text:
|Publication number
|Publication type
|Application number
|Publication date
|Sep 2, 1997
|Filing date
|Jun 6, 1995
|Priority date
|Jul 1, 1992
|Fee status
|Also published as
|Publication number
|Robert J. Goldman
|Original Assignee
|Goldman; Robert J.
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (22), Non-Patent Citations (23), Referenced by (40), Classifications (15), Legal Events (8)
|External Links: USPTO, USPTO Assignment, Espacenet
This is a divisional of application Ser. No. 07/908,121 filed Jul. 1, 1992, now U.S. Pat. No. 5,449,002.
This invention relates to a sensor to be used in medicine and more particularly to a sensor for providing biofeedback to help patients relearn physical function, or prevent complications that impede their physical function.
There are currently no such devices in general use by patients either in acute care, rehabilitation or at home in spite of the need for devices of this nature. However, the need for them is compelling.
For instance, biofeedback may be provided for monitoring the weight applied to a limb ("limb load monitoring"). A major example relates to limitation of weight bearing after leg fracture or hip replacement. Total hip replacements alone are performed 200,000 times per year in the United States. The surgeon prescribes "toe-touch" or "partial" weight bearing almost universally. Partial weight bearing continues until healing has progressed sufficiently to allow full weight bearing safely. In almost all rehabilitation centers there is no way to assure compliance with these orders except by observation. Due to the absence of suitable devices, the true cost of excessive weight bearing has never been systematically studied. However, it is reasonable to hypothesize that un-monitored weight bearing prolongs healing time, duration of hospitalization, and dependent status. Limb load monitoring could significantly reduce this apparent loss of time and money.
Additionally, there are applications to gait training. Limb load devices would prompt amputees to symmetrically bear weight on prosthetic limbs. (This is a prelude to progressive ambulation). Children with cerebral palsy could be taught to put weight on the heel first on planting the swing phase leg during gait. Time on an inpatient rehabilitation service could be reduced, and safe ambulation promoted.
Other potential applications include rewarding the patient for meeting strength goals, e.g., isometric grip strength and knee strength. Further, individuals with rheumatic diseases could greatly benefit from effective limb load monitoring.
These devices, if may generally available, could be used to teach the safe use of wheelchairs to prevent costly falls, or proper pressure relief methods to reduce incidence of gluteal decubiti. Further consideration also must be given to take into account the cost to hospitals of potential malpractice litigation involving medical complications. Overall, the cost saving due to reduced morbidity and length-of-hospital stay could be of great financial significance.
The remainder of this background outlines the prior art of rehabilitation force transduction. Capacitive, resistive, hydraulic, pneumatic and miscellaneous transducer types have all been proposed in the past. However, each has shortcomings that may explain their clinical scarcity and none are widely used by patients.
1. Capacitive Devices
Capacitive transducers have inherent benefits in monitoring loading on limbs or other body parts. They provide a relatively linear increase of capacitance with applied weight. (Therefore, one may use simple, low cost electronical circuitry). They measure total weight or force between complex, irregular surfaces. However, they have been ignored over most of the past decade by those skilled in the art in favor of resistive devices. This has been due to the inability of anyone in the field to find an acceptable dielectric.
The following paragraphs lists concepts and definitions important in evaluation of capacitive transducers.
The dielectric is the insulating middle layer between conductive plates of the sensor. Mechanistically, force applied compresses the dielectric, bringing plates closer together and increasing capacitance.
The effectiveness of such devices is dependent on the properties of the dielectric materials. Creep refers to the long term change of dielectric thickness after initial loading. Hysteresis compares the difference in compression (i.e., capacitance) at a given load with the load increasing, and the load decreasing. Sensitivity is the ratio of capacitance in the loaded and unloaded state, per application of unit weight. Dynamic response is the speed with which the sensor achieves steady state after weight is applied. Resiliency is the ability of the dielectric to retain its unloaded thickness (and baseline capacitance) after a long duration of static or dynamic loading. (A constant application of weight is static, and a ballistic application, dynamic). All these properties are a function of materials composition and structure (e.g., whether a foam or a solid). The ideal dielectric has zero hysteresis and creep. It also has very high resilience, sensitivity, and dynamic response.
The best known published prior art on capacitive sensors is the Krusen Limb Load Monitor. (Craik, R. and Wannstedt, G., Proceedings 2nd Conf. Devices and Systems for the Disabled, 1975, 19-24). The transducer consists of three layers of copper--Mylar(R) laminate separated by two layers of "closed cell foam tape". The whole device has the shape of an inner-sole. The characteristics of the dielectric, however, were not published. The device is manufactured by Electronics Quantification Inc., Plymouth Meeting, Pa. Disadvantages of the device include lack of adaptability to orthotics (i.e., braces), assistive devices or wheelchairs. Further, it is not designed for upper extremity use, and has a high cost ($600.00 per item).
Additionally, conventional polyethylene foam tape (the most likely dielectric material considering the limited public disclosure of the device) has significant hysteresis and creep, and also tends to "bottom out" with repeated compression. One further expects that the sensors need to be frequently replaced. Perhaps for these reasons the device is little used by the rehabilitation community after 15 years on the market. It is not well known even among rehabilitation engineers. Considering the size of the potential market, the device has been a failure.
After examination of this device, Miyazaki and Ishida published data on an apparent improvement. The dielectric is in this case two layers of Neoprene sponge, laminated between three layers of copper foil. (Miyazake, S., and Ishida A., Medical and Biological Engineering and Computing, 1984, pp. 309-316). However, Neoprene, like polyethylene, has limited resilience to static and dynamic loading. It also tends to "bottom out" with repeated use.
On initial testing there was good agreement between force plate and transducer data on patients with foot shapes that were not markedly abnormal. However, higher local pressures from deformed feet caused excessive error (i.e., overestimation of weight). Stiff plastic was therefore built into the laminate to spread out area of force application. As a result, the device was cumbersome, thick and stiff.
As published, it cannot be adapted to shoes, orthotics, assist devices or wheelchairs. No subsequent work using this device has been published. Apparently, it has been abandoned.
2. Resistive Devices
A second major category of sensors for rehabilitation are resistive devices. All of the recent public disclosure and most of the patented prior art is of this group.
Force Sensitive Resistors (FSR's) are marketed by Interlink Corp. They are thin (0.25 mm thick), physically flexible,-inexpensive devices available in 1 cm2 squares. Externally there are two adjacent Mylar (R) sheets. On one Mylar layer is a high resistance polymer or carbon film, and on the other is an interdigitated pattern of open ended conductors. Ruggedness appears to be good with extended use. It may be used with simple electronical circuitry.
For these reasons FSR's have been proposed for "everyday" rehabilitation pressure measurement in several applications. These include monitoring pressure under insensate feet in patients with diabetes and leprosy (M. Maalej, et. al., IEEE, 9th Conference of the Engineering in Medicine and Biology Society, 1987, pp. 1824, 1824). A finger and hand exercise device and "spot" high pressure sensor for decubitus prevention have also been described (Hyman, W. et. al., RESNA, 13th annual Conference, 1990, pp.201, 202). It additionally has been proposed for motorized prosthesis control (Heckathorne, C., RESNA, 12th annual Conference, 1989, pp. 224-225).
However, there are several significant drawbacks of these devices. The most significant is non-linearity; response is substantially exponential. Unloaded resistance is effectively infinite, and resistance falls rapidly as 10-20 pounds of weight is applied. Above this "breakpoint", resistance approaches a minimum asymptotically. Above 50 lbs resistance change may be obscured by noise. In terms of noise, the device demonstrates significant hysteresis that varies from device to device and trial to trial with the same device.
For processing, the response of each sensor must be linearized. Usually this is done by a microprocessor. However, it may be done (less successfully) with simple integrated circuits. To find the weight applied to a large irregular contour, the outputs of many linearized sensors must be summed. Using more sensors makes accuracy less dependent on variations of contour. However, this also increases complexity of necessary electronics, reduces reliability and dramatically increases current drain (i.e., decreases battery life in a portable device). Because of battery drain alone, FSR's would be unacceptable to use in a wheelchair embodiment.
To use a minimum number of sensors, FSR's must be placed at points of maximum pressure. For instance, for limb load monitoring sensors may be placed on the first and fifth metatarsals and the heel. However, many patients have irregular foot contours or may wear a cast. For them, sensor placement must be customized. Biofeedback for hand grip strength using FSR's presents an even more serious problem, as hand size and placement are difficult to control. Thus, FSR's lack flexibility where total pressure is to be obtained over complex surfaces. In contrast, capacitive devices are strong in this category, as they are in terms of current drain. Indeed, FSR's and other resistive devices are most reliably employed where local pressure are to be mapped and quantified over two dimensions, e.g. to prognosticate skin areas at risk for breakdown and ulceration. Calculated forecasts from resistive devices, readouts, may guide the prescription of custom shoes.
One such device is based on the flexible force sensor developed by Polchaninoff in 1984 (U.S. Pat. No. 4,426,884). Its electrical behavior (and disadvantages) is similar to the FSR. Twenty of these sensors are incorporated into the "multi-event notification system for monitoring critical pressure points on persons with diminished sensation of the feet" shown by Goforth in his 1987 patent (U.S. Pat. No. 4,647,918). Resistors are placed under the heel, lateral plantar surface, metatarsal heads, and toes. The information from each is transmitted to a microprocessor on the belt. However, drawbacks of complexity, cost, and fragility argue against the use of this device for limb loading after orthopedic procedure or for gait training. Indeed, many appropriate patients at risk for foot ulceration either cannot afford this device or cannot learn to operate it.
Tecscan Inc. (Boston, Mass.) has developed a pressure mapping system for gait analysis (Sensors, May 1991, pp. 21-25). Not 20 but 960 resistive elements are continuously sampled by a personal computer during ambulation. Graphics of excessive local pressure on the foot are displayed in real time. A similar device has been developed by Tecscan for determining gluteal pressure contours, and may be used for evaluating seating systems. Drawbacks of both are similar to the Polchaninoff device (i.e., high cost, lack of durability, fragility, complexity). The Tecscan limb load device is very expensive, exclusive of the cost of the required personal computer. The seating evaluator is at least this expensive.
3. Hydraulic Devices
Hydraulic devices easily determine total weight between irregular surfaces. They are similar in this respect to capacitive devices. However, by definition they do this by means of fluid pressure in a closed chamber. Thus, they tend to be large, heavy, bulky, thick or rigid. Hydraulic or (conceptually similar) pneumatic systems have been applied to limb load detection, grip strength detection, and seating pressure systems.
Two patented devices are designed to provide objective immediate feedback of limb loading (Sipe, U.S. Pat. No. 3,974,491; Pfeiffer, U.S. Pat. No. and 3,791,375). Both employ reservoirs below the heel and sole. Weight compresses these reservoirs and is monitored by a transducer on the ankle. There are several important disadvantages, including the need for a dedicated shoe. Also, the transducers are difficult to put on and take off, and the heights of these devices may impede ambulation in an individual whose walking is already impaired and possibly unsafe.
Hydraulic devices for measurement of grip strength are known as dynamometers. The devices are large and heavy. Force is read off a pressure gauge, which is difficult for the patient to read. Due to these limitations, they are not designed for use during exercise. They are certainly inappropriate for the elderly or frail arthritic who would most benefit from grip strengthening.
Pneumatic devices have been developed for measurement of pressure beneath bony prominences in evaluation of wheelchair seats. However, they are essentially flattened, modified balloons. Due to size, shape and fragility they are not appropriate to permanently monitor weight on wheelchairs. They are not adaptable to the wide variety of wheelchairs and wheelchair cushions now available.
4. Miscellaneous Devices
A mechanical switching device, patent granted to Gradisar, (Gradisar, U.S. Pat. No. 3,702,999) measures limb loading. Each of two switches has two metal discs separated by an "O" ring. As weight is applied to the top disk the "O" ring is compressed until contact is made, tripping a buzzer. By adjusting the set screw on one disk, the weight at which the buzzer sounds may be adjusted. This device is subject to mechanical problems and is difficult to adjust to the "trigger" weight, and only one type of shoe (a "cast boot") may be used.
Unpatented art that has been applied to biomedical force measurement include strain gauge and piezoelectric technologies. Strain gauges are made of coils of thin high-resistance wire or metal or silicon impressions on a substrate. Because of their high level of accuracy, these transducers are good for gait research. However, they are fragile, expensive and consume large amounts of power. Therefore they are unsuited for the demands of daily therapeutic use. Additionally, the signal from them is low, requiring extensive amplification. Thus, the complete system demands excessive space, a controlled environment and a large budget.
5. Summary
In summary, the prior art intended for weight bearing biofeedback has not gained commercial acceptance due to disadvantages of complexity, high cost, poor reliability, durability and concerns about safety. Force sensitive resistors are promising for certain limited applications. However, FSR's are substantially non-linear. Dedicated electronics must linearize and integrate outputs to measure weight. For large, irregular surfaces many FSR's must be used. This increases complexity, cost, and current drain. It decreases durability and battery life.
However, capacitive transducers inherently linearize and integrate force information. Capacitive transduction is independent of body contours and transducer contour. Devices need not be customized; they are readily interchanged between patients. They are simple, inexpensive, reliable, portable and have the best expected battery life.
One would think capacitive transducers are the best candidate for an "all-purpose" clinical weight bearing device discussed at the outset. However, capacitive transduction has apparently been abandoned by those skilled in the art due to non-existence of a suitable dielectric.
The present invention overcomes the disadvantages of the prior art. It is a capacitive transducer, consisting of an open cell polyurethane foam dielectric sandwiched between two conductors. This foam is a key feature of this invention. It is used in every embodiment. In most embodiments, the conductors are wire mesh, similar in appearance to window screen.
A second un-obvious feature of this invention is use of carbon impregnated silicone rubber (CISR) as the conductors in some embodiments. CISR is the material employed in transcutaneous electric nerve stimulation (TENS). It has high resistance to DC current. However, for measurement of capacitance it acts as a conductor. Embodiments with CISR have remarkable pliability and ruggedness.
The preferred polyurethane foam is characterized in terms of chemistry, density, firmness and morphology. Chemically, polyurethanes are formed as diisocyanates (having the general structure R--N═C═O) react with polyols (alcohols). Density is commonly referred to as pounds per cubic foot, or pcf. The preferred density range for the present invention is 5-50 pcf. Firmness, is defined as the compressive force (expressed as pounds per square inch, or psi) required to cause a 25% compression of the polyurethane foam. (A device causing a 0.2 inch/minute strain rate is used). The preferred firmness range for the instant invention is 0.1-100 psi. Morphologically, the foam made up of a plethora of approximately spherical cells. The preferred average cell diameter is 50-200 microns. Cells are in communication through a multitude of pores (thus, "open cell"). Unimpeded air flow through the foam substance helps establish resilience. On formal testing there is a loss of 5% thickness or less on an ASTM 1667 compression set at 73 degrees F. Foams of this type are preferably manufactured by Rogers Corp., Poron Materials Division, East Woodstock, Conn.
The preferred polyurethane foam of this class is marketed as PPT® (mnemonic for Personal Protective Technology), a trademark of Langer Biomedical Group. This open cell, high density, microcellular polyurethane foam has an average pore size of 100 microns, density of 20 pcf, and firmness of 10±3 psi at 25% compression. Polyurethane foams of the present invention are preferably constructed with PPT® dielectric. Thus, in the discussion below open cell polyurethane foam is referred to as PPT®. However, any member of the same class (with the above described components) may be used where appropriate). PPT® (unlike other open-cell foams) has been well-known in the field of rehabilitation for several years. The conventional application is as foam padding for protective use in shoes. Although well known in the field of rehabilitation for several years, no one has thought to apply it to solve the weight bearing dilemma. However, the same features that make it a good protective foam cushion with excellent energy absorptive properties also make it the best available capacitive dielectric. These include unsurpassed resiliency to both static and dynamic loading, and linearity with compression.
PPT® was found superior to neoprene and polyethylene foams in resiliency on extensive dynamic testing (Brodsky, et. al., Foot and Ankle, Vol. 9, December, 1988, pp. 111-116). PPT® showed no loss of thickness for 10,000 cycles of compression or shear compression. (10,000 cycles corresponds to 9 hours of walking at normal cadence), whereas Neoprene showed a 5-15% loss of thickness, and polyethylene foam 15 to 50% loss of thickness. Devices made with PPT® should thus last a very long time. Further, one expects devices made with PPT® be far more durable than polyethylene (Krusen Monitor) or neoprene (Miyazake and Ishida device).
Furthermore, testing reveals PPT® to have far better resiliency to static loading than polyethylene and neoprene foams. A square of PPT may be placed under a desk leg for a week and return immediately to its original shape. However, neoprene or polyethylene foams would remain permanently compressed.
The second advantage of PPT is improved linearity with compression. It is fortuitous but gratifying that this transducer is linear within narrow limits from zero to 75% compression of the dielectric. The range of linearity of pressure to compression is wider than the other two foam materials. Thus, deformed feet transferring high local areas of compression should yield equivalent capacitance change to a more spread out load. This is a clear improvement over the device described by Miyazake and Ishida.
Additionally, creep is imperceptible with greater than 75% compression for extended periods. This is in contrast to neoprene and polyethylene foams, which show significant creep at this level of compression.
Thus, this laminate is very promising as basis for a variety of weight biofeedback devices in the clinical setting. Seven specific embodiments are introduced in the following paragraphs:
The purpose of the cast boot embodiment is to improve compliance with toe touch or partial weight bearing order after bone surgery on the hip or leg. This assures the shortest, safest course of post-operative rehabilitation. A cast boot is an oversized, open, canvas shoe frequently worn by such patients during daily ambulation training. A thin embodiment may fit inside a cast boot. In the preferred embodiment the conductive layer is wire screen. However, carbon impregnated silicone rubber may be used.
The shoe embodiment is intended for a patient with an insensate extremity. The well known work of Bach-Y-Rita suggests that electrically augmenting sensory feedback helps such a patient re-learn and maintain safe, functional ambulation. Conditions that lead to insensate extremities include peripheral neuropathy (e.g., from diabetes) or traumatic spinal cord injury. A secondary benefit may be reduction in foot ulceration and deformities. Many prosthetic wearers would benefit from sensory feedback, especially the many amputees with diabetes.
Physically, the embodiment resembles an inner-sole and is concealed within a shoe. This device is best employed where cosmoses and high level functioning are paramount. Use is for an indefinite period due to a chronic condition. Low cost and long life are likely to create a market for this embodiment. In the preferred embodiment the conductive layer is wire screen. However, carbon impregnated silicone rubber may be used.
Children with diplegic cerebral palsy tend to ambulate with an equinus (i.e., downward pointing) ankle due to poor cerebral processing of motor information. They adopt normal heel strike only when constantly reminded (i.e., cued). This behavior persists despite use of an AFO (ankle foot orthosis, or ankle brace), specifically designed to prevent it. An embodiment is described which allows electronic cuing of heel strike. It is designed to be used with an AFO. Previous work (Seeger, B. R., Archives of Physical Medicine and Rehabilitation, 56: 237-240) suggests such a device would improve ambulation in these children. In the preferred embodiment the conductive layer is wire screen. However, carbon impregnated silicone rubber may be used.
Isometric strengthening (i.e., exercise of a muscle without change of its length) is preferred in treatment of arthritis and tendinitis of the hand. This is because joint and tendon movement tends to worsen inflammation. The grip strength embodiment measures grip pressure without perceptible movement of the device. This is due to the characteristics of the polyurethane dielectric in the capacitive transducer. It is designed to replace the heavy and cumbersome dynamometer that works on hydraulic pressure. The present invention is small and lightweight, and thus may be used by even a weak, frail patient without supervision. By giving continuous feedback about force applied, it will improve motivation and quality of exercise in these patients. It may also be used for strengthening and edema control in the weak hand of stroke patients, the hand that tends to be neglected (i.e., ignored) due to the cognitive effects of the stroke. Lastly, such a device could be used by the lay public for exercise of grip.
Axillary crutches are frequently used by patients that are partial weight bearing on one leg. They transmit weight through an arm and hand to the brace instead of through the leg. However, if the crutches are used incorrectly, the brachial plexus may be injured. (The brachial plexus is a bundle of nerves passing through the axilla to control arm movement and sensation). Months of pain, weakness and numbness may result from this not uncommon injury. A force transducer to detect excessive armpit pressure has been described which employs strain gauges (Ang, E., Archives of Physical Medicine and Rehabilitation, 70: 644-646). Strain gauges normally retail for $500.00 on up, so this device is not very cost-effective. However, the present invention has a simple construction that may enable its routine use as a safety feature within axillary crutches at relatively low cost.
An inexpensive, durable means to detect pressure on a wheelchair seat would have many uses. Paraplegics and patients with spina bifida, insensate below the waist, would benefit from a device to cue weight shifting at regular intervals. It might reduce incidence of sacral pressure ulcers in these populations. Malament, et. al. (Archives of Physical Medicine and Rehabilitation, 56: 161-165) demonstrated such behaviors could be learned by use of biofeedback. However, bulky hydraulic devices or expensive strain gauges have been used in the past for monitoring. Other devices are based on switches, which do not allow the patient to learn progressive weight reduction during the weight shift. The wheelchair embodiment is unobtrusive, relatively inexpensive, readily adaptable to any seat design, uses only simple electronics, and monitors gradations of pressure.
Such a device could also help reduce falls due to unassisted attempts to rise out of a wheelchair on the part of patients with dementia. An unassisted attempt would activate an alarm. It is currently very important in nursing home practice to use minimal restraint (e.g., wrist or waist straps). Thus, this device could reduce falls, a common cause of morbidity in nursing homes, and at the same time minimize physical restraint.
In terms of rehabilitation, many patients immediately after stroke are impulsive and unsafe. Pressure detection could electronically cue them to lock wheelchair brakes prior to rising, and thus promote safety awareness and reduce needless injury. All these issues could be successfully addressed by a wheelchair embodiment of my invention.
These consist of sets of circular, elliptical or rectangular laminates of carbon impregnated silicone rubber and PPT. They would be used for isometric strength training (i.e., other than grip strength) in the physical or occupational therapy gym. It may inspire creativity within these disciplines and be the source of further applications.
Thus, several objects and advantages of the present invention include:
1) Flexibility, such that it may be incorporated into weight measuring shoes, cast boots, wheelchairs, devices to measure grip or pinch, or directly incorporated within orthotics.
2) Low projected cost due to simplicity of construction.
3) Unobtrusiveness (it can be made less than 0.16 cm. or 1/8 of an inch thick).
4) Durability, as the only moving part is the compressible plastic foam, which returns to its original thickness after loading after even months of continuous use.
5) Simplicity of processing. Weight is integrated over any range desired to obtain a total weight.
6) Inherently linear output.
7) Simplicity of electronic processing due to 5 and 6.
8) Accuracy to the extent necessary in a clinical setting.
9) Ability to obtain weight applied through large, irregular surfaces as easily as through small, regular ones: That is, ability to integrate weight over area.
Because of these advantages many embodiments suggest themselves:
1) Foot Weight bearing feedback
a) after orthopedic procedures or fractures.
b) after stroke to facilitate re-learning of weight shifting, then ambulation.
c) facilitate paraplegic ambulation.
d) facilitate ambulation of amputees with prosthetic limbs.
2) Upper extremity feedback, such as to provide biofeedback or grip or pinch strength.
3) Application directly within orthotics.
a) correct tendency towards gait deviations in cerebral palsy patients.
4) Application within assist devices, such as crutches.
5) Wheelchair seat pressure feedback.
a) electronic cuing to weight shift in spinal injured patients prone to pressure ulcers.
b) alarm to nursing staff that patients on fall precautions have impulsively risen from the wheelchair.
c) cue patient to lock wheelchair breaks before rising.
6) Flexible "pressure pads" with general use within the physical or occupational gym, for muscle strengthening or endurance programs.
Other objects and features of the present invention will become apparent from the following detailed description considered in connection with the accompanying drawings. It is understood, however, that the drawings are designed for purposes of illustration only, and not as a definition of the limits of the invention for which reference should be made to the appending claims.
FIG. 1 is a enlarged perspective view of the cast boot embodiment within a cast boot and associated electronics module;
FIG. 2 is a top and side plan view of the cast boot embodiment;
FIG. 3 is an exploded perspective view of the cast boot embodiment;
FIG. 4 is a graph of weight versus capacitance for the cast boot embodiment;
FIG. 5 is an enlarged perspective view of the shoe embodiment being used in a typical application;
FIG. 6 is a top and side plan view of the shoe embodiment;
FIG. 7 is an exploded perspective view of the shoe embodiment;
FIG. 8 is an enlarged perspective view of the ankle brace embodiment;
FIG. 9 is an exploded perspective view of the ankle brace embodiment;
FIG. 10 is a perspective view of the grip strength embodiment being used in a human hand;
FIG. 11 is a perspective view of the grip strength embodiment outside the hand;
FIG. 12 is an exploded perspective view of the grip strength embodiment;
FIG. 13 is an enlarged perspective view of an axillary crutch and axillary crutch embodiment;
FIG. 14 is an exploded perspective view of the axillary crutch embodiment within the top of the crutch;
FIG. 15 is an exploded perspective view of the axillary crutch embodiment itself;
FIG. 16 is an enlarged perspective view of a wheelchair with the wheelchair embodiment within the seat;
FIG. 17 is a top and side plan view of the wheelchair embodiment;
FIG. 18 is an exploded perspective view of the wheelchair embodiment;
FIG. 19 is a perspective view of the pressure pad embodiment; and
FIG. 20 is an exploded perspective view of the pressure pad embodiment.
The cast boot embodiment is illustrated in FIGS. 1-4. FIG. 1 affords an overview of the device. The cast boot (52) is preferably an oversized canvas shoe with an open upper surface, and fastens over the foot or cast with a Velcro® strap (53). Also, it provides support and protection to the patient's foot.
The weight detecting element (50) sits on the upper surface of the cast boot. It is preferably very similar in shape to this platform, and not permanently glued or fastened to it. As such, it is easily installed and removed by the physical therapist. The device connects to a wire (62), then to an electronic module (64) placed on the cast boot, belt, pocket or other convenient place on the patient by means of a clip (82). Preferably, attached to the electronics module 64 is an adjustment knob (80), and an on/off switch (90). Additionally, there is preferably a piezoelectric buzzer within module 64 (not shown).
The general size and shape of the device is best shown with reference to FIG. 2 which shows top and side views. Device 50 is preferably flat, planar, flexible and rectangular with rounded ends. For a medium cast boot, it is preferably a maximum of 9.5 cm. wide at the forefoot (54), tapering to 8.5 cm wide at the hindfoot (56). Additionally, it is preferably 24.5 cm. long and 0.3 cm (1/8 inch) thick.
The method of fabrication is best described with reference to FIG. 3. It preferably consists of five layers. The top layer (66) is thin rubber. The next layer down (68) is steel mesh. The perimeter of this layer (69) is preferably recessed 0.7 cm from the perimeter of overall device (73). Mesh area and gauge are preferably similar to conventional window screen. Posteriorly, a wire (69) connects with the mesh, and is preferably soldered 2 cm. from one corner.
Below the mesh is a layer of polyurethane foam, as previously characterized, and preferably PPT® polyurethane foam (72), preferably 0.16 cm. (1/16 inch) thick. Below the PPT® is another layer of steel mesh (74), substantially identical to item 68. However, the wire (75) preferably connects 2 cm from the other rear corner of the mesh. Thus, the solder points of the wires are not directly beneath each other. Below this is the final layer (78), rubber sheet essentially identical to item 66.
All parts are preferably glued together with Contact Spray Adhesive (U.S. Pat. No. 4,401,272, 3M Company, St. Paul, Minn.). This neoprene-based product bonds well to rubber, plastic foam and metal. The five layers thus form a strong, but flexible interconnecting bond. This facilitates prolonged clinical use without loss of properties.
The capacitive transducer is created when the dense polyurethane foam (72) is placed between metal plates (68 and 74). As weight is applied the foam is compressed, decreasing inter-plate distance and increasing capacitance. Conversely, as the weight is removed the inter-plate distance and capacitance return to their original values. To demonstrate both stable baseline and linearity, a cast boot embodiment of the above specifications was tested for its pressure-capacitance profile. FIG. 4 is a graph describing these results. Zero to 160 lbs of weight were applied over 5 areas of weight application: 9, 16, 30.2 and 40.2 cm2. (For reference, the area of the average adult great toe is about 6 cm2, heel 30 cm2, and forefoot 45 cm2). Clearly, a high degree of linearity is demonstrated over a wide range of compression of the foam.
In a conventional manner, electronics module (64) converts capacitance linearly to voltage. Of note, top plate 68 is connected to ground to largely neutralize stray capacitance of the subject from affecting this signal. This voltage is internally compared with a trigger voltage preferably set by knob 80. This knob is set in conjunction with placing the patient's foot on a scale prior to ambulation. When weight exceeds this preset trigger a piezoelectric buzzer within the electronics module provides feedback to the patient and physical therapist. In this manner the patient is informed about toe touch, and partial or full weight bearing on the affected leg.
The shoe embodiment in a typical application is illustrated in FIGS. 5-7. In FIG. 5 a paraplegic is practicing reciprocal gait by weight shifting between extremities when both legs are anaesthetic. Transducer (100) is preferably beneath the inlay of an orthopedic oxford shoe on the right foot. Wires (114, 117) extend to the waist on the right side where electronic module (128) is hooked onto the belt. There is a similar configuration on the left side (not shown). The electronics module is similar in static design to the cast boot module, and has a trigger mode, and on/off switch. Inside, there is a small vibrating motor that imparts vibratory feedback to the patient. A wireless design could also be employed (not shown). This would further improve cosmoses.
The shape and thickness of the shoe embodiment is best illustrated with reference to FIG. 6. The device is preferably thin, planar, flexible and substantially flat.
The fabrication of this device is best described with reference to FIG. 7. It preferably consists of five layers. These are from top to bottom, gum rubber upper piece (104), top layer of steel mesh (106), layer of PPT (108), bottom layer of steel mesh (110), and gum rubber bottom piece (111). Steel mesh layers 106 and 110 are preferably recessed 0.7 cm inside the perimeter of the overall device. Steel mesh top plate (106) connects to a wire (114) near the front of arch (112). Similarly, steel mesh bottom plate (110) connects to a wire (117) near the rear of arch 112. These two wires connect to electronics module (128).
The calibration and operation of the shoe embodiment is similar to the cast boot embodiment. However, the feedback is preferably tactile instead of auditory. The rational is that the audible alarm may be distracting and embarrassing for a highly functioning patient in a work or social setting. Calibration of trigger level is done by the patient by setting a threshold in conjunction with putting the foot on a scale, and varying trigger adjustment knob. Vibration intensity is all or none above this threshold. Alternatively, it can be wired to continuously increase in intensity with weight increase above a threshold. The electronic design of this device is conventional. In preliminary studies an active adult male wore this device for several hours. A stable baseline capacitance, and stable capacitance increase per unit weight increase, were noted throughout the experiment.
FIGS. 8 and 9 refer to this device. FIG. 8 shows an overall perspective view. The ankle brace embodiment (152) preferably fits between the ankle foot orthosis (AFO) (150) and shoe (154). (Very often the patient with cerebral palsy has excess tone in the calf muscles. The AFO helps to control ankle movement in the face of this muscle imbalance). Wires (154, 156) go to an electronics module. This module is identical in appearance to the module 64 connected to the cast boot embodiment. Therefore, it is not specifically illustrated here. (As an alternative, transducer 152 and module 64 could be fabricated as integral parts of AFO 150).
FIG. 9 best illustrates construction of this device. As above, there are preferably five layers. They are (from top to bottom) thin rubber top piece (160), steel mesh layer (162), PPT® polyurethane foam layer (164), second steel mesh layer (166) and rubber bottom piece (168). Wires 154 and 156 go to the electronics module. Each layer is preferably of identical thickness and composition as the cast boot embodiment.
The goal of this device is to defeat the tendency of children with cerebral palsy to "toe walk" by constant reminder to plant the heel first after the swing phase of gait. Unfortunately, toe walking frequently persists even if AFOs are worn. Prior to an exercise session the device is calibrated to elicit a sound at or near the patient's full weight. During the exercise session normal heel strike causes a transient increase in capacitance above threshold causing an audible reward signal to be emitted. However, if the foot comes down in equinus (i.e., toe strike instead of heel strike) the capacitance increase does not exceed a preset trigger and no beep is elicited. This is because after toe strike weight is never concentrated in the heel. This invention will thus assist a cerebral palsy patient learn normal heel strike during gait.
FIGS. 10-12 refer to the grip strength embodiment. FIG. 10 illustrates the isometric grip strength detector embodiment (200) in use. It is preferably cylindrical of such dimensions as to fit snugly inside the clenched adult human hand (202). An electronics module (252) is preferably incorporated inside the cylindrical transducer. Threshold strength adjustment knob (254) has a built-in on/off switch. Threshold strength can be set preferably from 0 to 60 lbs on this embodiment using preset numerical settings (256). FIG. 11 shows the general configuration of the device. It is preferably 12 cm (4.25 inches) long and 3.7 cm (1.5 inches) in diameter.
Construction of the grip strength detector embodiment 200 is best described with reference to FIG. 12 where the five layers are shown. For clarity, the four outer layers are divided into upper and lower halves and projected above and below the central cylinder. However, the cylinders are preferably continuous and concentric. Housed within device 200 is the electronics module 252.
The innermost layer (204) of embodiment 200 is preferably cellulose butyrate tubing of inner diameter of 2.54 cm (1 inch) and outer diameter of 3.17 cm (11/4 inch). Going outward, the next layer (206) is a cylinder of steel mesh similar to window screen. It is preferably soldered at point 208 to a wire (210) and reinforced with epoxy cement. The mesh cylinder (206) preferably fits snugly over the cellulose tubing (20), and is 144 cm2 in area. The next layer (220) going outward is PPT® polyurethane foam preferably 0.16 cm (1/16 inch) thick. Above this is a second layer of steel mesh (226) slightly larger in diameter than item 206. However, there are preferably four grooves (228,230,232,234) about 1 mm wide that extend from edge 260 to 4 mm from the other edge of the cylinder. Thus at end 260 the mesh is preferably divided into four separate sections. However, at end 262 these sections are electrically continuous. Solder connection 246 connects this layer of mesh to wire (248). Solder connection 246 is also reinforced with epoxy cement. The outermost layer (250) is preferably made up of gum rubber. Wires 210 and 248 connect to electronics module 252 as indicated by the dashes. All layers are preferably glued together with Contact Spray Adhesive (3M company). (However, other contact adhesives may be used).
To use, the patient grasps transducer 200 and applies force centripetally (i.e., toward the center) with the fingers and thumb. Grooves 228 (etc.) allow the outer mesh element 226 to contract in diameter as distance between plates 226 and 206 decreases proportional to the force applied. Secondarily, capacitance increases. In testing, capacitance increases linearly and reproducibly from a stable baseline. This baseline is 78 pF. (no force applied). Capacitance increases to 148 pF. with 70 lbs. applied.
Grip strength is calibrated in pounds prior clinical use. The lbs. settings are preferably printed around the adjustment knob. The desired threshold force is set by the occupational therapist and patient uses the device as instructed. A sound emitted above a threshold serves as a reward for having achieved a given compression. (Alternatively, it could be an alarm to prevent exceeding a given compression). Experience will determine if subtle baseline shifts occur with heavy use. If so, periodic re-calibration may be required.
Although a cylindrical grasp is illustrated, there are several other important grasps. These include pinch (i.e, thumb against the lateral index finger) or ball grasp (i.e., hand grasping a ball). This invention, as a flexible laminate, may accommodate any of these shapes and provide feedback for nearly any type of grasp.
This embodiment is shown in FIGS. 13-15. An overview of the size, shape and placement of this embodiment is illustrated with reference to FIGS. 13 and 14. An axillary crutch (250) preferably has a rubber cap (252), which fits over a convex wood top-piece (254). From this, long, vertical, bowed wood members (256,258) converge on the rubber base (260). A transducer (262) is preferably interposed between rubber cap 252 and the top piece 254. The overall perimeter of transducer 262 is preferably the same as this top-piece, and it fits snugly inside the rubber cap. A set of two wires from the transducer (264, 266) go to an electronic module (268) fastened near the top of one of these two vertical wood members. In the preferred embodiment, module 268 has no external controls. Internally, there is an on-off switch, weight trigger control adjustment and piezoelectric alarm device similar to the cast boot electronics module (64).
The construction of this transducer is best described with reference to FIG. 15. There are preferably six layers from top to bottom. An uppermost layer (280) is preferably made of Orthoplast®, a 1/8 inch thick semi-rigid plastic the shape of which may be molded at the temperature of boiling water. Below this is preferably a copper foil layer (282), of the same length and width as the Orthoplast®. Wire 264 is soldered from foil 282 to the electronics module 268. Below this is a layer of polyurethane foam elements (284). Preferably there are 24 such rectangular pieces, about 0.95 by 0.50 cm. (0.375 by 0.2 inches), regularly spaced in a staggered arrangement. These preferably cover about 33% of the area between the adjacent layers. Below this is preferably an oversized layer of thin rubber insulation (286). Below this is copper foil (288) essentially identical to foil item 282. Wire 266 connects this layer to electronics module 268. The bottom layer is preferably Orthoplast® (290), of very similar dimensions to the corresponding top layer 280. All layers are preferably glued together with Neoprene contact adhesive. The overhanging edges of rubber layer 286 preferably fold under the Orthoplast® layer 290. This is so copper layers 282 and 288 will not short together.
In operation, a reference trigger weight (voltage) is set internally within the electronics module. An audible alarm sounds if the weight applied through the axilla exceeds a this reference weight. The accuracy is periodically checked by a physical therapist using an inexpensive scale, and re-adjusted if necessary.
By way of theory, the smaller area of polyurethane foam improves the sensitivity of the device. Sensitivity may be defined as the ratio of capacitance change to baseline capacitance. With one third of the area covered, sensitivity increases three times. Semi-rigid plastic members 280 and 290 are then used to more evenly transmit this weight over the set of compressible elements 284. By this method, capacitance increases linearly from a baseline of 70 pF. with 0 weight applied, to 140 pF. with 70 lbs. applied. A similar result would be obtained if a lower density of polyurethane foam were available. Then, the foam layer could cover the entire area between the plates, and eliminate the need for separate reinforcement and insulation layers.
FIGS. 16-18 refer to the wheelchair embodiment of the present invention. FIG. 16 affords an overview of this device. A pressure detector (300) is preferably placed in the sling seat of a standard wheelchair (301). This seat consists of and upper and lower vinyl member joined on the two sides. Transducer 300 fits easily between the vinyl panels that make up this seat. As it is preferably thin, flexible and smooth, it conforms to the curvature of the seat with the wheelchair folded, open or occupied, and it adds no irregularities to the seat. In other embodiments (not illustrated) the device may be placed between a sling or solid seat and special cushions, including a Jay® and Roho® cushion. Lead 302 connects transducer 300 to electronics module 304 located on the backrest. (Module 304 is physically equivalent to cast boot electronics module 64). In other embodiments leads may extend to toggle brake 306.
The size and shape of transducer 300 is more exactly illustrated in FIG. 17, which shows top and side views. It is preferably square, 33 cm. (14 inches) on a side and is 0.3 cm. (1/8 inch) thick. Moving inward, the dashed concentric outlines and perimeters are best presented in the exploded view, shown in FIG. 18.
The construction of transducer 300 is best illustrated in FIG. 18. As with previous embodiments, there are preferably five layers. The top layer (320) is preferably rubber sheet. Just below this is the upper layer of steel mesh (322). It is similar to window screen in appearance and is preferably 33.6 cm. (12.5 inches) on a side. It is connected to wire (324), thence to electronics module 304 (not shown on this figure). Preferably, below this is a layer of PPT® foam (328), 0.16 cm (1/16 inch) thick, and 31 cm (13 inches) on a side. Below this is a plethora of steel mesh elements (332), preferably forming a "daisy chain" of mesh elements. The chain preferably doubles back after every 3 elements to form 6 rows of elements. Preferably, square 336 and 16 others are connected by two wires, and the end square 340 and one other, are connected by one wire, to adjacent squares. All told, 18 substantially identical squares are connected electrically in series. Representative squares 336 and 340 are preferably 2.54 cm. (1 inch) on a side. Their centers are preferably 10.2 cm (4 inches) distant from their immediate neighbors along a row or column, and 7.0 cm along a diagonal. All other squares are similarly related to their neighbors. A wire (348) connects the set 332 to the module 304. The bottom layer is preferably another piece of thin gum rubber (352). As with other embodiments, the layers are preferably glued together with Neoprene Contact adhesive.
Device 300 is a capacitive transducer that preferably works in similar manner as cast boot device 50. However, weight is applied over a much larger in area than the cast boot. Because capacitance is directly proportional to plate area, a test device with both plates full size (i.e., the size of mesh element 322) had a baseline capacitance of 800 pf. This is compared to 215 pF. for the cast boot device (50) (see FIG. 4). Sensitivity is here defined as the ratio of capacitance change to baseline capacitance. Because of the larger baseline capacitance, the 20% sensitivity of the cast boot device decreased to only 5% for the initial wheelchair prototype. By sampling weight over 18 equidistant much smaller areas sensitivity improved to 35% for the preferred embodiment. The only caveat for sampling of discrete points is that weight be relatively evenly applied over the entire surface. For the gluteal region this assumption is reasonable.
In terms of function, electronics module 64 is modified internally. Here, an increase in weight (voltage) above the trigger set by knob 80 turns off an alarm. Conversely, a decrease in voltage (weight) below the trigger turns on the alarm. This is opposite the alarm sequence of the cast boot device.
To operate, the nurse or physical therapist has the goal of being notified if a patient is rising impulsively out of the wheelchair. The trigger is thus preferably set to about one quarter the patient's weight with knob 80. The device is engaged by turning on switch 90. Thus, if the patient rises, the alarm will sound and help summoned. (In some embodiments, a delay may be built in so the patient may weight-shift without sounding the alarm).
The other two uses mentioned in the beginning of this section may be accomplished by modification of electronics only. For paraplegics, a weight-shift off the gluteal region would reset an timer circuit. If a second weight-shift did not occur with a preset interval, an alarm would sound. For the stroke patient, setting both toggle brakes would disengage the alarm circuit that would otherwise sound on standing or transferring.
Lastly, the upper plate 322 is grounded to prevent stray capacitance from affecting weight measurement.
FIGS. 19 and 20 refer to this device. FIG. 19 is an overview. The pressure pad embodiment (400) is more flexible than those discussed above by virtue of carbon impregnated silicone rubber (CISR) as the conductive layer instead of wire mesh. Preferably, the pressure pad is circular. It is preferably connected by three wires (items 402, 404, and 406) to an electronics module. This module is otherwise identical to electronics module 62 (cast boot embodiment). Therefore, it is not shown.
FIG. 20 best illustrates construction of the pressure pad embodiment 400. A cloth upper surface 410 preferably fits above a layer of carbon impregnated silicone rubber (412). The latter is preferably 0.16 cm. (1/16 inch) thick. (However, 0.08 cm. may also be employed to increase flexibility). Below this preferably is a layer of PPT® polyurethane foam (414). Below this is preferably a central layer of carbon impregnated silicone rubber (416), and below this is another layer of PPT® (418). Below this is the lowest layer of carbon impregnated silicone rubber (420). Below layer 420 is the bottom cloth layer 424. Layers are preferably glued with Neoprene cement, after roughening apposing surfaces with sandpaper, or other appropriate abrasive. Wires 402, 404 and 406 are preferably glued to conductive layers 412, 414 and 416 respectively, by means of a conductive silicone rubber cement. PPT® layers 414 and 418 and cloth layers 410 and 424 are preferably of equivalent diameter and 0.5 cm greater in diameter than silicone rubber layers 412, 416 and 420.
Application of weight increases capacitance by compressing foam layers 414 and 418. This brings closer conductive layers 412, 416 and 420. Testing reveals a linear relationship between weight and capacitance. This is true for various areas of weight application, similar to that shown in FIG. 4.
In previously described embodiments there was only one capacitive layer, made up of two conductive layers with interposed dielectric foam. In those, only one surface, the layer closest to the patient, was electrically shielded. Only this side could face the patient without producing spurious readings. Those embodiments are said to be "polarized". For this embodiment there are essentially two capacitive layers in parallel. Both outer surfaces are grounded (i.e., shielded). Thus, either surface may face the patient without causing spurious readings. This embodiment is thus "non-polarized".
CISR is a suspension of small carbon particles within silicone rubber. Its conductivity is directly related to frequency; it is conductive to AC but not DC signals. Since electronics module 408 measures capacitance using AC signals, rubber layers behave as would metal conductors in this context. This sensor is therefore preferably made without metal, excluding the wires (402 404 or 406).
With respect to application, this embodiment is for general isometric strengthening (i.e., other than grip strengthening). The therapist may position the device in front of the ankle and behind an immovable surface for quadriceps strengthening, behind the ankle and in front of an immovable surface for hamstring strengthening, or between the knee and wall for gluteus medius strengthening. There may be occasions where the both surfaces are in contact with the patient, as if placed between the knees for adductor strengthening. In this case double shielding prevents spurious readings. The patient gains reward for reaching certain strength goals by auditory feedback.
In other embodiments, the device could be made a rectangular shape of various dimensions. Non-polarized pads could be used for strength and endurance training. Non-polarized rectangular pads could be made to fit various cast boot sizes, and take the place of the cast boot embodiment. Non-polarized inner-sole shaped devices could take the place of the shoe embodiment. Non-polarized circular devices could take the place of the AFO embodiment. Further, very thin (i.e., 0.8 cm. or less) conductive silicone rubber layers would make for very flexible, resilient devices between irregular surfaces.
Additional improvements could be made to the electronic module. Continuous digital display of weight in pounds is possible. This would be very helpful for weight training. Beyond this, a portable digital scale for the mass market could be made in the following way: Two rectangular or inner-sole shaped pressure pads could be positioned on a flexible plastic screen. These pads would be connected to a small electronics module with digital readout in pounds, also positioned on the screen. For a weight conscious person "always on the move", the device could be unrolled onto a hard surface such as a hotel bathroom floor and re-rolled for travel.
Accordingly, the reader will see that this invention is a capacitive transducer with wide potential application in the area of rehabilitation medicine and also the mass market for exercise or toning devices. In operation the device measures the capacitance change between two metal screens preferably separated by a open cell polyurethane foam dielectric known as PPT®. In some embodiments, the metal screen is replaced by elastic conductive silicone rubber. As weight is applied, the foam compresses and capacitance increases.
Although the concept of the capacitive transduction is not new, this device possesses many important improved properties. These including unsurpassed resiliency and durability of the dielectric. Response is linear and reproducible. It additionally has the ability to measure weight accurately when applied over a large or small area via irregular surfaces. Construction is simple, and readily available electronics may be used with very low current drain. Small batteries may thus power these devices over a long period of time.
The materials that create these improvements have been well known in the field for some time. However, no one has thought to apply them to solve the important problem of how to measure limb loading. Therefore these improvements are not obvious.
Additionally, the prior art has so far been totally unsuccessful in meeting the need for load monitoring. Very little of the prior art is actually in use. This is true even though the potential market is huge.
This invention is broad in application. It has many specific embodiments in the areas of shoes, prosthetics, orthotics, assist devices and wheelchairs. The invention would be of interest to the physiatrist, orthopedic or plastic surgeon, physical and occupational therapist. Its wide range of potential use stems from thin size, and ability to assume different shapes.
The device is simple for both patient and practitioner to operate. Thus, this invention may overcome the natural resistance of these non-engineers to "anything electronic". This is unfortunately another important reason why the prior art involving computers has been rejected by the rehabilitation and medical community.
Having gained wide acceptance, devices based on the present invention would be medically important. They would: 1) help insure accurate fracture healing by more accurate implementation of the weight bearing prescription; 2) improve safety of re-learned ambulation of paraparetics, hemiparetics, and amputees by improving sensory awareness; 3) improve ambulation of individuals with predominantly sensory loss; 4) improve upper extremity strength and reduce pain in arthritics and stroke patients; 5) reduce pressure sores in paraplegics due to prolonged wheelchair sitting; 6) reduce the rate of falls in stroke patients due to impulsively rising out of the wheelchair; and 7) improve effectiveness of strength or endurance training.
Although the description above contains many specificities, these should not be construed as limiting the scope of the invention but as merely providing illustrations of some of the presently preferred embodiments of this invention. For instance, any of the embodiments may be made wireless. The electronic modules may be made very small using surface mount or other technologies of manufacture. Thus, complete embodiments may be incorporated within orthotics, prosthetics or assist devices. There may be other means used to provide biofeedback to the patient besides sound or vibration.
Thus, the scope of the invention should be determined by the appended claims and their legal equivalents, rather than by the examples given.
|Cited Patent
|Filing date
|Publication date
|Feb 22, 1971
|Nov 14, 1972
|Ivan A Gradisar
|Partial weight bear warning device
|Sep 29, 1971
|Feb 12, 1974
|E Pfeiffer
|Device for sensing and warning of excessive ambulation force
|Jul 22, 1974
|Aug 10, 1976
|Smithkline Corporation
|Load signaling device for a patient's foot
|Jun 25, 1979
|Nov 4, 1980
|said Andre L. Haski
|Rate measuring device for joint and/or muscular performance
|Mar 10, 1980
|Mar 23, 1982
|Instrumentarium Oy
|Apparatus in medicine for the monitoring and or recording of the body movements of a person on a bed, for instance of a patient
|Feb 27, 1981
|May 3, 1983
|Douglas David W
|Method and apparatus for detecting apnea
|Feb 24, 1982
|Jan 24, 1984
|The Langer Biomechanics Group, Inc.
|Flexible force sensor
|Mar 14, 1983
|Mar 12, 1985
|The Langer Biomechanics Group, Inc.
|Flexible force sensor
|Jan 27, 1983
|Jul 23, 1985
|Imnetec Inc.
|Device for exercise and/or therapy
|Jan 30, 1984
|Oct 14, 1986
|Lloyd Stephen R
|Sleep posture monitor and alarm system
|Jan 16, 1985
|Mar 3, 1987
|Goforth William P
|Multi-event notification system for monitoring critical pressure points on persons with diminished sensation of the feet
|Oct 16, 1986
|May 24, 1988
|Chattanooga Corporation
|Force sensing insole for electro-goniometer
|Mar 24, 1986
|Aug 1, 1989
|Key Concepts, Inc.
|Capacitive pressure-sensing method and apparatus
|Mar 9, 1987
|Aug 21, 1990
|Benjamin Dextronics Pty. Limited
|Grip rate measurement
|May 23, 1990
|Oct 29, 1991
|Jo Jan Sportsequip Co.
|Device and method for fine tuning a compound archery bow
|Oct 19, 1990
|Jan 14, 1992
|Echols Wilford R
|Keep off your back alarm
|Feb 20, 1991
|May 11, 1993
|Baltimore Therapeutic Co.
|Device for inducing and registering imbalance
|Jan 14, 1993
|Jun 28, 1994
|Fullen Systems, Inc.
|System for continuously measuring forces applied to the foot
|Apr 13, 1988
|Oct 26, 1989
|Brunner Wolfgang
|Device to measure the shape of surfaces
|Oct 31, 1991
|May 6, 1993
|Anselm Dipl.-Ing. Futterer (Fh), 7483 Inzigkofen, De
|Electronic circuit for foot pressure warning appts. - generates warning signal when pressure detected by sensor mounted in heal of plaster cast exceeds limit
|Jul 9, 1990
|Mar 6, 1991
|AMOENA Medizin-Orthopädie-Technik GmbH
|Pressure detecting device responsive to load at different regions of the human body, especially at the sole of a foot
|Title not available
|"F-SCAN®: Introducing F-SCAN®," Tekscan, Inc. (Date unknown).
|Brodsky et al, "Objective Evaluation of Insert Material for Diabetic and Athletic Footwear", Foot & Ankle / vol. 9, No. 3 / Dec. 1988, pp. 111-116.
|Brodsky et al, Objective Evaluation of Insert Material for Diabetic and Athletic Footwear , Foot & Ankle / vol. 9, No. 3 / Dec. 1988, pp. 111 116.
|Combined Meeting of the Orthopaedic Research Societies of USA, Japan & Canada: "A New Flexible Sensor to Evaluate Dynamic In-Shoe Plantar Pressure During gait and Its Use to Measure the Effect of Shoe Accommodation Devices In Severe Metatarsalgia," Ambrey, et al., p. 1, date unknown.
|Combined Meeting of the Orthopaedic Research Societies of USA, Japan & Canada: A New Flexible Sensor to Evaluate Dynamic In Shoe Plantar Pressure During gait and Its Use to Measure the Effect of Shoe Accommodation Devices In Severe Metatarsalgia, Ambrey, et al., p. 1, date unknown.
|E.Q., Inc. 600 Galahad Road, Plymouth Meeting, PA, 19462, "Engineering tools for rehabilitation therapy", date unknown.
|E.Q., Inc. 600 Galahad Road, Plymouth Meeting, PA, 19462, Engineering tools for rehabilitation therapy , date unknown.
|Endicott et al. "Leg Load . . . handicapped", Medical and Biological Engineering, May 1974, pp. 318-320.
|Endicott et al. Leg Load . . . handicapped , Medical and Biological Engineering, May 1974, pp. 318 320.
|F SCAN : Introducing F SCAN , Tekscan, Inc. (Date unknown).
|Kirkwood et al., "A Flexible Printed Circuit Board for Monitoring Patterns of Foot Loading," ICAART 88-Montreal, pp. 488-489, date unknown.
|Kirkwood et al., A Flexible Printed Circuit Board for Monitoring Patterns of Foot Loading, ICAART 88 Montreal, pp. 488 489, date unknown.
|Maale, et al., "Pressure Monitoring Under Insensate Feet", Proceedings of the Ninth Annual Conference of the IEEE/Engineering in Medicine and Biology Society, The Boston Park Hotel, Boston, MA, Nov. 13-16, 1987.
|Maale, et al., Pressure Monitoring Under Insensate Feet , Proceedings of the Ninth Annual Conference of the IEEE/Engineering in Medicine and Biology Society, The Boston Park Hotel, Boston, MA, Nov. 13 16, 1987.
|Miyazaki et al., "Capacitive transducer for Continuous Measurement of Vertical Foot Force", Med. & Biolog. Eng & Computing, 1984, 22, pp. 309-316.
|Miyazaki et al., Capacitive transducer for Continuous Measurement of Vertical Foot Force , Med. & Biolog. Eng & Computing, 1984, 22, pp. 309 316.
|R. Craike et al., "The Limb Load Monitor: An Augmented Sensory Feedback Device", Devices and Systems for the Disabled, Apr. 29&30, 1975, Sponsored by Krusen Center for Research & Engineering at Moss Rehabilitation Hospital, pp. 18-24.
|R. Craike et al., The Limb Load Monitor: An Augmented Sensory Feedback Device , Devices and Systems for the Disabled, Apr. 29&30, 1975, Sponsored by Krusen Center for Research & Engineering at Moss Rehabilitation Hospital, pp. 18 24.
|Rose, et al. "A Method of Measuring Foot Pressures Using A High Resolution, Computerized Insole Sensor: The Effect of Heel Wedges on Plantar Pressure Distribution and Center of Force", Biomechanical Studies, p. 10, date unknown.
|Rose, et al. A Method of Measuring Foot Pressures Using A High Resolution, Computerized Insole Sensor: The Effect of Heel Wedges on Plantar Pressure Distribution and Center of Force , Biomechanical Studies, p. 10, date unknown.
|Sensors, May 1991, pp. 21 25 Tecscan Inc.
|Sensors, May 1991, pp. 21-25 Tecscan Inc.
|W. Hyman et al., RESNA, 13th Annual Conference, 1990, pp. 201, 202.
|Citing Patent
|Filing date
|Publication date
|Dec 2, 1997
|Mar 7, 2000
|Preventive Medical Technologies, Inc.
|Capacitative sensor
|Nov 6, 1998
|Nov 28, 2000
|Capita Systems, Inc.
|Headset for EEG measurements
|May 3, 1999
|May 15, 2001
|Promxd Technology, Inc.
|Ergonomic systems and methods providing intelligent adaptive surfaces and temperature control
|Jun 6, 2000
|Mar 26, 2002
|Technoskin, Llc
|Wearable effect-emitting strain gauge device
|Mar 15, 2002
|Apr 29, 2003
|Homayoon Kazerooni
|Device and method for wireless lifting assist devices
|Mar 20, 2002
|Jan 27, 2004
|Homayoon Kazerooni
|Device and method for wireless material handling systems
|Jun 30, 2001
|Aug 10, 2004
|Koninklijke Philips Electroncs N.V.
|Electronic assistant incorporated in personal objects
|Jan 24, 2003
|Aug 9, 2005
|Sensorpad Systems Inc.
|Method and system for detecting and displaying the impact of a blow
|Jul 13, 2005
|Jun 10, 2008
|Sensorpad Systems Inc.
|Method and system for detecting and displaying the impact of a blow
|Dec 15, 2009
|Debeliso Mark
|Grip force transducer and grip force assessment system and method
|May 8, 2007
|May 31, 2011
|Robert Bosch Gmbh
|Method for the diagnosis of, and control device for controlling a motor vehicle
|Jun 18, 2010
|Jul 12, 2011
|Cardiogrip Iph, Inc.
|Method for carrying out isometric exercise regimen
|Jul 18, 2008
|Sep 27, 2011
|össur hf.
|Wearable device having feedback characteristics
|Apr 15, 2005
|Apr 3, 2012
|University Of Massachusetts
|Device and method for quantifying edema
|Apr 17, 2008
|Apr 23, 2013
|University Of Massachusetts
|Device and method for quantifying edema
|Apr 27, 2010
|Nov 19, 2013
|Ec Service Inc.
|Systems and methods for providing a self deflating cushion
|Jul 18, 2008
|Feb 25, 2014
|össur hf.
|Wearable device having feedback characteristics
|Jul 6, 2012
|Jan 27, 2015
|Msys Ag
|Apparatus and method for detecting the hand force of the hand pressure
|Jan 28, 2014
|Aug 11, 2015
|össur hf.
|Wearable device having feedback characteristics
|Jan 24, 2003
|Nov 27, 2003
|Reinbold Kirk A.
|Method and system for detecting and displaying the impact of a blow
|Jan 7, 2004
|Jul 22, 2004
|Mccall Catherine A.
|Compositions and methods related to canine IgG and canine IL-13 receptors
|Jul 13, 2005
|Nov 10, 2005
|Reinbold Kirk A
|Method and system for detecting and displaying the impact of a blow
|Apr 15, 2005
|Feb 9, 2006
|Billiar Kristen L
|Device and method for quantifying edema
|May 29, 2002
|Jul 27, 2006
|Kotaro Hayashi
|Pressure-sensitive sensor and monitor using the pressure-sensitive sensor
|Jun 5, 2008
|Dec 11, 2008
|24/8 Llc
|Methods and apparatuses for measuring pressure points
|Jan 24, 2008
|Jan 29, 2009
|Debeliso Mark
|Grip force transducer and grip force assessment system and method
|Apr 17, 2008
|Feb 12, 2009
|University Of Massachusetts
|Device and method for quantifying edema
|Jan 8, 2009
|Jan 7, 2010
|Esoles, L,L.C.
|Intelligent orthotic insoles
|Feb 20, 2008
|Jul 1, 2010
|Koninklijke Philips Electronics N.V.
|Shear force and pressure measurement in wearable textiles
|Jun 18, 2010
|Oct 7, 2010
|Cardiogrip Iph, Inc.
|Method for carrying out isometric exercise regimen
|Jan 24, 2013
|Bernhard Brinkhaus
|Apparatus and method for detecting the hand force of the hand pressure
|May 5, 2000
|Mar 27, 2002
|Carl M. Smith
|Apparatus for monitoring and displaying exertion data
|Oct 8, 2004
|Oct 18, 2006
|Sierra Scientific Instruments, Inc.
|High resolution solid state pressure sensor
|Dec 9, 2008
|Mar 23, 2016
|Univ Tsukuba
|Detector for position of gravitational center and wearing-type motion assisting device equipped with detector for position of gravitational center
|Dec 15, 2015
|Jun 22, 2016
|Wireless connected indoors slipper and wireless connected footwear and associated detection methods
|Dec 1, 1998
|Jun 10, 1999
|Preventive Medical Technologies
|Capacitive force sensor
|Jun 20, 2002
|Jan 16, 2003
|Koninklijke Philips Electronics N.V.
|Electronic assistant incorporated in personal objects
|Jun 20, 2002
|May 1, 2003
|Koninkl Philips Electronics Nv
|Electronic assistant incorporated in personal objects
|Feb 7, 2008
|Aug 21, 2008
|Van Der Zanden Hendrikus Petro
|System, method and gripping element for giving tactile feedback
|Jun 5, 2008
|Dec 18, 2008
|24Eight, Llc
|Methods and apparatuses for measuring pressure points
|U.S. Classification
|600/595, 361/291
|International Classification
|Cooperative Classification
|European Classification
|Feb 20, 1998
Owner name: PREVENTIVE MEDICAL TECHNOLOGIES, INC., PENNSYLVANI
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNOR:GOLDMAN, ROBERT J.;REEL/FRAME:009005/0384
Effective date: 19980212
|Mar 2, 2001
|Fee payment
Year of fee payment: 4
|Mar 23, 2005
|Maintenance fee reminder mailed
|Aug 15, 2005
|Fee payment
Year of fee payment: 8
|Aug 15, 2005
|Surcharge for late payment
Year of fee payment: 7
|Mar 9, 2009
|Maintenance fee reminder mailed
|Sep 2, 2009
|Lapse for failure to pay maintenance fees
|Oct 20, 2009
|Expired due to failure to pay maintenance fee
Effective date: 20090902


text:
August 29, 2016
First PlaceTom Connell and Chris McGuire5 fish, 22.15 lbs.$1800(Tom Connell - Big Fish Winner - 7.48 lbs.)
Second PlaceMel Hodges and Brent Fortenberry5 fish, 16.08 lbs.$900
Third PlaceEric and John Plowman5 fish, 14.52 lbs.$500
Click Here to access the online Public Inspection File
Viewers with disabilities can get assistance accessing this station's FCC Public Inspection File by contacting the station with the information listed below. Questions or concerns relating to the accessibility of the FCC's online public file system should be directed to the FCC at 888-225-5322, 888-835-5322 (TTY), or email@example.com.

subset: commoncrawl


text:
$1,295—$1,695 1—2 bed 1511 South Mills Avenue, Lodi CA, 95242
from $1,295 1—2 bed
1511 South Mills Avenue, Lodi, 95242
Commute to Downtown Lodi
The rent at Lakeview Apartments ranges from $1,295 for a one bedroom to a $1,695 two bedroom. Apartments in this building are about 29% more expensive than the Lodi average. This is a pet-friendly apartment.
This apartment is in Lodi. Nearby parks include Beckman Park, Beckman Park and Kofu Park.
Lakeview Apartments has a Walk Score of 28 out of 100. This location is a Car-Dependent neighborhood so most errands require a car.
Beautifully Renovated Interiors *select apts, Wood-Burning Fireplaces *select apts, Washer & Dryer, or Connections, Central Heat and Air Conditioning, Two Sparkling Pools with Sundeck and Spa, Lake Access *select apts, Private Garages *select apts, Covered Carport Parking, Fitness Center, 24/7 Maintenance Emergency Services, Easy access to highway 12, 99 & I-5, Minutes away from Shopping, Dining & MORE!
Pets allowed:
Lodi, California, 95242
Most errands require a car.
About your score
Explore how far you can travel by car, bus, bike and foot from Lakeview Apartments.
Bus lines:
23 Stockton - Lodi
0.4 mi
93 Stockton - Lodi
0.4 mi
723 DTC - Kaiser - Lodi
0.4 mi
View all Lodi apartments on a map.
Popular apartment searches include single family, pool and fireplace.
Search for homes in Lodi or find a Lodi real estate agent who knows walkable neighborhoods. Links will open on our partner site Redfin.com.
Lakeview Apartments is in the city of Lodi, CA. Lodi has an average Walk Score of 55 and has 62,134 residents.Learn More About Lodi


text:
Bradford head coach Mick Potter described the decision to dock the Bulls six points for breaching the game's insolvency regulations as a "kick in the stomach".
The ailing Super League club had been bracing themselves for the announcement since entering administration on June 26 but Potter insists it is harsh on the players. The points deduction takes immediate effect and drops the Bulls out of the play-off positions ahead of Sunday's trip to Warrington.
Bradford have won three of their last four matches despite the massive uncertainty over the club's future to raise hopes of a top-eight finish for the first time in four years. Potter, who is working on a voluntary basis after being among 16 full-time staff made redundant by administrator Brendan Guilfoyle last month, said: "It's disappointing."
He went on: "It's no surprise. We understood that's what happens when you go into administration, it's part of the process for this type of thing.
"But it's a massive kick in the stomach for the players, who have done all they possibly could to get results, and also the fans, who have done over and above what they had to do. It's just disappointing, sitting on the sidelines waiting for your punishment, especially when you've not instigated any of it."
Bradford received the maximum penalty laid down in the Rugby Football League's operational rules.
The RFL board of directors exercised some flexibility by docking Wakefield and Crusaders four points when they entered administration on the eve of the 2011 season after taking into account the new owners' willingness to pay off some of the debts.
Although Guilfoyle has found a potential buyer, any deal is some way off being completed.
He has passed on details of a formal, written offer from the ABC consortium, a group of local Asian businessmen, to the RFL but it is conditional on them being able to buy back the lease to Odsal Stadium and receiving a guarantee on the Bulls remaining in Super League.
He has until this Friday to avoid putting the club into liquidation, although he has the option of a further extension of the deadline if he feels he can complete a deal.


text:
Definition of Malaya in English:
A former country in Southeast Asia that consists of the southern part of the Malay Peninsula and some adjacent islands (originally including Singapore) and that now forms the western part of the federation of Malaysia and is known as West Malaysia. The area was colonized by the Dutch, Portuguese, and the British, who eventually dominated; the several Malay states federated under British control in 1896. The country became independent in 1957, and the federation expanded and became Malaysia in 1963.
Words that rhyme with Malayaabaya, betrayer, conveyor, Eritrea, flayer, Freya, gainsayer, layer, Marbella, Maya, Mayer, Nouméa, obeyer, payer, player, portrayer, prayer, preyer, purveyor, slayer, sprayer, stayer, strayer, surveyor, waylayer, weigher
For editors and proofreaders
Syllabification: Ma·la·ya
Definition of Malaya in:
What do you find interesting about this word or phrase?
Comments that don't adhere to our Community Guidelines may be moderated or removed.
= trending


text:
The Insider: Gluten-Free Cafe
Published: Monday, May 5, 2014 at 11:45 p.m.
Last Modified: Monday, May 5, 2014 at 11:45 p.m.
Healthy Habits, a vitamin shop, plans on opening an organic, gluten-free cafe next to its current Lakeland location, 4760 S. Florida Ave., next week or even as early as Saturday, said Mary Jaramillo, an employee at Healthy Habits. The cafe, near Lake Miriam, will serve gluten-free soups and salads and would have a smoothie bar inside.
Reader comments posted to this article may be published in our print edition. All rights reserved. This copyrighted material may not be re-published without permission. Links are encouraged.


text: Hi everybody, i´m brand new to WordPress 1.5 and after a few searches here in the "calendar" tag, i didn´t found exactly what i want to know. I use Blix as my default theme. My question is: how i can make the calendar available in the main page, the index?


text:
Premier Plastic Surgery Center of New Jersey
Board Certified Plastic Surgeon Dr. Brian Glatt, and the staff of Premier Plastic Surgery Center of New Jersey, invite you to explore the possibility of looking as good as you feel. Making the decision to undergo cosmetic or reconstructive surgery is an important one. You deserve the outstanding, yet natural results experienced by Dr. Glatt's plastic surgery patients who travel to his Morristown, New Jersey practice from New York, New Jersey and beyond for his expertise and exceptional skill.
At Premier Plastic Surgery Center of New Jersey, you will find a warm and caring environment where the comfort and privacy of each patient is paramount. Dr. Glatt encourages honest and forthright discussion between himself and his patients in order to develop a specific treatment plan to create natural results that will exceed their expectations. After your consultation with Dr. Glatt, you will see why many patients return for additional procedures and frequently refer friends and family to him as well. Results are further enhanced by medically supervised skin care, and preoperative/postoperative vitamin supplements, all of which are provided on-site by Dr. Glatt's caring, professional staff.
Dr. Brian Glatt possesses exceptional technical skills and the expertise needed to produce your desired results. Equally important, he has the unique ability to help patients realize their vision, whether it be through improved body contouring, facial aesthetic balance and harmony, or reconstruction to restore a sense of wholeness. Contact us to schedule your consultation at Premier Plastic Surgery Center of New Jersey, where discretion and excellence go hand in hand. Let Dr. Glatt help you to look and feel your best!
Due to the high demand for CoolSculpting we are pleased to offer DualSculpting. The use of two CoolSculpting machines enables us to treat both sides of the body or limbs at the same time! Getting rid of unwanted fat without surgery is now possible to do in half the time.
Watch Dr. Glatt's
Informational Videos
Watch this collection of informational videos; learn more about Dr. Glatt, his procedures and his staff.


text:
Woman indicted for witness intimidationJuly 22,2013The Associated Press
ST. JOHNSBURY - A federal grand jury has indicted 21-year-old Vermont woman on a charge of witness intimidation for allegedly burning down a camp in retaliation against a drug informant.
Rebecca Ray of Barton was indicted Thursday for setting fire to the camp in Brownington on July 11, the day after her boyfriend, 34-year-old Matthew Pure of Barton, and her boyfriend's brother were arrested for allegedly selling heroin to the informant.
The Caledonian Record (http://bit.ly/12VzHO4 ) reports that state police Detective Jason Luneau wrote in an affidavit that Ray said she dumped gasoline on the house and lit it on fire but knew nobody was there at the time.MORE IN This Just InHARTFORD, Conn. Full StoryWHITE RIVER JUNCTION A Springfield man will spend 17 months in jail and the next decade on... Full StoryEvery year thousands of school children are transported in school buses. Full Story
- Most Popular
- Most Emailed


text:
- Report: #88308
Report - Rebuttal - Arbitrate
Complaint Review: Avlimil - Berkley Pharmaceudicals
Avlimil - Berkley Pharmaceudicals5649 Creek Rd. Cincinati, Ohio U.S.A.
Avlimil, Berkley Pharmaceudicals ripoff! I cancel billing, but they continue to ship and bill me. Cincinati Ohio
Ripoff Report
About you?
Ripoff Report
A business' first
line of defense
on the Internet.
If your business is
willing to make a
commitment to
customer satisfaction
Click here now..
Does your business have a bad reputation?
Fix it the right way.
Corporate Advocacy Program™
I called company several times, waited so long a few times that I finally hung up. finally got through and they cancelled me over phone, waiting to see if I receive more shipments.
west valley city, Utah
This report was posted on Ripoff Report on 04/20/2004 10:30 AM and is a permanent record located here: http://www.ripoffreport.com/r/Avlimil-Berkley-Pharmaceudicals/Cincinati-Ohio-45242/Avlimil-Berkley-Pharmaceudicals-ripoff-I-cancel-billing-but-they-continue-to-ship-and-b-88308. The posting time indicated is Arizona local time. Arizona does not observe daylight savings so the post time may be Mountain or Pacific depending on the time of year.
Ripoff Report has an exclusive license to this report. It may not be copied without the written permission of Ripoff Report.
Click Here to read other Ripoff Reports on Avlimil - Berkley Pharmaceudicals
If you would like to see more Rip-off Reports on this company/individual, search here:Search Tips
In order to assure the best results in your search:
- Keep the name short & simple, and try different variations of the name.
- Do not include ".com", "S", "Inc.", "Corp", or "LLC" at the end of the Company name.
- Use only the first/main part of a name to get best results.
- Only search one name at a time if Company has many AKA's.
Advertisers above have met our
strict standards for business conduct.


text:
Cal BRE#: 01314026
- (858) 692-0085 mobile
- (619) 460-6600 office
- La Mesa 9332 Fuerte Dr, La Mesa, CA 91941
- Set Brian as your preferred agent
Connect with Brian
Certified Residential Specialist (CRS), Certified Neighborhood Specialist (CNS), Senior Real Estate Specialist (SRES), NAR certification for online professionalism (e-PRO), Certified Distressed Property Expert (CDPE), REO Specialist Designation (RSD)
More about Brian Westre
San Diego "Native"
I have loved living in San Diego county for most of my life. I moved to San Diego in 1969 and I have either lived, or worked, in most every area of the county. I have sold homes as far north as Oceanside and Vista, and as far south as Chula Vista. Let me put that experience to work for you either to help you find the right home in the best neighborhood for you, or help you get the most for your home when selling.
Certified Residential Specialist and Relocation Specialist
I take great pride in the fact that I have earned the CRS designation. Only 4% of independent agents nationwide are able to carry that title. To qualify one must have accrued enough practical experience (sold homes) and taken several hard hitting courses, and successfully passed many tests along the way. I worked hard for it and would love to let it work for you. As they say, "Get Serious, Get a CRS"Certified Relocation Specialist: I have attained certification to handle all the corporate relocation needs of my clients by virtually any employer sponsored move. Years ago, when I was in the corporate world, I was relocated several times across the country and across the ocean. I know what a strain a transfer and move put on a family and know how to ease that transition.
CDPE, RCS-D, SRES, e-PRO and CNS
The real estate landscape changes all the time and the only way to stay current is to take continuing education courses and designations.
CDPE is Certified Distressed Property Expert. Someone that is in a financially distressed situation needs expert guidance on how to navigate the course of a potential short sale.
RCS-D is Real Estate Collaborative Specialist Divorce. Going through a divorce is very difficult personally, and financially. How to handle the number one asset for the family is critical.
SRES gave me insight into the unique personal and financial situations for the growing senior market in the country and how to best serve their needs.
e-PRO is dedicated to giving me the tools to best serve my clients with their home sale by maximizing the potential of the internet. The internet is a powerful tool that has surpassed print advertising as the vehicle to maximize exposure for homes for sale.
CNS is one of my favorites as negotiating takes place throughout the transaction, not just during the discussion of price.
Leadership and Achievements
I believe in giving back where possible and stay very active in the industry and the community. I was awarded the REALTOR of the Year for San Diego in 2011 and was a finalist in 2007 and 2009; I was Rookie of the Year for San Diego County my first year in the business. I am on the Board of Directors locally at the San Diego Association of Realtors and at the state level at the California Association of Realtors as well as serving on various other committees at both. I am past President of the Poway Real Estate Professionals and Past President of the Congregation Council at my church.
Real estate agents affiliated with Coldwell Banker Residential Brokerage are independent contractor sales associates and are not employees of Coldwell Banker.


text:
I have no idea why, but the 'Humanize' graph tab doesn't work for me. Clicking on it shows first an iframe of the wp-admin dashboard (the frame is shown below the graph, briefly) and then goes on to redirect me happily to... a blank page.
Not to disparage the kind opportunity granted so as to enjoy the ageless tranquility of a clutterless, pristine page, but I'm not sure I'm having an appropriate response compared to the intended effect, as it makes me restless, somewhat agitated even.
Any suggestions other than seeking prescription tranquilizers? (I suppose not clicking the Humanize link would be another, more high-tech solution but that's a notch above my comprehensive limitations.)


text:
There are no items in your cart at this time.
Celebrating 30 years of providing fashionable comfort footwear and excellent service— Footwear etc. • Established 1986 —
Sign up to receive new arrival, sale & special event info from us.
Note: we do not share your personal information with anyone. Period.
"Buying from Footwear etc. was quite pleasurable. The staff member was courteous, knowledgeable, and very helpful. The delivery was timely. I would recommend Footwear etc. to all shoppers."
©2016 Footwear Etc. All Rights Reserved. For any questions or assistance call 1-800-720-0572.


text:
Issue No.06 - November/December (2008 vol.25)
pp: 64-70
Željko Obrenovic , Technical University, Eindhoven
Dragan Gaševic , Athabasca University
Anton Eliëns , Vrije Universiteit, Amsterdam
DOI Bookmark: http://doi.ieeecomputersociety.org/10.1109/MS.2008.162
Using opportunistic software development principles in computer engineering education encourages students to be creative and to develop solutions that cross the boundaries of diverse technologies. A framework for opportunistic software development education helps to create a space in which students can combine systems that were never meant to work together or even to be reused, and thus produce innovative ideas and solutions. A case study involving students in a course on intelligent human-computer interaction design demonstrates the approach, and the authors discuss some lessons learned.
opportunistic software development, software engineering education, creativity support
Željko Obrenovic, Dragan Gaševic, Anton Eliëns, "Stimulating Creativity through Opportunistic Software Development", IEEE Software, vol.25, no. 6, pp. 64-70, November/December 2008, doi:10.1109/MS.2008.162


text:
Life Insurance
Life Insurance Tools
Term Life Insurance
Whole Life Insurance
Final Expense Insurance
Life Insurance Companies
Auto Insurance
Car Insurance Tools
Non Owner Car Insurance
Cheap Car Insurance
Car Insurance for High Risk Drivers
Car Insurance Companies
Health Insurance
Health Insurance Tools
Health Insurance Open Enrollment
COBRA Health Insurance
Domestic Partner Benefits
Health Insurance Companies
Home Insurance
Home Insurance Tools
Home-Auto Bundling
Renters Insurance
Condo and Co-op Insurance
Home Insurance Companies
Best Insurance Companies
Best Car Insurance Companies
Best Life Insurance Companies
Best Home Insurance Companies
Best Health Insurance Companies
Car Insurance Comparison
Car Insurance Rates by State
Car Insurance Discount Detector
Life Insurance Calculator
Financial Strength Ratings
En Español
Get disability insurance to protect you and your family.
Deal with top-rated companies
Get quotes online in just minutes
Choose the best option
Protect your assets
Site Reviews
"...we'd recommend you do your insurance shopping here..."
"The best Web site I've found..."
-The Dallas Morning News
"...this site is flush with useful features."
"...keeps a data base of about 350 insurance companies."
-Good Housekeeping
"You've cut my insurance premium in half!"
Katherine Havener
Toluca Lake, CA
About Us
Contact Us
Press Room
Privacy Policy
Terms of Service
Site Map
Copyright © 1984-2016 Quinstreet, Inc.
Insure.com is a part of the Insurance.com family
Disclaimer: The insurance products on Insure.com are from companies from which QuinStreet may receive compensation. Compensation may impact where products appear on Insure.com (including the order in which they appear). QuinStreet does not include all insurance companies or all types of products available in the marketplace.


text:
(775) 246-4477
10051 Hwy 50 • Mound House, NV
Closed Sunday
Cheese Burger with Fries
1/4lb patty with American cheese, lettuce & tomato
with your choice of French or curly fries.
Daddy D Dog
Big Nathans frank with mustard, onions & choice of
French or curly fries.
*No substitutes or add-ons
Sarah's Chips & Salsa – $3.95
Homemade tortilla chips & mild salsa
Jose's Nachos – $5.95
Chips topped with cheese, olives, salsa, jalapeño, guacamole, and sour cream
add seasoned beef or chicken – $1.50
Seasoned Curly Fries – $3.95
Six's Shoestring Fries – 3.95
Ava's Pickle Fries – $3.95
Strips of dill pickles breaded and deep fried
Garlic Fries – $3.95
Our shoestring fries tossed in garlic sauce
Chili Cheese Fries – $5.95
Your choice of shoestring or seasoned curly fries topped with our famous chili, cheddar, and jack cheese
Sweet Potato Tots – $3.95
Served with maple syrup
Jalapeño Poppers – $5.95
Cheese filled peppers fried & crispy
Mozzarella Sticks – $5.95
Served with our homemade marinara
Roxy's Onion Rings – $3.95
Your choice of sauce
Zucchini Sticks – $5.95
Served with ranch dressing
Potato Skins – $7.95
Stuffed with cheddar cheese, bacon bits, & green onions with sour cream on the side
Willow's Wings – 8pc. $6.95 / 16pc $12.95
Mild, Medium, Hot, Teriyaki, Honey BBQ, Garlic
Shrimp Cocktail – $5.95
Harley's Sriracha Shrimp – $6.95
Grilled Shrimp Skewers – $7.95
All sandwiches served with choice of bread, lettuce, tomato & choice of French fries, sweet potato tots, soup, or salad
Add Avocado 75¢
Add Cheese 50¢
Andy Kaufman Grilled Cheese – $4.95
Choice of cheese
Johnny Buss Lakers BLT – $5.95
Choice of applewood or jalapeño bacon
Sindy's Chicken Avocado Sandwich – $5.95
Grilled chicken, avocado on a potato bun
Honey's Buffalo Chicken Sandwich – $3.95
Chicken tenders tossed in buffalo sauce and served on a French roll
Caressa's Club Sandwich – $6.49
A triple decker with turkey & bacon
Molly's Grilled Chicken Club – $6.49
Grilled chicken with bacon, on a potato bun, choice of cheese
Hollywood's Pulled Pork – $6.95
Seasoned shredded pulled pork with homemade BBQ sauce. Served on a potato bun with fries and
cole slaw on the side
Clarissa's Veggie Sandwich – $5.95
Lettuce, avocado, tomato and cucumber and mushrooms
Entice's Salami Sandwich – $4.95
Dry salami & choice of cheese
Julia's Smoked Ham & Cheese Sandwich – $6.95
Ham, choice of cheese, choice of bread
Tom's Turkey Sandwich – $5.95
Smoked turkey breast
Kris's Tuna Salad Sandwich – $5.95
Tuna dill relish, red onion, bacon, mayo
Chase's Chicken Salad Sandwich – $5.95
Chicken, celery, red onion, bacon, mayo
Cumisha's French Dip – $6.95
Chicken, celery, red onion, bacon, mayo
Thin sliced roast beef on a French roll with AuJus
Add Cheese: 50¢
Daddy D only uses 100% grain fed beef, charbroiled to perfection with lettuce, tomatoes, pickles and onions.
Choice of French, curly seasoned fries, potato salad, cottage cheese or soup.
Onion Rings add $1.00 • Double Patty add $2.00
Add Avocado 50¢
Daddy D's Hof Burger – $7.50
Two 1/3 pound Angus beef patties topped with sautéed mushrooms, onions, swiss cheese, lettuce, and tomato served on a French roll served with choice of sides
Kandi's Hamburger – $6.45
Bridget's Cheeseburger – $6.95
Choice of cheese
Taylor's Protein Burger – $6.45
Our charbroiled burger wrapped in lettuce
Faith's Green Chili Burger – $7.25
Roasted sweet peppers
Rain's Bacon Cheeseburger – $7.25
Choice of applewood or jalapeño bacon. Choice of cheese
Holly's Western Burger – $7.25
Topped with onion rings, bacon & our homemade BBQ sauce & cheddar cheese
Freaky Geek's Mushroom & Swiss Burger – $7.25
Sautéed mushrooms, Swiss cheese
Heidi Fleiss Veggie Burger – $6.95
Garden veggie burger with choice of cheese
Randy's Chili Cheeseburger – $7.25
Scoop of Chili, choice of cheese
Mötley Crüe Bacon & Bleu – $7.25
Choice of bacon
Alice's Teriyaki Burger – $7.25
Grilled pineapple, teriyaki sauce, pepper jack cheese
Pantera's Patty Melt – $6.95
Patty covered in Swiss & grilled onions on rye or sourdough
Brooke Taylor Peanut Butter,
Pepper Jack & Bacon Burger – $7.50
Creamy peanut butter with choice of bacon
Stacy's Guacamole Burger – $7.25
Burger topped with pepper Jack cheese and our guacamole
Ivy Mae's Philly Melt – $6.95
Grilled peppers and onions with provolone on sourdough
Ron Jeremy's Big Combo – $7.50
Two 1/3 pound Angus beef patties topped with sautéed mushrooms, onions, swiss cheese, lettuce, and tomato served on a French roll served with choice of sides
Daddy "D" Dog – $2.69
All beef Nathan's frank with mustard & onions
Ruby Rae's Chili Dog – $2.99
Chili, cheese & onions
Domino Dog – $2.99
All beef Nathan's frank with bacon bits
Bob Zmuda's Chicago Dog – $3.99
Sweet relish, onions, dill pickle spear, tomato wedges, sport peppers & celery salt
Flavor Flav Dog – $3.99
Wrapped in applewood bacon & deep fried
Carlos Mencia Chihuahua Dog – $3.99
Wrapped in jalapeño bacon & deep fried
Italian Sausage & Peppers Dog – $4.95
Served on a French Roll with 7 peppers & provolone
Large 14" Cheese Pizza – $9.49
Additional Toppings 50¢
Chicken Alfredo Pizza – $11.95
White Alfredo sauce with cheese and chicken
Veggie Pizza – $11.95
Tomatoes, onions, bell peppers, mushrooms and onions
Meat Lover's Pizza – $11.95
Pepperoni, sausage, ham, bacon
Bob Zmuda's Chicago Dog – $3.99
Sweet relish, onions, dill pickle spear, tomato wedges, sport peppers & celery salt
Flavor Flav Dog – $3.99
Wrapped in applewood bacon & deep fried
Carlos Mencia Chihuahua Dog – $3.99
Wrapped in jalapeño bacon & deep fried
Italian Sausage & Peppers Dog – $4.95
Served on a French Roll with 7 peppers & provolone
Ground Beef
Bell Peppers
Banana Peppers
Artichoke Hearts
Diet Coke
Root Beer
Iced Tea
Iced Coffee
$2.00 Free Refills
Draft Beer – $2.25
Bottled Beer
Domestic $2.75
Specialty & Import $3.25
Wine by the Glass – $5.50
White Zinfandel
Well Drinks – $4.00
Call Drinks – $6.00
Wrapped in applewood bacon & deep fried


text:
- Report: #343111
Report - Rebuttal - Arbitrate
Complaint Review: Margaret Arrington
Margaret Arrington51 Bedford Street Bath, Maine U.S.A.
AFNI Collections Inc. Attempting to collectect a debt that is not legitimate. Bath Maine
*Consumer Comment: Advise.....
Ripoff Report
About you?
Ripoff Report
A business' first
line of defense
on the Internet.
If your business is
willing to make a
commitment to
customer satisfaction
Click here now..
Does your business have a bad reputation?
Fix it the right way.
Corporate Advocacy Program™
bath, Maine
This report was posted on Ripoff Report on 06/23/2008 09:24 AM and is a permanent record located here: http://www.ripoffreport.com/r/Margaret-Arrington/Bath-Maine-04530/AFNI-Collections-Inc-Attempting-to-collectect-a-debt-that-is-not-legitimate-Bath-Maine-343111. The posting time indicated is Arizona local time. Arizona does not observe daylight savings so the post time may be Mountain or Pacific depending on the time of year.
Ripoff Report has an exclusive license to this report. It may not be copied without the written permission of Ripoff Report.
Click Here to read other Ripoff Reports on Margaret Arrington
If you would like to see more Rip-off Reports on this company/individual, search here:Search Tips
In order to assure the best results in your search:
- Keep the name short & simple, and try different variations of the name.
- Do not include ".com", "S", "Inc.", "Corp", or "LLC" at the end of the Company name.
- Use only the first/main part of a name to get best results.
- Only search one name at a time if Company has many AKA's.
#1 Consumer Comment
AUTHOR: John - (U.S.A.)
SUBMITTED: Monday, June 23, 2008POSTED: Monday, June 23, 2008
Per the Fair Credit Reporting Act Act, I am requesting validation that [insert name] was a legal adult at the time that this alleged debt was incurred
Per the Fair Credit Reporting Act Act I am requesting written validation of this alleged debt
For AFNI:
Send them a letter via Certified Mail + Return Receipt stating:
Per the Fair Debt Collection Practices Act, I am requesting:
- written validation of this alleged debt and that said alleged debt is within the statute of limitations
- I am requesting validation that [insert name] was a legal adult at the time that this alleged debt was incurred
- Cease all verbal communications with me about this alleged debt.
Read up on the Fair Debt Collection Practices Act and understand your rights:
DO NOT pay AFNI one penny! If you do it will validate/affirm the debt as legitimate.
Are you an owner, employee or ex-employee with either negative or positive information about the company or individual, or can you provide "insider information" on this company?
Advertisers above have met our
strict standards for business conduct.