text:
US 7922732 B2 Abstract Medical devices for filtering fluids flowing through a lumen and a method of forming medical devices. The devices can be used in vascular channels, urinary tracts, biliary ducts and the like, and filter emboli and other debris generated at a treatment site.
Claims(21) 1. A system for filtering material from fluid flowing through a lumen defined by the wall of an anatomical structure comprising:
a guidewire;
a filter element carried on the guidewire, the filter element having proximal and distal ends and being formed of a self-expanding porous fabric material, the fabric material being configured to be expandable within the lumen of the anatomical structure from a collapsed configuration when the filter element is restrained to an expanded configuration when the filter element is unrestrained without the aid of other mechanically expandable or self-expandable structure, the filter element having a shape in the expanded configuration which defines a cavity having a proximally facing opening, the guidewire extending to at least the distal end of the filter element; and
a retrieval catheter having a distal portion defining a lumen, the retrieval catheter being configured and sized such that during use of the system the lumen of the retrieval catheter accommodates the guidewire and at least the proximally facing opening of the filter without significant loss of material during retrieval of the filter from the lumen of the anatomical structure.
2. The system of
3. The system of
4. The system of
5. The system of
6. The system of
7. The system of
8. The system of
9. The system of
10. A system for capturing thrombi removed from the wall of a vessel at a site in the lumen of the vessel comprising:
a guidewire;
an intravascular device carried on the guidewire, the intravascular device having pores and having proximal and distal ends and being formed of a self-expanding porous fabric material, the fabric material being configured to be expandable from a collapsed configuration when the intravascular device is restrained to an expanded configuration when the intravascular device is unrestrained without the aid of other mechanically expandable or self-expandable structure, the intravascular device having a shape in the expanded configuration which forms an interior portion having a proximally facing opening defined by a lip which is sized to expand to the size of the lumen of the vessel when the intravascular device is deployed in the vessel, the proximally facing opening being sized to allow the thrombi to enter the interior portion of the intravascular device, the pores being configured to retain the thrombi in the interior portion of the intravascular device; and
a retrieval catheter having a distal portion defining a lumen, the retrieval catheter being configured and sized such that during use of the system the lumen of the retrieval catheter accommodates the guidewire and at least the proximally facing opening of the intravascular device without significant loss of thrombi during retrieval of the intravascular device from the lumen of the vessel.
11. The system of
12. The system of
13. The system of
14. The system of
15. The system of
16. The system of
17. The system of
18. The system of
19. The system of
20. The system of
21. The system of
Description This application is a divisional of application Ser. No. 11/098,898, filed Apr. 4, 2005 now U.S. Pat. No. 7,404,820, which is a continuation of application Ser. No. 10/051,591, filed Jan. 18, 2002, now U.S. Pat. No. 6,989,019 B2, issued Jan. 24, 2006, which is a continuation of application Ser. No. 08/748,066, filed Nov. 12, 1996, now U.S. Pat. No. 6,605,102 B1, issued Aug. 12, 2003, which is a continuation of application Ser. No. 08/272,425, filed Jul. 8, 1994, now abandoned, the contents of each of which are hereby incorporated herein by reference. The present invention generally relates to intravascular devices for treating certain medical conditions and, more particularly, provides a method of forming intravascular devices and certain novel intravascular occlusion devices. The devices made in accordance with the invention are particularly well suited for delivery through a catheter or the like to a remote location in a patient's vascular system or in analogous vessels within a patient's body. A wide variety of intravascular devices are used in various medical procedures. Certain intravascular devices, such as catheters and guidewires, are generally used simply to deliver fluids or other medical devices to specific locations within a patient's body, such as a selective site within the vascular system. Other, frequently more complex, devices are used in treating specific conditions, such as devices used in removing vascular occlusions or for treating septal defects and the like. In certain circumstances, it may be necessary to occlude a patient's vessel, such as to stop blood flow through an artery to a tumor or other lesion. Presently, this is commonly accomplished simply by inserting, e.g. Ivalon particles, a trade name for vascular occlusion particles, and short sections of coil springs into a vessel at a desired location. These �embolization agents� will eventually become lodged in the vessel, frequently floating downstream of the site at which they are released before blocking the vessel. In part due to the inability to precisely position the embolization agents, this procedure is often limited in its utility. Detachable balloon catheters are also used to block patients' vessels. When using such a catheter, an expandable balloon is carried on a distal end of a catheter. When the catheter is guided to the desired location, the balloon is filled with a fluid until it substantially fills the vessel and becomes lodged therein. Resins which will harden inside the balloon, such as an acrylonitrile, can be employed to permanently fix the size and shape of the balloon. The balloon can then be detached from the end of the catheter and left in place. Such balloon embolizations are also prone to certain safety problems, though. For example, if the balloon is not filled enough, it will not be firmly fixed in the vessel and may drift downstream within the vessel to another location, much like the loose embolization agents noted above. In order to avoid this problem, physicians may overfill the balloons; it is not uncommon for balloons to rupture and release the resin into the patient's bloodstream. In still other procedures, it may not be necessary to permanently occlude a vessel, but it may be necessary to provide a filter or the like to prevent thrombi from passing a particular location. For example, rotating burrs are used in removing atheroma from the lumen of patients' blood vessels. These burrs can effectively dislodge the atheroma, but the dislodged material will simply float downstream with the flow of blood through the vessel unless steps are taken to capture the material. Some researchers have proposed various traps or filters for capturing the particulate matter released or created in such procedures. However, such filters generally have not proven to be exceptionally effective in actual use. Such filters tend to be cumbersome to use and accurate deployment is problematic because if they are not properly seated in the vessel they can drift to a more distal site where they are likely to do more harm than good. In addition, these filters are generally capable of only trapping relatively large thrombi and are not effective means for removing smaller embolic particles from the blood stream. The problems with temporary filters, which are intended to be used only during a particular procedure then retracted with the thrombi trapped therein, are more pronounced. Even if the trap does effectively capture the dislodged material, it has proven to be relatively difficult or complex to retract the trap back into the catheter through which it was delivered without simply dumping the trapped thrombi back into the blood stream, defeating the purpose of the temporary filter device. For this reason, most atherectomy devices and the like tend to aspirate the patient's blood during the procedure to remove the dislodged material entrained therein. Mechanical embolization devices, filters and traps have been proposed in the past. Even if some of those devices have proven effective, they tend to be rather expensive and time-consuming to manufacture. For example, some intravascular blood filters suggested by others are formed of a plurality of specially-shaped legs which are adapted to fill the vessel and dig into the vessel walls. In making most such filters, the legs must be individually formed and then painstakingly attached to one another, frequently entirely by hand, to assemble the final filter. Not only does this take significant skilled manpower, and hence increase the costs of such devices, the fact that each item must be made by hand tends to make quality control more difficult. This same difficulty and expense of manufacturing is not limited to such filters, but is experienced in many other intravascular devices as well. Accordingly, it would be desirable to provide a method for forming devices for deployment in a patient's vessel which is both economical and yields consistent, reproducible results. It would also be advantageous to provide a reliable embolization device which is both easy to deploy and can be accurately placed in a vessel. Furthermore, there is a need in the art for a trap or filter which can be deployed within a vessel for capturing thrombi, which trap can be reliably deployed; if the trap is to be used only temporarily, it should be readily withdrawn from the patient without simply dumping the trapped thrombi back into the blood stream. The present invention provides a method for forming intravascular devices from a resilient metal fabric and medical devices which can be formed in accordance with this method. In the method of the invention, a metal fabric formed of a plurality of resilient strands is provided, with the wires being formed of a resilient material which can be heat treated to substantially set a desired shape. This fabric is then deformed to generally conform to a molding surface of a molding element and the fabric is heat treated in contact with the surface of the molding element at an elevated temperature. The time and temperature of the heat treatment is selected to substantially set the fabric in its deformed state. After the heat treatment, the fabric is removed from contact with the molding element and will substantially retain its shape in the deformed state. The fabric so treated defines an expanded state of a medical device which can be deployed through a catheter into a channel in a patient's body. In accordance with the method of the invention, a distal end of a catheter can be positioned in a channel in a patient's body to position the distal end of the catheter adjacent a treatment site for treating a physiological condition. A medical device made in accordance with the process outlined above can be collapsed and inserted into the lumen of the catheter. The device is urged through the catheter and out the distal end, whereupon it will tend to return to its expanded state adjacent the treatment site. Further embodiments of the present invention also provide specific medical devices which may be made in accordance with the present invention. Such devices of the invention are formed of a metal fabric and have an expanded configuration and a collapsed configuration. The devices are collapsed for deployment through a catheter and, upon exiting the distal end of the catheter in a patient's channel, will resiliently substantially return to their expanded configuration. In accordance with a first of these embodiments, a generally elongate medical device has a generally tubular middle portion and a pair of expanded diameter portions, with one expanded diameter portion positioned at either end of the middle portion. In another embodiment, the medical device is generally bell-shaped, having an elongate body having a tapered first end and a larger second end, the second end presenting a fabric disc which will be oriented generally perpendicular to an axis of a channel when deployed therein. The present invention provides a reproducible, relatively inexpensive method of forming devices for use in channels in patients' bodies, such as vascular channels, urinary tracts, biliary ducts and the like, as well as devices which may be made via that method. In forming a medical device via the method of the invention, a metal fabric 10 is provided. The fabric is formed of a plurality of wire strands having a predetermined relative orientation between the strands. In the fabric of For example, in using a tubular braid such as that shown in The wire strands of the metal fabric used in the present method should be formed of a material which is both resilient and can be heat treated to substantially set a desired shape. Materials which are believed to be suitable for this purpose include a cobalt-based low thermal expansion alloy referred to in the field as Elgiloy, nickel based high-temperature high-strength �superalloys� commercially available from Haynes International under the trade name Hastelloy, nickel-based heat treatable alloys sold under the name Incoloy by International Nickel, and a number of different grades of stainless steel. The important factor in choosing a suitable material for the wires is that the wires retain a suitable amount of the deformation induced by the molding surface (as described below) when subjected to a predetermined heat treatment. One class of materials which meet these qualifications are so-called shape memory alloys. Such alloys tend to have a temperature induced phase change which will cause the material to have a preferred configuration which can be fixed by heating the material above a certain transition temperature to induce a change in the phase of the material. When the alloy is cooled back down, the alloy will �remember� the shape it was in during the heat treatment and will tend to assume that configuration unless constrained from so doing. One particularly preferred shape memory alloy for use in the present method is nitinol, an approximately stoichiometric alloy of nickel and titanium, which may also include other minor amounts of other metals to achieve desired properties. NiTi alloys such as nitinol, including appropriate compositions and handling requirements, are well known in the art and such alloys need not be discussed in detail here. For example, U.S. Pat. Nos. 5,067,489 (Lind) and 4,991,602 (Amplatz et al.), the teachings of which are incorporated herein by reference, discuss the use of shape memory NiTi alloys in guidewires. Such NiTi alloys are preferred, at least in part, because they are commercially available and more is known about handling such alloys than other known shape memory alloys. NiTi alloys are also very elastic�they are said to be �superelastic� or �pseudoelastic�. This elasticity will help a device of the invention return to a present expanded configuration for deployment. The wire strands can comprise a standard monofilament of the selected material, i.e. a standard wire stock may be used. If so desired, though, the individual wire strands may be formed from �cables� made up of a plurality of individual wires. For example, cables formed of metal wires where several wires are helically wrapped about a central wire are commercially available and NiTi cables having an outer diameter of 0.003 inches or less can be purchased. One advantage of certain cables is that they tend to be �softer� than monofilament wires having the same diameter and formed of the same material. Additionally, if the device being formed from the wire strands is to be used to occlude a vessel, the use of a cable can increase the effective surface area of the wire strand, which will tend to promote thrombosis. In preparation of forming a medical device in keeping with the invention, an appropriately sized piece of the metal fabric is cut from the larger piece of fabric which is formed, for example, by braiding wire strands to form a long tubular braid. The dimensions of the piece of fabric to be cut will depend, in large part, upon the size and shape of the medical device to be formed therefrom. When cutting the fabric to the desired dimensions, care should be taken to ensure that the fabric will not unravel. In the case of tubular braids formed of NiTi alloys, for example, the individual wire strands will tend to return to their heat-set configuration unless constrained. If the braid is heat treated to set the strands in the braided configuration, they will tend to remain in the braided form and only the ends will become frayed. However, it may be more economical to simply form the braid without heat treating the braid since the fabric will be heat treated again in forming the medical device, as noted below. In such untreated NiTi fabrics, the strands will tend to return to their unbraided configuration and the braid can unravel fairly quickly unless the ends of the length of braid cut to form the device are constrained relative to one another. One method which has proven to be useful to prevent the braid from unraveling is to clamp the braid at two locations and cut the braid to leave a length of the braid having clamps (15 in Alternatively, one can solder, braze, weld or otherwise affix the ends of the desired length together (e.g. with a biocompatible cementitious organic material) before cutting the braid. Although soldering and brazing of NiTi alloys has proven to be fairly difficult, the ends can be welded together, such as by spot welding with a laser welder. The same problems present themselves when a flat sheet of fabric such as the woven fabric shown in Once an appropriately sized piece of the metal fabric is obtained, the fabric is deformed to generally conform to a surface of a molding element. As will be appreciated more fully from the discussion below in connection with The molding element can be a single piece, or it can be formed of a series of mold pieces which together define the surface to which the fabric will generally conform. The molding element can be positioned within a space enclosed by the fabric or can be external of such a space, or can even be both inside and outside such a space. In order to illustrate one example of how such a mold may be configured and how it may be used in accordance with the method of the invention, reference will be had to In the molding element illustrated in The overall shape and dimensions of the center section can be varied as desired; it is generally the size of the central recesses 34 and the forming port 36 which will define the size and shape of the middle of the finished device, as explained below. If so desired, each half 32 may be provided with a manually graspable projection 38. In the embodiment shown in the drawings, this projection 38 is provided at a location disposed away from the abutting faces of the respective halves. Such a manually graspable projection 38 will simply enable an operator to more easily join the two halves to define the recesses 34 and forming port 36. The center section is adapted to cooperatively engage a pair of end plates 40 for forming the desired device. In the embodiment shown in One or more channels 48 for receiving bolts and the like may also be provided through each of the end plates and through the center section 30. By passing bolts through these channels 48, one can assemble the molding element 20 and retain the metal fabric in the desired shape during the heat treatment process, as outlined below. In utilizing the molding element 20 shown in A central portion of the length of the metal braid may be positioned within one of the two halves of the forming port 36 and the opposed halves 32 of the center section may be joined to abut one another to restrain a central portion of the metal braid within the central forming port 36 through the center section. The tubular braid will tend to have a natural, relaxed diameter which is defined, in large part, when the tubular braid is formed. Unless the tubular braid is otherwise deformed, when the wire strands are in their relaxed state they will tend to define a generally hollow tube having the predetermined diameter. The outer diameter of the relaxed braid may be, for example, about 4 mm. The relative size of the forming port 36 in the central section 30 of the molding element and the natural, relaxed outer diameter of the tubular braid may be varied as desired to achieve the desired shape of the medical device being formed. In the embodiment shown in If so desired, a generally cylindrical internal molding section (not shown) may also be provided. This internal molding section has a slightly smaller diameter than the inner diameter of the forming port 36. In use, the internal molding section is placed within the length of the metal fabric, such as by manually moving the wire strands of the fabric apart to form an opening through which the internal molding section can be passed. This internal molding section should be positioned within the tubular braid at a location where it will be disposed within the forming port 36 of the center section when the molding element is assembled. There should be a sufficient space between the outer surface of the interior molding section and the inner surface of the forming port 36 to permit the wire strands of the fabric 10 to be received therebetween. By using such an internal molding section, the dimensions of the central portion of the finished medical device can be fairly accurately controlled. Such an internal molding section may be necessary in circumstances where the natural, relaxed outer diameter of the tubular braid 10 is less than the inner diameter of the forming port 36 to ensure that the braid conforms to the inner surface of that forming port. However, it is not believed that such an internal molding section would be necessary if the natural, relaxed outer diameter of the braid were larger than the inner diameter of the forming port 36. As noted above, the ends of the tubular braid should be secured in order to prevent the braid from unraveling. Each end of the metal fabric 10 is desirably received within a cavity 46 formed in one of the two end plates 40. If a clamp (15 in As best seen in In the molding element 20 shown in It should be understood that the specific shape of the particular molding element 20 shown in Accordingly, the specific molding element 20 shown in Once the molding element 20 is assembled with the metal fabric generally conforming to a molding surface of that element, the fabric can be subjected to a heat treatment while it remains in contact with that molding surface. This heat treatment will depend in large part upon the material of which the wire strands of the metal fabric are formed, but the time and temperature of the heat treatment should be selected to substantially set the fabric in its deformed state, i.e., wherein the wire strands are in their reoriented relative configuration and the fabric generally conforms to the molding surface. The time and temperature of the heat treatment can vary greatly depending upon the material used in forming the wire strands. As noted above, one preferred class of materials for forming the wire strands are shape memory alloys, with nitinol, a nickel titanium alloy, being particularly preferred. If nitinol is used in making the wire strands of the fabric, the wire strands will tend to be very elastic when the metal is in its austenitic phase; this very elastic phase is frequently referred to as a �superelastic� or �pseudoelastic� phase. By heating the nitinol above a certain phase transition temperature, the crystal structure of the nitinol metal when in its austenitic phase can be set. This will tend to �set� the shape of the fabric and the relative configuration of the wire strands in the positions in which they are held during the heat treatment. Suitable heat treatments of nitinol wire to set a desired shape are well known in the art. Spirally wound nitinol coils, for example, are used in a number of medical applications, such as in forming the coils commonly carried around distal lengths of guidewires. A wide body of knowledge exists for forming nitinol in such medical devices, so there is no need to go into great detail here on the parameters of a heat treatment for the nitinol fabric preferred for use in the present invention. Briefly, though, it has been found that holding a nitinol fabric at about 500� C. to about 550� C. for a period of about 1 to about 30 minutes, depending on the softness or hardness of the device to be made, will tend to set the fabric in its deformed state, i.e. wherein it conforms to the molding surface of the molding element. At lower temperatures the heat treatment time will tend to be greater (e.g. about one hour at about 350� C.) and at higher temperatures the time will tend to be shorter (e.g. about 30 seconds at about 900� C.). These parameters can be varied as necessary to accommodate variations in the exact composition of the nitinol, prior heat treatment of the nitinol, the desired properties of the nitinol in the finished article, and other factors which will be well known to those skilled in this field. Instead of relying on convection heating or the like, it is also known in the art to apply an electrical current to the nitinol to heat it. In the present invention, this can be accomplished by, for example, hooking electrodes to the clamps 15 carried at either end of the metal fabric illustrated in After the heat treatment, the fabric is removed from contact with the molding element and will substantially retain its shape in a deformed state. When the molding element 20 illustrated in The vascular occlusion device 60 of The relative sizes of the tubular middle section and the expanded diameter portions can be varied as desired. In this particular embodiment, the medical device is intended to be used as a vascular occlusion device to substantially stop the flow of blood through a patient's blood vessel. When the device 60 is deployed within a patient's blood vessel, as detailed below, it will be positioned within the vessel such that its axis generally coincides with the axis of the vessel. The dumbbell-shape of the present device is intended to limit the ability of the vascular occlusion device 60 to turn at an angle with respect to the axis of the blood vessel to ensure that it remains in substantially the same position in which the operator deploys it within the vessel. Although the illustrated embodiments of this invention only have two expanded diameter portions, it should be understood that the device could have more than two such expanded diameter portions. For example, if the device has three expanded diameter portions, each expanded diameter portion is separated from at least one other expanded diameter portion by a tubular portion having a smaller diameter. If so desired, the diameters of each of the expanded diameter portions can be the same but they need not be the same. In order to relatively strongly engage the lumen of the blood vessel, the maximum diameter of the expanded diameter portions 64 (which occurs along the middle ridge 66 in this embodiment) should be selected so that it is at least as great as the diameter of the lumen of the vessel in which it is to be deployed, and is optimally slightly greater than that diameter. When it is deployed within the patient's vessel, the vascular occlusion device 60 will engage the lumen at two spaced-apart locations. The device 60 is desirably longer along its axis than the dimension of its greatest diameter. This will substantially prevent the vascular occlusion device 60 from turning within the lumen at an angle to its axis, essentially preventing the device from becoming dislodged and tumbling along the vessel with blood flowing through the vessel. The relative sizes of the generally tubular middle portion 62 and expanded diameter portion 64 of the vascular occlusion device 60 can be varied as desired for any particular application. For example, the outer diameter of the middle portion 62 may range between about one quarter and about one third of the maximum diameter of the expanded diameter portions 64 and the length of the middle portion 62 may comprise about 20% to about 50% of the overall length of the device. Although these dimensions are suitable if the device 60 is to be used solely for occluding a vascular vessel, it is to be understood that these dimensions may be varied if the device is to be used in other applications, such as where the device is intended to be used simply as a vascular filter rather than to substantially occlude the entire vessel or where the device is deployed in a different channel in a patient's body. The aspect ratio (i.e., the ratio of the length of the device over its maximum diameter or width) of the device 60 illustrated in The pick and pitch of the metal fabric 10 used in forming the device 60, as well as some other factors such as the number of wires employed in a tubular braid, are important in determining a number of the properties of the device. For example, the greater the pick and pitch of the fabric, and hence the greater the density of the wire strands in the fabric, the stiffer the device will be. Having a greater wire density will also provide the device with a greater wire surface area, which will generally enhance the tendency of the device to occlude a blood vessel in which it is deployed. This thrombogenicity can be either enhanced, e.g. by a coating of a thrombolytic agent or by attaching silk or wool fabric to the device, or abated, e.g. by a coating of a lubricious, anti-thrombogenic compound. A variety of materials and techniques for enhancing or reducing thrombogenicity are well known in the art and need not be detailed here. When the device is deployed in a patient's vessel, thrombi will tend to collect on the surface of the wires. By having a greater wire density, the total surface area of the wires will be increased, increasing the thrombolytic activity of the device and permitting it to relatively rapidly occlude the vessel in which it is deployed. It is believed that forming the occlusion device 60 from a 4 mm diameter tubular braid having a pick of at least about 40 and a pitch of at least about 30� will provide sufficient surface area to substantially completely occlude a blood vessel of 2 mm to about 4 mm in inner diameter in a suitable period of time. If it is desired to increase the rate at which the device 60 occludes the vessel in which it is deployed, any of a wide variety of known thrombolytic agents can be applied to the device. As explained more fully below in connection with The base 88 desirably flares out relatively rapidly to define a shoulder tapering radially outwardly from the middle 86 of the body. When the device is deployed in a vessel, this shoulder will abut the lumen of one of the vessels being treated. The forward end 84 is retained within the vessel and urges the base 88 of the body open to ensure that the shoulder engages the wall of the vessel to prevent the device 80 from becoming dislodged from within the shunt. As detailed above, in making a device of the invention it is desirable to attach the ends of the wire strands forming the metal fabric 10 to one another to prevent the fabric from unraveling. In the illustrations of The rearward ends of the wire strands are shown as being attached to one another by an alternative clamping means 90. This clamp 90 serves the same purpose as the schematically illustrated clamp 15, namely to interconnect the ends of the wires. However the clamp 90 also serves to connect the device 80 to a delivery system (not shown). In the embodiment shown, the clamp 90 is generally cylindrical in shape and has a recess for receiving the ends of the wires to substantially prevent the wires from moving relative to one another, and a threaded outer surface. The threaded outer surface is adapted to be received within a cylindrical recess (not shown) on a distal end of a delivery device and to engage the threaded inner surface of the delivery device's recess. The delivery device (not shown) can take any suitable shape, but desirably comprises an elongate, flexible metal shaft having such a recess at its distal end. The delivery device can be used to urge the PDA occlusion device 80 through the lumen of a catheter for deployment in a channel of the patient's body, as outlined below. When the device is deployed out the distal end of the catheter, the device will still be retained by the delivery device. Once the proper position of the device 80 in the shunt is confirmed, the shaft of the delivery device can be rotated about its axis to unscrew the clamp 90 from the recess in the delivery means. By keeping the PDA device 80 attached to the delivery means, the operator could still retract the device for repositioning if it is determined that the device is not properly positioned in the first attempt. This threaded attachment will also allow the operator to control the manner in which the device 80 is deployed out of the distal end of the catheter. As explained below, when the device exits the catheter it will tend to resiliently return to a preferred expanded shape which is set when the fabric is heat treated. When the device springs back into this shape, it may tend to act against the distal end of the catheter, effectively urging itself forward beyond the end of the catheter. This spring action could conceivably result in improper positioning of the device if the location of the device within a channel is critical, such as where it is being positioned in a shunt between two vessels. Since the threaded clamp 90 can enable the operator to maintain a hold on the device during deployment, the spring action of the device can be controlled and the operator can control the deployment to ensure proper positioning. A PDA occlusion device 80 of this embodiment of the invention can advantageously be made in accordance with the method outlined above, namely deforming a metal fabric to generally conform to a molding surface of a molding element and heat treating the fabric to substantially set the fabric in its deformed state. The molding element 100 generally comprises a body portion 110 and an end plate 120. The body portion 110 is adapted to receive and form the body 82 of the device 80 while the end plate is adapted to compress against the metal fabric to form the forward end 84. The body portion 110 includes an elongate, generally tubular central segment 112 which is sized to receive the elongate body 82 of the device. The central segment 112 of the molding element 100 optimally has an internal diameter slightly less than the natural, relaxed outer diameter of the tubular braid of which the device is formed. This compression of the braid will help yield devices with reproducibly sized bodies 82. The forward end of the body portion 110 includes a back plate 114 which has a generally annular sidewall 116 depending downwardly therefrom. The sidewall defines a recess 118 which is generally circular in shape. The end plate 120 of the molding element 100 has a generally disc-shaped face 122, which desirably has a clamp port 124 approximately centered therein for receiving a clamp 15 attached to the metal fabric, as noted above. The end plate also has an annular sidewall 126 which extends generally upwardly from the face 122 to define a generally cylindrical recess 128 in the end plate 120. The sidewall 116 of the body portion 110 is sized to be received within the recess 128 of the end plate. In use, the metal fabric is placed in the molding element and the body portion 110 and the end plate 120 are brought toward one another. The inner face of the back plate 114 will engage the fabric and tend to urge it under compression generally radially outwardly. The fabric will then be enclosed generally within the recess 118 of the body portion and will generally conform to the inner surface of that recess. If one prevents the entire clamp 15 from passing through the clamp port 124, the fabric will be spaced slightly away from the inner surface of the face 122, yielding a slight dome shape in the forward end 84 of the device, as illustrated in Once the fabric is compressed in the molding element 100 so that it generally conforms to the molding surface of the molding element, the fabric can be subjected to a heat treatment such as is outlined above. When the molding element is opened again by moving the body portion 110 and the end plate 120 away from one another again, the fabric will generally retain its deformed, compressed configuration. The device can then be collapsed, such as by urging the clamps 15, 90 generally axially away from one another, which will tend to collapse the device toward its axis. The collapsed device 80 can then be passed through a catheter for deployment in a channel in a patient's vascular system. The device can then be retracted so that the forward end 84 engages the wall of the pulmonary artery P. If one continues to retract the catheter, the engagement of the device with the wall of the pulmonary artery will tend to naturally pull the body portion 82 of the device from the catheter, which will permit the body portion to return to its expanded configuration. The body portion should be sized so that it will frictionally engage the lumen of the PDA's shunt. The device 80 will then be held in place by the combination of the friction between the body portion and the lumen of the shunt and the engagement between the wall of the pulmonary artery and the forward end 84 of the device. Over a relatively short period of time, thrombi will form in and on the device 80 and the thrombi will occlude the PDA. If so desired, the device may be coated with a suitable thrombolytic agent to speed up the occlusion of the PDA. This filter 180 has a generally conical configuration, tapering generally radially outwardly from its rearward end 182 to its forward end 184. A length of the device adjacent its forward end is adapted to engage the walls of a lumen of a channel. The maximum diameter of the filter device 180 is therefore at least as large as the inner diameter of the channel in which it is to be positioned so that at least the forward end will engage the wall of the vessel to substantially lock the device in place. Having a series of unsecured ends 185 of the wire strands adjacent the forward end of the device will assist in seating the device in the channel because the ends of the wires will tend to dig into the vessel wall slightly as the forward end of the device urges itself toward its fully expanded configuration within the vessel. The combination of the friction between the outwardly urging forward end of the device and the tendency of the wire ends to dig into the vessel walls will help ensure that the device remains in place where it is deployed rather than floating freely within a vessel to reach an undesired location. The method in which the device 180 of the invention is deployed may vary depending on the nature of the physiological condition to be treated. For example, in treating an arterio-venous fistula, the device may be carefully positioned, as described above, to occlude the flow of blood at a fairly specific location. In treating other conditions (e.g. an arterio-venous malformation), however, it may be desired to simply release a number of these devices upstream of the malformation in a vessel having a larger lumen and simply allow the devices to drift from the treatment site to lodge in smaller vessels downstream. The decision as to whether the device 180 should be precisely positioned at an exact location within the channel in a patient's body or whether it is more desirable to allow the device(s) to float to their final lodging site will depend on the size of the channels involved and the specific condition to be treated. This decision should be left to the individual operator to be made on a case-by-case basis as his or her experience dictates; there is no one right or wrong way to deploy the device 180 without regard to the conditions at hand. In the embodiment shown in The ends of the wire strands at the rearward end 182 of the device are secured with respect to one another, such as by means of a threaded clamp 90 such as that described above in connection with In the embodiment illustrated in When a tubular braid is used in forming this device, the tubular metal fabric may be applied to the molding element by placing the molding element within the tubular braid and clamping the ends of the braid about the molding element before cutting the braid to the desired length. In order to better facilitate the attachment of the clamps 90 to the ends of the tubular braid, the ends 206 of the molding element may be rounded, as shown, rather than tapering to a sharper point at the ends of the molding element. In order to ensure that the braid more closely conforms to the outer surface of the molding element 200, i.e. the molding element's molding surface, the natural, relaxed diameter of the braid should be less than the maximum diameter of the element, which occurs at its middle 204. This will place the metal fabric in tension about the middle of the element and, in combination with the clamps at the ends of the braid, cause the braid to generally conform to the molding surface. The two molding sections should be readily attached to one another with the larger, open ends 226 of the sections abutting one another. The mold sections can simply be clamped together, such as by providing a reusable jig (not shown) which can be used to properly position the sections 214, 216 with respect to one another. If so desired, bolt holes 228 or the like may be provided to allow a nut and bolt, or any similar attachment system, to be passed through the holes and attach the sections 214, 216 together. In use, a suitably sized piece of a metal fabric, optimally a length of a tubular braid, is placed in the recess 212 of the molding element and the two molding sections 214, 216 are urged toward one another. The fabric should have a relaxed axial length longer than the axial length of the recess 212 so that bringing the sections toward one another will axially compress the fabric. This axial compression will tend to urge the wire strands of the braid radially outwardly away from the axis of the braid and toward engagement with the molding surface of the element 210, which is defined by the surface of the recess 212. Once the metal fabric is deformed to generally conform to the molding surface of either molding element 200 or 210, the fabric can be heat treated to substantially set the shape of the fabric in its deformed state. If molding element 200 is used, it can then be removed from the interior of the metal fabric. If there is sufficient room between the resilient wire strands, the molding element can simply be removed by opening the web of wire strands and pulling the molding element out of the interior of the metal fabric. If molding element 210 is employed, the two molding sections 214, 216 can be moved away from one another and the molded fabric can be retrieved from the recess 212. Depending on the shape of the molding surface, the resulting formed shape may resemble either a pair of abutting hollow cones or, as noted above, a football, with clamps, welds or the like provided at either end of the shape. This shape can then be cut into two halves by cutting the wires in a direction generally perpendicular to the shared axis of the cones (or the major axis of the ovoid shape) at a location about midway along its length. This will produce two separate filter devices 180 substantially as illustrated in In an alternative embodiment of this method, the molding element 200 is formed of a material selected to permit the molding element to be destroyed for removal from the interior of the metal fabric. For example, the molding element may be formed of a brittle or friable material, such as glass. Once the material has been heat treated in contact with the molding surface of the molding element, the molding element can be broken into smaller pieces which can be readily removed from within the metal fabric. If this material is glass, for example, the molding element and the metal fabric can be struck against a hard surface, causing the glass to shatter. The glass shards can then be removed from the enclosure of the metal fabric. The resultant shape can be used in its generally conical shape, or it can be cut into two separate halves to produce a device substantially as shown in Alternatively, the molding element 200 can be formed of a material which can be chemically dissolved, or otherwise broken down, by a chemical agent which will not substantially adversely affect the properties of the metal wire strands. For example, the molding element can be formed of a temperature-resistant plastic resin which is capable of being dissolved with a suitable organic solvent. The fabric and the molding element can be subjected to a heat treatment to substantially set the shape of the fabric in conformance with the surface of the molding element, whereupon the molding element and the metal fabric can be immersed in the solvent. Once the molding element is substantially dissolved, the metal fabric can be removed and either used in its current shape or cut into separate halves, as outlined above. Care should be taken to ensure that the material selected to form the molding element is capable of withstanding the heat treatment without losing its shape, at least until the shape of the fabric has been set. For example, the molding element could be formed of a material having a melting point above the temperature necessary to set the shape of the wire strands, but below the melting point of the metal forming the strands. The molding element and metal fabric can then be heat treated to set the shape of the metal fabric, whereupon the temperature can be increased to substantially completely melt the molding element, thereby removing the molding element from within the metal fabric. It should be understood that the methods outlined immediately above for removing the metal fabric 10 from the molding element 200 can be used in connection with other shapes, as well. Although these methods may not be necessary or desirable if the molding element is carried about the exterior of the metal fabric (such as are elements 30-40 of the molding element 20 of This molding element 230 can be used with a metal fabric which is in the form of a tubular braid. If such a fabric is used and a clamp 15 (not shown in this drawing) or the like is provided to connect the ends of the wire strands adjacent one end of the device, a recess (not shown) analogous to the cavity 46 in the face of the compression disk 44 of molding element 20 ( However, the present molding element 230 can be used quite readily with a Flat woven piece of metal fabric, such as is illustrated in As can be seen by the discussion of the various molding elements 200, 210 and 230 in In the embodiment of The basket 270 (shown in its collapsed configuration in The body of the device is formed of a metal fabric, as explained above. The metal fabric of this embodiment is optimally initially formed as a tubular braid and the ends of the wires forming the braid can be attached together by means of the bands 272, 274 before the fabric is cut to length. Much like the clamps 15, 90 noted above, these bands 272, 274 will help prevent the metal fabric from unraveling during the forming process. (The method of forming the basket 270 is described below in connection with When the device is in its collapsed state for deployment in a patient's vessel (as illustrated in When the device is deployed in a patient's vascular system, the basket will take on an expanded configuration wherein it extends outwardly of the outer surface of the guidewire. As best seen in In moving from its collapsed state ( The precise dimensions of the metal fabric can be varied as desired for various applications. If the device 250 is to be used as a vascular filter to trap emboli released into the blood, for example, the pores (i.e. the openings between the crossing metal strands) of the fabric are desirably on the order of about 1.0 mm. This is generally deemed to be the minimum size of any particles which are likely to cause any adverse side effects if they are allowed to float freely within a blood vessel. One would not want to make the pores too small, though, because the blood (or other fluid) should be free to pass through the wall of the basket 270. If so desired, the basket may be coated with a suitable anti-thrombogenic coating to prevent the basket from occluding a blood vessel in which it is deployed. When a fabric having 1.0 mm pores is used to form the basket 270 of this embodiment of the invention, the forming process will reorient the wires relative to one another and in some areas (e.g. adjacent the proximal lip 286) the pores will be larger than 1.0 mm. However, because the basket's walls are formed of essentially two thicknesses 282, 284 of the fabric, the effective pore size of the device may be significantly reduced even at these locations. The device 250 may also be provided with tethers 290 for collapsing the basket 270 during retraction. The basket may include four independent tether wires, each of which extends proximally from the proximal lip 286 of the deployed basket. In a preferred embodiment, though, the four tether wires illustrated in the drawings are actually formed of two longer wires, with each wire extending peripherally about a portion of the proximal lip of the basket. These tether wires may be intertwined with the wires of the metal fabric to keep the tethers in place during use. When the tethers are retracted or drawn down toward the guidewire, the wires extending along the proximal lip of the basket will tend to act as drawstrings, drawing the proximal end of the basket radially inwardly toward the guidewire. This will tend to close the basket and entrap any material caught in the cavity 288 of the basket during use so that the basket can be retracted, as detailed below. The tether wires 290 may extend along much of the length of the guidewire so that they will extend outside the patient's body during use of the device 250. When it is desired to collapse the basket for retrieval, the operator can simply hold the guidewire 260 steady and retract the tethers with respect to the guidewire. This can tend to be relatively cumbersome, though, and may be too difficult to effectively accomplish without breaking the tethers if the device is deployed at a selective site reached by a tortuous path, such as in the brain. Accordingly, in the preferred embodiment shown in In the embodiment of An external catheter such as that referred to in the discussion of As best seen in its collapsed state (shown in The basket 320 of In the vascular traps 250 and 250′ of The cover 340 is shown in its deployed, expanded configuration in The proximal control hypotube 344 may extend along a substantial portion of the length of the catheter 310 so that it extends out of the patient's body when the device 300 is in place. By grasping the control hypotube and moving it relative to the guidewire 310, an operator can control the position of the cover 340 with respect to the basket 320, which is affixed to the guidewires. As explained in more detail below in connection with the use of the device 300, once the basket has been deployed and has been used to filter objects entrained in the fluid (e.g. emboli in blood), the cover 340 may be deployed and the trap may be drawn proximally toward the cover by moving the guidewire proximally with respect to the control hypotube 344. The inner diameter of the distal lip 358 of the cover is desirably slightly larger than the outer diameter of the proximal lip 328 of the basket. Hence, when the basket is drawn proximally toward the cover it will be substantially enclosed therein. The cover will therefore tend to trap any emboli (not shown) or other particulate matter retained within the cavity 330 of the basket. A retrieval sheath S may then be urged distally to engage the outer surface of the cover 340. This will tend to cause the cover to collapse about the basket, tightly engaging the outer surface of the basket. This somewhat collapsed structure can then be withdrawn from the patient's channel and removed from the patient's body. By enclosing the basket within the cover, the likelihood of any filtered debris within the basket being lost as the basket is retrieved will be substantially eliminated. The guidewire and the metal fabric can be of any diameter suitable for the intended application of the vascular trap 250, 250′ or 300. In a preferred embodiment, the guidewire is between about 0.014″ and about 0.038″ in diameter and the wires of the metal fabric used to form the basket (and the cover 340, if a cover is included) are between about 0.002″ and about 0.006″. The thickness of the metal bands (272, 274 or 322, 324) also is desirably in the range of about 0.002″-0.006″. In one particularly preferred embodiment intended to be used in narrower vessels such as those encountered in cerebral and coronary applications, the guidewire has an outer diameter of about 0.014″ and the wires of the metal fabric are about 0.002″ in diameter. The metal bands in this embodiment may also have a thickness of about 0.002″ so that they will not be substantially wider than the collapsed basket. When the device is collapsed for deployment through a catheter, it will have an outer diameter of about 0.018″, permitting the device to be used with catheters and other instruments adapted for use with a 0.018″ guidewire. The molding element 370 has an outer molding section 372 defining a curved inner surface 374 and an inner molding section 376 having an outer surface 378 substantially the same shape as the curved inner surface 374 of the outer molding section. The inner molding section 376 should be sized to be received within the outer molding section, with a piece of the metal fabric (not shown) being disposed between the inner and outer molding sections. In a preferred embodiment, the inner surface 374 of the outer molding element and the outer surface 378 of the inner molding section each include a recess (375 and 379, respectively) for receiving an end of the braid. The molding surface of this molding element 370, to which the fabric will generally conform, can be considered to include both the inner surface 374 of the outer molding section and the outer surface 378 of the inner molding section. In use, the two molding sections 372, 376 are spaced apart from one another and a length of a tubular braid of metal fabric (not shown in The method in accordance with the present invention further includes a method of treating a physiological condition of a patient. In accordance with this method, a medical device suitable for treating the condition, which may be substantially in accordance with one of the embodiments outlined above, is selected. For example, if a patent ductus arteriosus is to be treated, the PDA occlusion device 80 of Medical devices made in accordance with the method of the invention outlined above have a preset expanded configuration and a collapsed configuration which allows the device to be passed through a catheter. The expanded configuration is generally defined by the shape of the medical fabric when it is deformed to generally conform to the molding surface of the molding element. Heat treating the metal fabric substantially sets the shapes of the wire strands in the reoriented relative positions when the fabric conforms to the molding surface. When the metal fabric is then removed from the molding element, the fabric may define a medical device in its preset expanded configuration. The medical device can be collapsed into its collapsed configuration and inserted into the lumen of the catheter. The collapsed configuration of the device may be of any shape suitable for easy passage through the lumen of a catheter and proper deployment out the distal end of the catheter. For example, the devices shown in Once the medical device is collapsed and inserted into the catheter, it may be urged along the lumen of the catheter toward the distal end of the catheter. This may be accomplished by using a guidewire or the like to abut against the device and urge it along the catheter. When the device begins to exit the distal end of the catheter, which is positioned adjacent the desired treatment site, it will tend to resiliently return substantially entirely to its preset expanded configuration. Superelastic alloys, such as nitinol, are particularly useful in this application because of their ability to readily return to a particular configuration after being elastically deformed to a great extent. Hence, simply urging the medical device out of the distal end of the catheter tends to properly deploy the device at the treatment site. Although the device will tend to resiliently return to its initial expanded configuration (i.e. its shape prior to being collapsed for passage through the catheter), it should be understood that it may not always return entirely to that shape. For example, the device 60 of If the device is to be used to permanently occlude a channel in the patient's body, such as the devices 60 and 80 described above may be, one can simply retract the catheter and remove it from the patient's body. This will leave the medical device deployed in the patient's vascular system so that it may occlude the blood vessel or other channel in the patient's body. In some circumstances, the medical device may be attached to a delivery system in such a manner as to secure the device to the end of the delivery means, such as when the threaded clamp 90 shown in The devices of For example, any one of these devices are suitable for use in conjunction with a balloon angioplasty procedure. In such procedures, catheters having inflatable balloons at their ends, referred to as balloon catheters, are positioned within a blood vessel so that the balloon is positioned within a stenosis. These balloons are positioned by tracking the balloon catheter along a guidewire or the like; the balloons typically have a central bore therethrough. Once the balloon is properly positioned, it is inflated and urges radially outwardly against the stenosis. This will tend to squeeze the stenosis against the walls of the vessel, improving patency of the vessel. When the stenosis is treated in this fashion, though, there is a risk that some debris will break free and enter the blood flowing through the vessel. If left unchecked, this embolus can drift downstream and embolize a distal portion of the vessel. Depending on where the embolus comes to rest, the embolization can result in significant tissue or organ damage. This risk is particularly acute in cardiac and coronary applications because the embolization can result in a myocardial infarction or heart attack, and in neurovascular and interventional radiological procedures the embolization can lead to a stroke or damage to brain tissue. In order to prevent, or at least substantially limit, such embolization, a vascular trap 250, 250′ or 300 of the invention can be used with the balloon catheter. The device should be sized to permit it to be passed through the lumen of the particular balloon catheter to be used in the angioplasty. In one embodiment of a method for using such a vascular trap, the trap is deployed first. The basket (270 or 320) of the trap will be guided to a position located downstream of the desired treatment site through an introduction catheter (e.g. the catheter C in In an alternative embodiment of the present method, the balloon catheter can be used to perform the same function as performed by the introduction catheter in the preceding embodiment. In this embodiment, the balloon catheter is positioned in the patient's vessel so that the distal end of the balloon catheter is located downstream of the stenosis. The vascular trap (250, 250′ or 300) of the invention is then passed through the lumen of the balloon catheter and the basket is urged out of the distal end of the catheter. The basket will resiliently substantially return to its preferred expanded configuration, whereupon the balloon catheter can be retracted along the shaft of the device's guidewire until the balloon is properly positioned within the stenosis. If so desired, the balloon catheter can instead be provided with a length of standard catheter extending distally beyond the distal end of the balloon. The balloon can then be positioned within the stenosis and the basket can be urged out of the distal end of the distal extension of the catheter. In such an embodiment, the length of the distal extension of the catheter should be sufficient to properly position the basket with respect to the balloon when the basket exits the distal end of the catheter. This will eliminate the need to perform the separate step of retracting the balloon into position within the stenosis after the basket is deployed. The balloon can then be expanded, deflated and withdrawn as described above. Much the same procedure can be used to deploy a vascular trap of the invention for use in an atherectomy procedure. In such procedures, a cutting head is positioned at the distal end of an elongate, hollow shaft and the cutting head has a bore extending therethrough. The trap can be deployed in either of the methods outlined above, but it is anticipated that in most instances the first procedure will be used, i.e. the basket will be deployed with an introduction catheter, which will be removed so that the cutting device can be guided over the guidewire of the vascular trap. It should also be understood that the device 250, 250′ and 300 could also be used in other medical procedures in other bodily channels besides a patient's vascular system. Since the trap is positioned downstream of the stenosis, any debris released during the procedure will tend to drift distally toward the basket and be caught therein. In order to prevent any emboli from simply floating past the trap, it is preferred that the proximal lip (288 or 328) of the basket be at least as large as the lumen of the vessel. In a preferred embodiment, the natural dimension of the proximal lip (i.e. where the basket has fully returned to its expanded configuration) is somewhat greater than the vessel's inner diameter so that the basket will firmly engage the wall of the vessel. The method of retracting the basket will depend on which embodiment of the vascular trap is used, namely whether or not the device includes a cover 340. The devices 250 or 250′ of If so desired, a balloon catheter or like device can instead be used, with the balloon catheter being used to draw down the tethers 290 and collapse the basket. The vascular trap can then be withdrawn with the balloon catheter rather than having to separately introduce a removal catheter to remove the trap. In withdrawing the embodiment illustrated in Once the medical procedure is completed and any debris has been captured in the basket, the cover 340 is allowed to resiliently substantially return to its expanded configuration. Once it is deployed proximally of the basket, the basket 320 can be drawn proximally toward the cover 340 until it engages or is received within the cover, as noted above in connection with In actuality, the cover 340 may be unable to return to its full expanded configuration due to the confines of the vessel in which it is deployed. As explained previously, the cover 340 is desirably larger than the basket 320 so that the basket can be received within the cover. However, the basket is optimally sized to engage the walls of the vessel to prevent the unwanted passage of emboli or other debris around the edges of the basket. Accordingly, the distal lip 358 of the cover will engage the wall of the channel before it expands to its full size. The walls of most bodily channels, such as blood vessels, tend to be somewhat elastic, though. The cover 340 will therefore tend to urge harder against the wall of the vessel than the smaller basket and may stretch the vessel a little bit more than will the basket. In this fashion, the cover may still be able to expand to a dimension large enough to permit the basket to be received in the cavity 356 of the cover. If not, the distal lip 358 of the cover can simply be brought into close engagement with the proximal lip 328 of the basket to generally seal the basket. Once the cover 340 is brought into engagement with the basket 320, whether by receiving the basket within the cover or, less preferably, by engaging the lips 358, 328 of the cover and the basket, the device can be withdrawn proximally from the patient's vascular system. The cover will tend to prevent any emboli caught in the basket during deployment from being inadvertently lost during withdrawal. The vascular traps 250, 250′ and 300 of the present invention therefore have distinct advantages over other vascular traps or filters currently known in the art. As explained above, most prior art traps are difficult and expensive to form and cannot be readily collapsed for retrieval. The present invention, though, provides a method for making the vascular traps 250, 250′ and 300 which is both relatively inexpensive and less labor intensive, generally resulting in a more consistent product than prior art hand-forming methods. Furthermore, the structure of the device and the methods outlined above for removing the device will fairly reliably prevent the inadvertent dumping of trapped emboli back into the bloodstream while the device is being removed. Since most prior art traps and filters are much more difficult to use and are more likely to dump filtered debris back into the bloodstream, the present invention can be substantially safer than these prior art systems. While a preferred embodiment of the present invention has been described, it should be understood that various changes, adaptations and modifications may be made therein without departing from the spirit of the invention and the scope of the appended claims. Patent Citations
Non-Patent Citations
Referenced by
Legal Events
Rotate |


text:
Main Content
Are you an NHL® fan with a fierce sense of fashion? Bundle up in the G-III® women's bubble jacket with detachable hood this season! Totally spirited and totally cute, it features a team wordmark on the bottom left front and team logo on the center back; striped ribbing at the cuffs and waist adds a dash of sporty style, while the team-colored hood, shoulder and pocket accents offer a touch of shine.
- 100% Polyester
- Decorative piping
- Full-zip jacket
- Made in China
- Officially licensed
- Rib-knit cuffs, collar and waist
- Stripes on cuffs and waist
- Team graphics on lower left front and center back
- Touches of shine on front pockets, shoulders and detachable drawcord hood
Special event items are produced by manufacturers only after the outcome of a game
or event. These are advanced sale items and will ship immediately after they are
received in our warehouse.
Manufacturer direct items are shipped directly from the manufacturer. These items
are not available for international or expedited shipping.
Customized items can be personalized with options such as your name, your favorite
number, and/or designs. Some options may be limited by league rules.

subset: commoncrawl


text:
Paty EscobedoHairDark BrownDry HairPrimera VezNunca habia usado un peine para planchar mi pelo, pero esta vez compre una plancha Croc y decidi agregar al carro de compras este peine, que al principio no entendia como funcionaba, pero a los pocos dias me encanto, ya no me quemaba los dedos al tocar mi pelo...lo unico q no me agrado, es que el calor le afecto un poco y empezo a ponerse color blanco de la parte inferiorposted Jan 10, 2013Link to this post
EliHairMedium BrownNormal HairOily HairDry HairColor Treated HairBest Invention EverLove this thing. I straightened my hair in half the time and did a phenomenal job. I got so many compliments. Everyone thought I had it professionally done. Sent everyone over to folica.com to get one.posted Oct 4, 2011Link to this post
AlysonToronto, ONHairDark BrownDry HairFine or Thin HairThick or Coarse HairRelaxed HairChemically Treated HairI Wanted This To Work But It Didn't Boo!I have relaxed/texlaxed hair and I rarely straighten it because I am not a huge fan of heat. On the days I use my 'heat pass', I hoped that this tool would get my ends straight. This tool got STUCK IN MY HAIR:( It was really sad and I was really disappointed. However, it worked great on my sister's hair (she has a weave it). It gets her hair super straight. I wouldn't recommend this tool if you have tight, coily hair as this tool will probably get stuck in your hair.posted Sep 16, 2011Link to this post
YeseniaBrooklynHairBlackCurly HairGreat CombThis comb makes flat-ironing my hair an absolute breeze!!posted Mar 18, 2011Link to this post
Tove R.USAHairDark BrownCurly HairThick or Coarse HairBeautiful Results!I could tell the minute I held this flat iron that is was quality--it heats up in minutes and since I have course curly hair I actually take the time to blow it out (straight) first, but this iron works fabulously with only one pass. It has a professional, heavy duty cord, the black nano titanium plates heat evenly and hot! I don't want to have to make more than one pass with a flatiron and this one fits the bill, excellent! I really appreciate the craftsmanship of this product.posted Feb 22, 2011Link to this post
MalidaHairBlackNormal HairLove This CombI really like this comb. I think it works excellent with the hair straightener. Really easy to use and make your hair more straight help get rid of all the tangle before you used the straightener.posted Jan 19, 2011Link to this post
TrishAtlanta, GAHairBlondeCurly HairThick or Coarse HairNo NeedThis came free with the purchase of my Croc Wet to Dry Flat Iron and I am thankful I didn't pay for it. I don't get the need for this. It's not practical in my opinion and awkward to use.
It just seems to be one of those extra tools that you use once or twice then throw in a drawer never to use again.posted Jan 19, 2011Link to this post -
JCAHairMedium BrownDry HairColor Treated HairThick or Coarse HairLove This Comb!This comb is so handy - why didn't I buy this sooner? It came as a a free gift with a Croc flat iron purchase. I also purchased the same comb by Sedu. The Sedu version seemed inferior to the Croc one in terms of quality and handling so I sent it back and I ordered another Croc comb. It saves time and effort flat ironing your hair. The bottom piece allows you to part your hair at the same time you are getting ready to press down the top comb allowing you to do two things at once. It may take some a little getting used to if you're not very coordinated, but once you get the hang of it you'll see that it truly is a must have styling tool! Just remember to not press so hard on the comb when you're flat ironing your hair with your other hand...that's the habit I had to break when I first began using it.posted Jan 19, 2011Link to this post
Write Your Own Review
Have you tried this product? Help out by sharing what worked (or didn't)!
Write a Review


text:
Mon-Fri: 7am-1am EST / Sat-Sun: 7am-11pm EST
If you placed an order on ApplianceHelp.com the order is still valid and you can check on its status and make changes here on PartSelect.com. Click here for more details.
Enter your model number and our Instant Repairman will help diagnose and fix your appliance.
Need help finding your model number?


text:
Obama Honors Winners Of The Nat'l Medal Of Science, Technology, Innovation
Similar images
View allKeywords
President Barack Obama awards the National Medal of Science to Lucy... News PhotoAward,Barack Obama,Ceremony,Horizontal,International Landmark,Politics,President,Science,US President,USA,Washington DC,White House - Washington DCPhotographer Brendan HoffmanCollection: Getty Images News 2013 Getty ImagesWASHINGTON, DC - FEBRUARY 1: President Barack Obama awards the National Medal of Science to Lucy Shapiro in a ceremony at the White House on February 1, 2013 in Washington, DC. The National Medal of Science recognizes individuals who have made outstanding contributions to science and engineering, while the National Medal of Technology and Innovation recognizes those who have made lasting contributions to America's competitiveness and quality of life and helped strengthen the Nation's technological workforce. (Photo by Brendan Hoffman/Getty Images)


text:
United States > All 3M Products > Manufacturing > Industrial Adhesives & Tapes > Products
Email a friend
Your Email:
Your Name:
Friends Email:
Message (optional)
Privacy Policy
From 3M™ Masking Tapes that offer line sharpness and removal the way you want to 3M™ Duct Tape built to stick. 3M innovation and quality meets everyday taping challenges.
Let's Work Together
Quick Links
3M™ Specialty High Temperature Masking Tape 501+
3M™ High Performance Green Masking Tape 401+
3M™ Performance Masking Tape Yellow 301+
3M™ General Use Masking Tape 201+
3M™ Value Masking Tape 101+


text:
Practice Areas
Briefings & More
Search results
9 articles matched your search
- Firm:ApplebyRemove this filter
- Practice Area:EmploymentRemove this filter
- Section:BriefingsRemove this filter
Sort By: Newest first | Oldest first
Some factors to consider when becoming a director weblink
A company's business activities must ultimately be conducted by individuals — company directors — notwithstanding that a company has a distinct legal personality.
The Incentives for Job Makers Act 2013 weblink
This brief provides a condensed summary of the key changes brought about by the Incentives for Job Makers Act 2013.
Bermuda: scope of civil rights expanded by recent amendment weblink
In its changes to the Human Rights Act 1981, Bermuda's parliament has added sexual orientation to the list of defined characteristics protected from unlawful discrimination.
Bermuda can benefit from pensions restructuring weblink
Bermuda is well placed to benefit from increased life expectancy rates that have some employers seeking to restructure the pension entitlements of their employee members.
Guide to establishing an office in Bermuda weblink
This guide addresses the issues that arise when an exempted company seeks to establish a physical office in Bermuda.
Mauritius enhances the role of trade unions in the employer/employee relationship weblink
Four years ago, Mauritius brought about a new era in its employment laws by redefining the employer/employee relationship when it promulgated the Employment Rights Act.
New housing and work law legislation weblink
The Control of Housing and Work (Jersey) Law 2012 is expected to be brought into force on 29 April this year.
The Control of Housing and Work (Jersey) Law 2012 — seminar presentations download
The Control of Housing and Work (Jersey) Law 2012 is expected to come into force on 29 April 2013. Property lawyers from Appleby ran a seminar on 11 March outlining the key aspects of the legislation.
Mistaken dispositions into trust — a recent case heard before Royal Court of Guernsey weblink
Appleby recently acted in the matter of Dervan and MD Events v Concept Fiduciaries Ltd and Others for applicants to the Royal Court of Guernsey who sought to set aside two dispositions into an employee benefit trust on the ground of mistake.


text:
Classic Zip
Grey Heather
This product is out of stock!
Desperate? Find a shop selling etnies near you!
Save To WishlistAvailable Colors:
Product Overview
Part of etnies Classic Collection. 80% cotton 20% poly 240g French terry zip hooded sweatshirt.
Details & Features
- Drawcords on hood.
- Embroidered etnies Arrow logo on chest.
- Front pouch pockets.
- etnies classic label package and trim detail.
Item code: 4130002055/043
Reviews for Classic Zip
- Reviews
- 8
Quincey Rowe LOVE IT!
Dave The color is much lighter in real life than it looks in the photos.
I sent it back and got a black one.
More small logos would be nice (such as patterned inside the hood and/or at the ends of the drawstrings and/or near the sleeve cuff, etc.)
romain cozzani Veste assez cool pour le printemps et l'été car pas très épaisse! Pour le BMX, très agréable de rider avec quand il fait plus de 5°
firstname.lastname@example.com Good!
Chris cole Top zip hoody, definitely would buy another.
monica martinez overall good not too hot about stitching on pocket label.
email@example.com Very smooth comfort...
Good cut for long arms! -
email@example.com Great! Super comfy


text:
Windows 7, Windows Vista, Windows XP Service Pack 2, Windows XP Service Pack 3
Microsoft ActiveSync 4.5 is the latest sync software release for Windows Mobile-powered devices. ActiveSync provides a great synchronization experience with Windows®-powered PCs and Microsoft Outlook right out of the box. Note: Microsoft ActiveSync works only with Windows XP SP2 or earlier.
The Windows Mobile 6 SDK Refresh adds documentation, sample code, header and library files, emulator images and tools to Visual Studio that let you build applications for Windows Mobile 6.
Emulator Images, Gesture APIs, and Samples for Windows Mobile 6.5
Software Development Kit for Windows Mobile 5.0 software for Pocket PC.
Microsoft® ActiveSync® 4.2 is synchronization software for Windows Mobile®-based devices.
Loading your results, please wait...


text:
MERCS is a skirmish game of miniature combat taking place in the future. Games consist of two squads of 5, which have all special equipment and skills. The miniatures act simultanously, using a system of initiative and reaction to determine the exact order of turns.
MERCS uses unique cards as both reference and a means to measure. This streamlined approach means games are faster, with less rule-lawyering, providing a dynamic feel with combat that ebbs and flows. It also means that a game of MERCS is flexible; it can be played just about anywhere with any number of people. All that is needed is 1 fi g, 1 card, and 1d10.
MERCS tabletop is a fast-moving, high energy miniatures game. It is designed with idea that the game should work in a realistic manner. MERCS isn't typical. It is not a game players can sit back and shoot each other from across the table. Victory has more to do with the decisions a player makes, then the MegaCon he plays.
In MERCS, players must learn to think tactically, overcome and adapt on the fly, and move. Combat begins and ends with movement. Everyone can shoot; survivors know when to shoot and when to grab cover.
Welcome to the cause!
MERCS: Recon Game Rules - 20 MB
MERCS RPG Beta - 1 MB
MERCS Terrain Maps Release Form - 142 kB
MERCS Tabletop Cards - 10 MB
MERCS Campaign Maps Beta 1.1 - 14 MB
MERCS Campaign Beta 1.1 - 8 MB
Fan Kit - 7 MB
Store Kit - 42 MB
USCR Snow Terrain - 101 MB
CCC Urban Terrain - 126 MB
MERCS Quick Rules - 5 MB
USCR FAQ - 297 kB
Texico FAQ - 297 kB
sefadu FAQ - 325 kB
KemVar FAQ - 294 kB
Keizai Waza FAQ - 376 kB
ISS FAQ - 355 kB
FCC FAQ - 293 kB
Book FAQ - 293 kB
CCC FAQ - 320 kB
MERCS Campaign Maps - 14 MB
MERCS Campaign - 8 MB
Holiday Sale!
Amazing deals running now until December 25th in the MegaCon Games store! Save big on Myth and MERCS. First Edition BLOWOUT! MERCS 1st ED. Rulebook Retail Price: $35 Sale Price: $10 The Winter War 2 Factions of your choice & MERCS 1st ED. Rulebook Retail Price: $165 Sale Price: $80 Treasure Seeker's… Continue Reading
GECON: Demo Signup, Products available, EU Inc, Myth pickup, and MORE!
Here are some news and details about GenCon 2014. Our booth number is 1545. There are 5 big points to touch on! FIRST We will have EU, Inc starter kits at GenCon this year. This is a pre 2.0 run of the cards with metal models for the tabletop game. This package… Continue Reading


text:
Topic Contents
Cancer: Home Treatment for Sleep Problems
Skip to the navigationThings you can do
Home treatment may be all that is needed to relieve sleep problems caused by cancer or the side effects of chemotherapy or radiation therapy. If your doctor has given you instructions or medicines to treat sleep problems, be sure to follow them. Check with your doctor before using any nonprescription medicines to help you sleep.
Establish a sleep routine
- Set a bedtime and a time to get up, and then stay with those times, even on weekends. This will help your body get used to a regular sleep time.
- Get regular exercise but not during the 3 to 4 hours before your bedtime. Talk with your doctor about how much physical activity is okay for you. Walking, swimming, and yoga may be some good choices.
- Avoid caffeine after noon. This includes coffee, tea, cola drinks, and chocolate.
- Avoid drinking alcohol. It may make you sleepy but will probably also wake you up after a short time.
- Wind down toward the end of the day. Don't take on problem-solving conversations or challenging activities in the evening.
Make your bedroom a restful place
- Remove distractions such as a clock, telephone, television, or radio from your bedroom.
- Block out background noise in your bedroom throughout the night. You may want to use a fan or a white noise machine. Or try using a sleep mask and earplugs at night.
- Keep your bedroom dark, cool, and quiet.
- Reserve the bedroom for sleeping and sexual activities so that you come to associate it with sleep.
After you are in bed
- After getting into bed, make a conscious effort to let your muscles relax. Imagine yourself in a peaceful, pleasant scene. For more information, see the topic Stress Management.
- If you are still awake after 15 or 20 minutes, get up and read or do a boring task until you feel drowsy. Don't lie in bed and think about how much sleep you're missing. Do not watch TV in bed.
If you take medicines
- Review all of your prescription and nonprescription medicines with your doctor or pharmacist to see whether the medicines you take could be the cause of your sleep problem.
- If you take steroids (such as prednisone) or other medicines that may be stimulating, take them as long before bedtime as possible.
Complementary therapies
There is evidence that therapeutic massage improves sleep for people who are having cancer treatments. Massage may also reduce pain, anxiety, and other symptoms.1
Mind-body therapy, such as meditation, relaxation, and cognitive-behavioral therapy, often are helpful for people in cancer treatments. Movement-based mind-body therapy, such as yoga and tai chi, have been found to improve sleep quality.2
Watch for symptoms during home treatment
If one or more of the following symptoms occur during home treatment, contact your doctor:
- Your sleep problem lasts longer than 4 weeks.
- Your sleep problem becomes worse.
Related Information
- Freeman L (2009). Massage therapy. Mosby's Complementary and Alternative Medicine: A Research-Based Approach, 3rd ed., chap. 13, pp. 364–388. St. Louis, MO: Mosby Elsevier.
- Ulbricht C, et al. (2011). Complementary, alternative, and integrative therapies. In VT DeVita Jr. et al., eds., DeVita, Hellman and Rosenberg's Cancer: Principles and Practice of Oncology, 9th ed., pp. 2550–2561. Philadelphia: Lippincott Williams and Wilkins.
ByHealthwise Staff
Primary Medical Reviewer E. Gregory Thompson, MD - Internal Medicine
Specialist Medical Reviewer Michael Seth Rabin, MD - Medical Oncology
Current as ofNovember 14, 2014
- Top of Page
Next Section:
Related Information
Previous Section:
Things you can do - Top of Page
Next Section:
Previous Section:
Related Information - Top of Page
Next Section:
Previous Section:
References - Top of Page
Current as of: November 14, 2014
Author: Healthwise Staff
Medical Review: E. Gregory Thompson, MD - Internal Medicine & Michael Seth Rabin, MD - Medical Oncology
This information does not replace the advice of a doctor. Healthwise, Incorporated disclaims any warranty or liability for your use of this information. Your use of this information means that you agree to the Terms of Use. How this information was developed to help you make better health decisions.
To learn more, visit Healthwise.org
© 1995-2015 Healthwise, Incorporated. Healthwise, Healthwise for every health decision, and the Healthwise logo are trademarks of Healthwise, Incorporated.


text:
Decorative Embossing Wheelsby MercArt
These embossing wheels are ideal for decorating the backgrounds or borders of embossing projects. They are available in a variety of sizes and shapes:
Dot and Dash wheel creates a 1/16 in. wide line of alternating dots and dashes.
Diagonal Lines wheel makes a pattern of 1/8 in. wide diagonal lines.
Horizontal Lines wheel leaves a pattern of 1/8 in. wide horizontal lines.
Points wheel creates a pattern of small dots. The wheel is 1/8 in. wide.
Feathers wheel makes a feathered pattern of diagonal lines that is 5/16 in. wide.
Squares with Spaces wheel leaves a 1/8 in. wide line of alternating squares and spaces.
Diagonal Lines and Two Star Wheels wheel creates a pattern of diagonal lines flanked by lines of small dots that is 5/16 in. wide.
Big Wheel leaves a continuous line, 1/16 in. wide, of sall dots.
The Micro Wheel is 1/4 in. in diameter. This wheel creates an extermely fine line for textures and shadows.
Ask a question about Decorative Embossing Wheels Ask a question about this product Average User Rating: Based on (7 Reviews) luba s. from Atlanta, GA says: I got a set of all of them. They make great patterns on leatherhard clay. Well made, quality material. Very nice Please wait...
You May Also Like...
How to get VIP Club Prices
Anyone can be a VIP Member with:
Your email will not be shared. We will only use your email address to respond to your question.
When asking your question, please be as descriptive as possible, it will allow us to give you the most accurate answer possible.
Please wait...
Please wait...


text:
Solowheel: self-balancing last mile transport for the upstanding commuter
The Solowheel electric unicycle from Inventist - with foldaway foot platforms and carry handle
Article Summary
The fat wheeled eniCycle, the stylish and graceful U3 from Honda or the slightly scary prospect of the UnoMoto have all shared more in common than being one-wheeled, self-balancing personal transport solutions. They've all had somewhere for the user to sit. Inventist's Solowheel is a little different – you ride this electric unicycle standing upright, like a Segway or skateboard. It has a useful carry handle and fold-away foot platforms, is gyro-stabilized and the Li-ion batteries offer a range of about 12 miles between charges.
« Back to Solowheel: self-balancing last mile transport for the upstanding commuter
Related Articles
- Features
- Reviews
- Aircraft
- Architecture
- Around The Home
- Automotive
- Bicycles
- Children
- Computers
- Digital Cameras
- Electronics
- Environment
- Games
- Good Thinking
- Health and Wellbeing
- Holiday Destinations
- Home Entertainment
- Inventors and Remarkable People
- Laptops
- Marine
- Medical
- Military
- Mobile Technology
- Motorcycles
- Music
- Outdoors
- Pets
- Robotics
- Science
- Space
- Sports
- Telecommunications
- Urban Transport
- Wearable Electronics


text:
LSU set to honor 1993 championship baseball team
Along with Skip Bertman's legacy, LSU will honor the 20th anniversary of its 1993 national title team — a squad that thrived when the outcome looked bleak — in ceremonies Friday
For once, the 1993 LSU baseball team won't have to come from behind to earn its due.
This time the honors are coming right up front.
LSU's second College World Series national championship team will be honored Friday night shortly before the 7 p.m. first pitch of the middle game of the LSU-Ole Miss weekend series at Alex Box Stadium.
Also being honored is former LSU coach and athletic director Skip Bertman with the formal dedication of the field at Alex Box as Skip Bertman Field.
Bertman also coached CWS champions in 1991, 1996, 1997 and 2000, but none captured the crown in Omaha, Neb., quite like the 1993 team.
"They came out of the losers bracket three straight times, including in the SEC tournament," Bertman said Thursday.
"It was a high-character team," he said. "It was loaded with big leaguers, too. I was proud just to be a part of it."
About 20 players and coaches from the 1993 team are expected to attend Friday's game, including Ole Miss coach Mike Bianco. An assistant coach at LSU from 1993-97, Bianco also played for the Tigers in 1988-89.
"It's neat being here to honor a baseball legend (Bertman) and a Louisiana icon," Bianco said.
Among those not expected to attend is former LSU coach Ray "Smoke" Laval, now head coach at North Florida.
Laval was an assistant on the 1993 team and succeeded Bertman as head coach in 2002 but was fired in 2006 after a five-year record of 210-109-1 with two CWS appearances and an SEC title. He was succeeded by current LSU coach Paul Mainieri.
The Tigers' 1993 national championship came in the midst of a power run for LSU that saw the program win 50 straight games from 1989-93 with two CWS crowns, four CWS appearances and four Southeastern Conference titles.
The Tigers started that season, the 100th anniversary of the program, ranked No. 1 in the country and ended it by raising the CWS trophy, but it wasn't without a fight.
LSU battled back from an early loss to Mississippi State to win the SEC West Division tournament (the SEC had two divisional tournaments then), then rebounded from a 15-12 loss to Kent State in the NCAA South Regional at old Alex Box Stadium. Facing elimination, the Tigers had to beat South Alabama twice to advance, doing so by scores of 11-4 and 9-4.
In the CWS, LSU beat Long Beach State 7-1 and Texas A&M 13-8 before losing to Long Beach State 10-8.
With a trip to the national championship game on the line, LSU faced Long Beach again. The Tigers trailed 5-3 in the bottom of the ninth before Armando Rios tied the game with a two-run double and Todd Walker won it with an RBI single.
No such dramatics were needed in the winner-take-all national championship game against Wichita State, thanks to LSU pitcher Brett Laxton.
The Tigers freshman threw one of the most dominant games in CWS championship history, striking out a record 16 batters while allowing just three hits in an 8-0 victory.


text:
Phoronix: Fedora 18 Is Codenamed The Spherical Cow
Going up against Ubuntu 12.10's Quantal Quetzal will be Fedora 18, which is being developed under the codename of Spherical Cow...
Holy cow. Seriously, what is up with these silly names?
/me waits for Gentoo Foundation, Inc. to claim trademark violation.
While the codenames are unprofessional, it's pretty funny when people ask what OS I run and I reply with:
"I'm running a beefy miracle on my laptop"
"I'm running a spherical cow on my laptop"
Spherical Cow
Looks like a really fat cow is involved.
Linux bloated? Nah... its just a spherical cow O.o!
Also... http://en.wikipedia.org/wiki/Spherical_cow
the Linux distro grml has clearly better release names IMHO:
http://grml.org/changelogs/Release announcement for Grml 2011.12 - codename Knecht Rootrecht (similar to Knecht Ruprecht, in the German tradition that's the helper of St Nicholas, who punish bad children; rootrecht -> root permission)
Previous releases
grml 2011.05 - codename Just Mari
grml 2010.12 - codename Gebrüder Grml
grml 2010.04 - codename Grmlmonster (similar to Krümmelmonster was a monster in the German version of Sesame Street)
grml 2009.10 - codename Hello-Wien
grml 2009.05 - codename Lackdose-Allergie
grml 2008.11 - codename Schluchtenscheisser (a insulting name for Austrians, translation could be: chasm shithead)
grml 1.1 - codename Skunk
grml 1.0 - codename Meilenschwein (mile pig)
grml 0.9 - codename Dioptrienotto
grml 0.8 - codename Funkenzutzler
grml 0.7 - codename Bootenschnitzl (schnitzl -> cutlet)
grml 0.6 - codename Winterschlapfn (wintershoe)
grml 0.5 - codename Tokolytika
grml 0.4 - codename Eierspass (eggfun)
grml 0.3 - codename Hustenstopper (coughstopper)
grml 0.2 - codename Satura
grml 0.1 - codename OS04
grml is a system rescue distribution.


text:
Purchase a Used Car from Crown Auto
Group in Greensboro
Crown Auto Group has a
complete selection of used cars to choose from in New
Jersey, Virginia,
Carolina and South
Carolina. A member of our staff is ready to take you for a
test drive in your favorite used car, truck, SUV or van. If you are
looking for a used Acura, BMW, Chrysler, Dodge, Ford,
Honda, Jeep, MINI, Nissan, RAM or Volvo, Crown Auto Group
is the place to shop if you value low prices. We can be reached by
phone, or visit one of our dealership