text:
|Publication number
|Publication type
|Application number
|Publication date
|Aug 22, 2000
|Filing date
|Dec 28, 1998
|Priority date
|Dec 28, 1998
|Fee status
|Also published as
|Publication number
|Keith A. Miesel, Lee Stylos
|Original Assignee
|Medtronic, Inc.
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (24), Referenced by (167), Classifications (14), Legal Events (6)
|External Links: USPTO, USPTO Assignment, Espacenet
This invention relates to the field of implantable monitoring devices and is partially suitable for long term monitoring of physiologic parameters including most particularly blood pressure from small arteries.
Currently no device is available to automatically, continuously and chronically monitor blood pressure extravascularly. Sphygmomanometers are used to externally monitor blood pressure through the use of an external cuff. In common hospital parlance, stations having what are called inaccurately `dynamap` systems (probably this misnomer is derived from the Johnson & Johnson "DINAMAP" (R) patient monitor systems, which use various blood pressure cuffs) automate these measures for discrete data point analysis. In either system type, the inflatable cuff or other pressure monitoring apparatae are bulky, impede blood flow and require operating personnel to make measurements. As a result these external systems are limited in usefulness to less than 24 hours in common usage. Numerous intravascular and paracorporeal automated pressure sensors have been developed to automate blood pressure monitoring of critically ill patients in hospital settings and for surgical patients. Typically these are composed of a moveable or dispensable membrane surface that by its associations with an electrical sensor produces a signal indicative of the pressure on the membrane. Not only is the usefulness of such devices limited to acute situations where access to the insides of a blood vessel is available, but the mechanical configurations are complex. Accordingly such devices are only able to provide blood pressure readings for a very limited period of time, and in circumstances of intense patient care, like in a hospital setting.
Using currently available internal to the circulatory system systems, at present time there is no good method for chronically measuring blood pressure on the left side of the circulatory system without actually placing a device in contact with the blood. This of course presents a site for thrombogenesis and the possibility of subsequent stroke. If no sensor is placed directly in contact with the blood the challenge is then sensing blood pressure through an arterial wall without introducing errors due to the structure of the arteries supporting all or part of the pressure load.
Further, there is no good present method for continuously sensing over a long period of weeks or months, even including devices which activate a mechanical occluder or cuff so as to provide data beyond mere systolic and diastolic values.
Perhaps the most advanced pressure sensor at the present time is described in some detail in U.S. Pat. No. 5,564,434, incorporated herein by this reference in its entirety. However, this blood pressure sensor is designed for dwelling in the heart of the patient on the heart's right side. Thus any thrombogensis created by the artificial material presence in the blood stream would have limited capacity to damage the patient that is, could not cause a stroke because any thromboses thrown by the device would end up in the patients lung. Thus this patent highlights the difficulty in achieving left side blood pressure measurements.
Other prior systems for attempting to meet some of the requirements described just above include U.S. Pat. No. 4,256,094 issued to Kapp et al, and incorporated herein by this reference in its entirety. This Kapp teaching requires a fairly complex mechanical system including a liquid pump and reservoir and programmable controller for cuffing or occluding the blood flow in the artery, difficulties which are obviated completely by the present invention.
External devices include those described in U.S. Pat. No. 3,926,179, and three which describe a watch mounted external device, U.S. Pat. Nos. 4,802,488; 4,799,49, and 4,269,193 all four hereby incorporated herein by this reference in its entirety. Accordingly, there is a need for a device to enable the continuous chronic monitoring of patient blood pressure automatically and at low cost. To provide such a device without the requirement for vascular access and which could safely monitor the left side of the vascular system as well as the venous system would be of no small additional benefit. It is also an advantage to be able to provide additional devices for deploying sensors chronically to other vessels in the body for sensing other physiologic conditions(e.g. phlebitis).
Such devices as are described herein can be applied to monitoring patients with hypertension without cumbersome discrete point analysis over long terms with many hospital or clinic visits. Dosage accuracy can be enhanced with continuous long term monitoring of blood pressure for administration of hypertensive medications. It is also helpful for interventions to many disease processes such as diabetes, renal and neurologic disease, internal medicine and transplant patients, and any condition where surveillance of patient compliance could be helpful. Not only can continuous systolic, diastolic and pulse pressure monitoring be done with a minimally invasive implant, but additional or just different physiologic signals can be monitored, leaving wide open the scope of application of devices based on this invention.
Thus, the applicability of this invention is not limited to sensing blood pressure however. It should be recognized right from the start that the invention described herein can be used to obviate the use of many sensors which otherwise would require implantation into an active artery or vein system.
For example in U.S. Pat. No. 5,409 (also incorporated in its entirety by this reference) the blood flow is measured using permanently implantable lead within a vein adjacent to an artery. However, using our inventive fixture, flow, temperature, pressure, pulse rate, an even blood oximetry, as well as any other measurement that needs to be taken by a sensor next to the blood flowing through a artery or a vein can be accomplished using a device based on the inventive concepts described herein.
By providing accurate, continuous monitoring of patient blood pressure and other physiologic measures, a profile of patient status during activities of daily living and over prolonged intervals can accurately depict the patient's true hemodynamic health status over months of use. Such automatic monitoring circumvents patient compliance and operator compliance issues which are problematic in the use of external devices. Particularly, the fixture with sensors concepts described herein can be applied to peripheral, internal and neurologic blood and other fluid pressures, flow and so forth. Carotid arterial blood pressure, pulmonary venous blood pressure and cerebral ventricular pressure associated with hydrocephalous are just three examples. The device could be used to monitor the formation of transudate and exhudate fluids, such as in conditions of subcutaneous edema. The device can of course be useful as a research tool, and may find many uses in veterinary medicine and monitoring.
FIG. 1 is a perspective view of a preferred embodiment.
FIG. 2 is a perspective of another preferred embodiment of the invention.
FIG. 3 is a cut away view of FIG.2.
FIG. 4 is a cut away view of the upper portion of the embodiment illustrated in FIG. 2 taken at line 4--4 of FIG.2.
FIG. 5 is a frontal view of the embodiment described with respect to FIG. 2.
FIG. 6 is a top view of the device of FIG. 2.
FIG. 7 is a sectional view taken at line AA of FIG. 5.
FIG. 8 is a sectional of FIG. 6 taken at line BB.
FIG. 9 is a side view of one embodiment of the sensor fixture in accord with the invention.
FIG. 10. is a back view of the fixture of FIG. 4.
FIG. 11a is a side view of another embodiment of the invention surrounding an arteriole artery or other blood vessel which is pinched off to fit into an inventive fixture in accord with a preferred embodiment.
FIG. 11b is another side view of the fixture and blood vessel of FIG. 11 with the normal flow restored.
FIG. 12 is an illustration of am implantable medical device for monitoring data produced by the inventive fixture connected to it by a lead.
FIG. 13 is a view facing an inner sensor laden surface of the tunnel or passageway space for containing a living body's blood vessel as it may be constructed in accord with a preferred embodiment of the invention.
FIG. 14 illustrates a sensor array having an emitting element, a receiving element and their position within the inventive fixture relative to a blood vessel.
FIGS. 15a-e are illustrations of arterial blood vessel locations suitable fore use with one preferred embodiment of the invention.
FIG. 16A is a perspective view of a preferred embodiment fixture configuration and connection to a lead.
FIG. 16B is a side view of the top portion of the device illustrated in FIG. 16A.
FIG. 16C is a facing view of the top portion of the device of FIG. 16A.
FIG. 17 is a side view of a preferred embodiment similar to that of FIG. 16A.
FIG. 18 is a cross sectional view of a cable or lead in accord with a preferred embodiment of the invention.
FIG. 19 is a cross sectional view of an alternative embodiment of the inventive fixture.
FIG. 20 is a perspective view of another preferred embodiment of the invention.
FIGS. 20A-E are cross sectional slices of the interior surface of the internal space, tunnel, or passageway for containing blood vessels in accord with a preferred embodiment of the invention. each is taken from section lines A-E as illustrated on FIG. 20.
FIG. 21 is a perspective view of another preferred embodiment.
FIGS. 22 and 23 are partially sectioned perspective views of two preferred embodiments.
FIG. 24 is an exploded perspective view of a preferred embodiment of the invention, illustrating one manner of connecting two components in accord with the inventive concepts described herein.
FIG. 25 is an assembled structure of the exploded view of FIG. 24.
FIG. 26 is a side view of one form of the upper part of the fixture in accord with a preferred embodiment.
FIGS. 27A, 27B and 27C are a facing view, and two sectional views of the device of FIG. 26.
A number of general and specific objectives are set forth for what this invention can provide, and also a number of general and specific features. All of these features and objectives can be met individually or in combination by the description which follows.
In one preferred embodiment the sensor system consists of a side-sensing chronically-implantable absolute pressure sensor incorporated into the inventive structure. This sensor may be similar or identical to the one described in U.S. Pat. No. 5,564,434. To avoid unneeded prolixity, the just above mentioned and all subsequently mentioned patents are incorporated herein in their entireties by this reference sentence, and this sentence is operative to so incorporate all such mentioned patents at each such mention without the need of repeated recitation of this statement or its equivalent at the location of such mention.
The pressure sensor is packaged inside or in close and operative association with a clamp type structure which we often refer to as a fixture which holds an artery against the sensor diaphragm or against the pressure transferring medium (such as silicone or medical grade adhesive) capable of accurately transferring or transporting pressure from the artery side wall to the sensor diaphragm. The shape of the artery channel through the fixture's clamp-like structure is such that it slightly deforms the artery to provide a flattening of the artery (in cross section) in the region where it passes over the sensor diaphragm or the silicone adhesive between the sensor diaphragm and the artery. This flattening of the artery causes pressure interior to the artery to be supported by the sensor structure rather than the artery wall structure. This causes the arterial pressure to be passed through the artery wall to be detected by the pressure sensor.
This effect can be enhanced by varying the shape of the tunnel or channel through which the vessel is placed as described in some detail below.
Calibration can be effected if required once the device is in place which may account for variations in wall thickness or other features of the vessels or for other reasons known in the pressure sensor art. Another feature of the clamp structure is on the opposite side (in this preferred embodiment) from the pressure diaphragm. What can be described as a slit is provided along the length of the clamp structure, this slit being substantially narrower than the diameter of the artery. The artery is inserted into the clamp by squeezing the artery (or clamping it off upstream) and passing it's collapsed form through the slit and than releasing the artery and allowing it to spring out to conform to the shape of the insides of the clamp's channel. The pressure inside the artery will tend to keep the artery confined within the clamp channel without the need of any additional fastening hardware, thus reducing the likelihood of complications.
The artery can further be secured within the fixture (which we sometimes call an artery cuff) by forming the access slit in a serpentine shape, that is having a curved cut through which the filled vessel cannot easily slip out, once it has become secured within the fixture.
Thus, this invention allows for the measuring for pressure within a artery or other vessel (like a vein) through a vessel wall by causing a flattening of the vessel at the point of sensing. It teaches the method of inserting and retaining a vessel within the clamp's structure to hold the vessel against the pressure sensor. It provides a second pressure sensor on the backside of the arterial pressure sensor this second sensor not being in contact of the vessel which can be used to measure ambient pressure for calculation of "gage" pressure. This invention provides for chronic measurement for arterial pressure through the arterial wall without needing to contact the inside of the blood vessel.
In other preferred embodiments, other sensors for sensing other physiologic parameters could either supplement the pressure sensor or replace it as described elsewhere in more detail. For examples, temperature could be sensed and used for various purposes such as was described for rate control in pacing in U.S. Pat. No. 5,005,574 or U.S. Pat. No. 4,803,987 (incorporated herein by this reference), or temperature could be used as just another physiologic sense to monitor for therapeutic, research, drug titration and other medical purposes. Other sensors could provide additional data for similar medical uses like optical sensors for oximetry or pulse oximetry and even partial pressure of oxygen can be measured and stored or used for various purposes. An example of a sensor that could be used for this is described in a patent incorporated herein by reference; U.S. Pat. No. 5,596,986 (Blood Oximeter). Other sensors could be substituted or used in conjunction with the pressure sensor to provide other physiologic data for use in monitoring and treating patients. Merely supplying electrode surfaces in the fixture would provide the structure to make impedance measurements could be used for local flow characteristics, or piezocrystals and similar transducers for ultrasound generation could also or alternatively measure flow or viscosity of the fluid in the vessel held in the fixture in accord with this invention.
Many improvements in monitoring a patient's health can come about through the use of the inventive fixture described herein. Methods of construction and use are also described. Meeting any one of these described goals or objectives or any combination or all of them is the purpose of this invention.
Referring first to FIG. 1, a device 10 illustrates a preferred embodiment of the invention. The device 10 is constructed of an outer shell 11 continuous with the inner surface 15. It is essentially a tubular shape having a slit 14 therein extending along the length. The length is indicated by arrows 13 and the slit opening has two opposing surfaces 16 and 17 which are narrow relative to a deeper passage P which forms the main portion of the tunnel through the device 10 beneath the slit 14. In most preferred embodiments the sensors will be housed within the area between the passage P and the outer surface 11 indicated here by arrows 12. The surface 15 will be modified to accept the active end of any sensors employed, for operational effectiveness. Alternatively other areas of the surface of the tunnel passage P's walls may be used.
The device may be built of metal or plastic which is tolerated for long term implantation into a living body. Implantable medical devices are commonly constructed of titanium currently; and ceramic, plastic, and silicone are among other commonly used materials for implanted medical devices. Any biocompatitable material maybe used. Our first embodiments tested were constructed of titanium.
FIG. 2 illustrates another preferred embodiment of the invention. The device 20 has a passage there through, extending from the sides with openings 21 to area 28 in which the pressure sensor may be inserted and is preferably located. The pressure sensor is not visible from this view. The device 20 has a slit 22 with two opposing surfaces 23 and 24. These are rounded into the external shape of the device 20 and continuous with the external surface 25 and 26 as well as with the substantially flat internal surface 27.
Referring now to FIG.3, the internal details of the device can be better seen. The active diaphragm surface 1 is in this embodiment the functional equivalent or similar to the diaphragm 54 of the U.S. Pat. No. 5,564,434 patent referred to above. Any deflection of this diaphragm 1 will register a change in pressure by the pressure sensor. As with any sensor, this will produce a signal that has a value which can be stored, telemetered out, or used directly to affect the operation of an implantable medical device. The same is true with all the sensors in all the embodiments.
The aperture defined by wall 28 extends up to the surface 27a of device 20a. This aperture will be filled in by medical adhesive in the preferred embodiment to transmit pressure directly from the blood vessel, (an arteriole in the preferred embodiment). The blood vessel will be captured within the clamping structure 20a lying along surface 27a.
It is acceptable to build the inventive device such that the diaphragm (in the case of the pressure sensor, or other active surface of any other sensor) forms a surface area coextensive with surface 27a for direct contact with the vessel especially for embodiments using sensors requiring closer contact to the vessel such construction is advisable.
The construction of the FIG. 3 device may be illustrated more simply with reference to the cross section in FIG. 4 in which the sensor structure 41 is shown attached to the medical adhesive or pressure transfer medium Ps, through which the arterial pressure Pa, will be transmitted through the flattened arterial wall lying against the pressure transmission medium. Thus, the arms 42 and 43 provide the clamping structure that holds the artery in place for the sensor measurement to be made. In the alternative, embodiments employing for example oxygen sensors based on light transmittance, reflectance or absorption, these arms 42 may be covered with light reflective surfaces, or alternatively may also contain sensors instead of or in addition to the surface 27a. Note that in some embodiments the artery or other vessel wall 44 may bulge into the gap between arms 42 and 43.
The absolute size of the implanted sensor device can be adjusted to any specification so as to fit around the particular arterial, artery, vein, venuole, or other vessel through the wall of which one desires to make physiological measurements. In FIGS. 5-10, numerals indicate the size in centimeters of a preferred embodiment fixture similar to that shown in cross section in FIG. 3.
In FIG. 9, The inventive fixture 90 is shown in side view. The line 95 indicates the location of the flat surface and the line 96 indicates the location of the base of the slit or opposite to the flat surface. Taken together they define one diameter of the tunnel passageway that runs through the fixture and in which the vessel should rest when in use. A depression 91 is seen from this view. This depression is formed in the titanium block from which the fixture is formed in order to make the opening between the pressure sensor diaphragm on the surface coextensive with line 95. In FIG. 10, the outer edge of the generally flat, sensor bearing surface of the clamp's passageway can be seen at 107. The space 106 is the slit through the opposite side of said passageway. Openings 102 and 101 are the stop and entry ends for the pressure sensor which will be inserted into the fixture 100 to make the completed device.
FIG. 11a and 11b, illustrate the preferred method for having a body vessel (preferably a blood vessel) become surrounded and held by the clamp device, that is, the fixture, of this invention. In FIG. 11a, a clamp 92 can be used to prevent the continuing flow of blood through a vessel 91 thereby producing a collapsed vessel 93. This collapsed vessel can easily be slid through the slit at the top of device 90 so that it rests within the passageway as shown. Once the clamp 92 is removed (see FIG. 11b). The blood can flow through the vessel 91b thus clamping the vessel into the passageway in the clamp part 94 of device 90b. Additional physical features may be connected to the clamp device such as wings 95 or suture holes (not shown) which could be sewed onto tissue within the body to further stabilize the device, although when used with arterial vessels we have found no need for such further stabilization.
If desired, a clip, preferably of a deformable springy material like spring steel or other metal or plastic, may be fit over the top of the slit to help ensure that the vessel cannot escape the clamp. Such a clip is illustrated on FIG. 1 as clip 4, having flat stabilizing ends to keep it from rotating around the fixture 10, and the fixture 10 should have bumps or ridges 5 and 6 to keep the clip from sliding, or some other detent and ridge or other mechanical arrangement could be used to ensure the clip 4 does not slide once in place.
Refer now to FIG. 12, an implantable medical device 81 is illustrated having a body 87 containing a power source and appropriate electronics for powering and reading out data from the sensors in the device 85. It may also contain communications circuits for receiving signals from the inventive fixture and sensor device and/or from other devices in a living body, and for communication with devices external to the body as well. It may likewise contain the operative features and structures to be a drug pump, a cardiac pacemaker, a cardiodefibrillator, a nerve stimulator or just an implantable monitor, or any combination of these.
Device 85 is connected to the inventive structure by a lead 83 in the preferred embodiment. This lead may be similar in construction to leads currently used in pacemakers and cardio defibrillators and the like in that they contain a conductor for bringing electric power to and from the sensor device in the clamp 85. Here a vessel 84 is illustrated running through the clamp device 85 The device may contain its own power supply circuitry, communications circuitry and so forth as desired. Presently it is most preferred that only the sensor(s) and some electronics associated with processing the transducer signals for such sensor(s) be contained in a sensor capsule of the inventive device.
In FIG. 13, the lower surface of the passage 100 is shown lo separated from the device. The direction of blood flow is indicted by the arrows. In this setup two sensor areas 101 and 102 are provided to allow for differential readings upstream and down stream. In one preferred embodiment, a heater (preferably a resistive heater) would be included at area 101 or 101a or both, and a measurement of the temperature of the blood can then be taken at location 101 and 102 or at least at 102 and a determination made from these measurements relating to the viscosity of the blood itself or the speed of passage through the distance through the passage way between 101 or 101a and 102. This can be done using principles defined in U.S. Pat. No. 5,486,107 to Bonne et al hereby incorporated in its entirety by this reference hereto, which describes the use of specific heat and thermal conductivity to determine characteristics of a gas flowing past a sensor array.
Alternatively, Doppler sensors or ultrasound transducers may be used as were used in U.S. Pat. No. 5,409,009 patent cited above and incorporated herein by reference for determining flow or other physiologic parameters, using this or similar sensor array patterns or even located at a single point as is the pressure transducer of the first disclosed embodiment. An example is found in U.S. Pat. No. 4,227,407 (incorporated herein in its entirety by this reference) detailing how the transducers should be located relative to the conduit, here the contained blood vessel, to generate an electric signal proportional to fluid flow.
In FIG. 14, the internal passageway 110 of an inventive clamp device, may have an infra red sending and receiving array 111 and 112 mounted into the wall thereof so that the light L can penetrate the vascular wall VW and provide measurements for indications for the percentage of oxygen's satiation of the blood passing through the passageway 110.
Alternatively, these sensor locations can be adjusted to be on opposite sides of the fixture and the vessel, for example at points 113 and 114. Further, such configurations can be useful for impedance sensing to measure flow rate, viscosity and the like or for employing Doppler transducers for similar measurements. An example of placement of impedance electrodes relative to a flow duct (in this reference the Aorta), is found in U.S. Pat. No. 5,782,774, also incorporated by reference. Clearly to measure cardiac output over a period of days or months or even years by incorporating the electrode configuration around a less vital blood vessel downstream from the Aorta would provide significant benefit to medical science, and to the management of the health of a patient using our device.
Electromagnetic measurements can be made with electrode placement as indicated in U.S. Pat. No. 4,346,604 also, if desired, and the magnetic positioning can be adjusted to be a part of the fixture device itself.
Of course it must be recognized that a feedback loop can be created using the measurements taken by sensors in the clamp device to deliver therapy to a living body into which the clamp device has been implanted Thus, a fast acting blood pressure medication maybe monitored on a moment by moment basis throughout a patients daily activities by the use of such a clamp device associated with measurement circuitry and possibly processing circuitry and memory circuits implanted therewith. Data from the fixture device sensors can be used to modify programs for generating electrical pulses to stimulate nerves and to adjust pacing rate by coordination with an implantable pulse generator that has leads directing electrical energy to the appropriate bodily tissue.
This kind of closed loop feedback mechanism has long been established a potential use for implanted medical devices. In one example, in U.S. Pat. No. 5,199,428, Obel et al., incorporated by this reference, showed that a measure of ischemia could drive a cardiac pacemaker therapy and a vagal stimulator. Any of the physiologic signals measured by this device could provide the kind of input a closed loop control system would need for adjusting therapy, whether drug based via a drug pump, or stimulator based as in the rate delivered by a pacemaker or the frequency etc., delivered by a neurostimulator. Likewise, the signal data could be easily stored into historical memory in a device like that of 87 in FIG. 12, which has appropriate processing power and memory. Such a device can clearly have a telemetry circuit 88 which could transmit signals to an external device 89 which could communicate the signals to a hospital over phone lines or through wireless telemetry so that the doctor can manage his patients, be aware of changing conditions and even be alarmed about changes in patient condition without ever scheduling a visit. Examples of useful monitors include U.S. Pat. Nos. 5,807,336, 5,791,344, and 5,411,031.
It may be preferred to implant the invention's fixture into locations illustrated in FIGS. 15a-e. FIG. 15a illustrates an arm having three highlighted locations for preferable mounting of the inventive blood pressure fixture in accord with this invention. The Axilary Artery provides another apt location in FIG. 15b the Inguinal, and Femoral arteries as well as the posterior Tibial artery in FIGS. 15c and d, respectively provide additional useful locations as do the Radial Ulnar and Superficial Palmar arteries shown in FIG. 15e. Each of these locations provides particularly useful information and a combination of locations can be used so as to provide for a systemic or overall reading or to indicate that there is an imbalance in a given area or across the system.
Long term monitoring of any such arrangement of sensors or sensor structures can provide detailed measurement data that can be used for historical charting of the patient's conditions and long term patient management.
In FIG. 16, a perspective view of a fixture 130A has a cable 136 linking a pressure sensor, (not visible here)connected at a linking junction 137 to the fixture 130A. It is recognized that not all vessels to which the fixture will be used to surround will be of the same size and consequently, the inventive device will be made available in multiple sizes. The surgeon will select the appropriate sized fixture for the vessel and snap the pressure sensor into the one selected, either before or after the blood vessel is surrounded by the passageway in the fixture. The fixtures can be constructed of hylon or similar suitable biocompatible plastic, such as may be used in connector blocks for pacemakers or other implantable medical devices with electrical leads currently, or of ceramics or metals that may be acceptable.
A block of medical adhesive will have been formed in the well 134 prior to the surgeon connecting the pressure sensor into the fixture 130A. For construction purposes, if using a metallic, such as titanium block from which the device will be machined, the opening at 133 is what is left after wire electric discharge machining. It may be filled as desired in the completed device when prepared for use. To construct the entire device then, one would machine in the curves, the slit and the tunnel, then either the through hole at 133 or the space shown at 133a in FIGS. 16B and C to create the diaphragm. Then the assembly of the pressure capsule would be completed by bonding the sensor and electronics board to the lower surface of hole 133a so that the diaphragm contacts the sensor appropriately, and then sealing the sensor capsule. Next the through hole 133 can be filled with transmissive material up to surface 135 and the device connected to a lead if appropriate. Then the device can be sterilized and will be ready for use.
Details in the construction of the device of FIG. 16A can be more clearly seen with reference to FIGS. 16B and 16C. In FIG. 16B. The structure 130s is the same as the top portion of the device 137a. In one preferred embodiment of a titanium block that has been electric discharged machined to produce the outer configuration shown in these FIGS. 16A-C, including the tunnel like passage way 195 having the base 135 and a slit 134 at the top. To form this construction, the electric discharge machining first produces the slopped sides (if desired for the final design,) and then produces the tunnel like through passage way 195 and the slit 134. The rectangular passageway 133 must be machined very carefully because this step produces the pressure diaphragm itself. The lower surface of the pressure diaphragm which will be inside the pressure capsule is illustrated at line 139 in both FIGS. 16B and 16C. A circuit board CB containing the sensor package 139a will be inserted in the area 133a machined from the area of the block opposite the slit 134. For our purposes, the preferred thickness of the diaphragm formed by the lower surface of the hole provided at 133 and the upper surface provided in area 133a should be on the order of approximately 0.001 inches.
The cable 136 may be preferably manufactured with two conductors as illustrated in cross section 180 in FIG. 18, with one center wire 181 surrounded by a cabled or wound wire 183 spaced apart from each other by silicone or medical adhesive or other biocompatible and preferably non-conductive material in layer 182. Preferably also a sheath layer 184 surrounds the outer conductor. A current pacing lead construction with two coiled metal conductors would suffice as well, and many variations are possible, including merely twisted pair wires, straight conductors and the like. A coaxial configuration is preferred only because a less noisy signal will be obtained with such a structure. As fiber-optic communications channels and their uses with sensors become better able to handle the stresses of implant by technological innovation, such cables may also be employed, and perhaps preferred.
Referring now to FIG. 17, a side view of a structure like that of FIG. 16 is shown in cross section 140. The connecting cable 143 provides signal and power to what in the preferred embodiment is a titanium capsule 141 having two sensors at deformable membranes or plates 146 and 147, one at 147 being useful as a reference sensor. (If a reference sensor is used for any sensor, common mode noise can easily be canceled as is known in most sensor art fields. In the pressure field, by having an absolute measure of pressure available at a reference sensor protected from the environment of the measuring sensor, a value close to gage pressure can be obtained, as well as the differential represented by the value of the blood pressure signal. Accordingly, with such an embodiment we can cancel the effects of atmospheric pressure fluctuation as would be done in this art. So here, the sensor under plate 146 would sense the blood affected pressure through a window of preferably something that will transmit a regular approximation of the blood pressure in the vessel above it, such as cured medical adhesive, and the ambient or atmospheric pressure will be sensed above the plate 147, which faces an area of stable body tissue or is blocked off from it. Alternatively, a pressure sensor could be mounted within a capsule implanted in a more stable area of the body for use as the reference sensor.)
The sensor capsule 141 is fixedly mounted or formed into the fixture surrounding it which from this cross section is seen to be mainly comprised of the passageway supporting section 142 through which the vessel will be mounted when in operation and the retaining portion 148. To show one form of isolation structure for the reference sensor, a dotted section is shown at 149 which could be formed of the same hard plastic that preferably forms sections 148 and 147, which if perforated to allow fluid access to the window 144 will provide a good environment for reference pressure measurements. On FIG. 21, the outline of such a reference sensor is illustrated at 227, and the perforations at 228. An alternative embodiment 190 is viewed cut across the tunnel shaped passageway 195 into which the vessel would be mounted in FIG. 19. In this embodiment the slit or opening 192 is near the flat internal surface 191, which indicates that the inventors considered the relative location to be non-critical. That is, it is not believed that the opening and the flat surface must be on opposite sides to perform sufficiently well to function. In fact, if one does not use a pressure sensor, it may not be necessary to flatten one side or even a part of one side of the inner surface of the tunnel like passageway. Also illustrated in FIG. 19, is the potential for maintaining all electronics, power sources telemetry, memory and other processing circuitry in direct contact with the sleeve or tunnel like passageway which will surround the vessel. In FIG. 19, this is shown with an adjoining housing area 193, which can be of any convenient shape. It is expected that in most preferred uses, the smallness of the sensor fixture will be of paramount importance, so such circuitry will at least for the near future be mounted in a housing separated by a conductor containing lead from the fixture. For some applications a fiber-optic cable may be preferred or may supplement the electrical conductors.
For example, a sensor fixture could be mounted around the radial artery or vein with a pressure sensor mounted in the fixture, and a conductor bearing lead tunneled through the arm to an implantable pulse generator in the patient's chest. This sensor would provide feedback regarding changes in the pressure within the patient's body at the extremity and this measurement set could be used to modify stimulation pulses sent by the pulse generator mounted in the chest t, for example, by sending pulses of greater intensity or duration to the carotid sinus through another electrical medical lead connected to stimulate at that location. As mentioned earlier, data in the sensor measurements provided by sensors mounted in the fixture can provide the basis for adaptive closed loop programs that provide changes in heart pacing stimulation or drug delivery from implanted drug pumps.
Additionally, the fixtures could be mounted to any body vessel including but not limited to bile ducts, pancreatic ducts, seminal vesicles, lymph vessels or any of the tubes involved in the Kidney function or the excretion of urine downstream therefrom, in short, on any vessel through which a fluid may flow, so as to take measurements of that fluid. The squeezing off of the body vessel provides for a smaller outer diameter that can fit through the opening or slit in the fixture, and then the vessel upon release will fill or overfill the internal diameter of the tunnel shaped passageway becoming thereby fixed thereto so that repeatable reliable long term measurements can be taken by a sensor structure mounted into the internal surface.
Referring now FIG. 20 in which an alternate preferred embodiment 200 is illustrated, a thin lead 204 can be connected to an end of the pressures sensor capsule 202 if desired. This lead 204 may be much thinner than a cardiac pacing lead (for example) since no coiling would be required for a conductor in such a lead. Optical communication could of course take place over an optical fiber (in lead 204) if the circumstances of the device suggests such a design would be preferred. In fact, use of not connected telemetric pathways for communications with the implantable device are an option and such communications with implantable medical devices has been known for some time.
The fixture for holding the blood vessel 205 again is shown to contain a tunnel or passageway 201 having a slit 203 along one side of it between the arms 205a and 205b of the fixtures body 205.
For convenience, the internal diameter of the tunnel space 201 is taken in views of the sections A-E, illustrated in an abbreviated form in FIGS. 20A-20E. Referring to these FIGS. 20A-E, it can be seen that the slit width dA-dE does not vary substantially in this preferred embodiment, even though the internal configuration of the tunnel does. Each of the internal diameters taken from the view lines A-E of FIG., 20 can be described as two rough semi circles, concave sides facing but spaced apart the length of a relatively flat line on the bottom which joins the semi circles 210-214, this flat area, in the preferred embodiments, being a cross sectional of the preferred substantially flat area in which the sensor(s) are most efficaciously placed. (Some variation from flatness and placement are within the scope of the concepts described here for the placement of the various sensor configurations described, of course.) Thus it can be seen that at one end of the tunnel 201 the opening is nearly round where in the middle it is quite oblong and again round at the other side. The tunnel or interior space designed in this manner provides some resistance to length wise slippage of the fixture along the length of the vessel it's meant to surround as well as reducing the ability of the vessel to come out through the slit. It is also believed to enhance the sensing of pressure.
In FIG. 21, a convolved, curved or serpentine slit design is shown in slit 225 on device 220. It is certainly within the range of preferred embodiments to employ both the serpentine or curved slit design or something similar together with the design of the pinched tunnel illustrated in Figs, 20A-E in the same device, or not, if preferred. The curve of the slit shape could be more angular than that illustrated if desired as well.
In device 220, the pressure sensor capsule body 221 is again mounted to the fixture 223 but in this design, the lead 222 is built integrity with pressure capsule 221.
Refer now to FIG. 22. This holding fixture 238 and pressure capsule 231 can be designed as an integrated device or assembled in the operating room by the surgeon to form device 230. As in the device illustrated in FIG. 20, an attachment mechanism for attaching the lead to the pressure capsule 231 is also illustrated at 232. Thus, the device can be a 2 or 3 piece combination, that is, a capsule and lead joined to a fixture, or a capsule joined to the lead and the fixture separately. In this device design, either a pressure transmissible substance 235 (preferably medical adhesive although other similar substances can be used), or an actual pressure diaphragm surface should mate smoothly with the interior surface 237 of the device 230. Surround structure or tub 234 can be formed integrally with either the sensor and or pressure capsule 231 or with the material 233 of which the fixture is made, depending on whether it is easier to manufacture with the surround 234 holding the fixture to the pressure sensor capsule 231 or to the fixture 238.
In FIG. 23 the inventive device 240 is again shown to be built of two main parts; the fixture 248 and the capsule in which the sensors are located 241. In this particular design a pressure sensor diaphragm 246 is built below the upper surface of the mating tub 247 which will be filled with pressure transmissive substance in the preferred embodiment. In this design, it can be seen that the fixture 248 can simply be snapped over the mating tub 247 to employ a ridge of the same material as fixture 248 (not shown) which will mate with the detent 248 in the mating tub 247.
Currently since it is cheaper to manufacture the fixture part, if the reader wishes to accommodate the surgeon, a kit containing a plurality or multiplicity of fixtures of different sizes which can all mate to a single sensor capsule provides a most cost effective solution.
Thus, since presently configured pressure capsules like 241 are made of a metal like titanium, for this design it would be easiest to form the mating tub 247 with its mating detents 248 on the surface of that metal capsule. To manufacture, the next step would be to fill the tub with medical adhesive or other suitable pressure transmissive medium, wait for the pressure transmissive medium to cure, sterilize the whole thing and provide it to a physician or surgeon with a set of sized fixtures like that illustrated at 248. In this way, a kit could be provided so that the right sized fixture 248 could be picked from a kit of a range of such fixtures to mate with a single sterilized and prepared pressure capsule and lead combination. Of course this would be applicable to the pressure capsule without a lead or with a lead connector as illustrated in FIG. 22 for example, and to other sensor capsules containing sensors other than pressure sensors with minor accommodation. For examples, an oxygen sensor requiring that its sensor be touching the vessel would omit the steps of providing a pressure transmissive medium, and an impedance sensor may simply provide electrical connections to the fixture which may itself have electrodes mounted strategically around the tunnel formed by the fixture to contact the vessel most appropriately.
FIG. 24 shows the combination 250 of the pressure capsule 252 having the mating tub 257 filled with pressure transmissible substance 247a which can come from one sterile package. Selecting the correctly sized fixture 251 and pressing it over the sensor package 252 will provide a reassuring snap to the physician when the internal ridges or ridge mate with the detent 253 and produce the device of FIG. 25, 260. Other mechanical links which may be well known in the art can be substituted for a ridge with mating detent. Bumps and detents, set screws and glue describe just a few of these.
While it is assumed that a certain level of background is available to the reader, it is useful to mention and thus incorporate by reference a number of patents which describe features that may be employed in implementing the mentioned features of this invention including the use of multiple implantable devices having the inventive fixtures, which can communicate with each other or with other implanted devices with or without provision of conductors (for examples of background in this area see U.S. Pat. Nos. 4,987,897 and 5,113,859 issued to Funke), telemetry uplink and data storage for monitoring and communication with an external receiver generally (U.S. Pat. Nos. 5,411,031, 5,687,734, and 5,693,676), ultrasound transducers used in implantable devices (U.S. Pat. No. 5,188,106), and Drug pumps (U.S. Pat. Nos. 4,373,527, 5,681,285 and 5,586,629).
To further illustrate the use of tunnel narrowing along with a curved slit design, the illustrations of FIG. 26 and FIGS. 27A-C should be considered. FIG. 27B is a facing view looking into the tunnel 264. It should be noted that this part of the device structure when assembled will hold the pressure capsule or other sensor housing at location 263 where the fixture curves inward toward the central passageway or tunnel 264. It can be noted that the upper portion of the tunnel 266 can be seen narrowing past this entry way which is nearly round at perimeter 265 from this view. Note also that the sidewall 262 corresponds to sidewall 262 on FIG. 26.
Referring to the section taken at line CC from FIG. 26, one can see that the diameter 266 has only slightly started to become elliptical around the tunnel passage way 264. However, in FIG. 27A taken at section line AA from FIG. 26, a perimeter 265 of the tunnel can be seen as quite elliptical in shape and the curve of the slit 261 can be seen to have substantially moved in its location in FIG. 27B, the facing view taken at line B.
If desired of course, a clip similar to that illustrated in FIG. 1 may be used to further insure that no blood vessel may escape the tunnel passageway 264.
Certainly many variations on the features described herein and many cross combinations thereof may be made use of without leaving the ambit of this disclosure, which is deemed limited only as set forth in the following appended claims.
|Cited Patent
|Filing date
|Publication date
|Dec 27, 1977
|Feb 26, 1980
|Thoratec Laboratories Corporation
|Device for determining the patentcy of a blood vessel
|Jun 18, 1979
|Mar 17, 1981
|Kapp John P
|Arterial pressure control system
|Jul 14, 1980
|Aug 31, 1982
|Narco Bio-Systems, Inc.
|Electromagnetic flow probe
|Apr 27, 1979
|Feb 15, 1983
|The Johns Hopkins University
|Implantable, programmable medication infusion system
|May 30, 1985
|Jul 15, 1986
|Medex, Inc.
|Piezoelectric apparatus for measuring bodily fluid pressure within a conduit
|Nov 6, 1986
|Jan 24, 1989
|Sri International
|Blood pressure monitoring method and apparatus
|Jan 27, 1987
|Feb 7, 1989
|Sri International
|Blood pressure monitoring method and apparatus
|Jun 11, 1986
|Feb 14, 1989
|Intermedics, Inc.
|Temperature responsive controller for cardiac pacer
|Apr 24, 1989
|May 15, 1990
|Medex, Inc.
|Tubular pressure transducer
|Nov 28, 1989
|Apr 9, 1991
|Medical Engineering And Development Institute, Inc.
|Temperature-based, rate-modulated cardiac therapy apparatus and method
|Mar 8, 1991
|Feb 23, 1993
|Telectronics Pacing Systems, Inc.
|Method and apparatus for chronically monitoring the hemodynamic state of a patient using doppler ultrasound
|Mar 22, 1991
|Apr 6, 1993
|Medtronic, Inc.
|Implantable electrical nerve stimulator/pacemaker with ischemia for decreasing cardiac workload
|Mar 18, 1994
|Apr 25, 1995
|Medtronic, Inc.
|Methods for measurement of arterial blood flow
|Nov 24, 1993
|May 2, 1995
|Incontrol, Inc.
|Implantable cardiac patient monitor
|Feb 27, 1995
|Oct 15, 1996
|Medtronic, Inc.
|Implantable capacitive absolute pressure and temperature sensor
|Apr 13, 1995
|Dec 24, 1996
|Medtronic, Inc.
|Presure controlled system for filling an implantable drug infusion pump
|Feb 16, 1993
|Jan 28, 1997
|Scico, Inc.
|Blood oximeter
|Jun 19, 1996
|Oct 28, 1997
|Baxter International Inc.
|Infusion pump with an electronically loadable drug library and a user interface for loading the library
|Oct 20, 1994
|Nov 18, 1997
|Hewlett-Packard Company
|Flexible patient monitoring system featuring a multiport transmitter
|Jan 16, 1996
|Dec 2, 1997
|Medtronic, Inc.
|Compressed patient narrative storage in and full text reconstruction from implantable medical devices
|Oct 4, 1996
|Jul 21, 1998
|Imagyn Medical Technologies California, Inc.
|Apparatus and method of bioelectrical impedance analysis of blood flow
|Jan 4, 1996
|Aug 11, 1998
|Alfred E. Mann Foundation For Scientific Research
|Patient monitoring system
|Aug 2, 1996
|Sep 15, 1998
|Sabratek Corporation
|Apparatus for monitoring and/or controlling a medical device
|Sep 9, 1997
|Feb 23, 1999
|Lieber; Claude P.
|System and method for remote measurement of fluid flow
|Citing Patent
|Filing date
|Publication date
|May 10, 2001
|Dec 3, 2002
|Thermal Technologies, Inc.
|System for quantifying edema
|Aug 22, 2001
|Sep 16, 2003
|Scimed Life Systems, Inc.
|Two-piece sensor assembly
|Dec 12, 2000
|Mar 23, 2004
|Transoma Medical, Inc.
|Adjustable flow probe
|Mar 20, 2003
|Mar 22, 2005
|Medtronic, Inc.
|Method and apparatus for optimizing cardiac resynchronization therapy
|Apr 4, 2002
|May 10, 2005
|Matthew Joseph Fitz
|Implantable device for monitoring aneurysm sac parameters
|Oct 24, 2002
|Mar 7, 2006
|Furnary Anthony P
|Method and apparatus for monitoring blood condition and cardiopulmonary function
|Jul 8, 2002
|Dec 12, 2006
|Seiko Instruments Inc.
|Blood rheology measuring apparatus
|May 26, 2006
|Jul 10, 2007
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern California
|Implanted surgical drain with drain holes for monitoring internal tissue condition
|Feb 9, 2004
|Jul 17, 2007
|Alfred E. Mann Institute For Biomedical Engineering
|Implanted surgical drain with multiple sensing elements for monitoring internal tissue condition
|Feb 9, 2004
|Aug 7, 2007
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern California
|Implanted surgical drain with sensing and transmitting elements for monitoring internal tissue condition
|May 26, 2006
|Sep 4, 2007
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern California
|Method of utilizing a surgical drain with sensors for monitoring internal tissue condition
|Feb 9, 2004
|Sep 11, 2007
|Alfred E. Mann Institute For Biomedical Research At The University Of Southern California
|Surgical drain with sensors for monitoring fluid lumen
|Feb 9, 2004
|Jan 29, 2008
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern California
|Surgical drain with sensors for monitoring internal tissue condition by transmittance
|Feb 9, 2004
|Sep 2, 2008
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern California
|Surgical drain with positioning and protective features
|May 25, 2007
|Dec 23, 2008
|Cook Vascular Incorporated
|Adjustable tension cuff assembly
|Jun 6, 2006
|Apr 28, 2009
|Cardiac Pacemakers, Inc.
|Method and device for lymphatic system monitoring
|Feb 26, 2004
|May 19, 2009
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern California
|Implantable device with sensors for differential monitoring of internal condition
|Apr 25, 2007
|Feb 9, 2010
|Ethicon Endo-Surgery, Inc.
|System and method for determining implanted device orientation
|Jun 6, 2006
|Jun 8, 2010
|Cardiac Pacemakers, Inc.
|Method and apparatus for gastrointestinal stimulation via the lymphatic system
|Oct 23, 2006
|Sep 21, 2010
|Cvrx, Inc.
|Stimulus regimens for cardiovascular reflex control
|Oct 12, 2010
|Cvrx, Inc.
|Baroreflex stimulator with integrated pressure sensor
|Jul 20, 2005
|Nov 23, 2010
|Cvrx, Inc.
|Stimulus regimens for cardiovascular reflex control
|Feb 7, 2008
|Nov 30, 2010
|Ethicon Endo-Surgery, Inc.
|Powering implantable restriction systems using light
|Jun 6, 2006
|Feb 22, 2011
|Cardiac Pacemakers, Inc.
|Amelioration of chronic pain by endolymphatic stimulation
|Jan 29, 2007
|Apr 19, 2011
|Ethicon Endo-Surgery, Inc.
|External mechanical pressure sensor for gastric band pressure measurements
|Nov 10, 2009
|May 24, 2011
|Cvrx, Inc.
|Devices and methods for cardiovascular reflex control via coupled electrodes
|Mar 7, 2006
|Sep 13, 2011
|Ethicon Endo-Surgery, Inc.
|External pressure-based gastric band adjustment system and method
|Apr 6, 2006
|Sep 13, 2011
|Ethicon Endo-Surgery, Inc.
|Monitoring of a food intake restriction device
|Feb 26, 2008
|Oct 11, 2011
|Ethicon Endo-Surgery, Inc.
|Controlling pressure in adjustable restriction devices
|Feb 12, 2008
|Nov 15, 2011
|Ethicon Endo-Surgery, Inc.
|Automatically adjusting band system with MEMS pump
|Nov 15, 2011
|Cvrx, Inc.
|Baroreflex modulation to gradually decrease blood pressure
|Feb 12, 2007
|Nov 29, 2011
|Ethicon Endo-Surgery, Inc.
|Apparatus for adjustment and sensing of gastric band pressure
|Mar 3, 2006
|Dec 13, 2011
|Sensicor, Llc
|Method and apparatus for monitoring blood condition and cardiopulmonary function
|Oct 29, 2002
|Dec 27, 2011
|Cvrx, Inc.
|Devices and methods for cardiovascular reflex control
|Dec 14, 2007
|Jan 24, 2012
|Ethicon Endo-Surgery, Inc.
|Adjustable height gastric restriction devices and methods
|Feb 8, 2008
|Feb 14, 2012
|Ethicon Endo-Surgery, Inc.
|System and method of sterilizing an implantable medical device
|Jun 6, 2006
|Feb 28, 2012
|Cardiac Pacemakers, Inc.
|Method and apparatus for introducing endolymphatic instrumentation
|Dec 27, 2007
|Mar 27, 2012
|Ethicon Endo-Surgery, Inc.
|Controlling pressure in adjustable restriction devices
|Mar 17, 2008
|Apr 3, 2012
|Cvrx, Inc.
|Methods and devices for controlling battery life in an implantable pulse generator
|Feb 28, 2008
|Apr 10, 2012
|Ethicon Endo-Surgery, Inc.
|Physiological parameter analysis for an implantable restriction device and a data logger
|Oct 3, 2007
|Apr 24, 2012
|The Invention Science Fund I, Llc
|Vasculature and lymphatic system imaging and ablation
|Mar 6, 2008
|May 29, 2012
|Ethicon Endo-Surgery, Inc.
|Reorientation port
|Dec 10, 2007
|May 29, 2012
|Ethicon Endo-Surgery, Inc.
|Methods for implanting a gastric restriction device
|Jan 28, 2008
|Jun 5, 2012
|Ethicon Endo-Surgery, Inc.
|Methods and devices for measuring impedance in a gastric restriction system
|Feb 7, 2008
|Jul 17, 2012
|Ethicon Endo-Surgery, Inc.
|Powering implantable restriction systems using kinetic motion
|Feb 19, 2008
|Jul 31, 2012
|University Of Louisville Research Foundation, Inc.
|Perivascular pressure sensor and sensing system
|Mar 6, 2008
|Jul 31, 2012
|Ethicon Endo-Surgery, Inc.
|System and method of aligning an implantable antenna
|Nov 20, 2006
|Oct 2, 2012
|Seiko Instruments Inc.
|Blood rheology measuring apparatus
|Oct 4, 2007
|Oct 9, 2012
|The Invention Science Fund I, Llc
|Vasculature and lymphatic system imaging and ablation associated with a local bypass
|Oct 5, 2007
|Oct 9, 2012
|The Invention Science Fund I, Llc
|Vasculature and lymphatic system imaging and ablation associated with a reservoir
|Oct 10, 2011
|Oct 16, 2012
|Tandem Diabetes Care, Inc.
|Infusion pump system with disposable cartridge having pressure venting and pressure feedback
|Jul 7, 2006
|Oct 16, 2012
|Cvrx, Inc.
|Method and apparatus for stimulation of baroreceptors in pulmonary artery
|Oct 30, 2012
|Tandem Diabetes Care, Inc.
|Infusion pump system with disposable cartridge having pressure venting and pressure feedback
|Nov 6, 2012
|Pacesetter, Inc.
|Implantable pressure transducer system optimized for reduced thrombosis effect
|Jan 28, 2008
|Dec 25, 2012
|Ethicon Endo-Surgery, Inc.
|Methods and devices for diagnosing performance of a gastric restriction system
|Apr 26, 2011
|Jan 15, 2013
|Fresenius Medical Care Holdings, Inc.
|Medical temperature sensors and related systems and methods
|Oct 22, 2009
|Feb 5, 2013
|Cardiac Pacemakers, Inc.
|Method and apparatus for neural stimulation via the lymphatic system
|Dec 27, 2007
|Feb 19, 2013
|Ethicon Endo-Surgery, Inc.
|Constant force mechanisms for regulating restriction devices
|Dec 15, 2010
|Mar 12, 2013
|Cardiac Pacemakers, Inc.
|Transcutaneous neurostimulator for treating hypertension
|Jun 3, 2010
|Mar 19, 2013
|Medtronic, Inc.
|Capacitive pressure sensor assembly
|Oct 29, 2008
|Apr 2, 2013
|Tandem Diabetes Care, Inc.
|Flow regulating stopcocks and related methods
|Feb 26, 2009
|May 28, 2013
|Tandem Diabetes Care, Inc.
|Slideable flow metering devices and related methods
|Jun 10, 2011
|Jun 25, 2013
|Boston Scientific Scimed, Inc.
|Renal denervation and stimulation employing wireless vascular energy transfer arrangement
|Aug 5, 2010
|Sep 17, 2013
|Rainbow Medical Ltd.
|Enhancing perfusion by contraction
|Aug 26, 2010
|Oct 29, 2013
|Cardiac Pacemakers, Inc.
|Transcutaneous neurostimulator for modulating cardiovascular function
|Mar 8, 2010
|Nov 12, 2013
|Cvrx, Inc.
|Devices and methods for cardiovascular reflex control
|Jan 28, 2008
|Nov 26, 2013
|Ethicon Endo-Surgery, Inc.
|Gastric restriction device data handling devices and methods
|Feb 12, 2008
|Nov 26, 2013
|Ethicon Endo-Sugery, Inc.
|Automatically adjusting band system
|Jul 17, 2008
|Nov 26, 2013
|Cvrx, Inc.
|Baroreflex activation therapy with incrementally changing intensity
|Apr 13, 2007
|Dec 10, 2013
|Cvrx, Inc.
|System and method for sustained baroreflex stimulation
|Aug 16, 2011
|Jan 7, 2014
|Enopace Biomedical Ltd.
|Acute myocardial infarction treatment by electrical stimulation of the thoracic aorta
|Dec 1, 2010
|Jan 7, 2014
|Enopace Biomedical Ltd.
|Thoracic aorta and vagus nerve stimulation
|Dec 20, 2010
|Feb 11, 2014
|Rainbow Medical Ltd.
|Pacemaker with no production
|Sep 18, 2009
|Feb 18, 2014
|Tandem Diabetes Care, Inc.
|Solute concentration measurement device and related methods
|Aug 26, 2010
|Apr 1, 2014
|Cardiac Pacemakers, Inc.
|Percutaneous neurostimulator for modulating cardiovascular function
|May 24, 2012
|Apr 29, 2014
|Cvrx, Inc.
|Automatic baroreflex modulation responsive to adverse event
|Apr 19, 2010
|May 6, 2014
|Cvrx, Inc.
|Electrode structures and methods for their use in cardiovascular reflex control
|Aug 1, 2012
|Jun 24, 2014
|Fresenius Medical Care Holdings, Inc.
|Medical temperature sensors and related systems and methods
|Jul 29, 2010
|Jun 24, 2014
|Tandem Diabetes Care, Inc.
|Infusion pump system with disposable cartridge having pressure venting and pressure feedback
|Sep 27, 2006
|Sep 16, 2014
|Cvrx, Inc.
|Devices and methods for cardiovascular reflex treatments
|Nov 10, 2011
|Oct 7, 2014
|Enopace Biomedical Ltd.
|Detector-based arterial stimulation
|Feb 28, 2008
|Oct 28, 2014
|Ethicon Endo-Surgery, Inc.
|GUI for an implantable restriction device and a data logger
|Jun 25, 2013
|Nov 4, 2014
|Boston Scientific Scimed, Inc.
|Renal denervation and stimulation employing wireless vascular energy transfer arrangement
|Nov 30, 2012
|Nov 4, 2014
|Cvrx, Inc.
|Electrode structures and methods for their use in cardiovascular reflex control
|May 6, 2010
|Nov 25, 2014
|Cardiac Pacemakers, Inc.
|Method and apparatus for gastrointestinal stimulation via the lymphatic system
|Sep 1, 2006
|Dec 9, 2014
|Cardiac Pacemakers, Inc.
|Method and apparatus for endolymphatic drug delivery
|Jul 29, 2010
|Jan 6, 2015
|Tandem Diabetes Care, Inc.
|Infusion pump system with disposable cartridge having pressure venting and pressure feedback
|Feb 29, 2012
|Jan 27, 2015
|Vessix Vascular, Inc.
|Tuned RF energy and electrical tissue characterization for selective treatment of target tissues
|Nov 7, 2012
|Feb 10, 2015
|Boston Scientific Scimed, Inc.
|Ostial renal nerve ablation
|Oct 25, 2011
|Mar 10, 2015
|Boston Scientific Scimed, Inc.
|Renal nerve ablation using conductive fluid jet and RF energy
|Aug 7, 2009
|Mar 24, 2015
|Tandem Diabetes Care, Inc.
|Two chamber pumps and related methods
|Jan 31, 2008
|Apr 14, 2015
|Enopace Biomedical Ltd
|Intra-aortic electrical counterpulsation
|Nov 22, 2011
|May 5, 2015
|Boston Scientific Scimed, Inc.
|Renal ablation electrode with force-activatable conduction apparatus
|Dec 21, 2012
|May 12, 2015
|Vessix Vascular, Inc.
|Methods and apparatuses for remodeling tissue of or adjacent to a body passage
|Sep 23, 2011
|May 12, 2015
|Boston Scientific Scimed, Inc.
|Self-expanding cooling electrode for renal nerve ablation
|Feb 13, 2008
|May 19, 2015
|Virender K. Sharma
|Method and apparatus for electrical stimulation of the pancreatico-biliary system
|Dec 21, 2012
|May 19, 2015
|Vessix Vascular, Inc.
|Methods and apparatuses for remodeling tissue of or adjacent to a body passage
|Nov 18, 2011
|Jun 2, 2015
|Cvrx, Inc.
|Electrode structures and methods for their use in cardiovascular reflex control
|Dec 21, 2012
|Jun 9, 2015
|Boston Scientific Scimed, Inc.
|Off-wall electrode device and methods for nerve modulation
|Nov 9, 2011
|Jun 23, 2015
|Boston Scientific Scime, Inc.
|Catheter-focused magnetic field induced renal nerve ablation
|Dec 21, 2012
|Jul 7, 2015
|Vessix Vascular, Inc.
|Methods and apparatuses for remodeling tissue of or adjacent to a body passage
|Oct 16, 2012
|Jul 14, 2015
|Boston Scientific Scimed, Inc.
|Integrated crossing balloon catheter
|Jul 18, 2011
|Jul 21, 2015
|Boston Scientific Scime, Inc.
|Spiral balloon catheter for renal nerve ablation
|Nov 9, 2011
|Jul 28, 2015
|Boston Scientific Scimed, Inc.
|Renal denervation catheter with RF electrode and integral contrast dye injection arrangement
|Nov 15, 2012
|Sep 1, 2015
|Boston Scientific Scimed, Inc.
|Device and methods for renal nerve modulation monitoring
|Nov 16, 2012
|Sep 1, 2015
|Boston Scientific Scimed, Inc.
|Deflectable renal nerve ablation catheter
|Sep 28, 2007
|Sep 8, 2015
|Vessix Vascular, Inc.
|Selectable eccentric remodeling and/or ablation of atherosclerotic material
|Oct 18, 2007
|Sep 8, 2015
|Vessix Vascular, Inc.
|System for inducing desirable temperature effects on body tissue
|Jul 22, 2011
|Oct 13, 2015
|Boston Scientific Scimed, Inc.
|Sequential activation RF electrode set for renal nerve ablation
|Sep 28, 2012
|Oct 20, 2015
|Boston Scientific Scimed, Inc.
|Deflectable medical devices
|Sep 17, 2013
|Nov 3, 2015
|Boston Scientific Scimed, Inc.
|Self-positioning electrode system and method for renal nerve modulation
|Dec 21, 2012
|Nov 3, 2015
|Vessix Vascular, Inc.
|Methods and apparatuses for remodeling tissue of or adjacent to a body passage
|Jul 20, 2012
|Nov 17, 2015
|Boston Scientific Scimed, Inc.
|Nerve modulation system having helical guide
|Oct 10, 2012
|Nov 17, 2015
|Boston Scientific Scimed, Inc.
|Medical devices including ablation electrodes
|Jan 25, 2013
|Nov 17, 2015
|Boston Scientific Scimed, Inc.
|Methods and apparatuses for remodeling tissue of or adjacent to a body passage
|Nov 22, 2011
|Nov 24, 2015
|Boston Scientific Scimed, Inc.
|Renal denervation catheter with cooled RF electrode
|Apr 13, 2011
|Nov 24, 2015
|Boston Scientific Scimed, Inc.
|Focused ultrasonic renal denervation
|Jul 29, 2010
|Dec 15, 2015
|Tandem Diabetes Care, Inc.
|Infusion pump system with disposable cartridge having pressure venting and pressure feedback
|Oct 26, 2011
|Dec 29, 2015
|Boston Scientific Scimed, Inc.
|RF renal denervation catheter with multiple independent electrodes
|Jan 19, 2012
|Dec 29, 2015
|Boston Scientific Scimed, Inc.
|Guide-compatible large-electrode catheter for renal nerve ablation with reduced arterial injury
|Dec 29, 2011
|Feb 2, 2016
|Volcano Corporation
|Lumen based pressure sensing guidewire system with distortion correction
|Apr 4, 2002
|Dec 5, 2002
|Fitz Matthew Joseph
|Implantable device for monitoring aneurysm sac parameters
|Jul 8, 2002
|Feb 13, 2003
|Hiroyuki Muramatsu
|Blood rheology measuring apparatus
|Sep 26, 2003
|Apr 1, 2004
|Transoma Medical, Inc.
|Adjustable flow probe
|Oct 24, 2002
|Apr 29, 2004
|Furnary Anthony P.
|Method and apparatus for monitoring blood condition and cardiopulmonary function
|Feb 9, 2004
|Nov 18, 2004
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern Ca
|Surgical drain with sensors for monitoring internal tissue condition
|Feb 9, 2004
|Nov 18, 2004
|Alfred E. Mann Institute For Biomedical Engineering At The
|Surgical drain with positioning and protective features
|Feb 26, 2004
|Nov 25, 2004
|Alfred E. Mann Inst. For Biomedical Engineering At The Univ. Of Southern California
|Implantable device with sensors for differential monitoring of internal condition
|Feb 9, 2004
|Dec 16, 2004
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern Ca
|Surgical drain with sensors for monitoring internal tissue condition by transmittance
|Feb 9, 2004
|Dec 16, 2004
|Alfred E. Mann Institute For Biomedical Engineering At The Univ. Of S. California
|Surgical drain with sensors for differential monitoring of internal condition
|Aug 13, 2004
|Jul 28, 2005
|Turner Nicholas M.
|Medical devices
|Apr 21, 2005
|Dec 29, 2005
|Eigler Neal L
|Implantable pressure transducer system optimized for reduced thrombosis effect
|Mar 3, 2006
|Jul 6, 2006
|Furnary Anthony P
|Method and apparatus for monitoring blood condition and cardiopulmonary function
|May 26, 2006
|Sep 28, 2006
|Alfred E. Mann Institute For Biomedical Engineering
|Surgical Drain with Sensors for Monitoring Internal Tissue Condition
|May 26, 2006
|Sep 28, 2006
|Alfred E. Mann Institute For Biomedical Engineering
|Surgical Drain with Sensors for Monitoring Internal Tissue Condition
|Nov 20, 2006
|Aug 16, 2007
|Hiroyuki Muramatsu
|Blood rheology measuring apparatus
|Apr 17, 2006
|Nov 1, 2007
|Decampli William M
|Apparatus for regulating blood flow to treat cardiac abnormalities and methods of use
|May 25, 2007
|Dec 6, 2007
|Cook Vascular Incorporated
|Adjustable tension cuff assembly
|Jun 6, 2006
|Dec 6, 2007
|Shuros Allan C
|Method and device for lymphatic system monitoring
|Jun 6, 2006
|Dec 6, 2007
|Shuros Allan C
|Method and apparatus for gastrointestinal stimulation via the lymphatic system
|Jun 6, 2006
|Dec 6, 2007
|Shuros Allan C
|Amelioration of chronic pain by endolymphatic stimulation
|Jun 6, 2006
|Jan 10, 2008
|Shuros Allan C
|Method and apparatus for introducing endolymphatic instrumentation
|Sep 1, 2006
|Apr 24, 2008
|Shuros Allan C
|Method and apparatus for endolymphatic drug delivery
|Oct 31, 2007
|Jul 10, 2008
|Cvrx, Inc.
|Method and Apparatus for Providing Complex Tissue Stimulation Parameters
|Jan 31, 2007
|Jul 31, 2008
|Markowitz H Toby
|Systems and methods for monitoring effectiveness of congestive heart failure therapy
|Feb 13, 2008
|Aug 14, 2008
|Sharma Virender K
|Method and Apparatus for Electrical Stimulation of the Pancreatico-Biliary System
|Feb 19, 2008
|Aug 28, 2008
|University Of Louisville Research Foundation
|Perivascular pressure sensor and sensing system
|Feb 27, 2008
|Nov 20, 2008
|Cvrx, Inc.
|External Baroreflex Activation
|Mar 17, 2008
|Jan 15, 2009
|Cvrx, Inc.
|Methods and devices for controlling battery life in an implantable pulse generator
|Feb 27, 2009
|Jun 25, 2009
|Alfred E. Mann Institute For Biomedical Engineering At The University Of Southern Californ
|Implantable device with sensors for differential monitoring of internal condition
|Jan 31, 2008
|Aug 6, 2009
|Enopace Biomedical Ltd.
|Intra-aortic electrical counterpulsation
|Apr 27, 2009
|Sep 10, 2009
|Shuros Allan C
|Method and device for lymphatic system monitoring
|Nov 17, 2009
|May 20, 2010
|G&L Consulting, Llc
|Method and apparatus for reducing renal blood pressure
|Jun 2, 2010
|Dec 2, 2010
|Enopace Biomedical Ltd.
|Thoracic aorta and vagus nerve stimulation
|Aug 26, 2010
|Dec 23, 2010
|Imad Libbus
|Percutaneous neurostimulator for modulating cardiovascular function
|Aug 26, 2010
|Dec 23, 2010
|Imad Libbus
|Transcutaneous neurostimulator for modulating cardiovascular function
|Dec 15, 2010
|Apr 7, 2011
|Imad Libbus
|Transcutaneous neurostimulator for treating hypertension
|Nov 27, 2009
|Dec 1, 2011
|Royal United Hosptial Bath NHS Trust
|Method of measuring blood pressure and apparatus for performing the same
|Jul 5, 2012
|Volcano Corporation
|Lumen Based Pressure Sensing Guidewire System with Distortion Correction
|Apr 20, 2009
|Oct 21, 2010
|Albert-Ludwig-Universit�t Freiburg
|Implantable device i.e. reflective photoplethysmograph sensor, for extravascular detection of blood pressure of patient, has light absorbing optical barrier provided between solid body light source and solid body photodetector
|Sep 27, 2001
|Mar 14, 2012
|CVRX, Inc.
|Devices for cardiovascular reflex control
|Nov 9, 2001
|Jun 20, 2002
|Data Sciences Int Inc
|Adjustable flow probe
|Mar 27, 2003
|Oct 9, 2003
|Cvrx Inc
|Electrode structures and methods for their use in cardiovascular reflex control
|Feb 26, 2004
|Sep 10, 2004
|Alfred E Mann Inst Biomed Eng
|An implantable device with sensors for differential monitoring of internal condition
|May 16, 2006
|Nov 23, 2006
|Universitaetsklinikum Freiburg
|Implantable blood pressure sensor
|Jan 25, 2008
|Aug 7, 2008
|Medtronic Inc
|Systems and methods for monitoring effectiveness of congestive heart failure therapy
|Dec 30, 2010
|Jul 7, 2011
|Boston Scientific Scimed, Inc.
|Patterned denervation therapy for innervated renal vasculature
|U.S. Classification
|600/486, 600/488, 623/1.1
|International Classification
|Cooperative Classification
|European Classification
|Dec 28, 1998
Owner name: MEDTRONIC, INC., MINNESOTA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:MIESEL, KEITH A.;STYLOS, LEE;REEL/FRAME:009695/0215
Effective date: 19981222
|Oct 9, 2001
|Certificate of correction
|Jan 29, 2004
|Fee payment
Year of fee payment: 4
|Jan 7, 2008
|Fee payment
Year of fee payment: 8
|Feb 23, 2012
|Surcharge for late payment
Year of fee payment: 11
|Feb 23, 2012
|Fee payment
Year of fee payment: 12


text:
This product has additional required/recommended options. To configure, add to your cart.
Wall Switch Recommended
Trim Kit Recommended
Electric Heated Towel Warmer from 800 Series
Mr Steam Models 500 and up electric heated Towel Warmers are available in wall-mounted stationary and pivoting designs as well as free-standing cord connected models. Constructed of 100% Brass and equipped with stainless steel heating elements, they are ideal complements to any bathroom.
Product Features:
- Dimensions: 24-1/2" H x 23" W x 4-3/4" D
- Wattage: 100
- 120 Volts
- UL Listed
- Ship Weight: 25 lbs.
- Can be installed with wiring box on left or right side
Additional Mr Steam Links
This Mr Steam item can also be referenced by the following color/finish specific model #:
- Mr Steam W842AB*
Antique Brass
- Mr Steam W842BN
Brushed Nickel
- Mr Steam W842ORB
Oil Rubbed Bronze
- Mr Steam W842PB*
Polished Brass
- Mr Steam W842PC
Polished Chrome
- Mr Steam W842PN
Polished Nickel
- Mr Steam W842SG*
Satin Gold
*Discontinued model or finish
More Than Just a Store - Your Reasons to Shop
Discover the best online deals for the W842 and everything else for your home from the leading online home improvement store.
We're committed to providing you with the best assortment of Mr Steam products with over 400 warehouses across the country.
Our knowledgeable product specialists work to bring you the best prices on every product and provide unmatched customer service. Your satisfaction is our highest priority!
Installation Resource Center
Whether you are installing the product yourself or hiring a professional, our installation Resource Center has everything you need to get your project done. Visit the Installation Resource Center to get started.

subset: commoncrawl


text:
26 Spring Crafts for Kids
Bonus: Get our newsletter & special offers for free. We will not share or sell your email address. View our Privacy Policy
Spring Flower Pen
By: Linda Peterson
1 Comments
We are adding the craft to your Craft Projects.
The project was added to your Craft Projects.
Click here to view your Craft Projects.
This Spring Flower Pen from Linda Peterson is pretty and fun. Everyone will fall in love with this spring craft.
- 1 Bush of flowers (5-7 stems)
- Wire cutters
- Floral Tape
- Ink Pens (BIC Round Sticks work great)
- Scissors
- 4" clay pot
- Clear tape
- Package of black beans or beans in choice of color
- Cut stems apart with wire cutters and trim to length of ink pen.
- Place stem against ink pen and wrap with floral tape.
- Place a piece of clear tape in the bottom of the clay pot.
- Add beans.
- Press pens into the beans inside pot.
- Switch out blooms to coordinate with the season.
- Decoupage images or paint pot as desired
Your Recently Viewed Projects
Images from other crafters
Love these projects? Get MORE delivered to your Inbox – FREE Click Here!
Mar 11, 2011
I have seen this craft in various places and it always looks good if you have a lot of flowers to fill the pot. The instructions are easy to follow. A flower pen might make a nice inexpensive gift.
Report Inappropriate Comment
Are you sure you would like to report this comment? It will be flagged for our moderators to take action.
Thank you for taking the time to improve the content on our site.
What's Hot
- 24 Decorative Front Door Wreaths for Spring
- Decoupaged Clay Pot
- Easiest Yarn Wrapped Spring Wreath
- Mini Spring Bonnets
- Scrap Fabric Spring Chick
- 13 Spring Crafts: Easter Craft Ideas, Spring DIY Home Decor, and More free eBook
- Crochet Spring Celebration Wreath
- Rock Mosaic
- 5 Minute Spring Bouquet
- Bright Button Flowers
Something worth saving?
- SAVE all your favorite crafts
- ADD personal notes
- QUICKLY reference your crafts
Free eBooks
- 31 Halloween Craft Projects
- 8 Free Apron Sewing Patterns
- Wrap Up With 25 Afghan Patterns from Red Heart Yarn
- Quick & Thrifty Crochet Patterns
- Make It A Red Heart Holiday
- Quick and Easy Decoupage from Plaid
- Homemade Christmas Gift Guide
- 9 Thanksgiving Decorations to Make
- The Make My Own Holiday Gifts Guide
- Fun & Fabulous Mod Podge Projects From Plaid


text:
We want you to enjoy your visit to our website. That's why we use cookies to enhance your experience. By staying on our website you agree to our use of cookies.
Weymouth 5.5°c
BERNIE & JEANETTE HOSKINS Congratulations to a lovely brother and sister-in-law on your 45th Wedding Anniversary. Have a lovely day. Lots of love ~ Jan and Trish (The Bridesmaids) and all the
gang xxxx.
12:00am Tuesday 6th July 2004
This site is part of Newsquest's audited local newspaper network | A Gannett Company
Newsquest Media (Southern) Ltd, Loudwater Mill, Station Road, High Wycombe, Buckinghamshire. HP10 9TY|1350|Registered in England & Wales
This website and associated newspapers adhere to the Independent Press Standards Organisation's Editors' Code of Practice. If you have a complaint about the editorial content which relates to inaccuracy or intrusion, then please contact the editor here. If you are dissatisfied with the response provided you can contact IPSO here


text:
Select a category below to jump directly to that section... Click here for alternate link to faculty page.
Chapman, Barbara
Ph. D., Queen's University, Belfast, N. Ireland
Compiler Technology, Parallel Programming Languages, Tool Support for Application Development, Parallel Computing and High-Performance Computing, Computational Grids
Office: PGH 215
(713) 743-3354
Chen, Guoning
Assistant Professor
Ph.D., Oregon State University
Scientific Data Analysis and Visualization, Computational Topology, Geometric Modeling, Dynamical Systems
Office: PGH 566
Cheng, Albert
Ph.D., University of Texas at Austin
Real Time Systems, Distributed Systems, AI, Software Engineering, Computer Security, Computer Networks
Office: PGH 534
(713) 743-3353
Cheng, Kam-Hoi
Associate Professor
Ph.D., University of Minnesota
Artificial Intelligence, Object-Oriented Paradigm, Parallel Processing
Office: PGH 579
(713) 743-3357
Deng, Zhigang
Associate Professor and Director of Graduate Studies
Ph.D., University of Southern California
Computer Graphics, Computer Animation, Interactive Media and Gaming, Human Computer Interaction
Office: PGH 536
(713) 743-1018
Eick, Christoph
Associate Professor
Ph.D., Karlsruhe, Germany
Knowledge Discovery and Data Mining, Evolutionary Computing,
Machine Learning, Artificial Intelligence, Databases
Office: PGH 589
(713) 743-3345
Gabriel, Edgar
Associate Professor
Ph.D., University of Stuttgart, Germany
Message Passing Systems, High Performance Computing, Parallel Computing on Distributed Memory Machines, Grid Computing
Office: PGH 524
(713) 743-3857
Gnawali, Omprakash
Assistant Professor
Ph.D., University of Southern California
Wireless and Sensor Networks, Network Protocols and Architectures, Mobile Sensing, Social Networks, and Distributed Systems
Huang, Stephen S.-H.
Ph.D., University of Texas-Austin
Data Structures, Design and Analysis of Algorithms, Computer Security, Intrusion Detection
Office: PGH 594
(713) 743-3338
Johnson, Olin
Ph.D., University of California-Berkeley
Numerical Analysis, High Performance Computing
Office: PGH 596
(713) 743-3343
Johnsson, Lennart
Hugh Roy and Lillie Cranz Cullen University Professor of Computer Science, Mathematics and Electrical and Computer Engineering
Ph.D., Chalmers Institute of Technology, Gothenburg, Sweden
Parallel Computer Systems Architecture, Run-Time Systems, Data Parallel Programming and Large-Scale Applications
Office: PGH 592
(713) 743-3371
Kakadiaris, Ioannis
Hugh Roy and Lillie Cranz Cullen University Professor of Computer Science, ECE and Biomedical Engineering
Ph.D., University of Pennsylvania
Biomedical Image Analysis; Computational Biomedicine; Biometrics; Computer Vision; Pattern Recognition
Office: HBS 322
(713) 743-1255
Leiss, Ernst
Dr. Techn., TU Wien, Vienna
Vector and Parallel Computing, Data Security, Databases, Formal and Programming Languages, Geophysical Data Processing
Office: PGH 522
(713) 743-3359
Mukherjee, Arjun
Assistant Professor
Ph.D., University of Illinois at Chicago
Bayesian Inference, Data Mining, Natural Language Processing, Sentiment Analysis, Opinion Spam, Web Mining
Office: PGH 582
Ordonez, Carlos
Associate Professor
Ph.D., Georgia Institute of Technology
Database Systems, Data Mining
Office: PGH 571
(713) 743-1013
Pandurangan, Gopal
Associate Professor
Ph.D., Brown University
Algorithms, Distributed Computing, Networks, Large-scale Data, Computational Biology
Office: PGH 572
(713) 743-5556
Pâris, Jehan-François
Ph.D., University of California-Berkeley
Distributed Systems, Storage Systems, Performance Evaluation
Office: PGH 569
(713) 743-3341
Pavlidis, Ioannis
Eckhard Pfeiffer Professor
Ph.D., University of Minnesota
Computational Biomedicine, Biometrics, Computer Vision, Pattern Recognition, Software Engineering
Office: HBS 306
(713) 743-0101
Shah, Shishir
Associate Professor and Director of Undergraduate Studies
Ph.D., University of Texas, Austin
Quantitative Microscopy, Computational Biology, Image Analysis, Computer Vision, Pattern Recognition, Image Instrumentation
Office: 564 PGH
(713) 743-3360
Shi, Weidong (Larry)
Assistant Professor
Ph.D., Georgia Institute of Technology
Enabling System Technologies for Services, Computing Systems, Behavioral Computing
Office: 567 PGH
Solorio, Thamar
Associate Professor
Ph.D., National Institute of Astrophysics, Optics, and Electronics
Natural Language Processing, Information Extraction
Office: 584 PGH
(713) 743-5542
Subhlok, Jaspal
Professor and Chair
Ph.D., Rice University
Compiler and RunTime Support for Parallel and Distributed Systems, Network Based Computing, Scientific and Stream Processing Applications
Office: PGH 217
(713) 743-3340
Tsekos, Nikolaos V.
Associate Professor
Ph.D., University of Minnesota
Magnetic Resonance Imaging (MRI), Computer Technology, Robotics and Physiology
Office: 595 PGH
(713) 743-3358
Verma, Rakesh
Ph.D., State University of New York-Stony Brook
Formal Methods, Computational Logic, Text/Data Mining and Computer Security
Office: 532 PGH
(713) 743-3348
Vilalta, Ricardo
Associate Professor
Ph.D., University of Illinois at Urbana-Champaign
Machine Learning, Computational and Statistical Learning Theory, Neural Networks, Data Mining, Artificial Intelligence
Office: HBS 332
(713) 743-3614
Hilford, Victoria
Instructional Associate Professor
Ph.D., University of Houston
Programming Languages, Object-Oriented Analysis and Design, Data Structures, Database Management, Software Engineering, Fault Tolerant Computing Systems, Web Programming
Office: PGH 513
(713) 743-3342
Rizk, Nouhad
Instructional Associate Professor
Ph.D., University of Houston
Doctorate, Leicester University, UK
E-Learning, Information Retrieval, Parallel Programming, Web Designing
Office: PGH 529
(713) 743-3710
Yun, Chang
Instructional/Research Assistant Professor
Ph.D., University of Houston
Interactive Game Design/Development, Video Game Research, Serious Game, Computer-Aided Education, Human-Computer Interaction
Office: 514 PGH
(713) 743-3366
Fofanov, Yuriy
Research Professor
Ph.D., Kuibyshev (Samara) State University, USSR
Bioinformatics, Applied Statistics, Mathematical Modeling and Information Theory
Office: PGH 230
(713) 743-8553
Guillen, Pablo
Research Assistant Professor
Ph.D., Polytechnic University of Catalonia (UPC), Barcelona, Spain
Machine Learning, Statistical Learning, Pattern Recognition and Data Mining
Office: HBS 334
(832) 842-7096
Taha, Walid
Research Professor
Ph.D., Oregon Graduate Institute
Cyber-Physical Systems, Modeling and Simulation, Embedded Systems, Programming Languages, Software Engineering
Tsiamyrtzis, Panagiotis
Research Associate Professor
Ph.D., School of Statistics, University of Minnesota
Statistical Aspects of Computer Vision Problems, (Bayesian) Statistical Process Control, Bayesian Statistics
Zhao, Xi
Research Assistant Professor
Ph.D., Ecole Centrale de Lyon, France
Biometrics, Feature Extraction, Computer Vision, Pattern Recognition
Office: 348 PGH
Zheng, Rong
Research Associate Professor
Ph.D., University of Illinois at Urbana-Champaign
Wireless Networking, Distributed Systems, Network Simulation and Performance Modeling
Office: 565 PGH
(713) 743-2663
Huang, J.C.
Emeritus Professor
Ph.D., University of Pennsylvania
Software Engineering, Real-Time Computer Systems, Program Analysis and Testing
(713) 743-3350
King, Willis
Emeritus Associate Professor
Ph.D., University of Pennsylvania
Computer Architecture, Distributed Systems
(713) 743-3350
Garbey, Marc
Ph.D., Ecole Centrale de Lyon, France
Algorithms, Numerical Analysis, Parallel Computation and Scientific Computing
Office: PGH 501
(713) 743-3337
Gurkan, Deniz
Joint Associate Professor
Ph.D., University of Southern California
Computer networking, measurements, software-defined networking, interoperable data exchange protocols, distributed systems, optical fiber communications.
Office: TCB 230B
(713) 743-4037
Han, Zhu
Joint Professor
Ph.D., University of Maryland, College Park
Big Data Analysis, Wireless Networking, Smart Grid, Resource Allocation, and Security
Office: W302 (Engineering Building II)
(713) 743-4437
Merchant, Fatima
Joint Assistant Professor
Ph.D., The University of Texas at Austin
Computational Biology and Medicine
Roysam, Badrinath
Joint Professor of Computer Science
Professor of Electrical and Computer Engineering
Ph.D., Washington University
Biological image analysis High-performance computing Brain tissue mapping and Neuroprosthetic devices Computational Immunology Subsurface sensing and imaging systems
Office: Engineering 1 N325
(713) 743-4400
Anderson, David
Adjunct Professor
Ph.D., University of Wisconsin
Madison volunteer computing, distributed operating systems, real-times systems, computer graphics, and computer music
Collet, Christophe
Adjunct Professor
Ph.D., University of Toulon, France
Statistical Image Analysis, Multiscale Segmentation, Markov Models, Pattern Recognition and Classification, Bayesian Inference, Multigrid and Genetic Optimization
Hohl, Detlef
Adjunct Associate Professor
Ph.D., Technical University Aachen, Germany
Computational Physics & Chemistry, HPC, Seismic Processing and Inversion, Oil and Gas Reservoir Modeling, Big Data and Data Analytics for Oil and Gas, Reservoir Surveillance and 4D Seismic
Kojic, Milos
Adjunct Professor
Ph.D. Mechanical Engineering, Rice University
Finite Element Methods and software Development for Applications in Engineering and Bioengineering
Paliouras, Georgios
Adjunct Professor
Ph.D., University of Manchester, UK
Machine Learning, Knowledge Discovery, Artificial Intelligence, Personalization, User Modeling, Information Filtering, Information Extraction, Event Recognition, Ontology Learning, Grammar Learning, Web Mining
Perantonis, Stavros
Adjunct Professor
Ph.D., D.Phil. Comp. Physics, University of Oxford, UK
Computational Intelligence, Artificial Neural Networks, Image and Multimedia Processing, Pattern Classification, Pattern Recognition
Ramamurthy, Uma
Adjunct Assistant Professor
Ph.D., University of Memphis
Computer Architecture, Operating Systems, Database Systems, Modeling Cognitive Processes, Medical/Research Informatics
Schwarz, Thomas
Adjunct Professor
Ph.D., University of California, San Diego
Storage Systems, Reliability Algebraic Signatures and their Application in Security and Storage Systems, and Database Systems
Stotzer, Eric
Adjunct Assistant Professor
Ph.D., University of Houston
Compilers and Systems Software, Parallel Programming Languages, High Performance Embedded Computing, Computer Architecture, Speech and Language Development
Subramaniam, Venkat
Adjunct Professor and Instructional Professor
Ph.D., University of Houston
Software Development, Principles and Practices
Tao, Cui
Adjunct Assistant Professor
Ph.D., Brigham Young University
Conceptual Modeling, Ontologies, Semantic Web, Clinical and Biomedical Informatics
Baez-Franceschi, Jose
Assistant Professor, College of Technology
MFA, University of Houston
BFA, Ringling College
Computer Animation, Game Development, Digital Multimedia Development and Sustainable Renewable Art
Office: Fleming 127
(713) 743-6174
Lea, Robert
Ph.D., University of Houston
Artificial Intelligence, Numerical Analysis, Computer Architecture
Office: PGH 531
(713) 743-3350
Singh, Raj
Ph.D. University of Louisiana at Lafayette
Software Design, Software Engineering, Databases, Big Data, Business Intelligence, Data Warehousing, Data Mining
Office: PGH 531
Select a category below to jump directly to that section...
Click here for alternate link to faculty page.


text:
Baby Showers
> All Posts
Adorable Owl & Bird Cupcakes
I've finally found the perfect dessert inspiration for this Owl Baby Shower theme: these adorable red owl cupcakes from baking guru hello naomi! And the red robin design would be perfect for a "Love Birds" bridal shower or Birdie-themed baby shower. Swoon.
GIVING BACK: Operation Shower
- posted in:
- Baby Showers
I'm really excited to share this post from Amy with you today. Hope you'll take a few minutes to read on, get inspired, and learn how you can be a part of this celebration too! ***** Hello, fabulous HWTM readers. It's Amy Belle Isle from Stem Parties again. How have you been? Me? Great! Thanks […]
HWTM in Pregnancy Magazine: Owl-Themed Baby Shower
Have a baby shower in the works? If you're looking for a fun concept, here's an owl-themed baby shower that I created for this month's issue (April 2009) of Pregnancy magazine, featuring super-cute plush owls, colorful feathers, woodsy elements, and more! {view the article at a larger size} P.S. This could also be a fun […]
No matter which critter you're looking at, these colorful, felt toy party favors are pretty much the cuteness! Find out more about them over at Simply Stated today!
Adorable & Creative "Egg Creatures"
- posted in:
- Baby Showers
- Easter
- Kids Birthday
When it comes to decorating Easter eggs, I usually opt for the easy route, but these Egg Creatures from Martha Stewart are so ridiculously cute, I'm thinking they might be worth the extra effort! Especially the momma and baby pig design… how adorable is that?! (Cute decorating ideas for baby showers & kids parties too!)
How many of you have ever been to a "Sip and See"? A Sip and See is basically a casual type of baby shower that's held AFTER baby arrives instead of before. Friends and family are invited to drop by "open house style" within a certain time frame to meet the baby (see) and enjoy […]


text:
Connect With Us800-299-9470
A tall headboard and blocky design give our Eaton Modern Bedroom Set a commanding appearance but the simple design makes it approachable and easy to add to your home. Each of the five pieces included with purchase is made of eco-friendly solid rubber wood: the queen bed, six-drawer dresser, mirror, and two two-drawer nightstands. A black finish along with silver metal drawer pulls are a sophisticated way to finish off the bedroom furniture set. Both a mattress and box spring are required for the bed. Assembly required.
Features and Specifications:
Sign Up & Get Exclusive Deals and Rewards


text:
Place an Order
24 hours a day / 7 days a week
Customer Service
Monday thru Saturday
5am to 9pm (PT)
6am to 6pm (PT)
Technical problems
Monday thru Friday
7am to 4pm (PT)
Order Status
Returns and Exchanges
Double click on above image to view full picture
Recently Viewed
(based on 5 reviews)
Ratings Distribution
5 Stars
4 Stars
3 Stars
2 Stars
1 Stars
of respondents would recommend this to a friend.
Best Uses
Reviewed by 5 customers
Sort by
Displaying reviews 1-5
Back to top
Just what I needed
By Poindexter
from La.
About Me Avid Do-It-Yourselfer
Comments about HFT 5 ft. 6 in. x 7 ft. 6 in. Green/Farm All Purpose/Weather Resistant Tarp:
Had some unexpected cold weather come in and needed something to block the wind from my outdoor aviary. This tarp was just the ticket. Works great and a great price.....
Bottom Line Yes, I would recommend this to a friend
Was this review helpful? Yes / No
- You may also flag this review
Comment on this review
(0 of 1 customers found this review helpful)
Great Bargain Store
By Pearl the Singer
from Venice
Love Harbor Freight Tool store for the selection of useful products, great prices and knowledgeable staff. Tarps came in useful in our roof leak; great to keep them available in the garage.
By Boots
from Bend OR.
Great store. My tools are just fine
Working like a charm,,,!!! Eloy, AZ
By pazman
from eloy, az
Once charged up & 5-min later, it jumped my 90' Lariet...in just one try....Twice in one week.....
(2 of 2 customers found this review helpful)
OK for a Couple of Years
By Outdoor Enthusiast
from Olympia, WA
About Me Casual/ Recreational
This tarp was used to cover a small garden plot to prevent weed growth.Looked good and worked well the first 2 seasons.But after 2 years, it began to rot and split.For the money, a decent tarp.
Most orders will be processed and leave our warehouses within 1-2 business days from the time your order is placed. Orders placed on Saturday or Sunday will be processed on Monday.
*Overweight items are shipped via freight delivery, please allow up to 15 business days for shipping. Standard Ground Shipping rates apply to deliveries within the continental Unites States.
© Copyright 2016 Harbor Freight Tools. All Rights Reserved. MATERIAL AND PHOTOS ON THIS SITE ARE THE COPYRIGHTED PROPERTY OF HARBOR FREIGHT TOOLS AND MAY NOT BE USED WITHOUT THE PRIOR WRITTEN PERMISSION OF HARBOR FREIGHT TOOLS.
Harbor Freight Tools does not endorse any other business or organization or any technique in any customer and/or fan video. Always follow all of the instructions and warnings included with our products. We make no representation or warranty of any kind by including any customer and/or fan video on our website.
LIFETIME WARRANTYLifetime Warranty on Hand Tools! We guarantee our Hand Tools to be free from defects in material and workmanship for the life of the product. Limitations apply.
ABOUT HARBOR FREIGHT TOOLSFounded in 1977, Harbor Freight Tools is the leading discount tool retailer in the U.S. selling great quality tools at "ridiculously low prices" in over 600 stores nationwide and on its website www.harborfreight.com. Harbor Freight Tools stocks over 7,000 items in categories including automotive, air and power tools, shop equipment and hand tools. With a commitment to quality and a lifetime guarantee on all hand tools, Harbor Freight Tools is a favorite of automotive and truck repair shops, government agencies, schools, manufacturers, contractors and tool enthusiasts who want top-quality and great selection and value. The results speak for themselves - with over 30 million customers and thousands of people switching to Harbor Freight Tools every day.


text:
The Best Three Things We Learned
16 / 16
Beer drinkers are tougher
Cyclists who enjoy a postride pint average one more ride a week than those who opt for other recovery methods. And beer lovers are less likely to skip a ride if it's raining.
Mountain bikers are cheerier
If you ride the dirt, you're slightly more likely to wave at other riders, keep your group together, and feel happier and more social than cyclists who never hit the trails. (Possibly related: You're also more likely to get naked in a parking lot.)
The real reason to ride to work
Compared with those who ride only recreationally, cyclists who pedal for transportation were more likely to report that cycling led to more sex, better relationships, and clearer thinking.
16 / 16
Copyright © 2016 Rodale Inc.


text:
The World's Largest Coalition of Nontheists and Nontheist Communities!
All the news that shouldn't be news for July 12, including: - HEADLINES; - Go, Johnny, GO! - Microsoft Ventures Into Shipboard Furnishings
Views: 21
Tags: 214, 777, abuse, asiana, church, More…clergy, culpa, deen, dzhokhar, egypt, england, evil, god, have, ii, islam, john, lord, mea, mercy, miracle, my, of, oh, pagan, paul, paula, perry, pope, rick, saint, sfo, spirits, tonto, tsarnayev, xsfo
Join Atheist Nexus
Welcome toAtheist Nexus
Sign Upor Sign In
Or sign in with:
Nexus on Social Media:
© 2016 Atheist Nexus. All rights reserved. Admin: Richard Haynes.
Powered by
Badges |
Report an Issue |
Terms of Service
Please check your browser settings or contact your system administrator.


text:
Pack it, Man
Ironic, after all we expected and all he's been through and all he exhibited during training camp, that Pacman Jones' season is ended by this:
A muffed punt.
Back in Oxnard, Calif., Pacman dazzled us by catching six punts and holding all six balls without dropping a single one. In the end, because he couldn't catch just one, he suffered a neck injury that likely has finished his season (he definitely won't play Sunday against the Giants) and sealed his dramatic return to the NFL into one giant, not-even-close-to-worth-all-the-trouble disappointment.
Pacman, case you didn't hear, had some issues off the field. But on the field his biggest flop was as a crappy kick-returner.
He arrived in Dallas on the heels of three kick-return touchdowns in his last full NFL season with the Titans in '06 and a career punt-return average over 11 yards. But in Dallas he averaged a measly, paltry, harmless 4.6.
Blame Pacman. Blame Jerry Jones. Blame special teams coach Bruce Read.
But also save a little venom for the Steelers' Andre Frazier.
Pacman was likely already injured after he muffed a punt and fell on the ball and a pile amassed on top of him late in last Sunday's first quarter. But Frazier didn't help matters - he deserved a penalty, if not a fine - by yanking and dragging Pacman off the pile by his horse collar, all after the referees had already blown the whistle and signaled Cowboys' ball.
If you believe in karma, I guess one asshole deserves another. - Richie Whitt
Get the This Week's Top Stories Newsletter
Every week we collect the latest news, music and arts stories — along with film and food reviews and the best things to do this week — so that you'll never miss Observer's biggest stories.
Sponsor Content
More Blogs News
- After Club Signs Nathan, Moves Neffy to Rotation, Rangers Offer Wilson Arbitration
Upcoming Events
AT&T Stadium TourThu., Feb. 11, 12:00am
Tour of AT&T Stadium ArtThu., Feb. 11, 12:00am
North Texas Mean Green Womens Basketball vs. Old Dominion Lady Monarchs BasketballThu., Feb. 11, 7:00pm
POETRY SMASH #3Thu., Feb. 11, 7:30pm
- Tickets
- They Went and Stole Shawn Bradley's Bike!
- From Penn State to the Dallas Stars, the Good, Bad and Godawful News in This Day in Sports


text:
Drawing With Stickguy
A learning book by Marnie Joan Shick
door Marnie Joan Shick
Marnie is a 5th grader who's been making books since she was 4. This is her first one that her daddy helped her publish. She's really excited to share her love of art and storytelling with everyone.
Commentaar geplaatst.
Schrijf je in
of maak een account om commentaar te plaatsen.