text:
|Publication number
|Publication type
|Application number
|Publication date
|May 3, 2007
|Filing date
|Jan 27, 2006
|Priority date
|Nov 2, 2005
|Also published as
|Publication number
|David Licata
|Original Assignee
|David Licata
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (8), Referenced by (11), Classifications (9), Legal Events (2)
|External Links: USPTO, USPTO Assignment, Espacenet
Implants such as stents and occlusive coils have been used in patients for a wide variety of reasons. One of the most common "stenting" procedures is carried out in connection with the treatment of atherosclerosis, a disease which results in a narrowing and stenosis of body lumens, such as the coronary arteries. At the site of the narrowing (i.e., the site of a lesion) a balloon is typically dilatated in an angioplasty procedure to open the vessel. A stent is set in apposition to the interior surface of the lumen in order to help maintain an open passageway. This result may be effected by means of a scaffolding support alone or by virtue of the presence of one or more drugs carried by the stent to aide in the prevention of restenosis.
Various stent designs have been developed and used clinically, but self-expandable and balloon-expandable stent systems and their related deployment techniques are now predominant. Examples of self-expandable stents currently in use are the Magic WALLSTENT� stents and Radius stents (Boston Scientific). A commonly used balloon-expandable stent is the Cypher� stent (Cordis Corporation). Additional self-expanding stent background is presented in: "An Overview of Superelastic Stent Design," Min. Invas Ther & Allied Technol 2002: 9(3/4) 235-246, "A Survey of Stent Designs," Min. Invas Ther & Allied Technol 2002: 11(4) 137-147, and "Coronary Artery Stents: Design and Biologic Considerations," Cardiology Special Edition, 2003: 9(2) 9-14, "Clinical and Angiographic Efficacy of a Self-Expanding Stent" Am Heart J 2003: 145(5) 868-874.
Because self-expanding prosthetic devices need not be set over a balloon (as are balloon-expandable designs), self-expanding stent delivery systems can be designed to a relatively smaller outer diameter than their balloon-expandable counterparts. As such, self-expanding stents may be better suited to reach the smallest vasculature or to achieve access in more difficult cases.
One consideration pertinent to the delivery of self-expanding stent designs concerns frictional forces internal to the subject delivery system. Internal forces can be a significant issue with respect to system actuation. Testing by the assignee hereof has clearly demonstrated a loss of motive force available to actuate a distally located restraint when the delivery system is subject to conditions of or simulating tortuous anatomy. As such, systems that require little or no direct user actuation of components to effect stent release are desirable. Examples of such systems are presented in a number of patents.
U.S. Pat. No. 5,873,907 offers a system employing circumferential bands, each having an electrolytically erodable section, where the bands hold a stent in a collapsed configuration until the sections' joints are released. U.S. Pat. No. 5,980,514 discloses a system in which a self-expanding stent is held onto a wire directly by electrolytically erodable joints. U.S. Pat. No. 6,716,238 describes a system in which one or more tethers are connected to a stent to releasably restrain it for delivery or to stabilize its position during delivery. The tethers may work in conjunction with a sheath to hold a distal end of the stent, allowing withdrawal of the sheath from the stent to effect release. When the sheath is withdrawn, the stent expands radially, but is not fully released until the connections between the tethers and the stent are mechanically broken, electrolytically eroded or melted away. In another embodiment described in the '238 patent, one or more tethers is wrapped around the body of the stent to tie it down until released.
Upon closer examination, each of the referenced systems has serious limitations—either in terms of clinical practicality, basic mechanical feasibility or both. Issues variously presented by the referenced stent delivery systems include:
Other issues may be presented as well as appreciated by those with skill in the art. Accordingly, to maximize the potential benefits of electrolytic release for stents and/or other implants, there exists a need for improved systems.
Variations of the invention hold a radially-expandable, implantable prosthesis (such as a stent) in a collapsed configuration for delivery with the assistance of one or more members including an electrolytically erodable or disintegratable section. Unlike most electrolytic-based implant delivery systems, none of those in the subject invention employ a "joint" that joins or attaches the implant directly to the delivery member. Neither do the electrolytically erodible sections secure tethers or bands restraining the central diameter of the implant body.
Rather, either a twist imparted to the implant or a tubular sleeve holds the implant in a compressed profile. The sacrificial electrolytic member maintains the implant or restraining sleeve in a preloaded configuration until release. Upon release, the implant is freed to untwist or the sleeve withdrawn therefrom. In each case, the implant is then able to expand.
One aspect of the invention offers systems in which one or more ends of an implant or prosthesis such as a stent are held in a highly stable manner by a clinically and mechanically robust, yet compact holding/restraining and releasing means. In these variations of the invention, one portion of the means comprises pass-through or male-female interlocking features between an end of the implant and a delivery guide member body. The engaging features associated with the delivery member body may comprise as few as two features; more preferably, three, four or more features arranged around a ring. Most advantageously, the multiple members are symmetrically spaced apart.
The implants employed in connection with the present invention are often stents. Most typically, the stents employed are lattice, cage or successive linked ring-type structures. Often they are cut from tubing or they are mesh-like woven or assembled bodies. However, other means or modes of manufacture are possible as well.
In a number of variations of the invention, the configuration of the stent or other implant employed is critical to the operation of the system. In these systems, foreshortening of the implant (i.e., decreasing length of the body, or at least a portion thereof, in transitioning from a collapsed to an expanded state) plays an important roll in stent release. In those examples of the invention where foreshortening plays a role in implant release, the implant is often adapted to be twisted and it is this twisting that assists in holding the stent at a desired diameter.
In order to hold such an implant in a radially compressed and twisted state, the implant (e.g., a stent) optionally includes one or more extensions or projections from its main body. Such extensions are adapted to nest or otherwise interface with complimentary seat features set upon or retained by the guide body portion of the implant delivery system. The projections may simply comprise elongate members that offer a laterally stable interface or comprise hook-shaped forms (e.g., "J", "T", "L", "S", "V" etc. shapes) that also an axially stable interface. Alternatively, the implant body or projections therefrom may instead incorporate or define openings or receptacle features. As such, the prosthesis or the delivery guide seat features may extend partially to a certain depth (i.e., forming a pocket). Alternatively, the openings may extend through the entire member.
Stated broadly, either the implant and/or the delivery guide may include male and/or female interlocking features. A grasping form of interface may be employed to axially tension the implant and/or provide secure capture at one side of the implant to provide "bail-out" potential. The delivery guide side of the interface may be referred to as a "seat" or otherwise. Especially, where the members hook into one another, they may be regarded as "nesting" features. "Lock" and "key" terminology may also be used to describe the features.
Alternatively, the implant/delivery guide interface may be adapted for sliding receipt and release. Such configurations enable various self or automatic release approaches. For these, extension from the implant or even the crowns of adjacent stent struts may provide the interface with delivery guide component(s). For these types of interfaces, "key" and "way" terminology may be most appropriate. Still, the delivery device side may be regarded as a "seat" or "seating" region or portion.
In any of these systems employing a stent that is both radially compressed and twisted, various approaches to loading the implant onto or into the delivery guide may be employed. One highly advantageous method that forms an aspect of the present invention involves pre-loading the implant into a sleeve. The stent is compressed by hand, with an automated "crimper" such as produced by Machine Solutions, Inc., or otherwise, without a substantial twist and loaded into a tubular body close in diameter to its final size upon or in the delivery guide. By "close" in diameter, what is meant is that it is within at least 33%, or more preferably about 25%, or even about 10% of its final diameter. Then, with the stent so-constrained, it is twisted from either or both ends before or after partial or full attachment to the delivery guide.
The sleeve may comprise a plurality of separate pieces or segments (most conveniently two or three). As such, the individual segments can be rotated relative to one another to assist in twisting the stent. In addition, axial manipulation of the relation of thin individual segments can be employed to allow the implant to bulge outwardly over one section. The foreshortening caused by this action may then allow positioning and then axially loading end interface members by manipulating the segments to collapse the bulging.
As for the electrolytically releasable retention members, one class comprises wire or ribbon passing through at least one portion of the implant (including any extension or projection therefrom). A loop of material threaded through a receptacle or adjacent one or more crown portions of a stent offers a very strong, yet simple connection to the delivery guide. The loop may itself be erodable at a section, or an erodable bridging segment may be incorporated in the loop. An even more elegant connection employs a pin, post or rivet-like member received through the implant. The pin or rivet structure may be headed (i.e., have a bulbous or flattened head or mushroom-like shape) in order to prevent lift-off of the implant member. Erosion of the head and/or periphery of the structure will then release the implant.
Systems employing these pass-through type of securing features may work alone by holding an implant such as a stent in a twisted state. Alternatively, they may hold such an implant in a twisted and/or axially stretched configuration with a retractable tubular sheath or restraint sleeve over the implant. In this way, the electrolytically released components can partially restrain the implant so as to reduce sleeve hold-down and/or withdrawal forces. The pinned/threaded connections are particularly space-efficient and, thus, suited for use with supplemental restraining device components while maintaining small device crossing profiles.
When employed with a retaining sleeve, as few as one location on the implant can be secured by the electrolytically released component(s). For a stent, the location may be proximal or distal. A distal location may offer improved sleeve withdrawal characteristics; a proximal location may simplify electrical conduction and or performance issues. Employing both proximal and distal electrolytic retainer attachment locations enable axial and/or torque-based implant diameter reduction techniques to reduce the force required for restraint withdrawal.
Another class of retainers or retaining and release means or releaseable means comprises an overriding means or covering including at least one electrolytically erodable section that releasably secures ends of an implant in association with a seating portion or portions of a delivery guide. In one variation, the means comprises a wrap or band or plurality of bands. Upon release of one or more sections, the band opens or wrap loosens. In another variation, the radial retention member comprises a sleeve or casing with one or more electrolytically erodable sections that allow (or upon release, cause) the structure to open. Such a "flower petal" type design may be soldered or welded together in a pre-stressed state in order that it self expands upon eroding connecting webbing, solder, weld point(s), etc.
When either one of such structures is opened or relaxed, the captured projections of the implant can release from complimentary delivery guide side features. When the projection(s) and seat(s) offer a keyed interface, those on the implant can lift or be pushed out of plane to release at least one portion of the implant. When the projection(s) and seat(s) are adapted for sliding receipt and release, the latter will occur more readily upon release of the wrap or band and force holding parts together (if only by virtue of static friction).
As with the above variation of the invention in which retention members pass through one or more portions of the implant, those retention members that overlay one or more portions of the implant may secure the proximal and/or distal side the implant or point(s) between. They may be used in complimentary pairs or with other structures as is convenient. In the latter example, either one of the wrap-around or openable sleeve retainer may be set at the distal end of the stent with a mechanical release mechanism positioned at the proximal side of the stent. Examples of such mechanically-actuated systems including retractable mini-sheaths and wire or suture cut-down bands are presented in U.S. patent application Ser. No. 11/266,587, entitled, "Twist-Down Implant Delivery Technologies" filed on Nov. 2, 2005, which application is incorporated herein by reference in its entirety.
Furthermore, different ones of electrolytically releasable means may be set at opposite ends (or simply different locations) of the implant. Certain advantageous pairings are elaborated upon below, especially for stent release. For the sake of discussion here, however, one such combination employs the "flower petal" style means at the proximal side of the device and wrap-around means at the distal end. After implant delivery, a proximal-side open "flower petal" device will easily collapse when withdrawn into a delivery catheter. On the distal side, however, the open structure could get caught. In contrast, a loose filament released from a wrap-around means can easily be drawn into the delivery catheter regardless of location on the delivery guide. Yet, because the flower petal approach may be more easily slipped over the stent in a final loading step than a filament wound around the stent and secured, the former structure may be desired on at least one side of the system for ease of assembly. Of course, other exemplary systems employing the various retention and release means are possible in which the rationale for the combinations may be apparent upon further review of the subject specification.
Equally applicable to a mix-and-match approach are electrolytically releasable means that are specifically adapted to permit an implant's release by its mating portion(s) sliding out of a restraining configuration upon some triggering action. In other words, either one or both sides of a stent delivery system according to the present invention may employ this type of release approach.
As for the various device configurations that enable such action, a first one is advantageously used to initiate release of the implant. In one example, the releasable retention means comprises a covered interface held by an electrolytic latch. Once released, the interface rotates—allowing the stent to untwist and expand. The expansion causes the length of the stent to shorten because of its cell or strut geometry. As it shortens (foreshortens) the stent draws itself (e.g., end projections) out of the retaining interface. In one variation, the covering comprises a band. In another example, it comprises a wrap that loosens as the stent unwinds. Such a configuration will allow release of keyed projections, with or without substantial implant foreshortening.
In any case, all of the above-described release members are to be actuated by electrolytic erosion through application of voltage by user action (or computer programming) with a suitable power supply. Other variations of the present invention employ a second implant retaining and release means that is tripped or actuated by action of the first release means or by the resulting configuration of the implant. In these variations of the invention, the second or subsequent means automatically completes implant release. Such action may be desired as a matter of convenience in reducing user activity, it may offer improved release speed or reliability, or it may be desirable for yet one or more other reasons.
Of these auto-release systems, a first type is one in which a floating or non-secured tubular member is set over one end of the implant. Expansion of the implant—as in the example of a stent—drives the tube off of the stent in the direction of expansion. So-displaced, the adjacent captured end of the stent is released. An interlocking key or slip-type interface may be freed-up by virtue of moving the outer sleeve. In either example, the stent may lie upon a core member along its entire length or a region under the projections may be relieved or undercut. Such relief allows for steepening of the angle that the end of the implant can assume without flexing of material. This effect offers improved mechanical advantage for moving the outer sleeve.
Another example of an automatically releasable retention means employs a coil spring attached to or integrally formed with a sleeve. The sleeve is offered, again, to cap the implant member and the delivery guide interface features (i.e., the projection and seat members). Withdrawal of the sleeve permits release. The spring positively locks the sleeve in place in one state and actively withdraws it in a second state. In the locked state, the spring is tightly wound about the system body so that its position is bound-up by frictional forces. In the second state, the spring is opened by untwisting so as to be able to retract freely.
This unwinding of the spring is accomplished by releasing the pre-twist imparted to the adjoining implant. Until released, the spring is held twisted by the twisted implant interfacing with keys or ridge features underlying the sleeve. When the twisted implant (be it a stent or another medical device) is released, the spring also untwists and is freed from its cinched-down configuration—thereby allowing its retraction from a stretched-out/preloaded state.
As an alternative to the twist-lock approach described above, a spring member (in the form of a coil or otherwise configured) in an axially preloaded state can be secured by an anchor member. The anchor member may comprise an electrolytic latch. Alternatively, an "anchor" line in the form of a ribbon, suture, etc. can be set under the implant and held in place by its radial compression derived from the twist-down. When the twist to the implant is released and the body expands, so-too is the pressure securing the anchor line.
In another variation of the invention, no active release means is employed at a second side of the implant. Rather, the second end of an implant is held by an interface that can be released by simply withdrawing or advancing the delivery system. An advantageous configuration in this regard employs a covered slide-out implant/delivery guide interface at a proximal end of a stent. After distal release of the stent and expansion into contact with the body lumen, the delivery system is withdrawn from the stent, releasing the proximal end. In yet another approach where the proximal side of the delivery system includes the slide-out enabled architecture, an end of a catheter in receipt of the delivery guide is employed to abut the stent proximal/near side. Then advancement of the catheter or withdrawal of the delivery guide further into the catheter releases the near side of the stent.
In a manner related to the spring-back variations of the invention for releasing interface features on one side of a twisted-down implant, the concept can be applied to withdrawal of a sheath or a restraint covering at least a portion of the body of an implant. Specifically, an electrolytic latch can be employed for restraining a stretched-out spring element to withdraw a sheath from up to the entire length of a stent. In one example, the releasable trigger member is connected to a spring member that is only axially stretched. In another variation, the spring member is both axially stretched and twisted. The twisting in this case would be for binding-up the spring or at least assisting in its restraint. Upon release, the restraint untwists and is able to fully retract. When a twisted spring is employed, the implant to be delivered will rotate with the restraint as the restraint is withdrawn. In either of these approaches, end-capture features for the restraint/sheath may resemble or be identical to features described above in connection with releasably securing the end of an implant (i.e., the keyed projection and seat features).
As stated above, the inventive features may be used to deliver any of a variety of implants besides lattice or cage-type stents. Though stents are focused on in this regard, other examples of implants which may be employed with the subject invention include coil stents, embolic protection filters and other "clot-pulling" devices, embolic coils and even application to other technologies. However, in order to function properly in connection with certain variation of the invention, the implants must sometimes be configured such that at least a portion foreshortens during deployment. Still, other ones of the system variations—such as spring-back sheath variations discussed directly above—may work with any sort of deliverable medical device for permanent or temporary implantation. It is also contemplated that those improvements described herein pertaining to methodology, power profiles for electrolytic actuation, insulation and electrical architectures may be applied in delivery systems for any type of implant.
Regarding methodology, the subject methods may include each of the mechanical activities associated with implant release as well as electrical activity. As such, methodology implicit to the use of the devices described forms part of the invention. Such methodology may include that associated with completing an angioplasty, bridging an aneurysm, deploying radially-expandable anchors for pacing leads or an embolic filter, or placement of a prosthesis within neurovasculature, an organ selected from the kidney and liver, within reproductive anatomy such as selected vasdeferens and fallopian tubes or other applications. In some methods, the various acts of implant release are considered; in others, the power profiles, monitoring of power and other aspects of power control are considered.
More particularly, a number of methods according to the present invention involve the manner in which the delivery system operates in releasing an implant. In one such method in which the implant is restrained at two sides, a first side is released by an electrolytic latch, followed by automatic release of the second side, wherein the automatic action is initiated by release of the first side. In another method, two independently releasable ends are released one after the other. This action is accomplished not by the contemporaneous erosion of an electrolytic member on each side of the implant (with staged release controlled by different amounts of material to erode as is known, and suitable for certain variations of the invention disclosed herein) but, rather, by selecting and eroding material incorporated in separate electrical circuits. Yet another delivery method is contemplated in which an electrolytic component is released only on one side of the implant. Such action may be followed by a mechanical release action such as withdrawing or tearing a sheath member manually or activating a "muscle wire" to do the same. Another approach may involve withdrawing or advancing the delivery guide to effect final implant release from a second, slidably releasable means holding the other end of the implant.
Another aspect of the invention provides methods involving power profiles for electrolytic element erosion and hardware/systems and software to control the methods. While electrocoagulation (e.g., by thrombus formation in blood by blood cell attraction to a positive charge) can be a positive factor in embolic coil delivery, it has been appreciated that such action can severely hamper or interfere delivery of other types of implants. Rather than isolating the member to be eroded, however, an aspect of the present invention involves use of power profiles to ameliorate or eliminate any problems with electrocoagulation. Specifically, an AC signal is employed. Not to be bound by the theory, but it is believed that the negative charge of the cycle repulses blood cells. However, to effect the desired erosion, a net positive charge is required. Thus, a positive DC offset is applied to the AC voltage. The positive offset and amplitude of the AC signal yields higher peak voltages to drive electrolytic corrosion. Irrespective of electrocoagulation, these higher peak energies may result in increased erosion and consequent implant deployment. Numerous potential power profiles according to this general principle are detailed below. Furthermore, experimental results are provided which illustrate the effectiveness of such an approach.
Another approach according to the present invention involves system insulation in order to focus erosion, or at least maximize current density at the point of erosion. Generally, it is known that "fast acting" detachable systems can be produced by insulating or electrically isolating everything but that portion of the system where electrolytic erosion is desired. The present invention improves upon this concept in certain variations by producing system components with high-strength titanium or titanium alloy. In addition, they can be anodized to form an oxide layer insulative to DC voltage. The components are typically pre-anodized. Yet, when using higher voltages, the systems are "self-healing" such that any scratches, etc. exposing material to unintended corrosion/erosion will skin-over and maintain their integrity. Such a system is highly advantageous as it eliminates the need for polymeric insulation; further, such a system need not rely on the use of noble (or high Mendelev number) metal material selection to protect components that can be costly or cumbersome to apply, or cause failures if disturbed.
Yet another aspect of the invention concerns the electrical architecture or design of the subject systems. In order to minimize or eliminate any potential risk associated with the electrical signal carried by the system, certain variations of the invention are configured such that the release member(s) are positively charged at the point where erosion is desired and the section(s) are directly adjacent (i.e., within about 1 cm, preferably less) to a negatively charged element. In one variation of the invention, the subject retention and release means is located at a distal end of the delivery system. It is electrically connected to the (net) positive side of a power source via wiring running within a delivery guide body made from hypotubing.
In all, the implant retention member configurations, electrical connectivity and conductivity as well as the manner in which they are employed can vary in a number of ways according to the present invention. Furthermore, some variations of the invention are adapted to release an implant or prosthesis such as a stent without the use of any mechanically actuated components. In other variations of the invention, systems may include manually-actuated or activated components. However, it is envisioned that the former class of purely electrically-actuated systems may offer particular flexibility as subcomponents for systems requiring off-axis implant delivery.
Used as a subcomponent, a discrete electrically-actuated system as described in detail below could be attached in a non-axial or "V" shaped configuration to allow treating a crossing vessel in a "T" shaped site such as at or near the basilar artery junction. In which case, a tether underlying the implant or a distal sheath may be provided to re-constrain an open "V" configuration to allow its recapture and withdrawal into a catheter body. Alternatively, an electrically-actuated system could be provided at a distal end of a system that includes a drawstring or wire to force a dramatic prolapse in the system in order to treat such a vessel configuration as described above. Unless such features are provided, however, the subject systems typically comprise flexible linear or unidirectional delivery guide members without extensions, arms or fittings at the far end of the delivery guide body.
The delivery systems described herein offer a number of advantages in their efficient construction and ability to deliver implants with or without coatings in highly challenging applications. Those with skill in the art may appreciate further benefits or advantages of the subject inventive variations.
The term "stent" as used herein includes any stent, such as coronary artery stents, other vascular prosthesis, or other radially expanding or expandable prosthesis or scaffold-type implant suitable for the noted treatments or otherwise. Exemplary structures include wire mesh or lattice patterns and coils, though others may be employed in certain variations of the present invention. A "self-expanding" stent as used herein is a scaffold-type structure (serving any of a number of purposes) that expands from a reduced-diameter (be it circular or otherwise) configuration to an increased-diameter configuration. The mechanism for shape recover may be elastic or pseudoelastic. While it is generally desirable to employ an alloy (such as nickel-titanium, or Nitinol alloy) set for use as a superelastic alloy, it may alternatively employ thermal shape memory properties to drive expansion upon release.
A "wire" as used herein generally comprises a common metallic member such as made of stainless steel or another material. The wire may be at least partially coated or covered by a polymeric material (e.g., with an insulating polymer such as Polyamide, or a lubricious material such as TEFLON�, i.e., PolyTetraFluoroEthelyne or PTFE). Still further, the "wire" may be a hybrid structure with metal and a polymeric material (e.g., Vectran™, Spectra™, Nylon, etc.) or composite material (e.g., carbon fiber in a polymer matrix). The wire may be in the form of a filament, bundle of filaments, cable, ribbon or in some other form. It is generally not hollow. The wire may comprise different segments of material along an overall length.
A "hypotube" or "hypotubing" as referred to herein means small diameter tubing in the size range discussed below, generally with a thin wall. The hypotube may specifically be hypodermic needle tubing. Alternatively, it maybe wound or braided cable tubing, such as provided by Asahi Intec Co., Ltd. or otherwise. As with the "wire" discussed above, the material defining the hypotube may be metallic, polymeric or a hybrid of metallic and polymeric or composite material.
An "atraumatic tip" may comprise a plurality of spring coils attached to a tapered wire section. At a distal end of the coils typically terminate with a bulb or ball that is often made of solder. In such a construction, the coils and/or solder are often platinum alloy or another radiopaque material. The coils may also be platinum, or be of another material. In the present invention, the wire section to which the coils are attached may be tapered, but need not be tapered. In addition, alternate structures are possible. In one example, the atraumatic tip may comprise a molded tantalum-loaded 35 durometer Pebax™ tip. However constructed, the atraumatic tip may be straight or curved, the latter configuration possibly assisting in directing or steering the delivery guide to a desired intravascular location.
To "connect" or to have or make a "connection" between parts refers to fusing, bonding, welding (by resistance, by pressure, laser, chemically, ultrasonically, etc.), gluing, pinning, crimping, clamping or otherwise mechanically or physically joining, attaching or holding components together (permanently or temporarily). To "electrically connect" or to have or make an "electrical connection" between parts refers to providing a low-resistance path for current to flow upon the application of a voltage (AC and/or DC). The connection may be in the form of a wire, a trace of material such as cladding or plating overlying a selected area, conductive fill material (such as silver-loaded epoxy resin), adhesive, solder, etc.
The figures provided herein are not necessarily drawn to scale, with some components and features being exaggerated for clarity. Each of the figures diagrammatically illustrates aspects of the invention. Of these:
Variation of the invention from the embodiments pictured is, of course, contemplated.
Various exemplary embodiments of the invention are described below. Reference is made to these examples in a non-limiting sense. They are provided to illustrate more broadly applicable aspects of the present invention. Various changes may be made to the invention described and equivalents may be substituted without departing from the true spirit and scope of the invention. In addition, many modifications may be made to adapt a particular situation, material, composition of matter, process, process act(s) or step(s) to the objective(s), spirit or scope of the present invention. All such modifications are intended to be within the scope of the claims made herein.
Self-Expanding Stent Designs and Opportunities
In light of this framework,
Features of the present invention are uniquely suited for a system able to reach small vessels (though use of the subject systems s not limited to such a setting.) By "small" coronary vessels, it is meant vessels having an inside diameter from between about 1.5 to 2 mm and up to about 3 mm in diameter. These vessels include, but are not limited to, the Posterior Descending Artery (PDA), Obtuse Marginal (OM) and small diagonals. Conditions such as diffuse stenosis and diabetes produce situations that represent other access and delivery challenges that can be addressed with a delivery system according to the present invention. Other extended treatment areas addressable with the subject systems include vessel bifurcations, chronic total occlusions (CTOs), and prevention procedures (such as in stenting of vulnerable plaque).
It may be preferred to use a drug eluting stent (DES) in such an application to aid in preventing restenosis. A review of suitable drug coatings and available vendors is presented in "DES Overview: Agents, release mechanism, and stent platform" a presentation by Campbell Rogers, MD incorporated by reference in its entirety. However, bare-metal stents may be employed in the present invention.
Examples of various therapeutic agents that may be used in or on the subject prosthesis include, but are not limited to, antibiotics, anticoagulants, antifungal agents, anti-inflammatory agents, antineoplastic agents, antithrombotic agents, endothelialization promoting agents, free radical scavengers, immunosuppressive agents, antiproliferative agents, thrombolytic agents, and any combination thereof. The therapeutic agent may be coated onto the implant, mixed with a biodegradable polymer or other suitable temporary carrier and then coated onto the implant, or, when the implant is made from a polymeric material dispersed throughout the polymer. The agent can be directly applied to the stent surface(s), or introduced into pockets or an appropriate matrix set over at least an outer portion of the stent.
While some might argue that the particular role and optimal usage of self expanding stents has yet to be defined, they offer an inherent advantage over balloon expandable stents. The latter type of devices produce "skid mark" trauma (at least when delivered uncovered upon a balloon) and are associated with a higher risk of end dissection or barotraumas caused at least in part by high balloon pressures and related forces when deforming a balloon-expandable stent for deployment to account for recoil upon balloon deflation.
Yet, with an appropriate deployment system, self-expanding stents may offer one or more of the following advantages over balloon-expandable models: 1) greater accessibility to distal, tortuous and small vessel anatomy—by virtue of decreasing crossing diameter and increasing compliance relative to a system requiring a deployment balloon, 2) sequentially controlled or "gentle" device deployment, 3) use with low pressure balloon pre-dilatation (if desirable) to reduce barotraumas, 4) strut thickness reduction in some cases reducing the amount of "foreign body" material in a vessel or other body conduit, 5) opportunity to treat neurovasculature—due to smaller crossing diameters and/or gentle delivery options, 6) the ability to easily scale-up a successful treatment system to treat larger vessels or vice versa, 7) a decrease in system complexity, offering potential advantages both in terms of reliability and system cost, 8) reducing intimal hyperplasia, and 9) conforming to tapering anatomy—without imparting complimentary geometry to the stent (though this option exists as well).
At least some of these noted advantages may be realized using a stent 10 as shown in
In use, the stent will be sized so that it is not fully expanded when fully deployed against the wall of a vessel in order to provide a measure of radial force thereto (i.e., the stent will be "oversized" as discussed above). The force will secure the stent and offer potential benefits in reducing intimal hyperplasia and vessel collapse or even pinning dissected tissue in apposition.
Stent 10 preferably comprises NiTi that is superelastic at or below room temperature (i.e., as in having an Af as low as 15 degrees C or even 0 degrees C). Also, the stent is preferably electropolished to improve biocompatibility and corrosion and fatigue resistance. The stent may be a DES unit as referenced above. The stent may be coated with gold and/or platinum to provide improved radiopacity for viewing under medical imaging. It may be biodegradable.
For a stent able to collapse to an outer diameter of about 0.012 inches and expand to about 3.5 mm, the thickness of the NiTi is about 0.0025 inch (0.64 mm). Such a stent is designed for use in a 3 mm vessel or other body conduit, thereby providing the desired radial force in the manner noted above. Further information regarding radial force parameters in coronary stents may be noted in the article, "Radial Force of Coronary Stents: A Comparative Analysis," Catheterization and Cardiovascular Interventions 46: 380-391 (1999), incorporated by reference herein in its entirety.
In one manner of production, the stent in
Regarding the finer details of the subject stent, as readily observed in the detail view provided in
In certain variations of the invention, however, the bridge sections can be strategically separated or opened as indicated by the broken lines in
As for the optional double-concave profile of each strut bridge 12 shown, this form is advantageous in that it reduces material width (relative to what would otherwise be presented by a parallel side profile) to improve flexibility and thus trackability and conformability of the stent within the subject anatomy while still maintaining the option for separating/breaking the cells apart. Whether cut to provide rounded end portions or adjoined by a bridge section 12, strut junction sections 28 connect circumferentially or vertically adjacent struts (as illustrated). Where no bridge sections are provided, the junction sections can be unified between horizontally adjacent stent struts as indicated in region 30.
Further optional features of stent 10 are employed in the strut junction sections 28 of the design. Specifically, strut ends 20 increase in width relative to medial strut portions 22. Such a configuration distributes bending (during collapse of the stent) preferentially toward the mid region of the struts. For a given stent diameter and deflection, longer struts allow for lower stresses within the stent (and, hence, a possibility of higher compression ratios). Shorter struts allow for greater radial force (and concomitant resistance to a radially applied load) upon deployment.
In order to increase stent compliance so that it compresses as much as possible, accommodation is made for the stiffer strut ends 20 provided in the design shown in
In addition, it is noted that gap 24 an angle β may actually be configured to completely close prior to fully collapsing angle α. The stent shown is not so-configured. Still, the value of doing so would be to limit the strains (and hence, stresses) at the strut ends 22 and cell end regions 18 by providing a physical stop to prevent further strain.
In the detail view of
Additional features of interest in the stent design include near and far delivery system interface mating portions 92 and 94, respectively. These elements are formed within projections 90 that may be integral to the prosthesis (e.g., when the prosthesis is constructed from a single tube of material, or when a number of wire-type elements are woven to form the body where the ends of the elements form the prosthesis). Alternatively, the projections may be affixed or connected to the stent (e.g., via welding, adhesive bonding, fastening, etc.). In another variation of the invention, the projections may comprise polymeric material that is coated onto the prosthesis 82. Other modes of construction are possible as well. Further details of the projections and respective mating portions are discussed in detail below. Suffice it to say, the elements are sufficiently "floppy" and/or rounded to offer an atraumatic interface with opposing tissue.
The stent pattern 40 shown in
Furthermore, the bridge sections 42 of stent 82 can be separated for compliance purposes. In addition, they may be otherwise modified (e.g., as described above) or even eliminated. Also, in each design, the overall dimensions of the cells and indeed the number of cells provided to define axial length and/or diameter may be varied (as indicated by the vertical and horizontal section lines in
Like the previous stent design, strut ends 50 may offer some increase in width relative to medial strut portions 52. However, as shown in
The "S" curves defined by the struts are produced in a stent cut to a final or near final size (as shown in
For use in the present invention when employing a stent that is twisted down upon the delivery guide in order to reduce its diameter, it has been discovered that the design in
However derived, in order that the stent pack cleanly when twisted, it may be desired to pre-curve its shape. That is to say, the stent may be configured so that when it is twisted, its members go from a pre-twisted shape to a straightened configuration as shown. The amount of shaping to account for hold-down twist may be in the form of a simple bias or helix, S-curves or other shape(s).
Since each of the above stent designs account for problematic strain (and in the latter case actually uses the same to provide an improved compressed profile), very high compression ratios of the stent may be achieved from about 5� to about 10� or above. Moreover, they can be twisted a number of times to maintain a compressed delivery profile. The number of twists required for such action will vary depending on stent diameter and length. For a 28 mm stent sized to treat 3.0 mm vessels, between three and four twists may be desired. Similar diameter, shorter stents will require proportionally fewer rotations, as will generally smaller diameter stents.
Regardless of the design selected, it is noted that each of them exhibit a degree of foreshortening when expanding from a compressed profile. Essentially, the angle change of the struts relative to the central axis of the tubular body accounts for change in length. The amount of foreshortening experienced will, thus, depend on a combination of factors: strut length and angle as well as the number of repeating units within the design. The manner in which the resultant foreshortening is put to use in the present invention is elaborated upon below.
Before this discussion, however, it is noted that systems according to the present invention are advantageously sized to correspond to existing guidewire sizes. For example, the system may have about a 0.014 (0. 36 mm), 0.018 (0.46 mm), 0.022 (0.56 mm), 0.025 (0.64 mm), 0.035 (0.89 mm) inch crossing profile. Of course, intermediate sizes may be employed as well, especially for full-custom systems. Still further, it is contemplated that the system sizing may be set to correspond to French (FR) sizing. In that case, system sizes contemplated range at least from about 1 to about 2 FR, whereas the smallest known balloon-expandable stent delivery systems are in the size range of about 3 to about 4 FR. In instances where the overall device crossing profile matches a known guidewire size, they may be used with off-the-shelf components such as balloon and microcatheters.
At least when produced in the smallest sizes (whether in an even/standard guidewire or FR size, or otherwise), the system enables a substantially new mode of stent deployment in which delivery is achieved through an angioplasty balloon catheter or small microcatheter lumen. Further discussion and details of "through the lumen" delivery is presented in U.S. patent application Ser. No. 10/746,455 "Balloon Catheter Lumen Based Stent Delivery Systems" filed on Dec. 24, 823 and its PCT counterpart US824/008909 filed on Mar. 23, 824, each incorporated by reference in its entirety.
In larger sizes (i.e., up to about 0.035 inch crossing profile or more), the system is most applicable to peripheral vessel applications as elaborated upon below. Yet, even in "small vessel" cases or applications (where the vessel to be treated has a diameter up to about 3.0 mm), it may also be advantageous to employ a stent delivery system sized at between about 0.022 to about 0.025 inch in diameter. Such a system can be used with catheters compatible with 0.022 and/or 0.025 inch diameter guidewires.
While such a system may not be suitable for reaching the very smallest vessels, this variation of the invention is quite advantageous in comparison to known systems in reaching the larger of the small vessels (i.e., those having a diameter of about 2.5 mm or larger). By way of comparison, among the smallest known over-the-guidewire delivery systems are the Micro-Driver™ by Medtronic and Pixel™ systems by Guidant. These are adapted to treat vessels between 2 and 2.75 mm, the latter system having a crossing profile of 0.036 inches (0.91 mm). A system described in U.S. Patent Publication No. 2002/0147491 for treating small vessels is supposedly capable of downsizing to 0.026 inch (0.66 mm) in diameter. Furthermore, because certain of the subject devices can be used as a guidewire (in one fashion or another) after stent delivery, the present invention offers further advantages in use as elaborated upon below.
Still, as referenced above, it may be desired to design a variation of the subject system for use in deploying stents in larger, peripheral vessels, biliary ducts or other hollow body organs. Such applications involve a stent being emplaced in a region having a diameter from about 3.5 to 13 mm (0.5 inch). In which case, a 0.035 to 0.039 inch (3 FR) diameter crossing profile system is advantageously provided in which the stent expands (unconstrained) to a size between about roughly 0.5 mm and about 1.0 mm greater than the vessel or hollow body organ to be treated. Sufficient stent expansion is easily achieved with the exemplary stent patterns shown in FIGS. 2A/2B or 3A/3B.
Again, as a matter of comparison, the smallest delivery systems known to applicants for stent delivery in treating such larger-diameter vessels or biliary ducts is a 6 FR system (nominal 0.084 inch outer diameter), which is suited for use in an 8 FR guiding catheter. Thus, even in the larger sizes, the present invention affords opportunities not heretofore possible in achieving delivery systems in the size range of a commonly used guidewire, with the concomitant advantages discussed herein.
Angioplasty Balloon Lumen Based Stent Delivery
As for the manner of using the inventive system as optionally configured for "through the lumen use",
Turning to
As illustrated in
Next, for compatible systems (i.e., systems able to pass through a balloon catheter lumen) the balloon is at least partially deflated and passed forward, beyond the dilate segment 62′ as shown in
However, it should be appreciated that such an exchange need not occur. Rather, the original guidewire device inside the balloon catheter (or any other catheter used) may be that of item 80, instead of the standard guidewire 70 shown in
Alternatively, the exchange of the guidewire for the delivery system may be made before the dilatation step. Yet another option is to exchange the balloon catheter used for predilatation for a fresh one to effect postdilatation.
In addition, there may be no use in performing the step in
Once placement of the stent across from dilated segment 62′ is accomplished, stent deployment commences. The manner of deployment is elaborated upon below. Upon deployment, stent 82 assumes an at least partially expanded shape in apposition to the compressed plaque as shown in
Naturally, the balloon need not be reintroduced for postdilatation, but it may be preferred. Regardless, once the delivery device 80 and balloon catheter 72 are withdrawn as in
Furthermore, it is to be recognized that the subject invention may be practiced to perform "direct stenting." That is, a stent may be delivered alone to maintain a body conduit, without preceding balloon angioplasty. Likewise, once one or more stents are delivered with the subject system (either by a single system, or by using multiple systems) the post-dilatation procedure(s) discussed above are merely optional. In addition, other endpoints may be desired such as implanting an anchoring stent in a hollow tubular body organ, closing off an aneurysm, delivering a plurality of stents, etc. In performing any of a variety of these or other procedures, suitable modification will be made in the subject methodology. The procedure shown is depicted merely because it illustrates a preferred mode of practicing the subject invention, despite its potential for broader applicability.
Stent and Delivery System Overview
Returning to
A given implant may have a number of projections 90, each having various shapes rather than having a single configuration. For twist-down applications, at least two projections 90 will be provided on each side of the stent. When not every stent crown is capped by a projection, the projections are advantageously spaced substantially equally about the perimeter of the stent to evenly distribute loads upon the stent. In which case, the projections may be aligned with one another along the axis of the stent as shown in
The projections may vary in length, especially depending on the form of interface or mating portion it carries or forms. The projections advantageously have a length that allow for efficiently transition or transfer twisting load to the stent while occupying minimal space. Though not necessarily excluded from the invention, projections longer than about one cell's length may have a tendency to wrap or twist about the delivery device body in attempted use.
Now that stents as optionally used in the subject delivery systems have been described, an overview of an implant delivery system according to the invention is presented in
Alternatively, a power supply 116 connected by or to the delivery guide 102 by a cord 118 including one or more leads may be employed. In which case, the power supply is preferably also battery powered in that such an approach offers safety in control of current without resort to sophisticated electronics.
Irrespective of the approach employed, the electronics will be hard-wired or suitably programmed to operate to delivery a suitable power profile. Certain optional profiles as discussed below. As such, the subject power source may include such standard features as overrides and shutoffs as recognized by one with skill in the art as prudent. Further, the systems may be adapted to deliver power on one or more channels. In other words, where two electrolytic components are employed, it will be possible to power one component and corrode its member until deployment, check or monitor such deployment as evidenced by a drop-off in current, and then power another electrolytically erodable member in a staged or sequential fashion. Alternatively, every one or a selected number of electrolytic members may be powered simultaneously. However, a number of advantages as presented below are available as a result of staged deployment.
Still further, (in a configuration not shown) the handle may include one or more of a lever, slider, trigger, knob, wheel, etc. for release of any mechanical features as may be employed in a hybrid mechanical/electrical approach as referenced above. Furthermore, a removable interface member 120 may be provided to facilitate taking the handle off of the delivery system near or proximal end 122. The interface may be lockable with respect to the body and preferably includes internal features for disengaging the handle from the delivery guide. Once accomplished, it will be possible to attach or "dock" a secondary length of wire 124 on the delivery guide proximal end, allowing the combination to serve as an "exchange length" guidewire, thereby facilitating changing-out the balloon catheter or performing another procedure. Alternatively, a core member within the system may serve as an exchange-length (i.e., 300 cm) wire.
Irrespective of such options, common to the delivery guide systems is the inclusion of an electrolytically erodable release latch that releases at least one side of the implant carried by the delivery guide. A number of examples are provided below. Sections of systems are shown that can be mixed-and-matched with others (both in configurations shown and others as may be apparent to one with skill in the art).
Before describing these systems, however, it is noted that
In support of implant delivery, it is to be understood that various radiopaque markers or features may be employed in the system to 1) locate stent position and length, 2) indicate device actuation and stent delivery and/or 3) locate the distal end of the delivery guide. As such, platinum (or other radiopaque material) bands, use of such material in constructing various elements of the subject systems, and/or markers (such as tantalum plugs) may be incorporated into the system.
Stent Loading
In certain variations of the invention, the stent is loaded upon the delivery guide in a twisted configuration. Essentially, a cage or lattice/mesh type stent assumes a decreased diameter through twisting by pulling each portion of the body of the stent across its diameter inwardly compressing the body. When the stent and delivery system offer an interlocking or keyed interface, loading of the stent is easily accomplished by engaging the interface and then pulling (thereby causing the stent to "neck" down) and/or twisting the stent into a compressed profile by rotating the seat member in receipt of the mating portions of the stent about a core member of the delivery guide.
Another approach according to the present invention is suitable for use both in instanced in which a keyed or a slide-out stent and delivery guide interface is provided. In this method, the stent is compressed by hand, with an automated "crimper" such as produced by Machine Solutions, Inc., or otherwise, without a substantial twist imparted thereto. The stent may be compressed by virtue of the act of loading it into a tube, or loaded into a tube after being compressed by a machine. In any case, the tube or sleeve that it is loaded into will generally be close in diameter to its final size when secured upon or the delivery guide. By "close" in diameter, what is meant is that it is within at least about 33%, or more preferably within about 25% to about 10%, or even within about 5% or substantially at its final diameter. Then, with the stent so-constrained, it is twisted from either one or both ends before of after partial or full attachment to the delivery guide.
The sleeve may comprise a plurality of separate pieces or segments (most conveniently two or three). As such, the individual segments can be rotated relative to one another to assist in twisting the stent. In addition, axial manipulation of the relation of thin individual segments can be employed to allow the implant to bulge outwardly over one section. The foreshortening caused by this action may then allow positioning and then axially loading end interface members by manipulating the segments to collapse the bulging.
As for the specific example of loading,
Then, as shown in
The clamps may comprise part of a simple twist fixture supporting chucks aligned on bearings, etc. In any case, in
Following the twisting of the stent within the tube, the distal seat 202 is secured from counter rotation by glue or solder joint 170. Finally, clamps or chucks 164 and 166 are released and restraint 160 is cut or slid off of the delivery guide body to ready the system for stent deployment as shown in
Note, however, that the act of restraint 160 removal may take place even in the operating room as a final step prior to delivery guide use. Otherwise, it may occur as some step along the manufacturing process. When employed in the former manner, sleeve 160 will then do double duty as a loading and a storage sleeve.
Delivery Guide Implant Retention and Release Features
The delivery guide further comprises a number of wires or ribbons 182 passing through at least a portion of the implant 82 to secure it to the delivery guide. These members may form loops of material as shown. In any case, an erodible section 184 (indicated by the circled portion on each elongate member) serves as a releasable latch. When corroded away, the wire passing through receptacle portions at or adjacent the stent crowns are released, allowing the implant to expand.
The wire including the sacrificial material latch section may be steel or stainless steel wire. To define the sacrificial region, polyimide insulation or a protective layer of noble (or more noble) metal such as platinum or gold covered other portions of the material is stripped off (or never laid-down in the first place via a masking process) that section. The wire may be about 0.001 inch in diameter, with insulation or protective metal removed from about a 0.005 inch long section of the wire to provide the erodible section. Stainless steel wire will generally be selected for its strength and because it offers corrosion resistance "on the shelf" while being erodable in an electrolytic solution under power. Other material selection and construction options are discussed below.
As for configuration, the latch members may comprise portions of a loop of wire, as shown in
However situated, leads 186 may be employed for connection to discrete channels or circuits of a power supply to provide individual control over corrosion of the wires. Such a setup may be desired in order to first release the distal side of the implant and then release the proximal side.
Otherwise, leads 186 may be omitted and body 180 will itself serve as the electrical conductor, in which case it will likely be covered with an insulation layer 188 or be constructed of titanium as discussed below. When the body serves as the conductor, it will be connected to the live or "hot"/positive side of a power supply. Then a hypotube outer body member or separate catheter (such as the catheter described in U.S. Pat. No. 6,059,779 to Mills) in receipt of the delivery guide can will offer the ground or negative side of the circuit. Still further, an external ground pad could be applied to the patient as known in the art. See, e.g., U.S. Pat. No. 6,620,152 to Guglielmi. When the near and far erodable sections are not electrically isolated from one another, their release may still be staged by virtue of difference in the volume or type of material to be eroded as well as by proximity of such material to "ground".
In another delivery guide configuration, the elongate body/wire may be solid and/or comprise a number of pieces 182/182′ as shown in
For use in such a system, various stent end receptacle 190 configurations are shown in
Overall, the system shown in
The delivery guide portion 108 shown in
As in the previous examples, the ribbon includes an erodible section 184. It is not connected to the stent, but it may overlay the same. Upon release of the ribbon by erosion of the erodible section (exposed to corrode by stripped insulation, etc.) the ribbon at least partially unwinds or unravels to allow projections 90 to release and the stent body to untwist and expand.
The coil, ribbon or segment that cover or overlays the interlocking feature(s) of the stent and delivery guide seat(s) serve to secure the interlocking features in engagement such that those associated with the stent cannot lift out (or be driven out) of their mates. The wrap may subtend only a small arc across the body of the delivery system or it may wrap around several times. In those several wraps, the same member may cross over the engagement features multiple times. Still, only a portion of the wrap need overlay the engaged features. Conversely, sections of more than one capture member may wrap over one or more sets of engagement features.
By including a wrap over the interlocking or interfitting features, their orientation is stabilized relative to opposing surfaces. This characteristic is of particular value when the implant secured has a torsional load or preload upon it. Even where torsional load bearing characteristics are not required, the configuration offers potential for very low-profile assembly. Strong flat ribbon can be utilized as the capture member(s). Alternatively, multiple strands of wire wound in parallel can offer the requisite strength while maintaining a low profile.
Further, note that the system overview in
While in this variation of the invention at least one of the ends of the stent will be held by an electrolytically releasable latch, one side may be mechanically released. In the case shown in
In any case, one side of the implant may include features of one type and the other side another. There may even be instances in which a mix of different projection types may be desired on a single side of the implant.
As for special features that the projections may possess, certain ones may incorporate a tantalum or platinum plug 210 for radiopacity (such as the variation in
It is understood that the shape and configuration of the projection may vary for any number of factors (e.g., the particular application, the size of the stent, the tortuousity of the vasculature, etc.). As a result, the projections may comprise any hook, prong, opening, socket, key, grasper, tooth, bar or slot shaped configuration (whether the shape is nearly planar or the shape significantly extends in three dimensions) in addition to those shapes discussed herein. In accordance with the above, it follows that the corresponding seating features will be selected to receive or accommodate the shape of the respective projection. In any case, the shapes of the projections will be selected so as not to create undue risk of injury to the patient. For example, for vascular applications, the shape of the projections must be chosen so as not to create undue trauma to the vessel wall. On the other hand, non-vascular applications may not present the same risks. Accordingly, the design of the projections may be more aggressive for stents or other implants intended for such applications.
In all, the various implant/delivery guide interface features shown include captured types (via hook-in type or keyed interfaces or those that accommodate a member passing into or through an interior space) and slide-out types. The captured types offer an axial retention component at minimum. To a varying degree, they also offer lateral or radial support, where the interlocking interfaces excel in such regard. Elongate members adapted to slide out of an interface when desired are designed (together with their respective seat features) primarily to offer lateral or radial support—especially for twist-down type applications as follow.
As for more specific features pertaining to the construction of the wrapping covers to secure such relevant features,
As with the variations above, where the latch members are not isolated in discrete circuits, other parameters (e.g., ribbon thickness or width at the erodable section, etc.) may be varied to manipulate release timing. And, again, because more of the system is positively charged (as opposed to the releasable members, alone) the system may require insulation to focus erosion and/or avoid current leak resulting in higher power requirements, etc.
To electrically isolate seat 200, ribbon 206 and any uninsulated stent loaded onto the delivery guide from core member 180 and/or hypotube 210, insulation material 210 is employed. As shown, the insulation material may comprise a plurality of tube or sleeve segments "A, B, C" in order to facilitate twisting the stent in-situ into a minimal diameter. Regardless of the manner in which it is electrically isolated, the core member and/or hypotube can then be employed as "ground" or negative component in the delivery system.
Just as any of a variety of electrical architectures can be adopted, so too are there different approaches to constructing the cover over the interface of the mating portions of the stent.
As referenced above, the invention also contemplates systems with electrolytic release on one side of a stent and some other form of release on the other. One example is shown in
While hybrid mechanical systems may certainly be employed in certain variations of the invention, another class advantageously employs two electrolytically released assemblies at the near and far end of the implant to be delivered.
Here, a distal seat 202 is situated over a hypotube core member 250. An electrical lead wire 252 passes through the core member from a proximal end of the delivery guide to emerge to meet or define wrap section 254. Wire 252 may be a small gauge copper conductor and wrap section will comprise stainless steel with an insulative coating except at the erodible sacrificial section 184. Alternatively the same piece of stainless steel wire may serve as the electrical lead and length including the sacrificial section. The number of turns of wire 254 around the delivery guide body may vary from as few as a fraction of a turn to many (as shown). Still, at least one turn around the body may be desirable to direct force along the axis of the wire and/or take advantage of friction between the wire and delivery guide to hold position of the wire
Regardless, one end of the wire may be connected to an insulative layer 256 between core member 250 and band 260. In this manner, erodible section 184 is isolated from the delivery guide body, which in operation of this variation of the invention serves as "ground" or "negative" to the extent applicable in view of the power profile selected. Band 260 is secured to seat 202 by sleeve 262. The members may be metallic and laser-welded together as indicated or otherwise connected. A blocker 264 under sleeve prevents axial movement of untwisting assembly 266.
To release the stent received within seat 202, sacrificial latch section 184 is eroded thereby untwisting assembly 266 rotates. The rotation causes the associated stent to untwist and expand. The expansion results in foreshortening that pulls the stent's distal projections out of capped slot 268.
Untwisting assembly 266 is advantageous, among other reasons, because it requires no translation to operate; the assembly simply rotates to effect release. As such, it offers an extremely robust operation in even the most tortuous anatomy.
Still, a pair of sleeve sections 262 and 262′ may be provided as indicated by broken line. The near section 262′ would then be set to slide back for stent release in a manner similar to the approach taught in connection with
A blocker (like blocker 264) can be set within gap "G" to stabilize band 256 and allow seat 200 to travel in a distal direction to reduce the ribbon diameter upon withdrawal. Another blocker can be set distal to seat 200 to limit such travel as indicated in phantom line. However, it may be desirable to allow the member to "float" so that, upon withdrawal of the system, the distal movement of seat 200 pulls ribbon 206 into a reduced diameter.
Yet another class of delivery systems according to the present invention employing an untwisting assembly 266 is shown in
Spring-back assembly 280 includes a coil spring 282 or a similar member that is wound-down upon the delivery guide body portion 284 when the stent is fully twisted. As wound, the position of the spring is locked. However, when untwisting assembly 266 allows the stent to unwind, spring 282 follows, and its coils expand. As the coils expand, the spring withdraws seat 200. As such, it pulls seat 200 off of the stent some distance "D". One or more blocker members can be provided to maintain the proximal position of the stent for this withdrawal. Alternatively, the spring may withdraw sleeve 262 alone such that the seat is left uncovered by action of the spring.
As indicated by the broken line, the restraint 284 may cover the whole stent or only some portion thereof. Where only a partial restraint sleeve is used, the stent will be twisted down in diameter distal to the edge of the restraint. It may also lie twisted under the restraint. When a restraint covering the entirety of the stent is employed, the stent may or may not be twisted—though at least the restraint and coil assembly will be twisted relative to the delivery guide body and assembly 266 to effect the lock/unlock approach described.
An advantage of such a system is to reduce the force by which the stent pushes against the sleeve, thereby lowering (at least initial) sleeve withdrawal forces. After initial movement of the sleeve, the far end of the stent may be released so that the expansion of the stent helps drive the sleeve off the body 88 of the stent. In which case, blocker 286 will be necessary to stabilize the near end 84 of the stent in completing sleeve 290 withdrawal. In another approach, the ribbon is not electrolytically released until after complete sleeve withdrawal.
Erosion Control
In describing the various systems above, some discussion was offered as to which elements are insulated from one another or shielded from corrosion by noble metal. Certain details regarding the same are readily understood by those with skill in the art. Generally speaking, however, numerous ones of the above systems illustrate separate lead wires to actuate the electrolytically erodible release means. Likewise, many of the systems include separate lead wires to actuate the proximal electrolytically erodible release means. By providing separate circuits, especially in instances in which the body of the device serves as a return path, the latches can be actuated individually. Their action can be monitored. When current no longer flows on a given circuit, positive indication is offered that the subject latch has been released. Furthermore, the latches can be released individually without unintended early release of one relative to the other as is possible in systems which simply rely on the volume of material to be eroded. Another beneficial factor is that by eroding one latch at a time, current is limited relative to a system in which multiple sections of material would be eroded at once.
In certain variations of the invention, isolation of one member from another in order to isolate their respective polarities is accomplished through the use of interposed polymer tubing or coatings. However, one aspect of the invention contemplates systems in which little insulation is employed, and rather system components intended to be protected from corrosion are made from titanium. Those members which are to be set for erosion are to be made of stainless steel or another suitable material. The sacrificial material may be a section of material connected to or interposed between one or more titanium sections of material by laser welding, resistance welding, soldering, etc. Otherwise, some portion of the system may comprise titanium, while the ribbon, wire or other member including a sacrificial section consist substantially of one material (such as stainless steel) and the rest is covered by insulation or a noble metal protective coating.
Other components not to be eroded are made of anodized titanium. With the anodized layer, the material is far less susceptible to erosion. What is more, upon application of voltage, small nicks or scratches in the titanium will passivate (become anodized) given a voltage applied of about 3V or more. Some current drain will be experienced, but the major site for ion transfer will be at the section intentionally left bare.
Electrical Safety Architecture
In yet another aspect of the invention bi-polar systems are provided in which the referenced "hot" and ground sections are in close proximity to one another. By limiting the distance over which ion travel must occur to transfer charge / pass current, the spread of electrical lines of flux are minimized. The graphic of
Electrical Performance
Electrolytic erosion of a section of metal on a delivery system element to release an implant is driven by applying voltage to develop a positive charge on the element resulting in a strong motive force to cause current to flow to a negatively charged body. The mechanism by which current flows is ion transfer from the section to be eroded. Further discussion of this process is described and/or relates to the mode of release in various patents including U.S. Pat. No. 5,122,136 to Guglielmi; U.S. Pat. No. 6,716,238 to Elliot; U.S. Pat. No. 6,168,592 to Kupiecki, et al.; U.S. Pat. No. 5,873,907 to Frantzen and the multiplicity of continuation, continuations-in-part and divisional applications related to these patents.
In the present invention, a DC voltage component is likewise applied to effect corrosion/erosion of the implant release means. And while adding an AC voltage component for sensing purposes is known (e.g., as described U.S. Pat. No. 5,569,245 to Guglielmi, et al.;U.S. Pat. No. 5,643,254 to Scheldrup, et al.) the invention hereof does so and uses the AC voltage in a very different manner.
Specifically, it has been appreciated that use of significant AC component offset by a DC signal can dramatically improve the process of implant delivery through electrolytic corrosion. Not to be bound by a particular theory, but it is thought that efficiency gains are related to controlling blood electrocoagulation and to having periods of higher peak voltage during the upsweep of the AC signal.
This AC component is especially beneficial in coronary therapy because high frequency (e.g., 10 kHz to 100 kHz or greater) AC power does not effect heart rhythm unless the waveform becomes unstable. Controlling electro-coagulation is very important for safety reasons (in avoiding emboli formation that could lead to stroke or other complications) and also to increase the speed of corrosion.
Generally speaking, while corroding a positively charged section of metal, the positive charge attracts negatively charged blood cells which coagulate on the surface of the metal. The coagulated blood cells cover the corroding metal and slow the deployment process. Higher DC levels can be employed to push past this effect, but for safety considerations (especially in the vicinity of the heart) it is desirable to use lower DC voltages.
Instead, when an AC signal is employed that drops the trough of the waveform into the negative regime, an opportunity exists to repel the negatively charged blood cells. The resulting decrease or lack of electrocoagulation offers an efficiency increase so that DC voltage can be dropped while maintaining deployment times that are subjectively acceptable to a medical practitioner (e.g., less than about 1 minute or about 30 seconds).
Power is preferably delivered to the subject systems employing a custom battery-powered power supply. Still, various function generators may be employed for experimental purposes. A Fluke model PM 5139 Function Generator proved adequate in practicing the subject invention. A square wave function is most advantageously employed to maximize the time spent at peak and minimum voltage levels, but sinusoidal, saw-tooth, and other variations of these forms may be employed. Still further, frequency modulated waveforms in which more or less time is spent in the positive or negative regimes may be employed.
Regardless of how it is generated,
In porcine blood, it was determined that a peak waveform voltage of above 8V begins to cause electrocoagulation, even with trough voltages of −6 to −7V. The level of electrocoagulation varies with the level of the DC component and the size of the piece of metal to be eroded, but usually the peak voltage should remain below 9V and most often below 8V to avoid appreciable electrocoagulation.
In view of the above, and further for safety reasons—especially in the vicinity of the heart—it may be desirable to maintain the DC component of the power applied between about 1 and about 5V, and more preferably between about 1.75 and about 3V, and possibly most preferably between about 2 and about 3V. The AC waveform employed will generally then be selected to generate a peak below about 9V and usually below about 8V, with 7-7.5V being typical per the above. Accordingly, the resultant power profile may have a peak or maximum between about 4 and about 9V, and a minimum of about−0.5 to about −5V. Within this range (and in certain circumstances, outside the range, given situations where some amount of electrocoagulation is acceptable), more effective combinations exist as detailed herein and as may be apparent to those with skill in the art in review of the present disclosure.
The impact of AC voltage on actual erosion/corrosion rates during bench tests of tensioned 0.002" stainless steel wire was conducted. Setups were provided in which an insulated wire was equally tensioned and exposed along a 0.020 inch long section. The wires were placed in 38� porcine blood and power was applied. When applying 2V DC, it took 3-4 minutes to break the wire. When applying 2V DC and 10 Vpp AC, time to separation ranged from 20-30 seconds. The setups tested under DC-only conditions were observed to generate roughly 0.040 inch balls of electrocoagulation on the ends of the wire opposite the eroded section. In marked contrast, the AC/DC power driven setups showed no visible electrocoagulation.
The same test piece setup described in Example 1 was used with a lower DC voltage. With only 1V DC the wire would not break even after 15 minutes of applied power. When a 10 Vpp AC signal was added to the 1V DC signal, the sample test section broke in roughly 1 minute.
Tests were conducted to determine the improvement offered over the power supply provided by Target Therapeutics for detatching GDC� coils. First a comparative model was developed. The electrolytic "joint" in a GDC system was determined to be about a 0.005 inch long, 0.003 inch diameter stainless steel wire. In 38� porcine blood, with the Target Therapeutics power supply set at a 1 mA currently delivery setting, voltage metered by the power supply initially showed at 3V, rose to 6.5V for the majority of the deployment time, and then rose to 8V. Over a deployment time measured at 40 seconds, the average voltage observed was about 6.5V. In addition, a ball of electrocoagulation having about a 1/32 inch diameter was observed.
A "test joint" model was developed to compare a number of samples in performance. It employed a roughly identically sized exposed wire extension as described above, but no occlusive coil attached thereto. In eroding the wire extension with the Target Therapeutics power supply set at a constant 1 mA current, significant variability about the above referenced GDC test results was observed. Voltage varied over a greater range from 1.9V to 9V. Time to complete erosion ranged from 40-50 seconds irrespective of such variance. Average voltage was about 4V. Electrocoaguation was observed on all samples. In instances where voltage climbed to about 6V, similarly large 1/32 inch diameter balls of electrocoagulation were observed. When applying a 2.5V DC with 10 Vpp AC signal to test joint systems, current floated between 0.5 and 0.75 ma. Deployment times were consistently about 50 seconds. No electrocoagulation was visible upon inspection.
The invention includes methods that may be performed using the subject devices or by other means. The methods may all comprise the act of providing a suitable device. Such provision may be performed by the end user. In other words, the "providing" (e.g., a delivery system) merely requires the end user obtain, access, approach, position, set-up, activate, power-up or otherwise act to provide the requisite device in the subject method. Methods recited herein may be carried out in any order of the recited events which is logically possible, as well as in the recited order of events.
Exemplary aspects of the invention, together with details regarding material selection and manufacture have been set forth above. As for other details of the present invention, these may be appreciated in connection with the above-referenced patents and publications as well as generally know or appreciated by those with skill in the art. For example, one with skill in the art will appreciate that a lubricious coating (e.g., hydrophilic polymers such as polyvinylpyrrolidone-based compositions, fluoropolymers such as tetrafluoroethylene, hydrophilic gel or silicones) may be placed on the core member of the device, if desired to facilitate low friction manipulation. The same may hold true with respect to method-based aspects of the invention in terms of additional acts as commonly or logically employed.
In addition, though the invention has been described in reference to several examples, optionally incorporating various features, the invention is not to be limited to that which is described or indicated as contemplated with respect to each variation of the invention. Various changes may be made to the invention described and equivalents (whether recited herein or not included for the sake of some brevity) may be substituted without departing from the true spirit and scope of the invention. In addition, where a range of values is provided, it is understood that every intervening value, between the upper and lower limit of that range and any other stated or intervening value in that stated range is encompassed within the invention.
Also, it is contemplated that any optional feature of the inventive variations described may be set forth and claimed independently, or in combination with any one or more of the features described herein. Reference to a singular item, includes the possibility that there are plural of the same items present. More specifically, as used herein and in the appended claims, the singular forms "a," "an," "said," and "the" include plural referents unless the specifically stated otherwise. In other words, use of the articles allow for "at least one" of the subject item in the description above as well as the claims below. It is further noted that the claims may be drafted to exclude any optional element. As such, this statement is intended to serve as antecedent basis for use of such exclusive terminology as "solely," "only" and the like in connection with the recitation of claim elements, or use of a "negative" limitation.
Without the use of such exclusive terminology, the term "comprising" in the claims shall allow for the inclusion of any additional element—irrespective of whether a given number of elements are enumerated in the claim, or the addition of a feature could be regarded as transforming the nature of an element set forth n the claims. Except as specifically defined herein, all technical and scientific terms used herein are to be given as broad a commonly understood meaning as possible while maintaining claim validity.
The breadth of the present invention is not to be limited to the examples provided and/or the subject specification, but rather only by the scope of the claim language. That being said, we claim:
|Cited Patent
|Filing date
|Publication date
|May 10, 2001
|Apr 6, 2004
|Scimed Life Systems, Inc.
|Stent with detachable tethers and method of using same
|May 10, 2001
|Dec 12, 2002
|Elliott Christopher J.
|Stent with detachable tethers and method of using same
|Nov 1, 2002
|May 6, 2004
|Marc-Alan Levine
|Method and apparatus for caged stent delivery
|Dec 31, 2002
|Jul 1, 2004
|Marc-Alan Levine
|Bifurcated guidewire and methods of use
|Dec 24, 2003
|Sep 30, 2004
|Cardiomind, Inc.
|Balloon catheter lumen based stent delivery systems
|Dec 24, 2003
|Nov 4, 2004
|Cardiomind, Inc.
|Implant delivery technologies
|Dec 5, 2003
|Jun 9, 2005
|Scimed Life Systems, Inc.
|Detachable segment stent
|May 3, 2005
|Sep 1, 2005
|Boston Scientific Scimed, Inc.
|Sacrificial anode stent system
|Citing Patent
|Filing date
|Publication date
|Nov 18, 2004
|Jan 26, 2010
|Cardiomind, Inc.
|Corewire actuated delivery system with fixed distal stent-carrying extension
|Nov 2, 2005
|Aug 10, 2010
|Ton Dai T
|Twist-down implant delivery technologies
|Mar 23, 2004
|Aug 31, 2010
|Julian Nikolchev
|Implant delivery technologies
|Dec 20, 2005
|Jan 4, 2011
|Biosensors International Group, Ltd
|Indirect-release electrolytic implant delivery systems
|Dec 24, 2003
|Sep 13, 2011
|Biosensors International Group, Ltd.
|Guidewire-less stent delivery methods
|Nov 30, 2010
|Sep 25, 2012
|Biosensors International Group, Ltd.
|Indirect-release electrolytic implant delivery systems
|Jan 27, 2006
|Nov 12, 2013
|Biosensors International Group, Ltd.
|Untwisting restraint implant delivery system
|Jun 26, 2009
|Feb 25, 2014
|Biosensors International Group, Ltd.
|Implant delivery apparatus and methods with electrolytic release
|Jan 27, 2006
|Dec 2, 2014
|Biosensors International Group, Ltd.
|Covering electrolytic restraint implant delivery systems
|Jan 27, 2006
|Mar 10, 2015
|Biosensors International Group, Ltd.
|Pass-through restraint electrolytic implant delivery systems
|Sep 29, 2005
|Mar 29, 2007
|Chang Jean C
|Stent delivery system
|U.S. Classification
|International Classification
|Cooperative Classification
|European Classification
|May 5, 2006
Owner name: CARDIOMIND, INC., CALIFORNIA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNOR:LICATA, DAVID;REEL/FRAME:017580/0776
Effective date: 20060417
|Jul 6, 2010
Owner name: BIOSENSORS INTERNATIONAL GROUP, LTD., BERMUDA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNOR:CARDIOMIND, INC.;REEL/FRAME:024640/0037
Effective date: 20100617


text:
Hi All,Just curious if anyone uses the REI One gloves and/or Cloudveil Icefloe gloves and could tell me the pros and cons of each. I'm also wondering what the actual weight of the REI One Glove is as I can't find that anywhere on the web. Additionally, if anyone knows the weight of Joe Trailman Trail Gaiters I'd love to know that little piece of trivia too.
I'll throw mine on the scale and tell you tonight. I own the REI glove size XL. I don't own the cloudveil.
They're ok. I bought some when REI had them cheap, about $15/pair. I use to use them quite a bit but I prefer: Lightweight capilene glove liners, latex gloves, homemade goretex mittens, and homemade fleece mittens: all of which together weighs 104grams and provide far more flexibility and weather protection.Craig Shelley
My size large REI One gloves weigh 2.8oz for the pair. I don't really use them as they do not fit my hand well. The thumb is not "boxed" and so there is a bit of floppy fabric on the tip that gets in the way when I need dexterity. If they fit you though, the weight and fabric make for a great liner/ stand alone cool weather glove.Edit: I had the Icefloes and sold them. Very nice glove (twice as heavy as the Ones but burly as heck) but the OR Contact Gloves fit me better. Mine weigh 5.1 oz in XL and are what I wear under Alti-Mitt shells in winter. I can do just about everything in them from putting on crampons to lighting the stove. The Contacts and Icefloes are just about the same and will be more durable than the Ones with a bit more warmth. The Ones will work better as a liner inside of a pile mitten and shell though.
Edited by febbom on 10/13/2006 15:08:36 MDT.
I also have the rei gloves and find they do not dry quickly once wet thru [as a lot of soft shell materials seem to also].Even though you haven't asked about them I have a pair of the new or alibi gloves and I just want to mention they run a size small.
The XL are 3 ounces. I'm a big fan of the Pittards palm. The pittards lends the gloves good dexterity. They are low profile and don't interfere with the strap for my poles. The stretch allows for a snug fit with or without liners and I can tie my shoes/boots with them on. They don't have a ton of insulation, so they are not my first choice for severe cold. I primarily use them for aerobic activities.
Edited by Genetic on 10/13/2006 22:09:02 MDT.
To add some input for your decision.. I own the REI one, and they wet through quickly, and the thick seams running around each finger are annoyoing and interfere with dexterity. The leather palms and stretch fabric seemed like a good combination when I bought them. Can anyone suggest a reasonably water resistant, breathable glove which fits snugly? The goretex types Ive found have too much loft for technical finger work such as tying knots. Thanks in advance.
I've had the REI One gloves for a few years and I like them a lot. They give me more dexterity than any other glove of comparable warmth and wind/water resistance. The only problem I have is that they dry slowly, although that has more to do with the leather than the Power Shield fabric. My gloves are a bit different than the One Gloves that REI has sold for the last year or two.I also have a pair of OR Moraine gloves, which are good except that the fingers are too loose fitting for me to have much dexterity. This year I think I'm going to try the OR Airfoil gloves with or without a liner. The Airfoils are very water resistant but don't have much insulation and don't give me quite as much dexterity as the Ones.Of course, the fit of any pair of gloves is a completely individual matter. I have long, thin fingers, so a lot of gloves that otherwise fit well have too much room in the fingers for me to maintain good dexterity.
Edited by joe_r on 10/15/2006 21:49:27 MDT.
I tied a knot in the string from a teabag today while wearing the REI One glove. I can't imagine more dexterity in a glove. I've probably owned 20-24 pairs of gloves in the last 3 years if I count cycling specific (PI Gel Lite, etc). JMO.
You can buy youself a little more water resistance by working glove-pruf or snowseal into the palms.
I'd been thinking about trying snowseal on the leather of my gloves. Does it affect the grip at all? I was concerned that the leather might become a little bit slippery.
Edited by joe_r on 10/16/2006 19:54:49 MDT.
I usually work snoseal into a pair of deerskin gloves for work and I don't find it slippery handling conduit,tools,or climbing metal rungs. It is a lot less slippery then wet, slimy gloves. The other product glov-pruf is usually sold in ski stores [and I may not have that name quie right}.
I searched for over a decade for a glove shell with a grippy, durable palm to wear over a liner for cold weather rock climbing. OR finally perfected it with the 2.8oz medium (with wrist cords removed) Santa Ana and then true to form, discontinued it.I now find that I like the 2.3oz med REI One glove even more for the same application (I wear it alone w/o liner) although I'm sure it has many other uses also. I recently used them while clearing snow out of cracks on Whitney's East Buttress Route at 13,200' in October. I have very cold hands, so that's quite a testimonial.The $46 '05 version that I own is on sale for $27.93 at rei-outlet.com so I bought 3 more for spares.
This thread has been really helpful in thinking about my glove choice. I've gotten the REI One Gloves (on sale from REI-Outlet- $30) but I wonder if it is too much glove for the 40-60 degree temperature range. What do people wear when they want to keep hands from getting chilled in these temp ranges (say it's overcast)? I've been really impressed by reviews for OR Airfoils and Manzella Silkweight Windstopper gloves but are they about the same as the REI One gloves for temperature range? Would a simple pair of Icebreaker/Ibex/Arc'Teryx merino glove liners be a better choice? Thanks for your input.
I have the One Gloves and recently got a pair of Airfoils. The Ones are definitely the warmer of the two. If you just want protection from wind and moderate rain then the Airfoils would probably be fine. The way the Airfoils fit me, I can comfortably wear a light liner underneath when I need more insulation. The Ones don't allow me to wear a liner, but their tighter fit gives me a better grip and more dexterity.
I've use both REI ONE and Manzella Silkweight. The Manzella is my favorite, went through 2 pairs and will be getting a third as it seems I always lose them b/w seasons. The finger tips seem to wear thin though after a good 3 seasons of use, running, biking, climbing, snowboarding. I got the REI One because my hands got really cold in Jan/Feb winter running at night. The REI was toasty in these ocassions but less dexterous. They are less versitile for me, but work well as needed.
Gloves are the one component of my clothing system I just can not seem to 'dial in'. Maybe because they have demanding requirements such as pressurized contact with wet surfaces, dexterity, etc.. My requirement is simply for a waterproof breathable glove with some dexterity which I could combine with various inner layers. The REI one gloves wet through in a matter of seconds, and so did my Montbell Drytec gloves. I would rather avoid bulky insulated "ski" type gloves, but that seems to be the norm when you upgrade to goretex. I have a pair of seal-skinz on order to test out, but does anyone else have a suggestion for a thin, light, WP/B non-insulated glove? Thanks in advance, and sorry, this is very similar to a previous post of mine.Thanks.
Mountain Laurel Designs will be coming out with uninsulated eVENT mittens fairly soon.
Tee Hee!http://www.rei.com/product/787283
Another ancient thread resurrected.
You must login to post.
MEMBERSHIP IS REQUIRED TO POST: You must be a Forum, Annual or Lifetime Member to post messages in the backpackinglight.com forums.

subset: commoncrawl


text:
Event Registration
We offer full service registration or individual services including Online Registration, Advance Registration and Onsite Registration. Read More
Custom E-Vites
Had enough with those "free" e-vite systems that are more work than help? Our e-vite system offers truly custom designs and business class functionality. Read More
Lead Retrieval
Help your exhibitors get the highest return on their tradeshow investments by giving them tools to quickly and effectively follow up on their leads. Read More
E-vents Registration, a full service registration vendor. Our services include:


text:
By Aldo Panessidi
Today Apple can seemingly do no wrong. But if you think back a couple of years, you will likely recall the fuss made around the iPhone4 and the unheard of design flaw that had jaws flapping. Honestly, it had been ages since the incident had crossed my mind. And unfortunately for Apple, Antennagate has reared its ugly head once again.
The silver lining for the tech darling is that this time the reminder comes in the form of a bookend on the class-action lawsuit that had been levied against Apple after the 2010 iPhone 4 release that caused some serious problems for customers trying to actually use their iPhones to speak with others. Some users noticed that when the strip along the side of the handset was covered it would result in a loss of signal and eventually a dropped call.
Enhance your mobility and turn your device into a power tool! Discover select accessories such as; stylus & pens, micro SD memory cards, batteries, docking stations, Bluetooth headsets, data cables and glider gloves all specifically designed to make your devices functionally efficient and performance rich.
To stem the tide of a disgruntled public, Apple offered free Bumper cases for iPhone4 buyers. Alas, they were only available through September 30, 2010 which did not appease those who had waited in line and paid a premium for the long-awaited iPhone4 – hence, the multiple lawsuits filed against Apple.
So those iPhone customers who felt cheated because of the time limit on free bumper cases will be pleased to hear that the class-action lawsuit against Apple translates into either a free Bumper case or $15 cash to iPhone4 customers in the United States. Personally, my iPhone4 is already enshrined in the protective case of my choosing. But, the $15 cash is a great incentive to buy the new running accessories or headset I've had my eye on.
To claim the reward of your choice, visit www.iPhone4Settlement.com (it is not yet up and running). The website will provide information on how customers can receive their free case or get the cash in hand. In case you forget, Apple will be required to send reminder emails to iPhone4 users prior to April 30th.
Thankfully this design flaw was addressed with the iPhone 4S and the tech company continues to bring its signature innovation and style to consumers with products like the iPad that have made it the success story to date and an industry leader, not a follower.