solutes of very high molecular weights (e.g. 500,000), and as the energy imparted to the ions is low, very little, if any, fragmentation of the ions takes place. It is therefore well suited for the ionisation of thermally unstable molecules, such as those frequently encountered in biochemistry.
Electrospray ionisation differs from electrohydrodynamic ionization chiefly in the fact that in the former the solution is sprayed into a gas at atmospheric pressure, whilst in the latter, liquid is pumped slowly through a capillary which leads into an evacuated region so that most of the solvent evaporates before it leaves the capillary and ionisation takes place largely at the tip of the capillary tube. Electrospray type ion sources have been interfaced with mass spectrometers, and the use of such a combination for liquid chromatography--mass spectrometry is known. A typical system is described in U.S. Pat. No. 4,209,696.
A process which is related to electrospray ionisation has been developed by J. V. Iribarne and B. A. Thompson, and is described in U.S. Pat. No. 4,300,044. In this process, a solution containing the sample to be ionised is sprayed from a capillary into gas at atmospheric pressure, and the resultant jet of liquid is nebulized by means of a jet of compressed air flowing at right angles to the liquid jet. The droplets of liquid formed in this way are then electrically charged by induction from a high voltage electrode placed close to the nebulizing jet of air. This process is carried out in the mouth of a wide bore tube, into which the charged droplets are swept. The gas flow containing the drops is then directed across the surface of a plate containing a small hole which leads into the mass spectrometer analyser, and the ions formed as the droplets evaporate are caused to enter this hole by means of an electric field applied at right angles to the gas flow. A curtain of inert gas (e.g. carbon dioxide) flows between the plate which contains the hole and a second plate situated a short distance behind it. This is at a higher pressure than the gas in the remainder of the source and consequently escapes through the orifice in the first plate into the source region. This curtain gas serves to isolate the mass spectrometer vacuum system from excessive flows of solvent vapour, water vapour and other contaminants, and enables cryopumping to be carried out in the vacuum chamber of the mass spectrometer by ensuring that most of the gas that enters the vacuum chamber is the cryopumpable curtain gas. However, the process described is not a true electrospray source because the droplets are produced by a jet of air and are charged by induction, whereas in the true electrospray source they are produced by the action of an electric field on the jet of liquid, and no additional nebulization or charging is required.
The main disadvantage encountered with the prior art electrospray ionization systems, such as that described, is that, like electrohydrodynamic ionisation, the ions produced are usually clustered with a variable number of solvent molecules, although this number tends to be smaller than in electrohydrodyamic ionisation. It is thought that the clustering arises from the fact that during the last stages of evaporation the droplets are virtually stopped by collisions with the inert gas molecules, and the remaining collisions due to thermal motion of the gas molecules are not usually sufficiently energetic to remove the last remaining solvent molecules clustered round the sample ion. In U.S. Pat. No. 4,209,696, additional desolvation is achieved by accelerating the solvated ions after they emerge into the vacuum system in a region close to the orifice where the gas pressure is still fairly high. The increased energy of the collisions of the gas molecules is then sufficient to remove more of the solvent molecules from the ions. The ions are then decelerated again in a region further from the orifice where the pressure is lower and there is a much smaller probability of reassociation between the solvent molecules and the ions. A very similar process is described in U.S. Pat. No. 4,121,099, which also describes another problem encountered with prior art electrospray ion sources, that is the problem of providing an efficient focussing action which does not impart a significant energy spread to the ions and which would degrade the performance of the mass spectrometer. The patent suggests a possible solution to the problem by the provision of strong focussing fields in a region very close to the orifice in the free jet expansion where the pressure is high enough for collisions between the gas molecules and the ions to limit the amount of energy that can be imparted to the ions, thereby limiting the energy spread that is imparted to the ions during the focussing process. Additional lens elements may also be provided further from the nozzle to achieve the declustering effect discussed above. However, it is not possible to completely separate the focussing and declustering actions, and it is difficult to optimise both features simultaneously.
When a magnetic sector spectrometer is to be used, the potential of the inlet capillary of the electrospray source must in general be maintained at a value at least as great as the accelerating voltage required by the spectrometer, and the focussing problem is worsened because the ions still emerge through the orifice into the low pressure region with a very low kinetic energy. They must therefore be reaccelerated to the energy required by the spectrometer. It is difficult to construct electrostatic lenses which will achieve this without significant loss of transmission efficiency and broadening of the kinetic energy spectrum of the ions, and when the design of the acceleration step is further constrained by the desolvation requirements it becomes even more difficult. There is considerable advantage, therefore, in completely separating the desolvation stage from the focussing stage so that both processes can be optimised independently, and so that the need to accelerate the ions to cause desolvation can be eliminated. It is an object of the present invention, therefore, to provide means for desolvating the ions and effectively controlling the extent of the desolvation, before they leave the region of the source which is maintained substantially at atmospheric pressure.
In accordance with one aspect of this invention there is provided a method of generating ions for mass spectral analysis from a sample dissolved in a solution by electrospray ionization the improvement comprising:
(a) causing said solution to flow through a capillary tube or jet;
(b) maintaining a high electrical potential difference between said capillary tube or jet and a first conductive wall spaced from and disposed facing the outlet of said capillary tube or jet thereby causing said solution to be electrosprayed from said outlet;
(c) maintaining the ambient pressure between said outlet and said wall at substantially atmospheric pressure;
(d) providing between said wall and said outlet a flow of inert gas substantially counter to the flow from said outlet;
(e) providing an orifice in said wall in alignment with the axis of said capillary tube or jet at the outlet thereof; and
(f) maintaining the pressure on the side of said wall remote from said outlet at a reduced pressure whereby gas containing ions from said solution characteristic of said sample flows through said orifice.
In a preferred aspect of the method of the invention, the gas and ions flowing through said orifice away from said outlet are subsequently subjected to one or more additional pressure reduction, focussing on ion accelerating operations prior to the mass spectral analysis of the ion beam so generated.
In accordance with a further aspect of this invention there is provided an apparatus for generating ions for mass spectral analysis from a sample dissolved in a solution by means of electrospray ionisation, comprising:
(a) a capillary tube or jet through which the said solution may be caused to flow leading into a first chamber having a first conductive wall spaced from and disposed facing the outlet of the said capillary, means for maintaining a high electrical potential difference between said outlet and said first conductive wall whereby said solution may be electrosprayed from said outlet into said first chamber;
(b) and an inlet pipe or pipes through which an inert gas may be introduced into said first chamber between said wall and said outlet in a direction substantially counter to the flow from said outlet, and an outlet pipe or pipes leading from said first chamber to a region at substantially atmospheric pressure whereby the ambient pressure in said first chamber may be maintained at substantially atmospheric pressure or a slightly greater pressure; and
(c) an orifice in said first conductive wall and aligned with the axis of said capillary tube or jet at the outlet thereof and leading into a second chamber which is provided with means for maintaining therein a reduced pressure whereby a flow of ions from said solution characteristic of said sample into said second chamber may be maintained.
Preferably the apparatus additionally comprises one or more means for further preparing the ion beam emerging through said orifice into said second chamber for mass analysis said means for further preparing being selected from additional pressure reduction means, focussing means and ion accelerating means.
In accordance with a further aspect of the invention there is also provided a mass spectrometer containing ion beam generating means comprising an apparatus according to the invention.
Thus, in the present invention, the desolvation of the ions formed in the electrospray process takes place in the first chamber by virtue of the flow of gas maintained in that chamber. The solvated ions produced in the electrospray (M+)Sx exist in a dynamic equilibrium with the solvent molecules S and the desolvated ions M+, that is:
(M+)Sx ⃡M+ +xS
and this equilibrium is displaced to the right by removing the solvent molecules S. This is achieved by means of a flowing inert gas, in contrast with prior art electrospray systems (e.g. U.S. Pat. No. 4,209,696) in which the gas is essentially static and consists largely of the solvent vapour. The presence of solvent molecules in the vapour phase clearly displaces the equilibrium to favour the production of the solvated species (M+)Sx. The inert gas employed may be any gas which does not chemically react with the solvent or the expected samples, for example any of the noble gases, e.g., helium, neon, argon, or nitrogen or carbon dioxide. Nitrogen is generally used because it is relatively cheap, but other gases may be used if desired. Preferably the gas should be pure in order to prevent the formation of ions in the electrospray from impurities introduced in the gas, and to prevent the deposition of water and other condensable impurities in the electrospray chamber. The degree of desolvation of the ions can be varied by adjusting the flow of inert gas in the chamber. Substantially complete desolvation can be achieved in most cases with a flow of approximately 200 at.cc.s-1, but the flow rate required is dependent on the nature and the flow rate of the solution and the dimensions of the electrospray chamber. Typically the flow rate will be between 50 and 500 at.cc.s-1 of nitrogen.
The gas should be introduced into the electrospray chamber in a direction opposite to that of the flow from the capillary to ensure the most efficient desolvation of the ions. This can be done by introducing it close to the wall of the electrospray chamber which contains the exit orifice by means of a pipe or pipes which are positioned so that the flow is in the required direction, or by a series of pipes that pass through the wall itself. Alternatively, a perforated diaphragm, containing a number of small holes and a larger hole at its centre to allow the passage of the electrospray jet, may be positioned between the outlet of the capillary and the wall of the chamber, suitably approximately half way between the outlet and the wall, and the gas introduced between the diaphragm and the wall. The gas will then flow in the required direction through the remainder of the chamber from the perforations in the diaphragm. Preferably the outlet pipe of the chamber should be positioned in the end wall of the chamber through which the capillary passes, and should be substantially larger than the inlet pipes or pipes. In general, the outlet pipe should vent at atmospheric pressure so that the electrospray chamber remains substantially at, or slightly greater than, atmospheric pressure.
In many cases, the efficiency of the desolvation process is improved, and the maximum permissible solution flow rate increased, if the inert gas is heated prior to its introduction into the electrospray chamber. The additional heat input to the evaporating drops in the electrospray produced by the use of a heated gas helps to prevent their temperature falling as they evaporate due to the latent heat of vaporisation and consequently improves the efficiency of the solvent removal and increases the maximum solution flow rate which can be used. Any suitable means can be used to heat the gas prior to its introduction. Preferably the temperature of the gas should be between 25� and 100� C., and good results are usually obtained at about 60� C. The temperature should however be adjusted along with the flow rate of inert gas to obtain the desired degree of desolvation for a given solution and solution flow rate.
Although electrospraying of the solution can be effected simply by maintaining a potential difference between the inlet capillary and the opposite wall of the chamber, the process is more controllable if additional electrodes are placed in the electrospray chamber to focus the spray of charged droplets. These electrodes may conveniently take the form of a number (typically two or three) of cylindrical electrodes arranged between the capillary outlet and the first conductive wall and about the axis of the capillary and which form the walls of the electrospray chamber. The potentials applied to these electrodes should preferably be adjusted to maximise the generation of ions oy the electrospray source, and will lie between the capillary and wall potentials.
The electrospray source may be used to generate either positive or negative ions from the dissolved sample. Positive ions are formed if the capillary and focussing electrodes are positive with respect to the end wall, and negative ions are formed if they are negative.
It will be appreciated that a stable electrospray is only obtained for a particular solution under certain conditions of liquid flow rate and electrical potentials. Another feature of the invention is that it has been found that heating the inlet capillary stabilizes the spray in conditions where it would not otherwise be possible to produce a stable electrospray. Heating the inlet capillary therefore allows the use of a wider range of flow rates and potentials in the electrospray chamber, which allows better optimisation of the efficiency of the process and sometimes permits electrospray ionisation to be achieved with a particular solution at flow rates that would otherwise be impossible. Heating the capillary also increases the total ion currents generated by the source when a small bore capillary is used.
It has also been found that the introduction of oxygen in the region of the outlet of the inlet capillary sometimes supresses arcing in the electrospray chamber which occurs under certain conditions, especially when negative ions are being generated. Both these features of the invention will be described later.
Preferably the orifice in the end wall of the electrospray chamber should be formed as a small hole in the centre of a hollow conical frustum which forms part of the end wall and is orientated with its small diameter end facing the inlet capillary. The frustum serves as a skimmer and its use increases the total ion current generated by the source in comparison with that obtained with a planar orifice. The orifice itself should be as large as possible subject to the requirement of maintaining a sufficiently low pressure in the second chamber, and its diameter is therefore dependent on the speed of the vacuum pump used to evacuate the second chamber. Typically the pressure in this chamber should not exceed 10-3 torr.
In general, the potentials applied to the electrospray system will be such that the inlet capillary is at a high positive voltage (for positive ion production), and the ions will emerge with a low kinetic energy, e.g. 25 eV, as described in U.S. Pat. No. 4,209,696. The kinetic energy of the ions is then at a value suitable for their direct introduction into a quadrupole mass spectrometer, which might be positioned in the second evacuated chamber. However, it is more likely that the pressure in this chamber will be too high for the proper operation of the spectrometer, especially if the efficiency of the electrospray ionisation is to be maintained, and it is then desirable to place the spectrometer in a third region of still lower pressure, and use a second nozzle-skimmer arrangement to transmit the ions into the third region. Such arrangements are well known, and are described in U.S. Pat. No. 4,209,696. Additional electrostatic lenses can also be provided in the second and third chambers to optimise the transmission of ions into the mass spectrometer.
When a magnetic sector mass spectrometer is to be used, the inlet capillary of the electrospray source must in general be maintained at a potential at least as high as the accelerating voltage required by the spectrometer in order that they will have sufficient potential energy to permit them to be reaccelerated to the required energy after they emerge into the second chamber with low kinetic energy. The reacceleration can be achieved by provision of a suitable electrostatic lens system in the second chamber, preferably situated close to the orifice, as described in U.S. Pat. No. 4,121,099, but a better method is to use the invention disclosed in our copending U.S. patent application Ser. No. 486,642 entitled "Process and Apparatus for Changing the Energy of Charged Species Entrained in a Flowing Gaseous Medium." This application also discloses means by which the inlet capillary may be operated at ground potential, even when a magnetic sector mass spectrometer is used, and the use of this invention greatly simplifies the construction of efficient lens systems for adjusting the energy of the ions formed in the electrospray to the value required by the mass spectrometer.
The invention will be described in greater detail by reference to the drawings, in which:
FIG. 1 is a simplified sectional view of an electrospray ion source constructed according to the invention;
FIG. 2 is a detailed view of an electrospray inlet capillary suitable for use with the source shown in FIG. 1;
FIG. 3 is a drawing of the liquid jet as it leaves the tip of the inlet capillary under electrospray conditions and
FIGS. 4 and 5 are sectional views of part of an electrospray ion source constructed according to the invention showing alternative means for introducing the inert gas.
Referring first to FIG. 1, the electrospray source itself is mounted on end plate 1 which is supported at the end of tube 2 from flange 3. End plate 1 is also the first conductive wall of the electrospray chamber and contains a small orifice 4 in a conical frustum which also serves as a skimmer 5. The gas tight electrospray chamber itself is formed by four cylinders 6,7,8 and 9, and four externally flanged cylindrical electrodes 11,12, 13 and 14, which largely form its inner surface, all of which are attached to the end plate 1 by means of insulator 10. The cylinders 6-9 are made from an electrically insulating material such as PTFE, so that the conductive electrodes 11-14 are insulated from each other. Alternatively, the electrodes may be fastened together with insulating washers and securing means to obivate the need for the cylinders 6-9. The electrospray capillary 15 is fitted into the end electrode 14 by means of a gas tight union. "O" ring seals are incorporated between the various parts which comprise the electrospray chamber to ensure that it is gas tight.
In the embodiment shown in FIG. 1, a flow of inert gas is introduced into pipe 16 and emerges in the front section of the chamber between the skimmer 5 and a perforated diaphragm 17, which is supported between cylinders 6 and 7. The inert gas flow is directed by the outer surface of the skimmer 5 and the holes in plate 17 to flow through the electrospray chamber in the opposite direction to the flow emerging from capillary 15 to leave via tube 18 which vents at atmospheric pressure, so that the pressure in the electrospray chamber is maintained at, or slightly above, atmospheric pressure. Alternatively, diaphragm 17 may be omitted and inlet pipes 16 may be positioned as shown in FIGS. 4 and 5 to produce an inert gas flow substantially counter to the flow from capillary 15. Conveniently, the flow of inert gas is controlled using flow varying means 40 to control the degree of desolvation of the ions electrosprayed from capillary 15. The source may also be provided with heating means 41 for heating the inert gas before it enters the electrospray chamber.
Flange 3, which supports the electrospray chamber, is attached to vacuum housing 19 by means of insulator 20, and a vacuum pump is attached to port 21. This pump is typically a diffusion pump with a speed of aproximately 1000 l. s-1, but any pump capable of maintaining the pressure in housing 19 at less than 10-3 torr in the presence of the gas flow through orifice 4 can be used. Flange 3 is insulated from housing 19 in order that a variable electrical potential can be applied to the end plate 1 and skimmer 5 when housing 19 is at earth potential.
A second conical skimmer 22 containing an orifice 23 is positioned on the end wall of the housing 19 on a flange 24 which forms part of the second vacuum housing 26. The plate carrying the skimmer 22 is insulated from flange 24 by insulator 25, and all the joints between the flanges are made in a vacuum tight manner. Housing 26 is pumped by any suitable high vacuum pump such as a diffusion pump or a turbomolecular pump through port 27, so that a pressure of better than 10-4 torr, or preferably 10-5 torr, is maintained in housing 26. A series of electrostatic lens elements 28 serves to focus the ion beam emerging through orifice 23 into mass spectrometer 29, which in this case is a conventional quadrupole mass filter.
In the description given above, and in FIG. 1, many of the details concerning the construction of the vacuum seals and flanges, and the means used to fasten them together, incorporating electrical insulation where necessary, have been omitted. Such details are standard features of vacuum systems and are well known.
The electrospray capillary assembly 15 is illustrated in greater detail in FIG. 2. The needle 30 should be a good electrical conductor and be chemically inert, and is preferably made from stainless steel. Optimum performance is obtained when the tip 31 is ground into a sharp cone with steep sides. The inside diameter is preferably within the range 0.025 mm to 0.100 mm, and the outside diameter should be large enough to ensure mechanical rigidity. Hypodermic syringe needles can frequently be used for needle 30. The end of the needle may be heated by passing an electrical current through it by means of wire 32, which is silver soldered to the needle close to the tip at point 33. It has been found that heating the capillary can improve the stability of the electrospray, as explained above. It has also been found that the addition of oxygen close to the tip of the capillary can reduce the tendency for arcing in the electrospray chamber to occur, which is particularly troublesome in the negative ion mode. The oxygen can conveniently be supplied by enclosing the needle 30 in a larger bore open ended tube 34 and supplying oxygen through a side arm 35. Neither the heating or the addition of oxygen are essential, however, and both these features can be omitted if desired.
The capillary assembly 15 is secured by a union in the end electrode 14 of the electrospray chamber, and is maintained at a high potential relative to the end plate 1 of the chamber. The end plate 1 is supported on tube 2 from flange 3 which is insulated from the housing 19 so that it can be used as an electrostaic focussing element with a relatively low potential applied to it. Electrodes 11, 12 and 13 complete the inside surface of the electrospray chamber, as explained, and different variable potentials may be applied to the electrodes to optimise the electrospray process. The gap between the electrodes should be kept as small as possible in order to ensure that the walls of the chamber are electrically conducting over the maximum possible area so that the build up of electrical charge on insulated sections of the wall, which might otherwise destabilise the electrospray, is minimized.
As required by the invention, the optionally heated inert gas is introduced through pipe 16 into the space between end plate 1 and a perforated diaphragm 17, which also serves as a focussing electrode. Diaphragm 17 serves to direct the gas flow counter to the flow from the capillary outlet, and has a larger hole at its centre through which the electrospray jet passes. It is positioned midway between the end of capillary 15 and end plate 1. The surfaces and edges of all the electrodes in the electrospray chamber should be well polished to minimise the possibility of sparking. It will be appreciated that it is not essential to split the wall of the electrospray chamber in as many places as described, and in many cases satisfactory performance can be obtained with fewer electrodes. For example, electrodes 12 and 13 could be connected together, or made in one piece, as could electrode 11 and plate 17.
In order to produce positive ions, capillary 15 is maintained at a high positive potential relative to end plate 1, and the other electrodes at appropriate intermediate potentials, adjusted to optimise performance. Negative ions can be produced by changing the polarity of all the electrodes.
It has been found that there are two modes of electrospraying, dependent on the potential difference between the capillary and the end plate. If this is between 3 and 6 KV, the "low voltage" mode is observed, and the ionisation of the sample is very gentle resulting in very little, if any, fragmentation of the sample. If the potential difference is greater than 6 KV, a corona discharge is formed in the chamber, and the ionisation is more energetic, resulting in some fragmentation of the sample ions. Usually the source is operated at the upper end of the "low voltage" mode in order to achieve maximum sensitivity with minimum fragmentation. The electrospray is only stable given certain geometrical arrangements and potentials. The best results are generally obtained with the capillary 1.5 cm from orifice 4 in skimmer 5, but the optimum distance is dependent on the potentials, and the flow rate and composition of the solution, as well as the flow of inert gas. It is an advantage to make this distance variable. In all cases the diameter of the electrospray chamber should be at least twice the capillary to end plate distance, and preferably larger, but if the chamber is too large, the quantity of inert gas required becomes too great.
The voltages applied to the focussing electrodes are adjusted to optimum ion generation in the electrospray and to ensure a stable spray. Typical conditions for the generation of positive ions are:
where the subscript numbers refer to the electrodes identified by those numbers in FIG. 1, and the voltages are all relative to the housing 19 which is at earth potential.
The conditions over which stable electrospraying can be achieved can be extended by heating the capillary by passing an electrical current through it, as described. FIG. 3 is a drawing of the electrospray jet as it leaves the tip 31 of the needle 30. It consists of a small cone of liquid 37 which extends through the distance 36 to point 38. The true electrospray jet 39 starts at point 36, as shown. It is thought that heating the capillary reduces distance 36, which results in the formation of stable jets over a wider range of capillary internal diameters and flow rates, etc. It also results in greater ion currents when very small bore capillaries are used, and improves the overall efficiency of the process. However, it is possible that some thermal degradation of sensitive samples might be caused by heating the capillary.
There are some limitations on the flow rate and composition of the solution if successful electrospraying is to be obtained. Flow rates between 1 and 100 microliters per minute can be used, but in general the best results are obtained at the lowest flow rate. The solvent should be such that there is either electron or proton affininty with the expected samples, and it should be polar and preferably contain an ionic solute in addition to the sample. These requirements mean that when used as a liquid chromatograph--mass spectrometer interface, the electrospray source works well with highly polar solvents such as water or methanol, especially when the solvent contains additional dissolved salts such as buffers, in contrast with most other forms of LC-MS interfaces.
The inert gas introduced into the source should preferably consist of a fairly pure gas such as nitrogen, helium, or argon, etc. As explained, the degree of desolvation of the ions can be varied by adjusting the flow rate of the gas and its temperature. The flow rate required for a particular degree of desolvation is of course dependent on the liquid flow, and typically a flow of 200 at.cc.s-1 is used for a liquid flow rate of 10 microliters per minute, at a temperature of 60� C. to ensure complete desolvation. Flow rates from 50 to 500 at.cc.s-1 and temperatures from 25� C. to 100� C. can be used, however.
The ions generated in the electrospray process pass through orifice 4 in skimmer 5 into an evacuated region in housing 19 where the pressure is maintained at 10-3 torr or less by a vacuum pump connected to port 21. The electrical potential applied to end plate 1 is adjusted to obtain optimum transmission of the ions. Additional electrostatic lens elements can be provided behind orifice 4 for additional focussing of the ion beam if desired. Alternatively, means can be provided in this region to accelerate the ions to another kinetic energy, as described previously.
The focussed ion beam then passes through an orifice in a second skimmer 23 into a region of still higher vacuum which contains the mass spectrometer 29. Unless means are provided for reaccelerating the ions, this must be a mass filter capable of accepting ions with very low energies, such as a quadrupole analyser. The potential on plate 22 which supports the skimmer is also made variable to assist in focussing the ion beam, and more electrostatic lenses 28 are provided to ensure maximum efficiency of transmission of ions into the spectrometer. Typically, for a quadrupole analyser, plate 22 and the first element of lens 28 will be maintained at about +25 V, for the conditions given previously. This potential determines the energy with which the ions enter the spectrometer. The second skimmer is provided to allow a higher vacuum to be maintained in the region of the mass spectrometer whilst maintaining an efficient transfer of ions, and devices of this type are well known.
It will be appreciated that the source described above is intended by way of example only, and that it is possible to construct an electrospray source incorporating the invention in a variety of alternative ways, which will be apparent to those skilled in the art.
|Cited Patent
|Filing date
|Publication date
|Apr 25, 1977
|Oct 17, 1978
|The Governing Council Of The University Of Toronto
|Method and apparatus for focussing and declustering trace ions
|Feb 26, 1979
|Jun 24, 1980
|Fite Wade L
|Methods and apparatus for mass spectrometric analysis of constituents in liquids
|May 7, 1980
|Nov 10, 1981
|Iribarne Julio V
|Method and apparatus for the analysis of chemical compounds in aqueous solution by mass spectroscopy of evaporating ions
|"1st Workshop on LC/MS," Published in Journal of Chromatography, 1982, vol. 251, pp. 91-225, (Montreux, Oct. 1981).
|1st Workshop on LC/MS, Published in Journal of Chromatography, 1982, vol. 251, pp. 91 225, (Montreux, Oct. 1981).
|Adv. in Chem., (1973), vol. 125, p. 73 Dole, M., Cox, H. L., Jr., Gleniec, J.
|Adv. in Chem., (1973), vol. 125, p. 73-Dole, M., Cox, H. L., Jr., Gleniec, J.
|Analytical Chemistry, 1979, vol. 51, pp. 682A 701A, Arpino, P. J. and Guichon, G.
|Analytical Chemistry, 1979, vol. 51, pp. 682A-701A, Arpino, P. J. and Guichon, G.
|Biopolymers, 1971, vol. 10, pp. 821 826 Clegg, G. A. and Dole, M.
|Biopolymers, 1971, vol. 10, pp. 821-826-Clegg, G. A. and Dole, M.
|Journal of Chemical Physics, (1968), vol. 49, No. 5, pp. 2240 2249, Dole, M., Mack, L. L., Hines, R. L., Mobley, R. C., Ferguson, L. D.
|Journal of Chemical Physics, (1968), vol. 49, No. 5, pp. 2240-2249, Dole, M., Mack, L. L., Hines, R. L., Mobley, R. C., Ferguson, L. D.
|Journal of Electrostatics, 1978, vol. 5, p. 411, Stimson, B. A. and Evans, C. A., Jr.
|The Journal of Chemical Physics, vol. 52, (10), 1970, 4977 4986.
|The Journal of Chemical Physics, vol. 52, (10), 1970, 4977-4986.
|The Journal of Physical Chemistry, vol. 82, No. 6, (1978), p. 660, Stimpson, B. A. and Evans, C. A., Jr.
|Citing Patent
|Filing date
|Publication date
|Apr 17, 1985
|May 19, 1987
|Lagna William M
|Methods and apparatus for mass spectrometric analysis of fluids
|Sep 11, 1985
|Aug 18, 1987
|Georgia Tech Research Corporation
|Monodisperse aerosol generator
|Feb 24, 1986
|Mar 8, 1988
|Research Corporation
|Ion vapor source for mass spectrometry of liquids
|Mar 19, 1986
|Aug 9, 1988
|Georgia Tech Research Corporation
|Monodisperse aerosol generator
|Mar 25, 1987
|Mar 21, 1989
|Research Corporation
|Method and means for vaporizing liquids for detection or analysis
|Apr 6, 1987
|Jun 27, 1989
|Battelle Memorial Institute
|Combined electrophoretic-separation and electrospray method and system
|Sep 30, 1987
|Aug 29, 1989
|Cornell Research Foundation, Inc.
|Ion spray apparatus and method
|Mar 2, 1988
|Aug 29, 1989
|Research Corporation Technologies, Inc.
|Ion vapor source for mass spectrometry of liquids
|May 27, 1988
|Sep 5, 1989
|Interface for liquid chromatography-mass spectrometry systems
|Dec 16, 1988
|Nov 28, 1989
|Vestec Corporation
|Interface for coupling liquid chromatography to solid or gas phase detectors
|Aug 8, 1988
|May 8, 1990
|Georgia Tech Rss. Corp
|Monodisperse aerosol generator for use with infrared spectrometry
|Mar 20, 1989
|Oct 2, 1990
|Research Corporation Technologies
|Method and means for vaporizing liquids by means of heating a sample capillary tube for detection or analysis
|Aug 14, 1989
|Nov 6, 1990
|Extrel Corporation
|Method and apparatus for introduction of liquid effluent into mass spectrometer and other gas-phase or particle detectors
|Feb 19, 1988
|Dec 18, 1990
|Extrel Corporation
|Method and apparatus for introduction of fluid streams into mass spectrometers and other gas phase detectors
|Apr 24, 1990
|Mar 12, 1991
|Vestec Corporation
|Electrospray ionization interface and method for mass spectrometry
|Jun 28, 1989
|Apr 23, 1991
|Monsanto Company
|Apparatus for carrying out catalyzed chemical reactions and for studying catalysis
|Jun 1, 1990
|May 14, 1991
|Vestec Corporation
|Electrospray method for mass spectrometry
|Mar 15, 1989
|Nov 26, 1991
|Gte Products Corporation
|Manifold and method of batch measurement of Hg-196 concentration using a mass spectrometer
|May 18, 1990
|Jan 28, 1992
|Martin Marietta Energy Systems, Inc.
|Gas chromatograph-mass spectrometer (gc/ms) system for quantitative analysis of reactive chemical compounds
|Apr 26, 1989
|Apr 7, 1992
|Hitachi, Ltd.
|Mass spectrometer
|May 15, 1991
|May 19, 1992
|The University Of North Carolina At Chapel Hill
|Microelectrospray method and apparatus
|May 17, 1991
|Oct 20, 1992
|Finnian Corporation
|Method and apparatus for focusing ions in viscous flow jet expansion region of an electrospray apparatus
|Jan 25, 1991
|Nov 10, 1992
|Finnigan Corporation
|Method and apparatus for multi-stage particle separation with gas addition for a mass spectrometer
|Dec 20, 1991
|Dec 8, 1992
|Finnigan Corporation
|Electrospray ion source and interface apparatus and method
|Oct 8, 1991
|Jun 29, 1993
|Hewlett-Packard Company
|Apparatus for interfacing liquid chromatography-mass spectrometry systems
|Apr 14, 1992
|Jun 29, 1993
|Millipore Corporation
|Insulated needle for forming an electrospray
|Jan 24, 1992
|Aug 10, 1993
|Finnigan Mat, Inc.
|Probe-based electrospray adapter for thermospray equipped quadrupole based LC/MS systems
|May 6, 1992
|Sep 14, 1993
|The Rockefeller University
|Electrospray ion source for mass spectrometry
|Sep 30, 1991
|Sep 28, 1993
|Tsi Incorporated
|Electrospray apparatus for producing uniform submicrometer droplets
|Jan 25, 1993
|Nov 30, 1993
|General Electric Company
|Apparatus for interfacing liquid chromatograph with magnetic sector spectrometer
|Feb 2, 1993
|Mar 29, 1994
|Hitachi, Ltd.
|Mass spectrometer
|May 24, 1993
|Oct 25, 1994
|Hewlett-Packard Company
|Mass spectrometry with gas counterflow for particle beam
|Jan 27, 1993
|Dec 13, 1994
|Bruker-Franzen Analytik Gmbh
|Method and device for the mass-spectrometric examination of fast organic ions
|Sep 15, 1992
|Feb 28, 1995
|Finnigan Corporation
|Electrospray ion source and interface apparatus and method
|Mar 30, 1994
|Aug 8, 1995
|Labowsky; Michael J.
|Method for eliminating noise and artifact peaks in the deconvolution of multiply charged mass spectra
|Oct 19, 1994
|Sep 5, 1995
|Hewlett-Packard Company
|Method for interfacing liquid chromatography-mass spectrometry systems
|Nov 4, 1993
|Apr 2, 1996
|Hv Ops, Inc. (H-Nu)
|Electrospray ionization source and method for mass spectrometric analysis
|Apr 20, 1995
|Oct 1, 1996
|The United States Of America As Represented By The Administrator Of The National Aeronautics And Space Administration
|Analysis of supercritical-extracted chelated metal ions from mixed organic-inorganic samples
|Feb 4, 1994
|Oct 22, 1996
|Hitachi, Ltd.
|Mass spectrometer
|Sep 21, 1994
|Dec 3, 1996
|Fenn; John B.
|Method for determining molecular weight using multiply charged ions
|Mar 10, 1995
|Mar 4, 1997
|Bruker-Franzen Analytik Gmbh
|Electrospraying method for mass spectrometric analysis
|Feb 3, 1997
|May 12, 1998
|Hewlett-Packard Company
|Orthogonal ion sampling for APCI mass spectrometry
|Mar 3, 1997
|Oct 27, 1998
|Waters Investments Limited
|Ionization electrospray apparatus for mass spectrometry
|Sep 27, 1996
|Nov 17, 1998
|Hewlett-Packard Company
|Ionization chamber and mass spectrometry system containing an asymmetric electrode
|Jan 31, 1997
|Dec 1, 1998
|Whitehouse; Craig M.
|Electrospray and atmospheric pressure chemical ionization sources
|Sep 27, 1996
|Feb 9, 1999
|Hewlett-Packard Company
|Apparatus for forming liquid droplets having a mechanically fixed inner microtube
|Oct 14, 1997
|Feb 9, 1999
|University Of Washington
|Device and method for forming ions
|Dec 1, 1998
|Mar 7, 2000
|Hewlett-Packard Company
|Apparatus for forming liquid droplets having a mechanically fixed inner microtube
|Dec 10, 1997
|May 9, 2000
|Analytica Of Branford, Inc.
|Electrospray and atmospheric pressure chemical ionization sources
|Jan 17, 1997
|May 30, 2000
|Northeastern University
|Subatmospheric, variable pressure sample delivery chamber for electrospray ionization/mass spectrometry and other applications
|Feb 9, 1999
|Aug 22, 2000
|The University Of Washington
|Device and method for forming ions
|Jun 3, 1998
|Aug 22, 2000
|Battelle Memorial Institute
|Method and apparatus for directing ions and other charged particles generated at near atmospheric pressures into a region under vacuum
|Jul 29, 1997
|Sep 12, 2000
|Analytica Of Branford, Inc.
|Multiply charged ions and method for determining the molecular weight of large molecules
|Sep 28, 1998
|Jan 23, 2001
|Varian, Inc.
|Vortex gas flow interface for electrospray mass spectrometry
|Nov 12, 1998
|Aug 21, 2001
|Waters Investments Limited
|Electrospray for chemical analysis
|Jun 17, 1999
|Jun 25, 2002
|Micromass Limited
|Multi-inlet mass spectrometer for analysis of liquid samples by electrospray or atmospheric pressure ionization
|Dec 21, 2000
|Jul 9, 2002
|Kionix, Inc.
|Integrated monolithic microfabricated electrospray and liquid chromatography system and method
|Jun 14, 2000
|Oct 7, 2003
|Syagen Technology
|Atmospheric pressure photoionizer for mass spectrometry
|Jun 8, 2001
|Jun 1, 2004
|Edward W Sheehan
|Apparatus and method for focusing ions and charged particles at atmospheric pressure
|Apr 10, 2001
|Aug 17, 2004
|Science & Engineering Services, Inc.
|Time-of-flight/ion trap mass spectrometer, a method, and a computer program product to use the same
|May 31, 2003
|Nov 16, 2004
|Edward W. Sheehan
|Laminated lens for focusing ions from atmospheric pressure
|Feb 22, 2002
|Nov 30, 2004
|Agilent Technologies, Inc.
|Apparatus and method for ion production enhancement
|Apr 29, 2002
|Feb 22, 2005
|Agilent Technologies, Inc.
|Target support and method for ion production enhancement
|Apr 18, 2001
|Mar 1, 2005
|Kang Ho Ahn
|Apparatus for manufacturing ultra-fine particles using electrospray device and method thereof
|May 30, 2003
|May 3, 2005
|Edward W Sheehan
|Remote reagent chemical ionization source
|Sep 12, 2003
|Sep 27, 2005
|Edward William Sheehan
|Laminated lens for introducing gas-phase ions into the vacuum systems of mass spectrometers
|Jun 15, 2001
|Oct 18, 2005
|Bruker Daltonics, Inc.
|Means and method for guiding ions in a mass spectrometer
|Jul 9, 2004
|Mar 21, 2006
|Purdue Research Foundation
|Electrosonic spray ionization method and device for the atmospheric ionization of molecules
|Apr 7, 1999
|May 16, 2006
|Sproch Norman K
|Method for the characterization of the three-dimensional structure of proteins employing mass spectrometric analysis and computational feedback modeling
|Oct 15, 2004
|Jul 18, 2006
|Agilent Technologies, Inc.
|Apparatus and method for ion production enhancement
|Nov 15, 2004
|Jul 25, 2006
|Ross Clark Willoughby
|Laminated lens for focusing ions from atmospheric pressure
|Jun 9, 2004
|Aug 15, 2006
|Ionica Mass Spectrometry Group, Inc.
|Mass spectrometer interface
|Oct 15, 2004
|Aug 15, 2006
|Agilent Technologies, Inc.
|Apparatus and method for ion production enhancement
|May 2, 2005
|Aug 22, 2006
|Chem-Space Associates, Inc.
|Remote reagent chemical ionization source
|Feb 13, 2004
|Aug 29, 2006
|Mds Sciex
|Atmospheric pressure charged particle discriminator for mass spectrometry
|Dec 16, 2004
|Nov 7, 2006
|Agilent Technologies, Inc.
|Apparatus and method for ion production enhancement
|Jan 21, 2005
|Nov 14, 2006
|Agilent Technologies, Inc.
|Apparatus and method for ion production enhancement
|Dec 17, 2004
|Dec 5, 2006
|Varian, Inc.
|Atmospheric pressure ionization with optimized drying gas flow
|May 13, 2005
|Apr 3, 2007
|Thermo Finnigan Llc
|Electrospray ion source apparatus
|Feb 25, 2005
|Mar 25, 2008
|Kang Ho Ahn
|Apparatus for manufacturing ultra-fine particles using electrospray device and method thereof
|Jun 16, 2005
|May 13, 2008
|Agilent Technologies, Inc.
|Apparatus and method for ion production enhancement
|Apr 18, 2006
|Jul 29, 2008
|Ionics Mass Spectrometry Group, Inc.
|Mass spectrometer interface
|Jun 6, 2006
|Dec 9, 2008
|Mds Sciex
|Atmospheric pressure charged particle discriminator for mass spectrometry
|Nov 1, 2005
|Dec 16, 2008
|Bruker Daltonik, Gmbh
|Ionization by droplet impact
|Jun 19, 2006
|Aug 4, 2009
|Science Applications International Corporation
|Sample tube holder
|Oct 7, 2006
|Aug 4, 2009
|Science Applications International Corporation
|Remote reagent ion generator
|Nov 8, 2006
|Aug 18, 2009
|Science Applications International Corporation
|Non-contact detector system with plasma ion source
|Dec 3, 2007
|Sep 8, 2009
|Science Applications International Corporation
|Method and device for non-contact sampling and detection
|Jan 12, 2007
|Mar 30, 2010
|Ionics Mass Spectrometry Group
|High sensitivity mass spectrometer interface for multiple ion sources
|Jun 4, 2007
|Aug 31, 2010
|Microsaic Systems Limited
|Microengineered vacuum interface for an ionization system
|May 20, 2008
|Oct 19, 2010
|Chem-Space Associates, Inc.
|Laser desorption ion source
|Apr 30, 2007
|Jan 11, 2011
|Ionics Mass Spectrometry Group Inc.
|Mass spectrometer ion guide providing axial field, and method
|May 2, 2007
|Feb 1, 2011
|Dh Technologies Development Pte. Ltd
|Multipole mass filter having improved mass resolution
|Jan 18, 2008
|Jun 14, 2011
|Chem-Space Associates, Inc.
|Field-free electrospray nebulizer
|Dec 29, 2008
|Aug 30, 2011
|Science Applications International Corporation
|Ion transfer device
|Feb 12, 2010
|Aug 30, 2011
|Ionics Mass Spectrometry Group
|High sensitivity mass spectrometer interface for multiple ion sources
|Mar 10, 2009
|Dec 6, 2011
|Science Applications International Corporation
|Soft chemical ionization source
|May 16, 2008
|Feb 28, 2012
|Science Applications International Corporation
|Method and means for precision mixing
|May 19, 2009
|May 8, 2012
|Battelle Memorial Institute
|Low pressure electrospray ionization system and process for effective transmission of ions
|Dec 28, 2007
|Oct 16, 2012
|Griffin Analytical Technologies, Llc
|Analytical instruments, assemblies, and methods
|Jan 31, 2012
|Nov 13, 2012
|Science Applications International Corporation
|Method and means for precision mixing
|Sep 4, 2009
|Mar 5, 2013
|Seagate Technology Llc
|Deposition of lubricant onto magnetic media
|Jun 8, 2012
|Jun 11, 2013
|Sociedad Europea de Analisis Diferencial de Movilidad
|Ionizer for vapor analysis decoupling the ionization region from the analyzer
|Jun 27, 2008
|Oct 1, 2013
|Ionics Mass Spectrometry Group, Inc.
|Mass spectrometer interface
|Oct 12, 2010
|Dec 23, 2014
|Perkinelmer Health Sciences, Inc.
|Assemblies for ion and electron sources and methods of use
|Aug 29, 2013
|Feb 3, 2015
|Ionics Mass Spectrometry Group, Inc.
|Mass spectrometer interface
|Dec 21, 2014
|Feb 16, 2016
|Perkinelmer Health Sciences, Inc.
|Assemblies for ion and electron sources and methods of use
|Sep 7, 2012
|Sep 13, 2016
|Waters Technologies Corporation
|Techniques for automated performance maintenance testing and reporting for analytical instruments
|Dec 17, 2014
|Sep 20, 2016
|Perkinelmer Health Sciences Canada, Inc.
|Mass spectrometer interface
|Mar 3, 2009
|Nov 15, 2016
|Waters Technologies Corporation
|Interfacing with a digital microfluidic device
|Apr 10, 2001
|Oct 10, 2002
|Science & Engineering Services, Inc.
|Time-of-flight/ion trap mass spectrometer, a method, and a computer program product to use the same
|Apr 18, 2001
|Oct 31, 2002
|Ahn Kang Ho
|Apparatus for manufacturing ultra-fine particles using electrospray device and method thereof
|Sep 4, 2002
|Mar 20, 2003
|Yukimitsu Suzuki
|Arc welding quality evaluation apparatus
|Feb 22, 2002
|Aug 28, 2003
|Jean-Luc Truche
|Apparatus and method for ion production enhancement
|Apr 29, 2002
|Aug 28, 2003
|Jean-Luc Truche
|Target support and method for ion production enhancement
|Jun 15, 2001
|Aug 5, 2004
|Park Melvin A.
|Means and method for guiding ions in a mass spectrometer
|Feb 13, 2004
|Nov 4, 2004
|Mds Sciex
|Atmospheric pressure charged particle discriminator for mass spectrometry
|Jul 9, 2004
|Feb 10, 2005
|Zoltan Takats
|Electrosonic spray ionization method and device for the atmospheric ionization of molecules
|Jun 9, 2004
|Feb 17, 2005
|Charles Jolliffe
|Mass spectrometer interface
|Oct 15, 2004
|Apr 7, 2005
|Jean-Luc Truche
|Apparatus and method for ion production enhancement
|Oct 15, 2004
|Apr 14, 2005
|Jean-Luc Truche
|Apparatus and method for ion production enhancement
|Dec 6, 2004
|May 12, 2005
|Jean-Luc Truche
|Target support and method for ion production enhancement
|Feb 25, 2005
|Jun 30, 2005
|Ahn Kang H.
|Apparatus for manufacturing ultra-fine particles using electrospray device and method thereof
|Dec 16, 2004
|Jul 14, 2005
|Jean-Luc Truche
|Apparatus and method for ion production enhancement
|Dec 16, 2004
|Jul 14, 2005
|Jean-Luc Truche
|Apparatus and method for ion production enhancement
|Jan 21, 2005
|Jul 28, 2005
|Jean-Luc Truche
|Apparatus and method for ion production enhancement
|May 13, 2005
|Nov 24, 2005
|Thakur Rohan A
|Electrospray ion source apparatus
|Jun 16, 2005
|Dec 15, 2005
|Joyce Timothy H
|Apparatus and method for ion production enhancement
|Nov 1, 2005
|May 25, 2006
|Bruker Daltonik Gmbh
|Ionization by droplet impact
|Jan 12, 2006
|Jun 8, 2006
|Mds Sciex
|Atmospheric pressure charged particle discriminator for mass spectrometry
|Dec 17, 2004
|Jun 22, 2006
|Varian, Inc.
|Atmospheric pressure ionization with optimized drying gas flow
|Apr 18, 2006
|Aug 24, 2006
|Ionics Mass Spectometry Group, Inc.
|Mass spectrometer interface
|May 5, 2006
|Dec 7, 2006
|Sproch Norman K
|Method for the characterization of the three-dimensional structure of proteins employing mass spectrometric analysis and computational feedback modeling
|Nov 8, 2006
|May 24, 2007
|Karpetsky Timothy P
|Non-contact detector system with plasma ion source
|Jan 12, 2007
|Aug 9, 2007
|Charles Jolliffe
|High sensitivity mass spectrometer interface for multiple ion sources
|Jun 4, 2007
|Jan 3, 2008
|Richard Syms
|Microengineered vacuum interface for an ionization system
|Jun 27, 2008
|Oct 23, 2008
|Ionics Mass Spectrometry Group, Inc.
|Mass spectrometer interface
|May 19, 2009
|Oct 1, 2009
|Battelle Memorial Institute
|Low pressure electrospray ionization system and process for effective transmission of ions
|Nov 19, 2009
|Jun 24, 2010
|Roth Michael J
|Analytical Instruments, Assemblies, and Methods
|Mar 30, 2010
|Jul 22, 2010
|H-Nu Ops, Inc.
|General purpose experimental/computational analytical system
|Sep 4, 2009
|Mar 10, 2011
|Seagate Technology Llc
|Deposition of lubricant onto magnetic media
|Oct 12, 2010
|Apr 21, 2011
|Keith Ferrara
|Assemblies for ion and electron sources and methods of use
|Mar 3, 2009
|May 12, 2011
|Waters Technologies Corporation
|Interfacing With A Digital Microfluidic Device
|Dec 2, 2010
|Jun 9, 2011
|Lisa Cousins
|Mass spectrometer ion guide providing axial field, and method
|Mar 13, 2012
|Sep 13, 2012
|Excellims Corporation
|Apparatus and method for ion mobility spectrometry and sample introduction
|Jun 8, 2012
|Oct 25, 2012
|Sociedad Europea De Analisis Diferencial De Movili
|Ionizer for vapor analysis decoupling the ionization region from the analyzer
|Dec 26, 2011
|Oct 31, 2013
|Bio Chromato, Inc.
|Mass spectrometry method, mass spectrometer, and mass spectrometry system
|Jan 31, 1997
|Oct 3, 2000
|Agilent Technologies
|Orthogonal ion sampling for electrospray .[.LC/MS.]. mass spectrometry
|Jun 9, 1989
|Dec 14, 1989
|Hitachi Ltd
|Mar 10, 1994
|Sep 14, 1995
|Bruker Franzen Analytik Gmbh
|Verfahren zur Ionisierung von gel�sten Atomen oder Molek�len aus Fl�ssigkeiten durch elektrisches Verspr�hen
|Nov 25, 1994
|Sep 14, 1995
|Bruker Franzen Analytik Gmbh
|Schaltbare Elektrospr�h-Ionisierung f�r getaktet arbeitende Massenspektrometer
|Apr 25, 1997
|Oct 29, 1998
|Bergmann Thorald
|Orthogonal ion accelerator for time of flight mass spectrometer
|Sep 10, 1997
|Jun 30, 1999
|Analytica Of Branford, Inc.
|Improvements to atmospheric pressure ion sources
|Sep 10, 1997
|Nov 10, 2010
|PerkinElmer Health Sciences, Inc.
|Improvements to atmospheric pressure ion sources
|Sep 15, 2000
|Apr 18, 2001
|Hitachi, Ltd.
|Mass spectrometer, mass spectrometry, and monitoring system
|Sep 15, 2000
|May 3, 2006
|Hitachi, Ltd.
|Mass spectrometer, mass spectrometry, and monitoring system
|Mar 25, 2010
|Oct 6, 2010
|Vidal de Miguel, Guillermo
|Method and apparatus to produce steady beams of mobility selected ions via time-dependent electric fields
|Sep 15, 2000
|Mar 16, 2011
|Hitachi, Ltd.
|Ion source, mass spectrometer, detector and monitoring system
|May 21, 1992
|Nov 26, 1992
|Analytica Of Brandford, Inc.
|Method and apparatus for improving electrospray ionization of solute species
|Jan 9, 1995
|Jul 20, 1995
|Mds Health Group Limited
|Ion spray with intersecting flow
|Jan 9, 1995
|Oct 26, 1995
|Joseph F Anacleto
|Ion spray with intersecting flow
|Mar 7, 1995
|Sep 14, 1995
|Analytica Of Branford, Inc.
|Improvements to electrospray and atmospheric pressure chemical ionization sources
|Nov 26, 1996
|Jun 4, 1998
|Anglo American Research Laboratories (Proprietary) Limited
|May 17, 1997
|Nov 26, 1998
|Analytica Of Branford, Inc.
|Configuration of an atmospheric pressure ion source
|Jun 4, 1999
|Dec 9, 1999
|Advanced Cardiovascular Systems, Inc.
|Method and apparatus for concentrating a solute in solution with a solvent
|Feb 22, 2000
|Aug 31, 2000
|Isis Innovation Limited
|Improvements in or relating to microfluidic sample preparation and mass spectrometry
|Feb 22, 2000
|Dec 7, 2000
|Isis Innovation
|Improvements in or relating to microfluidic sample preparation and mass spectrometry
|Jun 14, 2001
|Dec 20, 2001
|Syagen Technology
|Atmospheric pressure photoionizer for mass spectrometry
|Jan 28, 2003
|Feb 5, 2004
|Agilent Technologies Inc
|Target support and method for ion production enhancement
|Jan 12, 2007
|Jul 19, 2007
|Ionics Mass Spectrometry Group
|High sensitivity mass spectrometer interface for multiple ion sources
|Jan 13, 2010
|Jul 22, 2010
|Sociedad Europea De An�lisis Diferencial De Movilidad, S.L.
|Improved ionizer for vapor analysis decoupling the ionization region from the analyzer
|Sep 7, 2012
|Mar 21, 2013
|Waters Technologies Corporation
|Techniques for automated performance maintenance testing and reporting for analytical instruments
|U.S. Classification
|250/288, 250/282, 250/286, 250/281
|International Classification
|Cooperative Classification
|European Classification
|Dec 22, 1983
Owner name: YALE UNIVERSITY, NEW HAVEN, CT., A NON-PROFIT CT C
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST.;ASSIGNORS:LABOWSKY, MICHAEL J.;FENN, JOHN B.;YAMASHITA, MASAMICHI;REEL/FRAME:004204/0258;SIGNING DATES FROM 19830508 TO 19830716
Owner name: YALE UNIVERSITY,, CONNECTICUT
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:LABOWSKY, MICHAEL J.;FENN, JOHN B.;YAMASHITA, MASAMICHI;SIGNING DATES FROM 19830508 TO 19830716;REEL/FRAME:004204/0258
|Dec 27, 1983
Owner name: BOEING COMPANY THE SEATTLE WASHINGTON A DE CORP
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST.;ASSIGNORS:GRATZER, LOUIS B.;GILLETTE, WALTER B.;REEL/FRAME:004204/0485
Effective date: 19831220
|Mar 18, 1986
|Certificate of correction
|Jan 23, 1989
|Fee payment
Year of fee payment: 4
|Feb 12, 1993
|Surcharge for late payment
|Feb 12, 1993
|Fee payment
Year of fee payment: 8
|Oct 4, 1996
|Fee payment
Year of fee payment: 12


text:
Rated 4.1 out of 5 by 7 reviewers.
Rated 3 out of 5 by Catherine77 An excellent service An excellent product but still more exzpensive than other retailers despite the half price. I didn't realise this was 330ml rather than the litre cartons. 21 December 2016
Rated 5 out of 5 by cferro Delicious and came next day The title says it's all. Drinks are delicious and I order on Sunday afternoon and it can Monday morning. Stock up! 28 June 2016
Rated 5 out of 5 by Gracie85 Amazing product! Couldn't get enough of this drink, so much so that I've ordered another 2! Great tasting and very refreshing - curbs the sugar cravings too. 26 March 2014
Rated 2 out of 5 by ljbeauchamp I love coconut water and have had this brand many times but unfortunately the carton I received had been leaking upon arrival so think it must have gone of as it tasted a bit rancid and bits floating in it which I have never had before. So had to throw away! 04 March 2014
Rated 4 out of 5 by Merrymommy Good quality as with all your goods. Unusual taste at first, but it is refreshing. I also find that it is better in small quantities so that the pineapple does not separate. I bought two cartons to try and will buy again. 13 February 2013
Rated 5 out of 5 by rmcs Would definintely recommend! Great taste, really hydrating; would recommend to the busy, sporting and thirsty!! 05 September 2012
Rated 5 out of 5 by Squatlover Tastes like a Pina Colada This tastes incredible! Really nice when drank cold after training. 16 July 2012
1-7 of 7

subset: commoncrawl


text:
You can count on your nearby Nordstrom department store when it comes to Adidas kids' shoes in Oakdale, Pennsylvania. Our selection is always available for anytime, anywhere online shopping—whether you're at work, on the go or waiting for the kids to come home from school. Once you've selected the pairs you like best, you can purchase them online and get them shipped to you for free.
Or you can view what's available at your local store in Oakdale, Pennsylvania and visit the store to pick them up. We'd be happy to help ensure those little shoes are fitting right and looking great.


text:
You can contact us by phone, email, fax, or chat for immediate assistance.
A technical expert will call you shortly.
Where to Buy
Black Box products can be purchased through qualified distributors and resellers globally.
Government Purchasing
TAA Products
Black Box Catalogs
Get the complete Black Box catalog for all your IT solution needs.
Special Programs
Special programs for Government, Education, First-time Buyers and more.
Our Giga, Giga2, and Giga Plus and systems feature jacks, wallplates, surface-mount boxes, and other accessories. Components of each system are designed to work together. And they all work with our GigaTrue® CAT6 and GigaBase® CAT5e cable. Here are the differences between the systems so you can make the right decision when choosing hardware.
•Giga products are our original line of jacks, wallplates, etc.
•Giga products, such as jacks and wallplates, are designed to work with Giga products.
•To meet the needs of existing Giga systems, we continue to carry Giga products.
•Giga2 products are a newer line. They offer the same quality but are priced economically.
•Giga2 products, such as jacks and wallplates, are designed to work with Giga2 products.
Giga Plus
•Giga Plus is our newest line and is entirely made in the U.S. So if you need to buy American-made products, choose this line.
•Giga Plus products are designed to work with Giga2 products.
You're closing chat, but you can still access it under "Talk with Us" if needed.
Do you still want to close it?


text:
Gift Premium Subscription To TheMarquis
Join Date:12-09-2009
Last Activity:02-14-2014 9:28 PM
- Female
- Single
Posted in thread: Light IPA? on 11-08-2011 at 02:39 PM
I used to brew IPAs exclusively until I came up with this. I like it just as much as my 7% IPAs
and it's cheaper, faster to brew and has less alcohol and calories.6 lbs. extra light pilsen
extract.75 ...
Posted in thread: Switched the fermentation vessel, now what? on 10-27-2011 at 05:21 PM
I definitely didn't tighten the spigot enough; I'm always cautious not to overtighten it. I
think I leaned the bucket on the spigot and that's what started it. Oh well.I figured a little
extra oxygen ...
Posted in thread: Switched the fermentation vessel, now what? on 10-27-2011 at 02:50 PM
So I brewed a batch of pale ale last night and I used a bucket with a spigot on it for the
fermenter (yeah I know). Pitched the dry yeast as I normally do. This morning I woke up to the
spigot leaking...
Posted in thread: Carapils vs. carahell on 08-30-2011 at 08:36 PM
I have a blonde ale extract recipe that calls for 1 lb carapils for steeping grains and I
happen to have 1 lb of carahell on hand. Will substituting make a big difference in the beer?
Posted in thread: Brown threads on cones on 09-10-2010 at 03:53 PM
What are the brown threads on my hop cones? Should they be pulled off when I harvest?
Posted in thread: Steeping grains for my extract IPA on 07-16-2010 at 04:06 PM
I've brewed 7 or 8 IPAs in my short brewing career. My base recipe uses 8 lbs extra light DME
and 1 lb crystal 40L for steeping. I started out with all amarillo as the hops and have tried
to change ju...
Posted in thread: Help me name my brewery on 07-13-2010 at 03:36 AM
3 man weaveCount The Basket And A Foul Brewery
Posted in thread: Help me name my brewery on 07-12-2010 at 08:02 PM
Hank Baskett's BreweryPut the Dog Biscuit in the Basket BreweryUnderwater Basket Weaving 101
BreweryGoing to Hell in a Hand Basket BreweryLittle Yellow Basket Brewery
Posted in thread: Hops still good after being left out? on 07-12-2010 at 03:21 AM
Smell them. If they smell cheesey, I wouldn't use them. I'm sure they've lost a little
potential, but they're still most likely usable.Literally like cheese? They definitely smell
like hops, maybe che...
Posted in thread: Hops still good after being left out? on 07-12-2010 at 01:34 AM
I left some hops out on the counter in my 70-75 degree basement for about a week. They were in
the foil package they came in which was folded over tightly at the opening but not completely
air-tight. ...


text:
You are not currently logged in.
Access your personal account or get JSTOR access through your library or other institution:
Reviewed Work: Political Repression in Modern America: From 1870 to the Present by Robert Justin Goldstein
Review by: Mary S. McAuliffe
The Journal of American History
Vol. 69, No. 4 (Mar., 1983), p. 1044
Published by: Oxford University Press on behalf of Organization of American Historians
Stable URL: http://www.jstor.org/stable/1901298
Page Count: 1
- Item Type
- Article
- Thumbnails
Preview not available
Preview not available
Note: This article is a review of another work, such as a book, film, musical composition, etc. The original work is not included in the purchase of this review.
Page Thumbnails
The Journal of American History © 1983 Organization of American Historians


text:
Hi everyone,
I just finished this piece but still is not working completely for me, any advice would be much appreciated. Thanks a lot!
Hi everyone,
I just finished this piece but still is not working completely for me, any advice would be much appreciated. Thanks a lot!
blog: http://almuredondo.blogspot.com/
portfolio: http://www.almuredondo.com/
facebook: https://www.facebook.com/almuredondoart
twiter: https://twitter.com/AlmudenaRedondo
Ermmm... what is this thing? A combat robot flower? A mini shiny tank with a snail inside? I am at lost, please, tell us what it's.
Really neat design ! I've noticed that it looks like there is some type of creature in it, Is it using it similar to how something would use a mech ? pretty neat ! You might want to try adding some lights and give it a more textured appearance to the plating ? The textured plating on the right one's metal fin looks pretty good, it might look good if you were to expand that texture on to a few plated areas. Try adding some types of logo or number decals on some parts of it ? Looseness of the wiring looks good. It may look a bit better color wise if you were to experiment with some back/frontal lighting to make the forms pop out a little more. Finally it might be a good idea to do a side view to show off the creature a bit more and get an over all feel for it's complete silhouette to communicate to the audience what it exactly is, you want to be sure that who ever is in charge knows what they are looking at and what it dose.
Oh, as another detail, maybe add some exhaust with some smoke/steam coming out for some special effects ? Not a ton but just a bit.
Very interesting concept ! It'd be really cool to see these creatures running around in this.
Great Work !
Practice Makes Prefect
SB - http://www.conceptart.org/forums/sho....php?&t=278703
Lacks fundamentals, which is why it is so hard to read/understand. Get Norling: Perspective made easy & Robertson: How to Draw. Both of those books combined will give you the necessary toolset to execute a design like this.
Developed Actively by the makers of the Best Amazon Podcast