text:
|Publication number
|Publication type
|Application number
|Publication date
|Mar 25, 2003
|Filing date
|Oct 6, 2000
|Priority date
|Oct 7, 1999
|Fee status
|Also published as
|Publication number
|Alexander K. Mills
|Original Assignee
|Alexander K. Mills
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (16), Referenced by (108), Classifications (39), Legal Events (5)
|External Links: USPTO, USPTO Assignment, Espacenet
This application claims benefit of provisional application Ser. No. 60/158,097 filed on Oct. 7, 1999.
The present invention relates generally to noninvasive methods of quantitatively determining various physiologic parameters relating to cardiovascular and respiratory function. More particularly, the invention relates to a method and apparatus for continuous, noninvasive determination of: arterial blood pressure, venous pressure, arterial oxygen saturation, venous oxygen saturation, arterial pulse wave velocity, aortic pulse wave velocity, aortic pulse flow velocity, cardiac stroke volume, cardiac output, heart rate, and respiratory rate.
Critically ill and seriously injured patients require constant care and attention. Doctors, nurses, and hospital technicians need a continuous flow of information about the many patients under their care. Heart rate and blood pressure measurements are two primary vital signs that indicate the health of patients under their care. When these two common indices of wellness fall below normal readings, a patient is usually in distress and requires immediate attention.
Dangerous conditions brought about by a cardio-vascular or pulmonary disease, severe trauma, or drug abuse may bring about a failure of the lungs and heart to supply the bloodstream with life-giving oxygen. Such a fatal deficiency can be detected by continually gauging the amount of hemoglobin in the bloodstream that is carrying oxygen. This third vital sign, which manifests oxygen saturation of the blood, is especially critical because a rapid decline in oxygen in the bloodstream is associated with increased risk of patient mortality.
It is well known that blood pressure can be directly measured by placing a fluidfilled catheter directly into the vessel and coupling this to an electromechanical transducer. This is the most accurate means, but has all the disadvantages of invasive measurement, including pain on insertion, risk of infection or disease transmission, risk of bleeding or thrombosis, and great expense. A further disadvantage is the creation of toxic medical waste (needle, gloves, skin dressing, etc).
Blood pressure measurement can also be measured indirectly using an occlusive cuff (with either auscultation or oscillometry to make the determination). This is the most common means of blood pressure measurement. Illustrative are U.S. Pat. Nos. 5,582,179, 5,048,533, 5,152,296 and 4,793,360.
A further occlusive cuff apparatus is disclosed in U.S. Pat. No. 5,766,130. According to the invention, the apparatus includes multiple "pressurized pneumatic cuffs" that are used to "plot blood pressure and/or volumetric blood flow wave forms from a plurality of separate digits and/or extremities of a patient so that circulatory parameters may be measured rapidly and recorded from a great number of the patient's digits or limbs".
Although commonly employed, the occlusive cuff also has numerous disadvantages, which include discomfort, intermittent readings, and poor reliability An additional means of determining blood pressure is through an assessment of "pulse wave velocity". Several prior art references disclose methods and/or apparatus employing such means. Illustrative is U.S. Pat. No. 5,649,543.
There are also several prior art references that disclose methods and/or apparatus for determining blood pressure through a "pulse wave amplitude" assessment Illustrative are U.S. Pat. Nos. 4,735,213, 4,872,461, 4,793,360, and 5,385,149.
Although most of the noted noninvasive blood pressure methods and apparatus, particularly the occlusive cuff, have been employed for many years by health care personnel, the conventional methods and apparatus have one major, common drawback-the need for separate calibration.
Accordingly, there is a need for noninvasive methods and devices for determining various physiological characteristics, such as central venous pressure and cardiac output, without separate calibration. There is also a similar need for noninvasive methods and devices for determining various blood parameters including pulse amplitude, pulse delay, pulse velocity, pulse contour, flow velocity and flow delay.
As will be appreciated by one having ordinary skill in the art, the present invention satisfies these and other needs.
The present invention includes a device for the noninvasive monitoring of a physiologic characteristic of a patient's blood. In one embodiment, the device comprises a tissue probe having a radiation emitter and a radiation detector configured to receive the radiation after absorbance through the patient's blood; a position sensor for determining the relative height of the probe compared to a level corresponding to the patient's heart; and a controller for computing the physiologic characteristic of the patient's blood based on the absorbance of the first wavelength of radiation and the relative height of the probe. The radiation emitters of the invention can utilize a single wavelength or a plurality of discrete wavelengths and may include visible light, infrared light, and ultraviolet light. The probes are adapted for use with hands, fingers, feet, toes, ears, earlobes, nares, lips, tongue and the like. Additional radiation emitters and detectors may also be used. Preferably, the probe further comprises ECG leads.
An alternative embodiment of the device of the invention comprises a tissue probe and controller in conjunction with a movement generator for inducing a position change of the probe with respect to a level corresponding to the patient's heart. Preferably, the movement generator induces a known position change of the probe and moves the probe to positions above and below a level corresponding to the patient's heart.
The invention also comprises method for determining a physiological characteristic of a patient's blood noninvasively. In one embodiment, absorbance characteristics of the blood are measured at varying positions relatively to the level of the patient's heart. By comparing blood parameters such as pulse amplitude, pulse velocity, pulse delay, pulse contour, flow velocity and flow delay to hydrostatic pressure differences induced by the position changes, characteristics such as arterial and central venous blood pressure and cardiac output can be determined. Alternatively, two probes are used to compute pulse delays between coupled tissues or opposing tissues.
The subject invention relates novel methods for noninvasive determination of physiologic characteristics. The first new and unique method and device utilizes changes in hydrostatic pressure induced by positional changes to facilitate measurements. A second new and unique method and device for noninvasive determination of cardiac output by measuring delays in pulse arrival times in coupled organs or members on opposite sides of the body is also described. The two methods are such that they can advantageously be used together.
By varying the hydrostatic pressure in an extremity, one can not only perform self-calibration for a blood pressure determination, but also change the pulse wave velocity and pulse propagation delay with respect to the opposite extremity. With this information pulse wave velocity, and consequently flow wave velocity at the aortic root can be determined.
Similar techniques of varying hydrostatic pressure can be used to assess venous pressure and saturation. The technique of repetitious determinations made while altering position or other variables allows a multitude of additional analyses to be made. The determinations can be made intermittently or continuously.
Further objects of the invention are exemplified by the following potential applications:
(a1). A patient is anesthetized for a surgical procedure. Probes are attached to the index fingers of each hand, and a movement generator is placed on one arm. A complete set of vital signs and physiologic characteristics is generated continuously, including: arterial blood pressure, venous pressure, arterial oxygen saturation, venous oxygen saturation, arterial pulse wave velocity, aortic pulse wave velocity, aortic pulse flow velocity, cardiac stroke volume, cardiac output, heart rate, and respiratory rate. Other characteristics can be calculated if desired.
(a2). A patient is anesthetized for a cardiac surgical procedure. As access to the arms is difficult, probes are attached to the patient's temples. A complete set of vital signs and physiologic characteristics is continuously generated.
(a3). A patient is anesthetized for a cardiac surgical procedure; this time the procedure includes valvular repair or replacement. Since the cardiac output and other characteristics can be continuously computed, the adequacy of the surgical repair can be judged immediately.
(a4). As the number of endoscopic or minimally invasive cardiac surgical procedures is expected to increase, the demand for less invasive monitoring will also increase. The device described herein provides noninvasive, continuous monitoring of essentially all cardiovascular characteristics.
(a5). Cardiac catheterization procedures are often done on critically ill patients. As the procedures are usually relatively brief and accomplished without general anesthesia, invasive monitoring methods are often not desired despite the illness of the patients. The device described herein will provide the necessary monitoring that is typically provided by much more invasive, expensive, and time consuming monitors
(a6). A patient is hospitalized in the intensive care unit of a hospital after a heart attack. Probes are attached to the index fingers of each hand, and a movement generator is placed on an arm or a leg. A complete set of vital signs and physiologic characteristics can be continuously generated. In addition, arrhythmias can be detected and diagnosed.
(a7). The patient noted above is now moved to a "step-down" or telemetry unit from the intensive care unit. Because the device described herein eliminates the need for invasive monitoring lines, a complete set of vital signs and physiologic characteristics can still be continuously generated. As the patient has mobility of arms and legs, a movement generator is no longer needed, as the patient's spontaneous motion, even during sleep, will generate hydrostatic pressures in the limbs, allowing all computations to be made. In addition, the probes may be made wireless, and connected to a central nursing station by means of infrared or radio frequency communication.
(a8). The patient noted in applications 6 and 7 above is now moved to a regular hospital bed, and does not require continuous monitoring. However, vital signs can still be recorded by a technician moving the device from bedside to bedside on a cart. The device does not require highly trained nursing personnel to operate.
(a9). The patient noted in applications 6, 7, and 8 above has now been discharged from the hospital, and now presents to his physician's office for follow-up. The same device can be used in physician's offices, as it provides better care at lower cost. (a10). Ambulances, emergency vehicles, and military vehicles can also employ this device as it is very simple to operate, and provides data that currently is impossible for them to obtain. In addition, the information can be transmitted to central stations where medical personnel are available for help and advice.
(a11). The device and methods of the invention will provide means of monitoring patients or checking vital signs for extended care facilities, nursing homes, and other health-related facilities
(a12). Blood pressure screening clinics and drugstores will have a greatly proved means of determining patient's blood pressures and other vital signs. Airports d airplanes are able to purchase medical equipment, but often do not have personnel trained to operate the equipment. The device is simple and quick to operate.
(a13). The patient noted in applications 6 through 9 above can also monitor his heart disease and health care at home. The operation of the device is straightforward enough to be used by the layman with minimal instruction, and inexpensive enough for personal home use. The patient can measure his cardiovascular characteristics daily, or as frequently as he and his physician desire. A communication means, such as a modem can easily be incorporated into the device. This, with appropriate software and support, would allow essentially instantaneous communication with a physician's office, clinic, or hospital. In addition, a permanent record can be made and stored electronically. If desired, the device could automatically "sign on" to the Internet or other network, and link to the appropriate website or other address. The ability to participate more fully in their own health care will improve the welfare of individuals.
(a14). The patient of above presents to the emergency room of a hospital with chest pain. The ER physician can access, via the Internet or other means, the patient's vital sign history, including ECG. This allows the physician to determine if abnormalities are new or chronic. Changes, such as dysrhythmias, can be identified as to when they first occurred, perhaps to within a time frame of hours or less.
(a15). People without diagnosed cardiovascular disease can use the device to allow themselves to participate in their own health care. This will allow virtually immediate diagnosis of any problems, allowing early intervention. In addition, a permanent record can be created if desired.
(a16). The device will impact fitness and physical training for everyone from lay people to military personnel to professional athletes.
(a17). The device can be employed in the diagnosis and management of peripheral vascular disease. Measurement of pulse wave velocity in the extremities, and particular differential pulse wave velocities in the lower extremities, can be used to diagnose peripheral vascular disease. Since measurements are real time and continuous, they can also be used in management. For example, if balloon angioplasty of an artery is performed, the clinician can tell immediately if flow has improved. In the case of angioplasty of coronary arteries, the clinician can follow cardiac characteristics on a beat-by-beat basis.
(a18). In addition to peripheral vascular disease, other diseases, such as abdominal aortic aneurysm, can be diagnosed and managed. Changes in pulse wave velocity and waveform can be followed for years if desired.
(a19). Some of the most important potential uses of the device relate to the health care of neonates and young children. For these patients, the measurement of common characteristics such as blood pressure can be difficult even for highly trained personnel in well-equipped facilities. The simple placement of probes on fingers will alleviate this. The device will also allow noninvasive diagnosis of congenital cardiac defects and anomalies. Analysis of differential pulse wave velocity and blood pressure will allow rapid, accurate, and specific diagnosis of many disorders, including Tetralogy of Fallot and transposition of the great vessels. The ability to distinguish both arterial and venous saturations and pressures will allow diagnosis of patent ductus arteriosus, truncus arteriosus, atrial septal defect, and ventricular septal defect. Differential arm and leg pulse wave velocities and pressures will confirm diagnosis of coarctation of the aorta. Because of its continuous measurements, the device can be used for only for diagnosis but confirmation of adequacy of repair, including intraoperatively. As the device is inexpensive and easy to operate, it may become a screening tool for newborns and infants.
(a20). The device can be used in conjunction with intra-aortic balloon pump (IABP) counterpulsation. Beat-by-beat analysis of effectiveness and ability to wean from counterpulsation can be made.
(a21). The device can be used in conjunction with placement of cardiac pacemakers, to set proper rate and timing intervals. In addition, efficacy of pacemakers can be checked as frequently as desired, and scheduling of reprogramming or replacement made automatically.
(a22). It is straightforward to incorporate other devices, such as the electroencephalogram (EEG) or electromyogram (EMG), into probes of the invention. As a general-purpose monitor, the device will invite the addition of specialized add-ons.
(a23). Many enhancements are included in the invention. For example, addition of chest (horizontal) leads allows full diagnostic ECGs to be performed.
(a24). Under some circumstances, such as severe hypotension, the pulse cannot be identified in the periphery. In such cases, many of the determinations claimed herein cannot be made. However, the ability of the device to identify venous blood can still give important information.
(a25). Forces other than gravity can be used. In a microgravity environment such as a space station orbiting the Earth, a device such as the one described could be constructed to perform all indicated determinations using acceleration caused by movement in place of gravitational acceleration.
(a26). As mentioned in the examples above, an anticipated use is in the field of home health care, with the possibility of automatic sign-on and direction to a website. As the user is already participating in his or her health care, the extension of providing access to related health or other information via the Internet� is a natural one.
(a27). A verification means, such as fingerprint scanning, can be incorporated into a personal-use device, to ensure that any medical information gathered belonged to the individual using the device.
(a28). The device will be used in conjunction with the Penaz technique or other methods, such as calibration with a cuff or other means, as desired.
Further features and advantages will become apparent from the following and more particular description of the preferred embodiments of the invention, as illustrated in the accompanying drawings, and in which like referenced characters generally refer to the same parts or elements throughout the views, and in which:
FIG. 1 is a diagram of the central cardiovascular system, showing the asyrmmetry of origins of the vessels off the aortic arch;
FIG. 2 shows a representative probe of the invention with a single emitter-detector pair;
FIG. 3 shows an alternative embodiment of a probe of the invention with a single emitter-detector pair;
FIG. 4 shows a probe of the invention with two emitter-detector pairs spaced a known distance apart. This can be used to measure the velocity of the pulse wave within the probe itself;
FIG. 5 shows a probe with a single emitter-detector pair and a single electrocardiogram (ECG) electrode;
FIG. 6 shows a probe with a single emitter-detector pair and two ECG electrodes;
FIG. 7 shows a probe with a two emitter-detector pairs and two ECG electrodes;
FIG. 8 shows a probe of the invention further comprising a position sensor;
FIG. 9 shows an embodiment of the invention with probes placed on opposite digits of a subject;
FIG. 10 shows an embodiment of the invention with probes placed on opposite temples of a subject;
FIG. 11 shows a circuit schematic of the invention comprising a photoplethysmogram;
FIG. 12 shows a circuit schematic of the invention comprising a photoplethysmogram with an ECG amplifier;
FIG. 13 shows a circuit schematic of the invention comprising a photoplethysmogram with an ECG amplifier and a level signal;
FIG. 14 shows a circuit schematic of the invention comprising a photoplethysmogram with two independent channels;
FIG. 15 shows a circuit schematic of the invention comprising a photoplethysmogram with two independent channels and an ECG amplifier;
FIG. 16 shows an embodiment of the invention with probes placed on the digit and on the arm near the brachial artery;
FIG. 17 shows an embodiment of the invention with probes placed on a finger and on a toe;
FIG. 18 shows an embodiment of the invention with probes placed on opposite fingers and on a toe;
FIGS. 19 and 20 show embodiments of the invention with probes placed on opposite digits of a subject positioned at differential heights relative to the patient's heart;
FIG. 21 shows an embodiment of the invention with probes placed on opposite fingers positioned at differential heights and on a toe; and
FIGS. 22-25 are graphical representations of an oscilloscope screen showing recordings using methods of the invention.
Functionally the heart is divided into two sides or sections. The right or pulmonary circulation section that receives blood from the veins of the body and pumps it through the lungs and the left or systemic circulation section that receives the blood from the lungs and pumps it to the body. Th blood is then collected in the veins to be returned to the right side of the heart. This anatomy is generally shown in FIG. 1. The arterial system begins at the aorta 1, to which the left ventricle of the heart pumps. The first three branches of the aorta are the brachiocephalic or innominate artery 2, the left (common) carotid artery 3, and the left subclavian artery 4. The brachiocephalic artery branches into the right subclavian 5 and right (common) 6 carotid arteries. These arteries provide the blood supply for the head and upper extremities. The aorta then passes down (caudad) through the body, continuing to provide arterial branches to organs, terminating as a bifurcation creating the iliac arteries. The brachiocephalic or innominate artery is the first branch of the aorta. It in turn branches into the right subclavian and right carotid arteries. In contrast, the left subclavian and left carotid arteries originate directly off the aortic arch. Thus, the subclavian and carotid arteries and any of their branches have different paths from their counterparts on the opposite side of the body.
Because of the different origins from the aorta and different branching pattern of the arterial tree, it can be appreciated that blood ejected from the left ventricle will not follow symmetrical pathways to opposite arms or opposite sides of the head. Similarly, the pressure pulse wave associated with left ventricular ejection will follow different pathways, and can be expected to arrive at different times for paired organs or members of the upper body.
Measurements performed by the inventor have shown this delay can range from less than one millisecond to several milliseconds, depending on the subject and circumstances. In addition, the inventor has found that this delay can be altered by several methods disclosed herein. This propagation delay, its alterations, and other factors make possible several determinations heretofore not possible by noninvasive means.
Blood pressure is the pressure exerted by the blood within a vessel upon the wall of the vessel. It is measured in units of force per unit area. Central venous pressure is the pressure within the large veins in the chest and the right atrium, which is the common emptying point for the venous system. Cardiac output is the amount of blood pumped by the heart, expressed in units of volume per time.
Central venous pressure (CVP) is defined as the distending pressure present in the veins in the chest (proximate to the heart), and is considered equal to the pressure in the right atrium (which is the emptying point for the venous system). Pressure should be the same throughout the venous system, but there are valves to ensure that the blood does flow back toward the heart (for example, when standing the venous blood must flow uphill, and there is no pump as on the arterial side).
As discussed in detail below, the present invention generally includes a radiation emitter having at least one wavelength being applied through a patient's tissue to the patient's blood; a radiation detector which detects reception of the at least one wavelength after absorbance through the blood, a movement generator for inducing position changes in the tissue; and a controller for computing the various characteristics based on the absorbance of the at least one wavelength of radiation at various position levels. In a preferred embodiment, the radiation emitter and detector are inserted in a probe which can be placed about the tissue/blood to be measured. A number of suitable configurations for probes are shown in FIGS. 2-8.
For example, FIG. 2 shows a representative probe 10 with a single emitterdetector pair 12. The emitter and detector are placed such that transmittance through a body member, such as a finger 13, is measured. Generally, any part of the body that can be successfully transilluminated with the radiant energy used can be utilized. Thus, toes, ears, etc. could also be used. In addition, pulse oximetry can be accomplished with this and all of the following embodiments. FIG. 3 shows a representative probe 14 with a single emitter-detector pair 16 placed such that reflectance of a body member, such as a finger, is measured. Further, FIG. 4 shows a probe 18 with two emitter-detector pairs 20 and 22 spaced a known distance apart. This can be used to measure the velocity of the pulse wave within the probe itself.
In certain embodiments of the invention, the probe comprises one or more electrocardiogram (ECG) electrodes in conjunction with the emitter-detector pairs. For example, FIG. 5 shows a probe 24 with a single emitter-detector pair 26 and a single electrocardiogram (ECG) electrode 28. Similarly, FIG. 6 shows a probe 30 with a single emitter-detector pair 32 and two ECG electrodes 34 and 36 and FIG. 7 shows a probe 38 with a two emitter-detector pairs 40 and 42 and two ECG electrodes 44 and 46. Such probes, if placed on opposite extremities of a patient, can be used to measure central and peripheral pulse wave velocity as well as ECG. Other configurations, such as double emitter-detector pairs and single ECG electrode, can be envisioned.
In yet other embodiments of the invention, the probe further comprises a position sensing or measuring device together with the emitter-detector pairs and/or ECG electrodes. FIG. 8 shows a probe 48 similar to that shown in FIG. 7 with the addition of a position sensor 50. This position sensor could be used in conjunction with a position sensor placed at heart level in order to determine the hydrostatic pressure difference between the two position sensors.
As discussed in detail herein, the invention employs hydrostatic pressure to enable precise self-callibration of the devices in a completely non-invasive manner. Hydrostatic pressure affects all liquids. Gravity or other acceleration will affect both the arterial and venous sides of the circulation. It affects all aspects of the blood pressure equally—mean, systolic, diastolic. For example, an increase in height which causes a change of 10 torr will change every pressure measurement during the cardiac cycle by this amount.
For example, if the "true" blood pressure (taken level with the heart) is 120/80, when the arm is raised an amount needed to decrease the measured pressure by 10 torr, the measured pressure in the arm will be 110/70. The pulse pressure will be the same, but the transmural pressure will be 10 torr lower at all times. In addition, the vessel will be smaller at all points.
The heart is taken to be the center of the circulatory system, and all values are in reference to it. This is not necessary for the practice of the invention, but serves as reference points for values in the current medical literature.
The electromagnetic radiation in this description will refer to light in the visible and infrared range although, as noted in the attached claims, it is conceivable that other forms could be used.
Similarly, while the present invention primarily describes the use of transillumination, it will be appreciated that reflectance spectrophotometry may alternatively be employed.
It is well known that Incident radiation passing through a body part is attenuated (absorbed) in the tissue. The theoretical basis for spectrophotometic techniques is Beer's law (the Beer-Lambert-Bouguer law) which expresses the incident intensity in terms of transmitted intensity and extinction coefficients of the tissue compartments through which the radiation has passed. The equation can be written as:
Io=the incident intensity of the source radiation;
I=the transmitted intensity of the source through the sample;
E=the extinction coefficient of the component of interest;
C=the concentration of the component in the tissue itself;
L=the optical path length (distance) through the absorber; and
Beer's law and the practice of spectrophotometry and oximetry have been exhaustively reviewed in the literature. Generally, pulse oximetry in effect filters out signals other that pulsating (AC). In the body, it can be assumed that the pulsatile component of the signal is arterial blood, while all other tissue absorbers should be non-pulsatile (DC).
An additional feature of this invention, not found in any previous disclosure, is the use of hydrostatic pressure changes to vary the amount of venous blood within a body member such as a finger. Thus, hydrostatic changes can be used in a similar manner to the pulse to perform measurements on both arterial and venous blood. If a finger is contained within a probe, raising the probe will lower the hydrostatic pressure of all vessels in the finger, both arterial and venous. Both arteries and veins (and arterioles and venules) will be smaller due to lower pressure distending their walls. Most change will occur on the venous side of the circulation due to lower pressure. Total absorbance of the finger will decrease. As the arterial oxygen saturation can be measured by pulse oximetry, the venous oxygen saturation can be calculated in a similar manner.
A light signal of a known intensity and wavelength is produced by means of light-emitting diodes (LEDs) as in currently used oximeters or, as in one possible embodiment, a broad-band light source whereby wavelengths are isolated by a rotating filter or diffusion grating. In the latter case, the emitted light is distilled through a filter which allows a known wavelength and intensity of light to penetrate. Use of tunable lasers or other equipment is also possible. If the light source is proximate to the point of use, no further mode of transmission will be needed. If it is not, the light will be transported to the desired point by means such as a fiber optic cable, preserving the wavelength and intensity.
Several means of motion induction are possible. Various means of position measurement are also possible. For example, a liquid filled tube with an end open to the atmosphere can be employed. Other position sensors are known to those having skill in the art, and include electromagnetic, spectroscopic, and chemical means. A broad-band photo detector (in the case of visible or infrared light) or other means will be utilized to measure the quantity of transmitted light.
To generate a single data point, the movement induction means is used to bring the finger (or other space of interest) to a known position relative to the heart. Light of known wavelength and intensity is emitted (and transmitted if necessary) on the surface of interest. Detection of the light signal at a distinct point (normally opposing surface) is made and the relative absorbance and extinction of the signal is calculated. Signal processing is used to determine the pulsatile portion of the signal. The arrival time of the pulse is recorded, as is the amplitude and waveform. This measurement may be repeated one or more times to ensure the accuracy of the measurement; this can be done within a very short time frame (less than a millisecond).
To generate multiple data points, the process outlined in the previous step will be repeated at the next chosen wavelength, while still at the same predetermined position. The range and number of wavelengths can be selected, and changed for different applications.
Once the desired number of wavelengths has been examined, the movement induction means would bring the finger or other volume to a predetermined second position, and the data collection of steps would be repeated. At the completion of measurements and determinations for this second position, the movement induction means will bring the space to a third predetermined position, and the measurements and determinations repeated. This process would be continued until the desired range of positions has been scrutinized.
In order to make computations of pulse propagation delay, identical measurements would be made simultaneously with a probe on the same member on the opposite side of the body. For example, if one probe were placed on the index finger of the right hand, the other probe would be placed on the index finger of the left hand.
Because the arterial path to the arm is essentially identical after the second part of the subclavian artery, any differences in pulse wave velocity and pulse wave propagation time must occur prior to this point; that is, very close to the root of the aorta. In any case, pulse wave velocity increases rapidly as the pulse wave propagates down the aorta and into the periphery (Fung). Thus, any timing differences in the periphery will be greatly reduced by the high wave velocity, leaving central effects as the most prominent.
The apparatus of the invention can be operated intermittently or continuously. In the intermittent mode, a single set of calculations can be used for analysis to produce the determinations claimed. However, the device can also be easily operated in continuous mode, with the process outlined above repeated as often as wished (constantly if desired). In addition, a rapid ("stat") mode can be offered with the minimum number of measurements made that will provide an accurate estimation of correct values. Such a rapid mode would be useful in emergency situations.
While this methodology should give precise values, further adjustment may be desired to compensate for any discrepancies between theoretical and in vivo measurements. Contemporary oximeters in fact use a calibration curve when determining oxygen saturation, with the curve being generated with data from normal volunteers.
The following algorithms are further examples of the use of the present invention. Some variables have degrees of co-dependence. In these cases, values are calculated by iterative computational techniques.
Generally, measurement of pulse wave amplitude and timing is made using probes such as that shown in FIG. 2, using methods similar to standard oximetry described in the prior art. As shown in FIG. 9, a first probe 52 is placed on a finger and set at a known position relative to the heart. Another, simultaneous measurement of pulse wave amplitude and timing is made by a second probe 54 placed on a finger on the hand opposite that of the first probe. The pulse delay occurring between the two measurements is made. Alternatively, as shown in FIG. 10, probes 52 and 54 can be placed on opposite temples of the patient to measure pulse wave values and delay. The probes can also be placed on the patient's ears.
From this information alone, an estimate of pulse wave velocity at the aortic root could be made, by utilizing a table of normal values for the distance of the central anatomical difference.
If a measurement of blood pressure is then made, one can perform the following calculation:
c=pulse wave velocity;
u=flow wave velocity; and
ρ=the density of the blood (approximately 1.055 grams/cm3).
According to the invention, p and c have been measured, ρ is known. This allows one to solve for u, which is the flow wave velocity at the aortic root. This by itself is a measure of cardiac output. If one makes an estimate of aortic root diameter, one can then compute cardiac stroke volume.
Techniques described by O'Rourke and others describe reconstruction techniques that can be used to convert or "transform" peripheral blood pressures and waveforms to the corresponding pressure and waveform at the aortic root. Ideally, the blood pressure at the aortic root should be used as the pressure term in Fung's equation.
One can improve on the above determination in several ways. The first way is by additionally measuring the peripheral pulse wave velocity. To do this, measurement of pulse wave amplitude and timing is made by a first probe such as that shown in FIG. 5. The probe is at a set known position relative to the heart. Another, simultaneous measurement of pulse wave amplitude and timing is made by a second probe placed on a finger on the hand opposite that of the first probe. The pulse delay occurring between the two measurements is made. The respective peripheral pulse wave velocities are also computed. If the peripheral pulse wave velocities are different, it can be assumed that this is because of the different central anatomies from which the respective pulses traveled. This information alone may be enough to compute central pulse wave velocity from a table of normals. However, when combined with the pulse wave delay information, this data enables one to construct a function of pulse wave speed from the periphery back to the aortic root, thus giving another measure of central pulse wave velocity.
Another method of the invention is to vary the position of the probes relative to the heart. If the first probe is at heart level and the second probe is raised above (with respect to the earth) heart level, the hydrostatic pressure of the blood vessels within the second probe will be lower than those within the first probe. In turn, in accordance with Fung's equation stated above, this means that the pulse wave velocity of the arterial vessels within the second probe will be lower than that in the arterial vessels within the first probe. This will change both the measured pulse delay between the two probes, and the measured peripheral pulse wave velocities. This creates additional measurements by which to compute central pulse wave velocity.
According to the invention, changes in hydrostatic pressure are controlled by the following equation:
ρ=blood density;
g=gravitational acceleration; and
h=height above a reference point (with respect to the earth).
The difference in hydrostatic pressure between the vessels in two probes is thus governed completely by their difference in heights relative to the heart (referenced to the surface of the earth). Therefore, a known change in position produces a known change in hydrostatic pressure.
According to the invention, the above measurements can be employed to derive a number of physiological properties. Preferably, the probes of the invention are connected to a controller to aid the data collection and analysis used to make the desired determination. The controller includes a computing device or standard personal computer (PC) with a monitor. Included within the controller are algorithms for the calculation of variables not measured directly.
For example, FIG. 11 shows a circuit schematic for a one or two wavelength photo-plethysmograph. Emitters 56 and 58 and detector 60 are positioned adjacent the tissue being measured, such as a finger 61. Emitters 56 and 58 are driven by drive circuitry 62, which is in turn governed by control signal circuitry 64. Detector 60 is connected to amplifier 66. The signal from amplifier 66 is sent to demodulator 68, which is also synched to control signal circuitry 62. The signal from the demodulator 68 is sent to analog-digital converter 70. The desired computations are performed on the output from the converter 70 by signal processor 72 and the results sent to display 74. Emitters 56 and 58 operate specific wavelengths, such as 805 nm, and may comprise light emitting diodes (LEDs) or laser diodes. Detector 60 preferably comprises a silicon photodiode. Such emitter-detector pairs are shown in FIGS. 2 and 3.
FIG. 12 shows a schematic of an alternate embodiment of suitable circuitry. As with FIG. 10, emitters 76 and 78 are connected via LED drive circuitry 79 and control signal circuitry 80 to demodulator 82. Signal from detector 84 is amplified at circuit block 86 and sent to demodulator 82. Output from demodulator 82 is sent to A/D converter 88. In addition, ECG leads 90 are connected to differential amplifier 92 and the signal is sent to converter 88. Output from converter 88 is processed at block 94 and the results sent to display 96. A probe such as those shown in FIGS. 5 and 6 may be used with the circuitry. The ECG leads are preferably silver/silver chloride or stainless steel.
Yet another embodiment of the invention is shown in FIG. 13. Emitters 98 and 100 are connected via LED drive circuitry 101 and control signal circuitry 102 to demodulator 104. Signal from detector 106 is amplified at circuit block 108 and sent to demodulator 104. Output from demodulator 104 is sent to A/D converter 109. ECG leads 110 are connected to differential amplifier 112 and the signal is sent to converter 109. Digit level sensor 114 and heart level sensor 116 are connected to amplifier 118 and the signal is sent to converter 109. Output from converter 109 is processed at block 120 and the results sent to display 122.
FIG. 14 shows a circuit schematic suitable for use with a probe having two physically independent channels, such as the one shown in FIG. 4. A first emitterdetector pair comprising emitters 124 and 126 and detector 128 are positioned adjacent the tissue being measured, such as a finger. A second pair comprising emitters 132 and 134 and detector 136 are positioned a selected distance from the first pair. Emitters 124, 126, 132 and 134 are driven by drive circuitry 138, which is in turn governed by control signal circuitry 140. Signal from detector 128 is amplified by block 142 and sent to demodulator 144. Independently, signal from detector 136 is amplified and demodulated at blocks 146 and 148, respectively. Output from demodulators 144 and 148 is sent to analog-digital converter 150. The desired computations are performed on the output from the converter 150 by signal processor 152 and the results sent to display 154.
An alternative embodiment configured for use with a probe having two physically independent channels and an ECG lead, such as the one shown in FIG. 7, is schematically shown in FIG. 15. A first emitter-detector pair comprising emitters 156 and 158 and detector 160 are positioned adjacent the tissue being measured, such as a finger. A second pair comprising emitters 164 and 166 and detector 168 are positioned a selected distance-from the first pair. Emitters 156, 158, 164 and 166 are driven by drive circuitry 170 which is in turn governed by control signal circuitry 172. Signal from detector 160 is amplified by block 174 and sent to demodulator 176. Independently, signal from detector 168 is amplified and demodulated at blocks 178 and 180, respectively. Output from demodulators 176 and 180 is sent to analog-digital converter 182. ECG leads 184 are connected to differential amplifier 186 and the signal is also sent to converter 182 The desired computations are performed on the output from the converter 182 by signal processor 188 and the results sent to display 190.
As one of ordinary skill in the art will appreciate, the placement of the various probes discussed above will effect the types of measurements that can be taken. As discussed above, FIGS. 9 and 10 show probes placed on opposite extremities to enable measurement of pulse wave delay. FIG. 16 shows an embodiment of the invention with probe 52, such as in FIG. 1, placed on the digit, and a probe 54, such as in FIG. 2, placed on the arm near the brachial artery. This could measure the pulse wave velocity in the arm (as well as pulse oximetry). A similar embodiment could measure pulse wave velocity in the leg. FIG. 17 shows probes 52 and 54 placed on a finger and on a toe to measure the pulse wave delay. FIG. 18 shows probes 52 and 54 placed on opposite digits and probe 55 placed on a toe. This allows measurement of the differential pulse wave delay between the fingers and toe, and allows calibration of the toe probe to be used in place of a finger probe (if only one finger probe could be used, such as in hand surgery). The use of appropriate probes also allows a diagnostic-quality ECG. FIGS. 19 and 20 show probes 52 and 54 placed on opposite digits. One arm of the subject is placed at the level of the heart, while one arm is moved to different positions, both above and below the level of the heart. By generating different hydrostatic pressures in the vessels, the pulse velocity and hence pulse wave delay changes. In addition, the amplitude of the pulse wave, and amplitude of venous absorbance changes. This allows the additional computations of arterial blood pressure and venous pressure. FIG. 21 shows probes 52 and 54 placed on opposite digits and probe 55 placed on a toe. The differential hydrostatic pressures in the vessels allow measurements of pulse wave velocity and pulse wave delay, as well as arterial blood pressure and venous pressure. Use of probes with suitable ECG leads will also allow the invention to perform a diagnostic-quality ECG. In addition, heart rate and respiratory rate can be calculated, and cardiac output and several other cardiovascular characteristics computed.
As discussed above, the controllers of the invention preferably output the results of the measurements and computations to a display. FIG. 22 shows an oscilloscope screen. The two tracings are from pulse oximeter probes, such as those shown in FIG. 1, placed on the index fingers of both hands. The pulse wave delay is visible as the slight phase difference between the two tracings. As the probes are at the same level, the pulse amplitudes are essentially identical. FIG. 23 shows the oscilloscope screen after the hand with the probe displayed as the top tracing has been placed at a level higher than the heart and the hand with the probe displayed as the bottom tracing has been placed at a level lower than the heart. The induction of a pressure differential between the two probes effects a change in the pulse delay. The change in pressure also correspondingly alters the pulse amplitudes. FIG. 24 shows the oscilloscope screen after the hand with the probe displayed as the top tracing has been placed at a level lower than the heart and the hand with the probe displayed as the bottom tracing has been placed at a level higher than the heart. Here, the pulse delay has substantially reversed as have the pulse amplitudes. FIG. 25 shows an oscilloscope screen displaying an electrocardiogram in conjunction with a pulse waveform.
The algorithms outlined below serve as examples, but modifications are possible to arrive at the indicated results, and are meant to be included within the spirit of this application. Various additional components of the device will be discussed in more detail below with reference to the following examplary determinations.
(d1). Determination of Arterial Blood Pressure
A probe such as that shown in FIG. 1 is placed on an extremity, and that extremity is moved in relation to the heart. As mentioned above, the hydrostatic pressure within the arteries and arterioles changes as a function of height with respect to the heart. Because of this, both the pulse wave velocity and pulse wave amplitude change as a function of probe height. These two parameters can be mapped against known distance above or below the heart. In this way, function curves of pressure vs. pulse wave amplitude and pressure vs. pulse wave velocity can be drawn. For example, a full excursion of the arm in a standing adult produces hydrostatic changes of greater than 50 cm of water in both directions. Using an arm and a leg, a gradient of well over 200 cm of water can be generated. This is a significant portion of the normal blood pressure range, and certainly enough to produce the function curves mentioned above.
There is a huge amount of medical literature describing arterial behavior, so the curves can be extrapolated if necessary. These curves serve as calibration.
It can thus be determined if "recalibration" is necessary—if either pulse amplitude or pulse wave velocity changes, and the other parameter does not change correspondingly. In other words, a shift on one curve should matched by a corresponding shift on the other curve. If this shift does not occur as predicted, recalibration is required. Of course, the process of recalibration is the simple procedure outlined above.
In a preferred embodiment, a first probe having a position sensor is placed level with the patient's heart. A second probe, such as one shown in FIG. 8, having a position sensor and a pulse detector is placed on the patient's finger. The patient's arm is held out level with the heart so there is zero displacement between probes. Pulse amplitude is recorded from probe. The patient's arm is slowly raised, while pulse amplitude and relative displacement of probe are recorded. The hydrostatic pressure difference between probes is also computed. By comparing the recorded pulse amplitude to the hydrostatic pressure difference, a mathematical function relating pressure to pulse amplitude can be derived. Preferably, circuitry similar to that shown in FIG. 13 is used to aid the process. This process is repeated while lowering the arm back to heart level, then lowering the arm to below heart level and, finally, raising the arm back to heart level. Similar steps can be applied to measure pulse delay, pulse velocity and pulse contour.
(d2). Determination of Cardiac Output
Cardiac output can be determined by measuring delays in pulse arrival times in coupled organs or members on opposite sides of the body. In a preferred embodiment of the invention, probes such as those shown in FIG. 1, having sensors for detecting a patient's pulse are placed on opposite fingers of the patient. The patient positions both arms straight out from the side. The blood pressure of the patient can be determined either through conventional means or by the methods of the invention. The pulse delay between the two probes can be measured utilizing circuitry such as that shown in FIGS. 14 or 15, for example. The dicrotic notch of the pulse may be determined by standard methods, and used to calculate the ejection time based on the timing. The size of the aortic root can be estimated by standard means and the consequently the pulse distance differential at the aortic root. This allows the calculation of the pulse velocity c at the aortic route by the following equation:
The value of c can then be used to solve for flow wave velocity based on the following equation:
c=pulse wave velocity;
u=flow wave velocity; and
ρ=density of the blood (approximately 1.055 grams/cm3).
According to the invention, cardiac stroke volume can be determined by multiplying the aortic root area by the flow wave velocity and by the cardiac ejection time. Cardiac minute output can be calculated by multiplying the cardiac stroke volume by the pulse rate. These steps may be augmented by raising and lowering the patient's arms with respect to each other to vary the pressure and the pulse wave velocity.
Alternatively, cardiac output can be determined by placing probes such as those shown in FIG. 5 on a patient's finger and toe. The probes measure oxygen saturation at each pulse. The oxygen saturation for each pulse at the first probe is compared to the oxygen saturation of that pulse and subsequent pulses at the second probe. With continuous monitoring, this allows the determination matching oxygen saturation, within given tolerance limits, of the pulses from the probes. The patient's blood volume and the physical separation of the probes can be determined by standard methods. This allows the computation of caridac stroke volume by dividing the blood volume displaced by the number of pulses. Then, the cardiac minute output can be calculated by multiplying the cardiac stroke volume by pulse rate. Circuitry such as that shown in FIGS. 11 or 12 is suitable for use with this embodiment.
(d3). Determination of Venous Saturation and Pressure
Determination of arterial oxygen saturation can be determined by pulse oximetry and techniques well delineated in both the patent and medical literature. Hydrostatic changes as described in this application allow the determination of venous saturation and pressure as well.
Place a probe such as that shown in FIG. 1 on a finger. Make measurements of both total absorbance and pulsatile absorbance. Raise the probe a known distance. Again measure both total absorbance and pulsatile absorbance. Both will be decreased. This is because the pulse amplitude is less because the arterial blood pressure within the probe is less (due to decrease in hydrostatic pressure). However, the total absorbance will also decrease, as the distending pressure in the venous system is less, and hence the veins and venules are smaller. All changes in absorbance can be assumed to be due to changes in blood volume. Saturation is calculated using the ratios of absorbance of distinct wavelengths.
In one embodiment, the central venous pressure (CVP) can be estimated. A probe containing a position sensor is place level with a patient's heart. A second probe, such as the one shown in FIG. 8, also comprising a position sensor is placed on the patient's finger. The patient positions the arm so that the second probe is initially lower than the first probe. The total absorbance measured at the second probe is continuously monitored. The patient's arm is slowly raised, and the rate of change of absorbance of the second probe is computed with respect to the relative displacement to the first probe. When the rate of change changes by a predetermined amount representing an abrupt decrease, the arm position corresponding to the point of central venous drainage has been reached. The CVP can then be calculated by computing the hydrostatic pressure difference between the first probe and the second probe at that arm position. The circuitry shown in FIG. 13 is suitable for use with this embodiment.
(d4). Determination of Heart Rate
According to the invention, heart rate can be determined by counting the pulsatile arterial signal for a known length of time, or by the ECG impulse.
(d5). Determination of Respiratory Rate
The impedance changes of the chest due to filling and emptying can be measured from the electrocardiogram tracing. During normal breathing, negative pressure is created within the chest by lowering of the diaphragm and expansion of the rib cage. This negative pressure causes blood to empty more rapidly from the peripheral into the central veins. This is also the case when respiration is assisted by a negative-pressure device such as the "iron lung".
During modem mechanically-assisted ventilation (with "ventilators"), positive pressure is created within the chest by forcing air into the lungs. For both positive- and negative-pressure ventilation, expiration is passive. This respiratory variation by itself can be used as an estimate of cardiac filling, giving left heart pressures. This determination can be assisted by the use of the hydrostatic techniques described above.
(d6). Diagnosis of Congenital Heart Disease and Anatomic Anomalies
Diagnosis of many disorders with anatomic anomalies can be made by the detection of unexpected propagation times, and abnormal propagation delays between right- and left-sided organs.
The ability to measure both arterial and venous saturation, as well as arterial and venous pressures, can aid further in investigations.
(d7). Diagnosis of Dysrhthmias
By measuring blood pressure and the electrocardiogram simultaneously, the diagnosis of dysrhythmias can be aided greatly. Both arterial and venous pressure are recorded with the ECG, allowing differentiation of atrial vs. ventricular arrhythmias.
(d8). Determination of Additional Cardiovascular Characteristics
By measuring blood pressure and the electrocardiogram simultaneously, many additional characteristics, such as systolic and diastolic pressure time indices, can be determined.
An enormous amount of information can be gleaned from the use of probes on opposite sides of the body combined with hydrostatic perturbations. It is important to realize that the time of arrival of a pulse to paired members is different, but the velocity of the pulse is also different. Examination of pulse propagation time, pulse propagation phase or delay, pulse velocity, and pulse amplitude yields four parameters that may change in different ways for each perturbation. Particularly, raising and lowering an arm by the same amount may give different changes. Raising and lowering the other arm by the same amount may give still different changes. Further, raising an arm by a given amount, then raising again by the same amount, may give different changes. Raising the other arm by the given amount, then raising again by the same amount, may give still different changes. Similar effects can be obtained by lowering the extremity.
Without departing from the spirit and scope of this invention, one of ordinary skill can make various changes and modifications to the invention to adapt it to various usages and conditions. As such, these changes and modifications are properly, equitably, and intended to be, within the full range of equivalence of the following claims.
|Cited Patent
|Filing date
|Publication date
|Aug 20, 1986
|Apr 5, 1988
|Omron Tateisi Electronics Co.
|Device and method for determining systolic blood pressure
|Jul 16, 1987
|Apr 18, 1989
|Nihon Seimitsu Sokki Co., Ltd.
|May 31, 1988
|Oct 10, 1989
|Omron Tateisi Electronics Co.
|Electronic blood pressure meter having improved cuff repressurization means
|Aug 30, 1989
|Sep 17, 1991
|Nicolay Gmbh
|Method and apparatus for a non-invasive examination of blood circulation in a living organism
|Mar 16, 1987
|May 12, 1992
|Nasiff Roger E
|Indirect continuous blood pressure method
|Mar 1, 1990
|Oct 6, 1992
|Hewlett-Packard Company
|Dual-finger vital signs monitor
|Jul 16, 1991
|May 10, 1994
|Avl Medical Instruments Ag
|Blood pressure measuring device and method
|Apr 3, 1991
|May 31, 1994
|Casio Computer Co., Ltd.
|Measurement of electrocardiographic wave and sphygmus
|Sep 23, 1993
|Jan 31, 1995
|Samsung Electronics Co., Ltd.
|Maximum pulse wave amplitude calculating system and operation method for an electronic blood pressure measuring device
|May 17, 1994
|Dec 10, 1996
|Omron Corporation
|Device to measure vascular function
|Jun 6, 1995
|Jul 22, 1997
|Nihon Kohden Corporation
|Pulse-wave propagation time basis blood pressure monitor
|Dec 12, 1996
|Jun 16, 1998
|Selmonosky; Carlos A.
|Vascular testing method
|Dec 5, 1996
|Sep 22, 1998
|Essential Medical Devices
|Non-invasive carboxyhemoglobin analyer
|Feb 28, 1997
|Feb 8, 2000
|Seiko Epson Corporation
|Blood pulse wave detecting apparatus and motion intensity measuring apparatus
|Nov 19, 1998
|Jan 1, 2002
|Seiko Epson Corporation
|Method of detecting pulse wave, method of detecting artery position, and pulse wave detecting apparatus
|Feb 24, 1992
|Dec 19, 1995
|Nellcor Incorporated
|Method and apparatus for detecting optical pulses
|Citing Patent
|Filing date
|Publication date
|Aug 22, 2002
|Nov 4, 2003
|Bci, Inc.
|Combination fingerprint and oximetry device
|Nov 6, 2002
|Apr 26, 2005
|Nec Corporation
|Fingerprint input devices and electronic devices provided with the same
|Aug 9, 2001
|May 3, 2005
|Thomas Dietiker
|System and method for a self-calibrating non-invasive sensor
|Feb 6, 2004
|Jul 26, 2005
|Woolsthorpe Technologies, Llc
|Device and method for noninvasive continuous determination of physiologic characteristics
|Jul 30, 2003
|Sep 27, 2005
|Yu-Yu Chen
|Watch-typed heartbeat sensing device
|Dec 8, 2004
|Oct 17, 2006
|Elekon Industries Usa, Inc.
|System and method for a self-calibrating non-invasive sensor
|Apr 5, 2004
|Dec 12, 2006
|Lumidigm, Inc.
|Multispectral biometric sensor
|Apr 24, 2001
|Feb 20, 2007
|Arkray, Inc.
|Pulsimeter and measuring system comprising the same
|Nov 8, 2005
|Feb 27, 2007
|Woolsthorpe Technologies, Llc
|Pulse amplitude indexing method and apparatus
|Jul 8, 2004
|Dec 25, 2007
|Orsense Ltd.
|Device and method for non-invasive optical measurements
|Jun 7, 2005
|Feb 12, 2008
|Chem Image Corporation
|Non-invasive biochemical analysis
|Dec 27, 2006
|Oct 20, 2009
|General Electric Company
|Patient monitoring device with multiple sensors
|Oct 21, 2004
|Oct 27, 2009
|Martis Ip Holdings, Llc
|Test authorization system
|Mar 9, 2006
|Nov 10, 2009
|Aframe Digital, Inc.
|Mobile wireless customizable health and condition monitor
|Mar 1, 2006
|Jan 12, 2010
|Masimo Laboratories, Inc.
|Multiple wavelength sensor equalization
|Sep 1, 2005
|Feb 23, 2010
|Lumidigm, Inc.
|Comparative texture analysis of tissue for biometric spoof detection
|Aug 17, 2007
|Mar 9, 2010
|Massachusetts Institute Of Technology
|Wearable pulse wave velocity blood pressure sensor and methods of calibration thereof
|Oct 6, 2004
|Mar 23, 2010
|Martis Ip Holdings, Llc
|Patient identification system
|Mar 1, 2006
|Jun 1, 2010
|Masimo Laboratories, Inc.
|Configurable physiological measurement system
|May 17, 2006
|Jun 15, 2010
|Lumidigm, Inc.
|Biometric sensor
|Jul 19, 2006
|Jul 6, 2010
|Lumidigm, Inc.
|White-light spectral biometric sensors
|Mar 1, 2006
|Jul 20, 2010
|Masimo Laboratories, Inc.
|Multiple wavelength sensor substrate
|Mar 1, 2006
|Jul 27, 2010
|Masimo Laboratories, Inc.
|Multiple wavelength sensor emitters
|Apr 24, 2006
|Sep 21, 2010
|Lumidigm, Inc.
|Multispectral biometric sensors
|Jul 31, 2006
|Sep 21, 2010
|Lumidigm, Inc.
|Biometrics with spatiospectral spoof detection
|Jul 31, 2006
|Sep 28, 2010
|Lumidigm, Inc.
|Spatial-spectral fingerprint spoof detection
|May 18, 2006
|Oct 26, 2010
|Lumidigm, Inc.
|Multispectral biometric sensor
|Oct 3, 2008
|Nov 9, 2010
|Lumidigm, Inc.
|Multispectral imaging biometrics
|Oct 29, 2008
|Nov 16, 2010
|Lumidigm, Inc.
|Multispectral imaging biometrics
|Jul 19, 2007
|Mar 1, 2011
|Lumidign, Inc.
|Multibiometric multispectral imager
|Mar 1, 2006
|Jun 7, 2011
|Masimo Laboratories, Inc.
|Physiological parameter confidence measure
|Jun 10, 2008
|Aug 9, 2011
|Lumidigm, Inc.
|Contactless multispectral biometric capture
|Mar 1, 2006
|Nov 1, 2011
|Masimo Laboratories, Inc.
|Multiple wavelength sensor drivers
|Mar 1, 2006
|Mar 6, 2012
|Masimo Laboratories, Inc.
|Noninvasive multi-parameter patient monitor
|Sep 10, 2009
|Apr 24, 2012
|Lumidigm, Inc.
|Two camera biometric imaging
|Jun 10, 2008
|May 8, 2012
|Lumidigm, Inc.
|Whole-hand multispectral biometric imaging
|Jun 14, 2010
|May 22, 2012
|Lumidigm, Inc.
|White-light spectral biometric sensors
|Mar 1, 2006
|May 29, 2012
|Masimo Laboratories, Inc.
|Noninvasive multi-parameter patient monitor
|Mar 1, 2006
|Jul 17, 2012
|Masimo Laboratories, Inc.
|Noninvasive multi-parameter patient monitor
|Apr 9, 2008
|Jul 24, 2012
|Lumidigm, Inc.
|Hygienic biometric sensors
|Jul 19, 2010
|Aug 28, 2012
|Cercacor Laboratories, Inc.
|Multiple wavelength sensor substrate
|Jun 15, 2007
|Sep 18, 2012
|Medtor Llc
|System and method for a non-invasive medical sensor
|Mar 19, 2008
|Oct 9, 2012
|Lumidigm, Inc.
|Biometrics based on locally consistent features
|Oct 30, 2012
|Cercacor Laboratories, Inc.
|Multiple wavelength sensor emitters
|Sep 3, 2008
|Nov 6, 2012
|Seiko Epson Corporation
|Pulse meter, method for controlling pulse meter, wristwatch-type information device, control program, storage medium, blood vessel simulation sensor, and living organism information measurement device
|Mar 20, 2009
|Nov 20, 2012
|Massachusetts Institute Of Technology
|Calibration of pulse transit time measurements to arterial blood pressure using external arterial pressure applied along the pulse transit path
|Apr 10, 2008
|Jan 15, 2013
|Lumidigm, Inc.
|Biometric detection using spatial, temporal, and/or spectral techniques
|Sep 28, 2009
|Feb 19, 2013
|Aframe Digital, Inc.
|Mobile wireless customizable health and condition monitor
|Apr 13, 2009
|Feb 26, 2013
|Cercacor Laboratories, Inc.
|Multiple wavelength sensor emitters
|Oct 31, 2011
|Jul 9, 2013
|Cercacor Laboratories, Inc.
|Multiple wavelength sensor drivers
|Feb 11, 2008
|Sep 10, 2013
|ChemImage Technologies, LLL
|Invasive chemometry
|Dec 13, 2007
|Oct 8, 2013
|Robert A. BARUCH
|Monitoring respiratory variation of pulse pressure
|Mar 14, 2011
|Oct 29, 2013
|Lumidigm, Inc.
|Biometric imaging using an optical adaptive interface
|Mar 5, 2012
|Nov 12, 2013
|Carcacor Laboratories, Inc.
|Noninvasive multi-parameter patient monitor
|Aug 25, 2010
|Dec 31, 2013
|Aframe Digital, Inc.
|Mobile wireless customizable health and condition monitor
|May 18, 2010
|Jan 21, 2014
|Cercacor Laboratories, Inc.
|Configurable physiological measurement system
|Jun 3, 2011
|May 6, 2014
|Cercacor Laboratories, Inc.
|Physiological parameter confidence measure
|Aug 26, 2010
|May 20, 2014
|Lumidigm, Inc.
|Multiplexed biometric imaging
|Sep 26, 2006
|Jun 17, 2014
|University Of Florida Research Foundation, Inc.
|Optimized gas supply using photoplethysmography
|Feb 24, 2011
|Jul 15, 2014
|Lumidigm, Inc.
|Contactless multispectral biometric capture
|Mar 26, 2008
|Jul 15, 2014
|Cercacor Laboratories, Inc.
|Multiple wavelength optical sensor
|Jan 5, 2011
|Jul 22, 2014
|Lumidigm, Inc.
|Multispectral barcode imaging
|Dec 3, 2010
|Aug 12, 2014
|Masimo Corporation
|Calibration for multi-stage physiological monitors
|May 17, 2011
|Sep 9, 2014
|Lumidigm, Inc.
|Contactless multispectral biometric capture
|Feb 25, 2013
|Sep 30, 2014
|Cercacor Laboratories, Inc.
|Multiple wavelength sensor emitters
|Aug 26, 2010
|Oct 28, 2014
|Lumidigm, Inc
|Dual-imager biometric sensor
|Nov 11, 2013
|Dec 16, 2014
|Cercacor Laboratories, Inc.
|Noninvasive multi-parameter patient monitor
|May 29, 2012
|Dec 16, 2014
|Lumidigm, Inc.
|Optical biometrics imaging with films
|Jul 8, 2013
|Jan 6, 2015
|Cercacor Laboratories, Inc.
|Multiple wavelength sensor drivers
|Jan 13, 2006
|Jan 13, 2015
|Welch Allyn, Inc.
|Vital signs monitor
|Jun 15, 2011
|Feb 3, 2015
|Fujitsu Limited
|Biometric authentication device and biometric authentication method
|Feb 11, 2013
|Feb 24, 2015
|Cercacor Laboratories, Inc.
|Tissue profile wellness monitor
|Oct 11, 2013
|Sep 15, 2015
|Cercacor Laboratories, Inc.
|Noninvasive multi-parameter patient monitor
|Feb 6, 2004
|Aug 19, 2004
|Mills Alexander K.
|Device and method for noninvasive continuous determination of physiologic characteristics
|Apr 30, 2003
|Nov 4, 2004
|Raul Garcia
|Medical needle guard
|Jul 30, 2003
|Feb 3, 2005
|Yu-Yu Chen
|Watch-typed heartbeat sensing device
|Feb 6, 2004
|Jun 16, 2005
|Mills Alexander K.
|Device and method for noninvasive continuous determination of physiologic characteristics
|Mar 26, 2004
|Sep 29, 2005
|Triage Data Networks
|Cuffless blood-pressure monitor and accompanying web services interface
|Apr 25, 2005
|Dec 8, 2005
|Lumidigm, Inc.
|Multispectral imaging biometrics
|Jul 8, 2005
|Jan 5, 2006
|Lumidigm, Inc.
|Liveness sensor
|Jul 8, 2005
|Jan 5, 2006
|Lumidigm, Inc.
|Noninvasive alcohol sensor
|Jul 8, 2004
|Jan 12, 2006
|Orsense Ltd.
|Device and method for non-invasive optical measurements
|Sep 1, 2005
|Mar 23, 2006
|Lumidigm, Inc.
|Comparative texture analysis of tissue for biometric spoof detection
|Oct 6, 2004
|Apr 6, 2006
|Martis Dinesh J
|Patient identification system
|Oct 21, 2004
|Apr 6, 2006
|Martis Dinesh J
|Test authorization system
|Dec 8, 2004
|Jun 8, 2006
|Thomas Dietiker
|System and method for a self-calibrating non-invasive sensor
|Feb 16, 2006
|Jun 29, 2006
|Triage Data Networks
|Wireless, internet-based, medical diagnostic system
|May 18, 2006
|Sep 14, 2006
|Lumidigm, Inc.
|Multispectral biometric sensor
|Mar 9, 2006
|Sep 14, 2006
|Cindy Crump
|Mobile wireless customizable health and condition monitor
|May 17, 2006
|Sep 21, 2006
|Lumidigm, Inc.
|Biometric sensor
|Mar 1, 2006
|Sep 21, 2006
|Ammar Al-Ali
|Multiple wavelength sensor substrate
|Mar 1, 2006
|Sep 21, 2006
|Ammar Al-Ali
|Multiple wavelength sensor equalization
|Mar 1, 2006
|Sep 21, 2006
|David Dalke
|Multiple wavelength sensor emitters
|Mar 1, 2006
|Sep 21, 2006
|Ammar Al-Ali
|Configurable physiological measurement system
|Mar 30, 2005
|Oct 5, 2006
|Dailycare Biomedical Inc.
|Integrated physiological signal assessing device
|Mar 1, 2006
|Oct 26, 2006
|Ammar Al-Ali
|Multiple wavelength sensor drivers
|Apr 24, 2006
|Nov 2, 2006
|Lumidigm, Inc.
|Multispectral Biometric Sensors
|Sep 26, 2006
|Feb 1, 2007
|Melker Richard J
|Optimized gas supply using photoplethysmography
|Jul 19, 2006
|Feb 8, 2007
|Lumidigm, Inc.
|White-light spectral biometric sensors
|May 26, 2006
|Nov 29, 2007
|Triage Wireless, Inc.
|System for measuring vital signs using bilateral pulse transit time
|Sep 3, 2008
|Jan 1, 2009
|Seiko Epson Corporation
|Pulse meter, method for controlling pulse meter, wristwatch-type information device, control program, storage medium, blood vessel simulation sensor, and living organism information measurement device
|Aug 4, 2009
|Feb 10, 2011
|Nellcor Puritan Bennett Llc
|Methods and apparatus for using multiple sensors to measure differential blood transport time in a patient
|Jun 16, 2006
|Jul 15, 2009
|Process of calibrating no-cuff arterial blood gauge
|Apr 5, 2004
|Jul 11, 2012
|Multispectral biometric sensor
|Aug 28, 2008
|Mar 20, 2013
|Pressure and emotion adjustment system and a physiology signal acquisition device
|Aug 14, 2003
|Mar 4, 2004
|Bci Inc
|Combination fingerprint and oximetry device
|Apr 5, 2004
|Oct 21, 2004
|Robert M Harbour
|Multispectral biometric sensor
|Jul 14, 2006
|Mar 1, 2007
|John Koblanski
|Methods and apparatus for monitoring heart impulses
|U.S. Classification
|600/481, 600/323, 600/500, 600/502, 600/501, 600/324, 600/503, 600/485
|International Classification
|Cooperative Classification
|European Classification
|Sep 13, 2004
|Aug 9, 2006
|Fee payment
Year of fee payment: 4
|Nov 1, 2010
|Maintenance fee reminder mailed
|Mar 25, 2011
|Lapse for failure to pay maintenance fees
|May 17, 2011
|Expired due to failure to pay maintenance fee
Effective date: 20110325


text:
(forgot?) or Register
Keep Your Site Happy, add Khush Captcha Anti-SpamBot Technology to your WordPress Registration and Comment Forms.
Download Version 0.1
A new, unique type of captcha. This plugin will display the Khush Captcha Anti-Bot filtering technology on your websites registration and comment forms.
Requires: 3.0.1 or higher
Compatible up to: 3.7.11
Last Updated: 2013-12-7
Active Installs:
Got something to say? Need help?
Translate Khush Captcha
0 people say it works.
0 people say it's broken.
Log in to vote.

subset: commoncrawl


text:
O winter! frozen pulse and heart of fire,
What loss is theirs who from thy kingdom turn
Dismayed, and think thy snow a sculptured urn
Of death! Far sooner in midsummer tire
The streams than under ice. June could not hire
Her roses to forego the strength they learn
In sleeping on thy breast. No fires can burn
The bridges thou dost lay where men desire
In vain to build.
O Heart, when Love's sun goes
To northward, and the sounds of singing cease,
Keep warm by inner fires, and rest in peace.
Sleep on content, as sleeps the patient rose.
Walk boldly on the white untrodden snows,
The winter is the winter's own release.
Still lie the sheltering snows, undimmed and white;
And reigns the winter's pregnant silence still;
No sign of spring, save that the catkins fill,
And willow stems grow daily red and bright.
These are the days when ancients held a rite
Of expiation for the old year's ill,
A beautiful book of poem to match the Months and Seasons of the year.
Cara Devon has always suffered curiosity and im... Read more
When the economic downturn ends Matty Cruzs co... Read more
A telekinetic teenager. A telepathic child. A p... Read more
A century after an apocalyptic war reduces all... Read more
Fall in love with the Kensington siblings in th... Read more
See it as donating a moment of your social media time, every little thing helps us improve and stay online.
The list of books below is based on the weekly downloads by our users regardless of eReader device or file format.
See more popular titles from this genre.


text:
King Attacks Obama at Breakfast
New York, NY —
On Day Two of the Republican National Convention, Long Island Congressman Peter King addressed the New York State delegation breakfast.
He says Sen. John McCain's support of key anti-terror legislation helped keep New Yorkers safe from another attack.
KING: If Barack Obama had been in there, there would be no Patriot Act. There would be no surveillance of phone calls being made into the us by foreign terrorists. And I don't know what we would be doing as far as all the relationships set up with overseas allies, and countries that are not even our allies who are working very closely in our war against terror.
REPORTER: Congressman King says New York State should involve more Iraq war veterans, to help revamp the party.


text:
Vancouver's North Shore is the place to stay to discover Vancouver from a whole different perspective. Staying mountainside you are minutes away from some of Vancouver's premier attractions. Outdoor adventures abound with world-class mountain biking trails, rainforest hiking trails and more.
This popular attraction that offers a complete British Columbia experience — a must-see when visiting Vancouver.
Outdoor Activities
The mountainside of Vancouver offers endless outdoor recreation options for all ages and fitness levels.
Need a a place to stay? We've got a variety of accommodation options, all within minutes of downtown Vancouver and popular attractions.
Culinary Month
October is culinary month. Discover the culinary experiences on the North Shore, and enter to win your own culinary adventure.
Top 10 Things to Do
If you are planning a visit to the North Shore, check our top 10 lists for ideas about things to do and see in spring, summer, fall or winter.
Special Offers
Find the latest hotel packages and special offers in North Vancouver. Find information on pet-friendly accommodation options.
Visitors' Guide
Check out the 2015 Visitors' Guide, and discover things to do in North and West Vancouver including festivals, arts and culture and more!


text:
Testing, which was performed as part of a partnership with the National Research Council's Advanced Materials Division, involved coating of several commonly utilized device materials, including nylon, polypropylene (PPE), and polyethylene terephthalate (PET) nonwoven scaffolds with hECM using several common coating methods. The hECM-coated and uncoated scaffolds were then surgically implanted in the subcutaneous space of SCID mice and histological samples of excised implants were assessed for inflammatory response, cellular infiltration, foreign body giant cells and capsule formation.
"Coating polymers with a naturally-produced, all-human ECM masks the foreign device material and offers a physiological surface which supports healthy tissue infiltration and interaction," said Dr. Michael Zimber, Director of Applied Research at Histogen. "The hECM-coated polymers promoted a two-fold increase in normal cell proliferation as compared to uncoated polymers, as well as causing a significant reduction in the host inflammatory and fibrotic response to surgically implanted polymers."
"Coating Polymers with a Human Extracellular Matrix Significantly Improves Implant Biocompatibility" will be presented by Dr. Zimber at the ASAIO Annual Conference, taking place May 27-29, 2010 in Baltimore.
About Histogen
Histogen, launched in 2007, seeks to redefine regenerative med
|Source: PRWeb
Copyright©2010 Vocus, Inc.
All rights reserved


text:
Ratings based on 599 Verified Reviews
They paid and stayed. We double-checked.
4.6 Room cleanliness
4.4 Service & staff
4.5 Room comfort
4.4 Hotel condition
They paid and stayed.
We double-checked.
We double-checked.
for Everyoneby A verified traveler from arkansas
Posted Oct 5, 2015
Rough start, had to purchase an extra room for two nights, because they over booked! I booked 3 rooms, 2 being doubles and one king . They replaced with all king rooms and I had two couples and 4 in another and king for my husband and I. Another $600, not good. The rest was good and everyone loved the hotel. Wish I had coffee in my room. Hard to get coffee free downstairs. It was free, but the bartender had to retrieve for you and he did not like doing this task. Maid, bellman, front desk, and valet were great. Love the location! Could walk everywhere and if I couldn't, Horace would take you. If you stay there ask for Horace or Landmark Limo for travel! Problems there at least the staff tried to help you .
for Business Travelersby A verified traveler from Arizona
Nice hotel, great locationPosted Sep 27, 2015
The Stoneleigh is a nice smaller hotel in a location that allows you to walk many places. The rooms are comfortable and on the larger side. The beds are very cozy. The walls are very thin and it can be very loud, especially in the morning. There is a small lobby bar which has a great bar menu and friendly bartenders. The new pool is blah. Overall the hotel is fine and fit my needs for several business trips.
for Everyoneby A verified traveler
Wonderful stay! We loved the hotel!Posted Sep 14, 2015
Pros: Valets were terrific, as was the front desk staff!
Location: Great Uptown Dallas location!
We were looking for a upscale weekend get away, and the Stoneleigh was ideal for this!
for Everyoneby A verified traveler from Louisiana
Awesome location!Posted Sep 2, 2015
Wonderful time at the hotel. Special thanks to Angelina front desk and Mark bartender for their helpful and friendly attitudes. Wish spa was onsite.
for Everyoneby A verified traveler from Columbia, SC
Great Hotel, Great LocationPosted Sep 1, 2015
Pros: Beautiful lobby, beautiful views, beautiful room, comfortable bed, great food/drinks, awesome gym
Location: There are tons of restaurants and bars within a few blocks, definitely walking distance.
We had a wonderful experience at The Stoneleigh for our wedding weekend! The hotel it beautiful, in a great location close to so many restaurants and bars. The bar in the lobby has great cocktails and the hotel restaurant was great for breakfast. We had the included breakfast package and were given vouchers worth $40 for each day. The one thing I thought was odd, and this might be my fault for not noticing/researching better, was that our Junior Suite did not have a wall separating the bed and sitting area. It was basically one giant, open room. We were very impressed by the gym - it was big, clean and had more equipment than I'm used to seeing in hotel fitness centers. And they even stock water bottles down there.
for Everyoneby A verified traveler
GetawayPosted Aug 15, 2015
Had a great time. Pool was nice. Restaurant was quiet.
for Everyoneby A verified traveler
Great hotel with one small problemPosted Aug 4, 2015
This is a nice, quaint and ecclectic hotel. I loved it. With one exception. There is a blond male bartender - late 20's - that just rubbed me the wrong way. You know how you can just be repelled by someone's energy but you don't know why? One instance, I ordered some cheesecake. 20 minutes later no cheesecake. I asked him about it and he just gave me a smirk and said something smart Alec and then laughed and said "we will look into it." I saw that the guy from the kitchen and taking the cheesecake up 10 minutes earlier but I thought it was for someone else and it had been sitting behind the bar the whole time. The bartender in the morning for breakfast was really friendly and had a much better attitude. This would be a great place to go but they need to get rid of that blonde bartender .
for Everyoneby A verified traveler from Dallas, TX
Nice Uptown LocationPosted Aug 4, 2015
This hotel takes you back in time...the Art Deco style is reminiscent of a time of dinner clubs, when folks dressed to the 9s for dinner. The only issue with the property is the inflated prices for internet and valet parking. I had a friend stay over one night and got charged for 2 nights of valet parking ($50). Watch the charges at check out for accuracy.
for Everyoneby A verified traveler
GorgeousPosted Aug 2, 2015
Lovely hotel! Would stay again although I found out a little too late that it's haunted!
for Everyoneby A verified traveler
Nice hotel in UptownPosted Jul 27, 2015
We decided to book a hotel at the last minute to celebrate our anniversary. When they found out it was our anniversary, they upgraded our room to a suite and sent us champagne and chocolate covered strawberries. Everyone was very nice and did their best to accommodate us comfortably.


text:
Obituary & Guest Book Preview for JOHN C. FOUCHÉ Jr.
JOHN C. FOUCHÉ Jr. Obituary
Published in Knoxville News Sentinel from July 27 to July 28, 2013
First 25 of 165 words
: FOUCHÉ, JOHN C., JR. - age 89, of Maryville, passed away Thursday, July 25, 2013, at Blount Memorial Hospital. He was a member of Broadway...
View Full Obituary & Guest Book for JOHN FOUCHÉ Jr.
JOHN FOUCHÉ Guest Book | View 1 of 1 Entry:
View Full Obituary & Guest Book for JOHN FOUCHÉ Jr.
Ways to View
Restore Obituary & Guest Book
Keep the Obituary and Guest Book online for one of these time periods:
Permanent Sponsorship
$ 79.99
One-Year Sponsorship
$ 29.99
24-Hour Restoration
$ 2.95
Print Obituary & Guest Book
Order a professionally printed copy, starting at $39.00:
Deluxe Hardcover Book
$ 79.00
Laminated Hardcover Book
$ 59.00
Laminated Softcover Book
$ 39.00
Didn't find the Obituary you were looking for?
Return to today's Obituaries for the KnoxNews.com.
Powered By
Search Obituaries
How To
Questions? Contact Us
Privacy Policy
© 2015 Legacy.com All Rights Reserved


text:
We have a great range of women's tops for sale... From plain t-shirts to statement blouses, you're sure to get it for less.
Results 1-5 of 5
Topshop size8 striped orange top
River Island - Size: L - Orange - T-Shirt
New Look cotton top Size 16
BNWT-Fat Face - Size: 16 two peach coloured sleeveless tops
BNWT Topshop Size 8 Peach Lace Detailed Crop Top
Page 1 of 1 | Showing 1-5 of 5
What we do
Get involved
Make a donation
© 2015. Oxfam is a registered charity in England and Wales (no 202918) and Scotland (SC039042).
Oxfam GB is a member of Oxfam International.


text:
First Baptist Church brings back 118-year-old bell
Joyce McCollaum, a member of the First Baptist Church's historical committee, stands with Pastor Dave Miller behind the 118-year-old cast-iron bell. (Jill McLaughlin Photo)
The First Baptist Church in downtown Roswell is preparing to ring-in Christmas this year by reviving its past.
A cast-iron bell that was first rung in 1895 by the church's founding members has been restored and is ready to serve again.
A Bible study class at the church started the idea of bringing back the bell, said Joyce McCollaum, a member of the church's historical committee.
"They started talking about why they hadn't heard the bell in a long time," McCollaum said.
After many years of sitting idle, the bell was corroded and had lost its color. With the help of a local handyman, it was scrubbed and restored in time to ring Login to read more
Related Posts
Leave a Reply
You must be logged in to post a comment.
« 3 drive-by shootings take place in one night AWA fundraiser set for Saturday »


text:
Resolution Analysis of Acoustic Tomographic Imaging with Finite-Size Apertures Based on Spatial-Frequency Coverage
ABSTRACT The objective of this paper is to present a comprehensive analysis of resolving capability of acoustic diffraction tomography
using finite-size receiving apertures based on the span of spatial spectral coverage. The analysis is conducted on both the
transmission and reflection modes, and can be generalized into various data acquisition configurations. In addition, the analysis
provides an important link to other tomographic imaging modalities such as the classical X-ray tomography and synthetic aperture
radar systems.
- References (7)
- Cited In (0)
- The Journal of the Acoustical Society of America 12/1988; 84(6):2103-2108. DOI:10.1121/1.397055 · 1.50 Impact Factor
Article: Modern Acoustical ImagingThe Journal of the Acoustical Society of America 01/1989; 85(5). DOI:10.1121/1.397840 · 1.50 Impact Factor
- [Show abstract] [Hide abstract]
ABSTRACT: The scanning tomographic acoustic microscope (STAM) was proposed in 1982 as a method of improving the resolution capability of the scanning laser acoustic microscope (SLAM) based on the principles of tomography. By modifying the SLAM with a quadrature detector, tomographic projections that contain both the amplitude and phase information of the scattered wavefield can be acquired. Subsequently, multiple projections acquired with different incident waves are combined using the "back-and-forth" propagation algorithm to form the tomographic reconstruction. The first STAM reconstructions have been obtained to experimentally demonstrate the superior resolution capability of the STAM over the SLAM. In this paper, the implementation of the STAM is described, and experimental reconstructions of multiple-layer specimens are demonstrated.IEEE Transactions on Image Processing 02/1995; 4(3):358-69. DOI:10.1109/83.366483 · 3.63 Impact Factor
Similar Publications
Data provided are for informational purposes only. Although carefully collected, accuracy cannot be guaranteed. The impact factor represents a rough estimation of the journal's impact factor and does not reflect the actual current impact factor. Publisher conditions are provided by RoMEO. Differing provisions from the publisher's actual policy or licence agreement may be applicable.


text:
Email this article from PlasticsNews.com
Dart Container planning $47 million headquarters expansion
MASON, MICH. (Nov. 29, 11:20 a.m. ET) — Dart Container Corp. is planning a $47 million expansion and renovation of its headquarters and facilities in Mason. The project includes a new 110,000-square-foot corporate headquarters, a new 500,000-square-foot warehouse, and the renovation of existing buildings. The expansion will create 325 jobs.
More Events > Upcoming Events
Most Popular
More plastic packaging can mean less waste, say experts
NEWS Monday, October 05
Kullman leaving DuPont; former Tyco exec Breen named interim CEO
NEWS Monday, October 05
Franz retires from Haitian executive roles
NEWS Monday, October 05
Taking 3-D printing beyond the prototype
NEWS Monday, October 05
Plastic Bank aims to reduce marine debris, help people
NEWS Tuesday, October 06


text:
Alerts loading...
Please note:
PC Designer
License Wizard
Software Downloads
NOTE: the information below is provided for your convenience only and we cannot guarantee its accuracy. If necessary, please verify with us before purchasing.
Belkin 3m LC / SCLC / SC 50/125µm, MM, 3m
Belkin cables ensure the performance and reliability of servers, storage, and network switches. Built using the highest-quality fibre cable and connectors, Belkin fibre-patch cables come with the satisfaction guarantee of Belkin's Lifetime Warranty. All Belkin fibre cables are factory-terminated and -tested to ensure the highest level of performance and reliability.
shopping Cart
0.00Total in cart:
0.00Total ex.VAT:
LambdaTek win the Queen's Award for Enterprise
|LambdaTek Home | About Us | Frequently Asked Questions | Terms and Conditions | Privacy Policy | Weee Directive | RSS Daily Deals


text:
Antique Pyrography wood toy piano candy container. The front top lifts up. Children on the front, decorations on the sides. Very good condition. 2 d x 4 1/4 w x 4 1/4 t". Unusual piece. 1915 written on the base in pencil.
Another shopper has this item in their Cart or Wish List.
Quality Antiques ~ Something for every antique collector ~ Specializing in antique toys
Fine examples of our inventory. A lot more available, please inquire. All items normally ship next day after payment
|Shipping & Handling :
Price for shipping to USA
change location
USPS First-Class Mail Parcel


text: http://bugs.freedesktop.org/show_bug.cgi?id=16697 --- Comment #2 from Eric Anholt <eric@...> 2008-10-08 23:59:23 PST --- Better fix in commit 9aec1288eeae8e87adc9a99f377be536892941b2 Author: Eric Anholt <eric@...> Date: Wed Oct 8 23:34:38 2008 -0700 i915: Accelerate depth textures with border color. but it needs to be ported to 965 as well. -- Configure bugmail: http://bugs.freedesktop.org/userprefs.cgi?tab=email ------- You are receiving this mail because: ------- You are on the CC list for the bug.


text:
Sorry for the service hiccup, we are back.
I may have some posts about project economics and Net Present Value, materials handling, pressure/vent calculations, or other items in the future.
Sorry for the service hiccup, we are back.
I may have some posts about project economics and Net Present Value, materials handling, pressure/vent calculations, or other items in the future.
In this post I'll list a few resources to find fouling factors for your heat exchanger designs. But first, let's take a step back and review the definition and purpose of fouling factors in heat exchanger design.
Recall that in the heat exchanger sizing equation Q = U * A * LTMD, the "U" factor was a representation of all the resistances of heat exchange between the two sides. "U" was influenced by the types of fluids in the exchanger and also to a lesser degree by the material of construction of the heat exchanger. The fouling factor helps us add some additional detail, by representing the extra resistances that appear on the inside and outside of tubes after an exchanger has been operating awhile: the caked on products of fouling. There are different types of fouling, from crystalline scale to literal pieces of gunk. Fouling introduces a wrinkle into our simple equation, because now the resistance to heat exchange varies over time. But we still have to pick one exchanger design and one "A" value for the entire cycle of operation.
A properly chosen fouling factor will inform the detailed design of heat exchangers by vendors and/or software. As always, the best values to use are real-world values based on experience or tests in your own plant. Nevertheless, it's helpful to have literature values for a starting point.
Note that some of these links are PDF files: Continue reading
After a little downtime we are back! I have added a new category, Cost Estimation, to our store of suggested books. The collection of books by John Page, also called "Page and Nation" by some, are a great way to get manhour requirements for installing piping, HVAC systems, cable trays, and other items.
With the growing opportunities in Latin America, I've found myself having to deal with the occasional document in Spanish, despite never having taken so much as a high school class in Español. (I ended up with another language instead.) In some cases I've had to sift through getting everything in Spanish: the process description, P&IDs, you name it. Of course "you'll need to start learning the process before we've hired the translator." How to cope with a language you've never studied?
Here are some tools and tricks for steps 2 and 3: Continue reading


text:
Property Search Results
Office Space for rent in Habshiguda, Hyderabad
- 1200 SqFeet Office Space @ Rs.17/SqFeet
- ID: P584110272 — Posted: by B Prasad
- 1200 sft office space on rent on Habsiguda Main Road
- 1200 sft office space on rent on Habsiguda Main Road Prime property, prime location. 1 min walkable distance from bus stops and metro stations. Very closeby to banks. more details
Still couldn't find the property of your choice?
- Try a broader search like: Commercial Properties for Rent in Hyderabad
- Signup to receive free alerts by SMS and email when properties matching this search are posted.
- Post your requirement so that owners with matching properties may contact you directly.


text:
The object of the Editors of this series is a very definite one. They desire above all things that, in their humble way, these books shall be the ambassadors of good-will and understanding between East and West--the old world of Thought and the new of Action. In this endeavour, and in their own sphere, they are but followers of the highest example in the land. They are confident that a deeper knowledge of the great ideals and lofty philosophy of Oriental thought may help to a revival of that true spirit of Charity which neither despises nor fears the nations of another creed and colour. Finally, in thanking press and public for the very cordial reception given to the "Wisdom of the East" Series, they wish to state that no pains have been spared to secure the best specialists for the treatment of the various subjects at hand.
ow merely after material possessions in the belief that we can retain them, we are pursuing an object which does not really exist." . . . And this idea of transitoriness prevails so
generally among the Arabs that the salad-seller recommends his transitory wares to pious folk by calling, "God is that which does not pass away!" So, too, the Arab pictures as a bird, a thing of transience, the human soul. In Syria the dove is often carved upon their ancient tombstones. And the Longobards among their graves erected poles in memory of kinsfolk who had died abroad or had been slain in battle; on the summit of the pole was a wooden image of a dove, whose head was pointed in the direction where the loved one lay buried. With us, as with Abu'l-Ala
(_quatrain_ 26), the soul may metaphorically be imagined as a bird, but for the European's ancestor it was a thing of sober earnest, as it is to-day to many peoples. Thus the soul of
Aristeas was seen to issue from his mouth in the shape of a raven.[6] In Southern
Cara Devon has always suffered curiosity and im... Read more
When the economic downturn ends Matty Cruzs co... Read more
A telekinetic teenager. A telepathic child. A p... Read more
A century after an apocalyptic war reduces all... Read more
Fall in love with the Kensington siblings in th... Read more
See it as donating a moment of your social media time, every little thing helps us improve and stay online.
The list of books below is based on the weekly downloads by our users regardless of eReader device or file format.
See more popular titles from this genre.


text:
2 Bedroom Apartment / Flat for sale in Weavind Park
28 Pitts Street, Weavind Park
A warm and friendly 2 Bedr Flat with full Bathroom. (Bath & Shower). Open-Plan L/Areas. Balcony with view to die for! For the young ...
3 Bedroom Apartment / Flat for sale in Arcadia
Damaza, 463 Edmond Street, Arcadia
Good investment block in quiet environment!! 70sqm 2 bedroom flat with rooms that has built in cupboards, enclosed balcony which is ...
3 Bedroom House for sale in Magalieskruin
68 Fairleigh Estate, Braampretorius Str 564, Magalieskruin
Don't miss your chance to make this Lovely HOUSE...Your home! Call us now to make THIS HOUSE YOUR HOME!!! Don't hesitate to contact ...
1 Bedroom Apartment / Flat for sale in Montana
A 14 La-mon Villa, 531 6th Road, Montana
Lovely studio/Batchelor unit in Montana. The complex offers security at the gate, covered parking (carports) and low low levies and rates ...
Development in Annlin
From R 375 000 to R 735 000
Phase 1 of Bateleur will be obtainable on 1 September 2014, being a high quality and well located apartment block. Phase II is available from 1 ...
Be the first to see new properties for sale in Pretoria Note: You can cancel at any time
If you are using Internet Explorer 8 or higher, please verify that your Internet Explorer compatibility view settings are not enabled.
For the best browsing experience, update to the latest Version of Internet Explorer or try out Google Chrome or Mozilla Firefox.
Please contact our Property24 Support Team for further assistance. Tel. +27 (0)861 111 724