text:
A system and method are provided for both recharging and communicating with a stimulator having a rechargeable battery, which stimulator is implanted deeply in the body, in particular where the stimulator is a microstimulator, the system includes a base station and an external device, for instance a chair pad. The chair pad may contain an antenna/charging coil and a booster coil. The antenna/charging coil can be used for charging the rechargeable battery and also for communicating with the stimulator using frequency shift keying and on-off keying. The booster coil can be used to recharge a battery depleted to zero volts. The base station connected to the chair pad may be used to power the antenna/charging coil and the booster coil.
1. A system for recharging and communicating with a body implanted stimulator having a rechargeable battery comprising:
a chair pad;
a base station that is electrically connectable to the chair pad; and
an antenna/charging coil that is used to inductively charge the rechargeable battery with the implanted stimulator and to transcutaneously communicate with the stimulator,
wherein the antenna/charging coil is placed inside the chair pad.
2. The system of
circuitry for accomplishing forward and backward frequency shift keying (FSK) telemetry with the implanted stimulator,
wherein the antenna/charging coil is configured and dimensioned to enable FSK telemetry.
3. The system of
circuitry for accomplishing forward on-off keying (OOK) telemetry with the implanted stimulator using the antenna/charging coil.
4. The system of
a current measuring circuitry for determining power consumption in the antenna/charging coil.
5. The system of
an automatic power shut-off circuitry for automatically shutting off power to the antenna/charging coil when the power consumption through the antenna/charging coil exceeds a predetermined level.
5. The system of
a printed circuit board (PCB) contained in the chair pad; and
a sensing circuitry for sensing temperature included on the PCB.
6. The system of
an automatic power shut-off circuitry for automatically shutting off power to the antenna/charging coil when the sensed temperature through the antenna/charging coil exceeds a predetermined level.
7. The system of
a booster coil inside the chair pad that is used for zero-volt battery recovery (ZVR).
8. The system of
9. The system of
a power sensing circuitry for determining power consumption in the booster coil; and
an automatic power shut-off circuitry for automatically shutting off power to the booster coil when the power consumption through the booster coil exceeds a predetermined power level.
10. The system of
a printed circuit board (PCB) contained in the chair pad;
a sensing circuitry for sensing temperature included on the PCB; and
an automatic power shut-off circuitry for automatically shutting off power to the booster coil when the sensed temperature through the booster coil exceeds a predetermined power level.
11. The system of
12. The system of
a compliant chair pad housing made of polyurethane foam;
a chair pad printed circuit board (PCB); and
a coil assembly housing which contains a booster coil, the antenna/charging coil and the chair pad PCB,
wherein the polyurethane foam housing encapsulates the coil assembly housing.
13. The system of
padding that surrounds the polyurethane foam housing; and
an exterior slipcover that surrounds the padding.
14. The system of
a booster coil that is placed in a coil assembly with the antenna/charger coil, wherein the booster coil and antenna coil are wound over a spool coil in a configuration to present at least one substantially flat side; and
a coil shield which is grounded and which shield is placed as part of the coil assembly to substantially cover the antenna/charger coil and the booster coil,
wherein the coil assembly is fully encapsulated in a-housing, which housing is contained within the chair pad.
15. The system of
16. The system of
a chair pad cable that connects the chair pad to the base station; and
detection circuitry for automatically detecting disconnection of the chair pad cable from the chair pad.
17. The system of
a speaker for generating an audible sound to signal a significant system event.
18. The system of
a booster coil for use in zero volt battery recovery (ZVR); and
first and second impedance matching networks,
wherein a first amplifier power supply to the antenna/charging coil is impedance matched with the first impedance matching network; and
wherein a second amplifier power supply to the booster coil is impedance matched with the second impedance matching network.
19. The system of
20. The system of
21. The system of
a sensor for detecting power levels in the antenna/charging coil; and
a variable output power supply that automatically adjusts downwards when the power levels detected by the sensor detects power levels that exceed a predetermined levels,
wherein the variable output power supply is contained within the base station.
22. The system of
23. The system of
24. A system for communicating and recharging an implantable stimulator comprising:
an external interface device;
an antenna/charging coil; and
a booster coil, wherein the antenna/charging coil and the booster coil are contained within the external device;
wherein the antenna/charging coil is configured to facilitate radio-frequency telemetry communication with the implanted stimulator; and
wherein the booster coil is configured for use in zero-volt recovery operation for reviving a completely depleted rechargeable battery in the stimulator.
25. The system of
26. The system of
27. The system of
circuitry for accomplishing forward and backward frequency shift keying (FSK) telemetry with the implanted stimulator,
wherein the antenna/charging coil is configured and dimensioned to enable FSK telemetry.
28. The system of
circuitry for accomplishing forward on-off keying (OOK) telemetry with the implanted stimulator with the antenna/charging coil.
29. The system of
a current measuring circuitry for determining power consumption in the antenna/charging coil; and
an automatic power shut-off circuitry for automatically shutting off power to the antenna/charging coil when the power consumption through the antenna/charging coil exceeds a predetermined power level.
30. The system of
a printed circuit board (PCB) contained in the external interface device;
a sensing circuitry included on the PCB for sensing temperature; and
an automatic power shut-off circuitry for automatically shutting off power to the antenna/charging coil when the sensed temperature exceeds a predetermined level.
31. The system of
32. The system of
33. The system of
a coil shield which is grounded and which shield is placed as part of a coil assembly to substantially cover the antenna/charger coil and the booster coil,
wherein the booster coil that is placed in the coil assembly with the antenna/charger coil, wherein the booster coil and antenna coil are wound over a coil spool in a configuration to present at least one substantially flat side; and
wherein the coil assembly is fully encapsulated in a polyurethane housing, which housing is contained within the external interface device.
34. The system of
a base station;
a connection extension cable that connects the external interface device to the base station; and
detection circuitry for automatically detecting disconnection of the extension cable from the external interface device.
35. The system of
first and second impedance matching networks,
wherein a first amplifier power supply to the antenna/charging coil is impedance matched with the first impedance matching network; and
wherein a second amplifier power supply to the booster coil is impedance matched with the second impedance matching network.
36. The system of
a sensor for detecting power levels in the antenna/charging coil; and
a variable output power supply that automatically adjusts downwards when the power levels detected by the sensor detects a power level that exceeds a predetermined level,
wherein the variable output power supply is contained within the base station.
37. The system of
38. The system of
circuit means, located in the base station, for reducing all voltages and currents inside the external interface device to below about 4.5 Amperes and below about 25 VDC.
39. The system of
40. A system for communicating and recharging an implantable stimulator having a rechargeable battery, the system comprising:
an external interface device;
an antenna/charging coil contained in the external interface device, which antenna/charging coil is used for radio-frequency communication with the stimulator; and
a charging circuit for inductively recharging the rechargeable battery, wherein the charging circuit is capable of recharging the battery in the stimulator which is implanted in a patient's body up to an implant depth of about 15 centimeters.
41. The system of
a booster coil for implementing zero volt recovery (ZVR) for the rechargeable battery in the stimulator,
wherein the rechargeable battery is of a type that can be revived even after the battery has been depleted down to zero volts.
42. The system of
a base station; and
circuit means for converting supply voltages and currents in the base station to levels of voltages and currents in the external interface device that are about or below 0.5 amperes and 25 VDC.
43. A method of communicating with and recharging an implantable stimulator having a rechargeable battery, the method comprising:
(a) providing an antenna/charging coil;
(b) providing a booster coil;
(c) positioning the antenna/charge coil and the booster coil near a patient's skin over the implanted stimulator; and
(d) determining whether the rechargeable battery is in the following condition (1) completely depleted to zero volts or (2) needs recharging, but is not at zero volts,
wherein if case (1) applies, performing the steps:
(e) applying a 1.2 MHz through the booster coil in order to reset the stimulator recharging circuitry to accept 127 KHz;
(f) turning off the power to the booster coil; and
(g) applying power to the antenna/charge coil to commence trickle charging of the rechargeable battery,
wherein if case is (2) applies, performing step (g) directly, without performing steps (e) and (f).
[0001] The present application claims the benefit of U.S. Provisional Application Serial No. 60/392,475, filed Jun. 28, 2002, which application is incorporated herein by reference in its entirety.
[0002] The present invention relates to stimulation systems and, more particularly, systems for transcutaneously charging and communicating with a body-implanted stimulator having a rechargeable battery.
[0003] Radio-frequency powered implantable stimulators and battery powered, implantable microstimulators are described in the art. See, for instance, U.S. Pat. Nos. 5,193,539 (�Implantable Microstimulator); 5,193,540 (�Structure and Method of Manufacture of an Implantable Microstimulator�); 5,312,439 (�Implantable Device Having an Electrolytic Storage Electrode�); 6,185,452 (�Battery-Powered Patient Implantable Device�); 6,164,284 and 6,208,894 (both titled �System of Implantable Device for Monitoring and/or Affecting Body Parameters�). The '539, '540, '439, '452, '284, and '894 patents are incorporated herein by reference in their entireties.
[0004] Microstimulators to prevent or treat various disorders associated with prolonged inactivity, confinement or immobilization of one or more muscles are taught, e.g., in U.S. Pat. Nos. 6,061,596 (�Method for Conditioning Pelvis Musculature Using an Implanted Microstimulator�); 6,051,017 (�Implantable Microstimulator and Systems Employing the Same�); 6,175,764 (�Implantable Microstimulator System for Producing Repeatable Patterns of Electrical Stimulation�); 6,181,965 (�Implantable Microstimulator System for Prevention of Disorders�); 6,185,455 (�Methods of Reducing the Incidence of Medical Complications Using Implantable Microstimulators�); and 6,214,032 (�System for Implanting a Microstimulator�). The '596, '017, '764, '965, '455, and '032 patents are incorporated herein by reference in their entireties.
[0005] Implantable stimulators having rechargeable batteries have specific requirements. In general once a stimulator is implanted within a patient's body, it is intended to stay there permanently. When a rechargeable battery is used to power an implantable stimulator, there must be a transcutaneous means to recharge the rechargeable battery in the stimulator without explanting the stimulator. In addition, there must also be a means to communicate with the stimulator after it has been implanted in order to transmit and receive control signals from and to the stimulator, as well as to transfer data from and to the stimulator. An important technical issue is how a rechargeable battery in an implanted stimulator may be revived when the rechargeable battery is completely depleted, i.e., to zero volts.
[0006] A specific form of an implantable stimulator is a microstimulator. Microstimulators present advantages over conventionally sized stimulators in that microstimulators are more easily implanted, with less surgical trauma. Advantageously, microstimulators may be injected through a large bore needle or placed via a small incision in the skin. In addition, microstimulators may be implanted in locations that do not offer enough space to contain larger, conventional-sized stimulators and their associated extension leads.
[0007] One application that is particularly suited for using a microstimulator is the treatment of urinary urge incontinence. The BION� microstimulator (Advanced Bionics� Corporation, Sylmar, Calif.) is currently being used in patients to treat urinary urge incontinence by stimulating the pudendal nerve. In prior art implantable systems for treating incontinence, a conventional-size stimulator is attached to a lead having an electrode or electrodes on the distal lead tip. The lead, having a substantial length, can be tunneled to the target nerve deep inside the body, while the conventional-sized stimulator can remain implanted just beneath the surface of the skin. Because the conventional stimulator is intended to be placed just below the surface of the skin, the technical requirements for designing a telemetry communication system using such a conventional stimulator is, relatively speaking, easier to accomplish. Moreover, because the conventional-sized stimulator is in a comparatively large housing, such a stimulator can contain a primary, one-time-only-use battery and thus, no recharging is required.
[0008] An implantable microstimulator, in contrast, generally does not use an extension lead, as the electrodes are often placed directly on the body of the microstimulator. Because the electrodes are placed on the body of the microstimulator, it must usually be placed very close to the target tissue (usually a nerve) being stimulated. In the case of the urinary incontinence application, therefore, the microstimulator is implanted deep inside the body near the pudendal nerve and not, as in the case with conventional stimulators, just below the surface of the skin. In addition, when a microstimulator is used, the small housing puts the use of space at even a greater premium and effectively prohibits the use of a primary battery and, instead, necessitates the use of a rechargeable battery.
[0009] Thus, the technical challenges presented by a microstimulator that is implanted deep in the body are highly complex because (a) the microstimulator uses a rechargeable battery and (b) the microstimulator can be implanted more deeply and at variable depth within the body than a conventional stimulator. The variability of implantation depth can be problematic because the components that are used to charge the battery and communicate with the stimulator must accommodate this variability. Consequently, the technical requirements needed to fulfill the dual operations of transcutaneously recharging a battery and communicating with an implantable microstimulator are more daunting than using a conventional implantable stimulator, which merely requires near-distance telemetry communication and no charging circuitry. There are additional technical challenges to overcome when a microstimulator is placed deep in the body, for instance, in the lower part of the torso, as is the case for the urinary incontinence application. In particular, the external device configuration or configurations must be determined which can best recharge the battery in the microstimulator and communicate with the microstimulator. This set of technical requirements presents unique challenges in designing a charging and communication system which the present novel invention addresses.
[0010] The present invention addresses the above and other needs by providing an external charging and communication system that can be used with a deeply implanted stimulator or microstimulator having a rechargeable battery.
[0011] In accordance with one aspect of the invention, there is provided a charging and communication system that includes a chair pad (for sitting on) and a base station. The chair pad may include an antenna/charging coil for inductively charging the rechargeable battery and a booster coil for recharging the rechargeable battery when it has been inadvertently depleted to zero volts. The recharging from a base of zero volts is hereinafter termed �zero volt recovery� or �ZVR.� A polyimide film substrate with copper traces is used as a grounded coil shield to cover the booster coil and antenna/charging coil in order to reduce the capacitively coupled currents to a patient and to prevent the occurrence of electric shock.
[0012] The booster coil and antenna/charging coil may be designed for charging and communications functions with a microstimulator that is implanted between 12 to 15 centimeters deep in the body with up to a 30� off-axis misalignment from the microstimulator with the charging field. Inductive charging may be accomplished, for example, using a preferred frequency of 127 KHz:
[0013] In addition, the antenna/charging coil may be used for communication with the stimulator either employing frequency shift keying (FSK) or on-off keying (OOK) communication. FSK is akin to frequency modulated (FM) radio signals in that a radio frequency signal is conveyed through frequency modulation of a carrier signal. OOK may be compared to amplitude modulated (AM) radio signals in that the amplitude of a signal is modulated. However, unlike true AM signals, OOK communication permits only two amplitude levels: on or off, which amplitudes can convey the binary numbers 1 or 0. Data transfer with OOK communication is slow and therefore it is generally used only as a back-up when FSK communication is not available.
[0014] When the antenna/charging coil is used for FSK telemetry communication, it is necessary to tweak the bandwidth of the coil circuit to flatten or �dampen� the frequency response in order to broaden the bandwidth. This may be accomplished by connecting a damping resistor in parallel to the charging coil or by other circuit means. The damping resistor or other circuit means may be included in the base station.
[0015] The chair pad may contain a chair pad printed circuit board (PCB). The chair pad PCB may contain a temperature sensor such as a thermocouple to sense the temperature of the PCB and to send a signal to microcontroller to cut off power to the antenna/charging coil. The chair pad PCB may also contain a tuning circuitry for the antenna/charging coil to impedance match the charging coil with the amplifier in the base station and, similarly, the chair pad PCB may contain a tuning circuitry for the booster coil to impedance match the driving amplifier for the booster coil. Such impedance matching (tuning) is necessary to optimize the transfer of power from the base station amplifier to the coils and to reduce excess heat dissipation from the coils and, hence, from the chair pad, which excess heat would be undesirable. In addition, the impedance matching also contributes to reducing voltages and currents that are 1000 Volts and 4.5 Amperes, respectively in the base station, to a safer 0.5 Amperes and 25 volts in the antenna/charging coil, located in the chair pad. Thus, although in one embodiment of the present invention, the coils and amplifier may all be contained in a single, external interface device, the use of a separate base and external interface device (the chair pad) advantageously promotes a safer charging/communication system, by isolating the higher voltages and currents to the base station only.
[0016] The PCB may also contain protection circuits that include a temperature sensor and/or booster coil voltage sensing. The temperature sensor may be used to sense PCB temperature and when a predetermined temperature is reached, and automatic shut-off circuit may disconnect the power to the booster coil or antenna/charging coil to prevent overheating. Similarly, a voltage (or current) sensor may be used to sense the current and voltage in the booster or antenna/charging coils to calculate power dissipation and cut off power if it exceeds a predetermined trigger level. Such temperature and power monitoring features are desirable in the system, since the coils and circuitry are all contained in a coil assembly and encapsulated in a polyurethane foam housing, such that the circuitry cannot be directly accessed.
[0017] The base station is connectable via a chair pad cable to the chair pad. The base station is powered by an AC adapter. The base station may contain a microcontroller that can interface with visual displays such as LED and LCD displays mounted on the base station or with an audio alarm. The audio speaker may be used to produce sounds or tones to indicate when charging is complete or, alternatively, as a signal for other significant events. The microcontroller can be electrically connected to a coil driver amplifier, which microcontroller also controls the zero volt recovery charging and FSK and OOK communications. The base station contains a power converter section for generating several different values of D.C. voltages to supply power to the antenna/charging coil, the booster coil, the chair pad PCB, as well as circuitry in the base station.
[0018] The amplifier circuitry is provided in the base station in which a lower value +3.3 VDC is used in most cases, such as during charging, but is temporarily and briefly switchable to a +4.5 VDC during telemetry reception. Advantageously, this amplifier design reduces power consumption and thereby reduces the heat generated in the base station and also reduces switching regulator noise. A variable output +7 to +20 VDC power supply can be produced from the base station, which variable power may be continuously adjusted during inductive charging to prevent overheating.
[0019] It is thus a feature of the present invention to provide a charging and communication system that can be used both to charge the battery in the stimulator and to communicate with a stimulator implanted at a depth of between about 10 to 15 centimeters, while the stimulator is positioned up to 30� off-axis relative to the inductive charging or RF telemetry field.
[0020] It is a further feature of the invention to provide impedance matching of the primary antenna/charging and booster coils with the driving amplifiers and also to adjust for misalignment of the primary coil with the secondary coil in the microstimulator to reduce heat dissipation in the base station and chair pad and to maximize power transfer.
[0021] It is another feature to include a zero-volt battery recovery system that can revive and recharge a battery that is fully depleted to zero-volts.
[0022] It is yet a further feature of the present system to provide fail-safe features to sense currents or voltages in the antenna/charging coil and booster coil or to directly sense temperatures on the chair pad PCB or on the base station PCB to automatically trigger power shut down to a coil to prevent overheating.
[0023] The above and other aspects, features and advantages of the present invention will be more apparent from the following more particular description thereof, presented in conjunction with the following drawings wherein:
[0024]FIG. 1 is a block diagram of an exemplary microstimulator system, including an implantable microstimulator and various external components for charging the rechargeable battery in the microstimulator and communicating with the microstimulator;
[0025]FIG. 2 (2-1; 2-2) is a view of a system that includes a base station and a chair pad, which system is used to charge the battery in the microstimulator and to communicate with the microstimulator;
[0026]FIG. 3A is a simplified schematic diagram representing the electrical circuitry in the chair pad, including a printed circuit board;
[0027]FIG. 3B is a mechanical representation of the exterior of the chair pad with a chair pad cable;
[0028]FIG. 4A is a view of the coil assembly in the chair pad;
[0029]FIG. 4B is an exploded view of the coil assembly of FIG. 4A;
[0030]FIG. 5 is a block diagram of the power converter and shut down circuitry sections contained in the base station; and
[0031]FIG. 6 is a block diagram of power supply drivers for the antenna/charging coil and booster coil contained in the base station.
[0032] Corresponding reference characters indicate corresponding components throughout the several views of the drawings.
[0033] The following description is of the best mode presently contemplated for carrying out the invention. This description is not to be taken in a limiting sense, but is made merely for the purpose of describing the general principles of the invention. The scope of the invention should be determined with reference to the claims.
[0034] A fully assembled battery-powered microstimulator may operate independently, or in a coordinated manner with other implanted devices, or with external devices. For the purpose of describing the present invention, an exemplary microstimulator, the battery-powered BION� microstimulator (Advanced Bionics� Corporation, Sylmar, Calif.), will be referred to interchangeably as a �microstimulator.�
[0035] The microstimulator is a pulse generator which includes a rechargeable battery. The battery is recharged, as required, from an external battery charging system, typically through an inductive link. The microstimulator is preferably a substantially cylindrical shape and at least portions of it are hermetically sealed. The microstimulator may include electronic circuitry that allow it to generate stimulus pulses that are applied to a patient through electrodes.
[0036] The microstimulator circuitry, battery capacity, cycle life, hermeticity and longevity preferably provide implant operation for at least five years at typical settings. Battery control circuitry protects the battery from overcharging, operates the implant in a safe mode upon battery depletion and avoids any potentially endangering failure modes, with a zero tolerance for unsafe failure or operational modes. The microstimulator accepts programming only from compatible programming devices.
[0037] The publications and patents listed in the table below, which are all incorporated herein by reference, describe various uses of the implantable microstimulator for the purpose of treating various neurological conditions:
[0038]FIG. 1 shows an exemplary, implantable, microstimulator system which includes internal and external components, as well as surgical components. Internal components 10′ are implanted in the target tissue area of the patient and external components 20 are used to recharge and communicate with the internal components. The components shown in FIG. 1 represent, as a whole, an implantable BION� microstimulator system 100 (also known as a Battery-Powered BION or �BPB�). As shown in FIG. 1, the various components may be subdivided into three broad categories: (1) implantable components 10′, (2) external components 20, and (3) surgical components 30.
[0039] The microstimulator 10 includes a case 12; rechargeable battery 16; microstimulator electronic subassembly 14, which includes microstimulator coil 18 and a stimulating capacitor, CSTIM, 15; indifferent/reference electrode 24; and an active/stimulating electrode 22.
[0040] The external components 20 include a charging system 39, which consists of a chair pad 32 and a base station 50; a remote control 40; and a clinician's programmer 60. The chair pad 32 has an antenna/charging coil 34 which is electrically connectable to the base station 50 with extension 36 (or, alternatively, a chair pad cable) and communicates with the microstimulator electronic subassembly 14 with a bidirectional telemetry link 48. The base station 50 has an external, medical grade AC adapter which receives wall AC power 52 through an extension 54. The remote control 40 can send and receive communication from and to the base station 50 through an Infrared Data Association or IrDA interface 42. (IrDA is a standard for transmitting data via infrared light.) The remote control 40 also communicates with the clinician's programmer 60 through an IrDA interface 44 and communicates with the microstimulator electronic subassembly 14 with an RF telemetry antenna 46 through the bidirectional telemetry link 48′. The clinician's programmer 60 may also communicate with the microstimulator electronic subassembly 14 through the bidirectional telemetry link 48″. The base station 50 also communicates with the clinician's programmer 60 through an IrDA interface 45. The bidirectional telemetry link 48 is also known as the FSK (Frequency Shift Key) telemetry link, or RF telemetry link. In addition, the charging system 39 has a forward telemetry link 38 which is also known as OOK-PWM (On/Off Keying�Pulse Width Modulation), or inductive telemetry link.
[0041] The surgical components 30 illustrated in FIG. 1 include the BPB implant tools 62 and an external neurostimulator 64. The implantable BPB microstimulator 10 is inserted through the patient's tissue through the use of appropriate surgical tools and, in particular, through the use of tunneling tools, as are known in the art or as are specially developed for purposes of implantable BPB stimulation systems.
[0042]FIG. 2 (2-1; 2-2) shows, in accordance with the present invention, a block diagram of a base station 50 and chair pad 32 charging and communication system. The base station is connected to the chair pad via a chair pad cable 414.
[0043] The base station/chair pad system is used to carry out four operations: (a) transcutaneously charging the rechargeable battery within the body implanted BPB stimulator using inductive coupling (b) transcutaneously providing recovery to the rechargeable battery in the BPB stimulator when the battery has been depleted down to zero volts, i.e., zero volt recovery (ZVR) (c) transcutaneously communicating with the BPB stimulator using frequency shift keying (FSK) and (d) transcutaneously communicating with the BPB stimulator using on-off keying (OOK).
[0044] The difference between operations (a) and (b) is that, in case (b), the rechargeable battery has been depleted down to zero volts. In that case, normal recharging will not work and it is necessary to use the ZVR procedure.
[0045] In order to enable the various operations, the chair pad 32, as shown in FIG. 2-2, has two coils: (a) an antenna/charging coil- 34 and (b) a booster coil 419. The antenna/charging coil 34 is used for inductively charging the rechargeable battery in the stimulator and also for performing forward and backward FSK or forward OOK telemetry communication with the stimulator. The booster coil 419 is used to perform the ZVR procedure. The chair pad 32 further includes a coil shield 417 over the antenna/charging coil 34 and booster coil 419. Also included in the chair pad 32 is a chair pad printed circuit board (PCB) 416.
[0046] Referring to FIG. 2-1, an external AC adapter 400 is used to power the Base Station 50. The base station includes a battery backup 403 and a microcontroller 402 which can control the real-time clock (RTC) 409 and user interface displays such as user buttons/keys, LCD display(s), LED display(s) or an audio alarm. The audio alarm may be used to indicate when charging is complete or the alarm may be use as an signal indicator for other significant system events or modes. The microcontroller 402 is also connected to a coil driver amplifier 408 which controls the ZVR charging via connection 412 and the chair pad cable 414. In addition, the microcontroller 402 controls another coil driver 406 for controlling forward FSK and OOK telemetry communication and charging via connection 413 which is further connected to the chair pad cable 414. Through the same connection 413, reverse RF telemetry signals are received and amplified by telemetry receiver 407. These signals are then processed by the microcontroller 402. The microcontroller can communicate with an external clinician programmer 410 (or remote control) via an IrDA converter 405. The microcontroller may also receive data from various sensors via connection 411, which sensors may be located on the chair pad PCB 416. For instance, one type of sensor is a temperature sensor or a thermocouple.
[0047]FIG. 3A shows, in accordance with the present invention, a diagrammatic representation of the chair pad, electrical architecture. The representation shows the following chair pad components: a PCB board 416, the charging coil 34, the booster coil 419, and coil shield 417. The printed circuit board (PCB) 416 carries most of the chair pad electronics including a temperature sensor 418. The temperature sensor is used to ensure that the exterior surface of the chair pad 32 (shown in FIG. 3B) is at 41� C. or less. The interior of the chair pad may, however, vary in temperature ranging from 0� to 65� C.
[0048] All of the electronic components in the chair pad are contained on the chair pad PCB 416, with the exception of the antenna/charging coil 34, the booster coil 419 and coil shield 417. To ensure against electrical short circuit from moisture and liquids, the chair pad housing 633 (shown in FIG. 3B) contains all of the chair pad electrical components including the charging coil, the booster coil, and the chair pad PCB.
[0049]FIG. 3B shows, in accordance with the present invention, an illustration of the outer shape of the chair pad 32, which is approximately rectangular. The recharge coil assembly 427 (shown in FIG. 4A) contains the antenna/charging coil, the booster coil and a coil shield, which are all fully encapsulated in polyurethane foam housing 633 of chair pad 32. Because the coils are fully encapsulated by the foam housing 633, there are no access test points to verify that current is flowing in the charging and booster coils. The polyurethane foam housing provides a number of functions. For instance, the polyurethane foam housing 633 is UV resistant and flame retardant and forms a outer skin to keep Water and dust from the recharge coil assembly electronics. The foam housing also insulates the user from heat generated by the embedded electrical components in the chair pad and prevents the surface temperatures exterior to the polyurethane housing 633 (and the chair pad surface) from exceeding allowable surface temperatures. Further, the polyurethane housing protects the chair pad components from shock and vibration and contributes, in some measure with the outer padding, to provide a comfortable and compliant sitting surface. The chair pad housing 633 and electronics are preferably designed to meet a target chair pad surface temperature (including an outer padding around the chair pad foam housing 633) that is less than about 41� C., without employing forced air-cooling or ventilation holes.
[0050] Padding may be placed around the polyurethane foam housing, which padding may be fabricated from a washable fabric or batting sewn together in a loose quilting format and be from about 0.75 inches to 1.00 inches in thickness in an uncompressed state. An exterior chair pad slipcover (not shown) may go over the padding. The slipcover may be constructed from a durable, washable upholstery grade fabric and should provide a relatively comfortable surface to sit on for extended periods of time, for instance, up to several hours. An opening in the slipcover should permit the cable to exit allow insertion of additional or replacement padding, when needed. Thus, although the foam housing, as shown in FIG. 3B, and coil assembly inside is, for the sake of simplicity, referred to herein as the �chair pad,� it is emphasized that the actual �chair pad� used by a patient will usually include the extra padding and the slip cover.
[0051] The chair pad cable, 420 which is preferably longer than 150 cm, extends from the chair pad 32 and can be connected to the base station. The end of the cable can have a 24 pin male connector 423. The chair pad cable 420 can be molded into the polyurethane foam housing 633 to provide strain relief to the cable at the point of extension from the foam housing 633. In one embodiment, the chair pad may have dimensions which are about or smaller than 50 cm by 50 cm by 15 cm and weigh less than 5 kilograms. These are preferred dimensions and weight for the chair pad that work well for the average patient, although other pad shapes and dimensions may also work.
[0052] In use, the chair pad 32 may be placed on a chair and a patient, who has an implantable BPB microstimulator, can sit on the chair pad between about 15 minutes to half an hour a day to recharge the battery in the microstimulator. Generally, however, 15 minutes of charging may be sufficient.
[0053]FIG. 4A shows a mechanical illustration of a recharge coil assembly 427 with chair pad cable 420 having connector 423.
[0054]FIG. 4B shows a more detailed, exploded view of the components of the recharge coil assembly shown in FIG. 4A contained in the chair pad. The various components of the recharge coil assembly 427 include a cover 421 that may be made from injection molded polypropylene, a chair pad cable assembly 420 that links the chair pad to the base station, a coil spool 424, a shield flex circuit 425 (�coil shield�) and a nest 426. Another coil shield 417 is used with shield 425 to �sandwich� the coil spool 424. The antenna/charging coil and booster coils are both contained in the coil spool 424 that is preferably made from injection molded polypropylene. Polypropylene is used for its low cost and excellent resistance to deformation under repetitive stress conditions.
[0055] The nest 426 is constructed from an injection-molded housing made from a polycarbonate-acrylonitrile butadiene styrene (PC/ABS) alloy. The nest is used to house the chair pad FR4 PCB assembly 422. Because it is necessary to minimize mechanical deflections of the PCB 422 to avoid damaging or loosening any of the circuit components, the nest 426 must be constructed of high strength and stiffness material to withstand compression/flexion loading of the recharge coil assembly 427. The box shape of the nest 426 enhances its stiffness. The PCB 422 is preferably at least about 0.093 inches thick to enhance its stiffness. In the complete coil assembly 427, the PCB is located between the nest 426 and the complementary rectangular portion of the cover 421.
[0056] The cover 421 is used to position the nest 426 and the spool 424 relative to each other. The assembled coil spool 424, cover 421 and nest 426 protect the coil spool 424 and the coils from being damaged by external objects that might penetrate through the polyurethane foam housing.
[0057] The coil shield flex circuit assemblies 417 and 425 (�coil shields�) consist of polyimide film substrate with copper traces placed on one side and pressure sensitive adhesive on the opposite side. The coil shields 417 and 425 are preferably used to reduce the capacitively coupled currents to the patient and minimize the possibility of electric shock. The coil shields are grounded to provide a low impedance pathway to ground for the capacitively coupled currents generated in the antenna/charger coil and the booster coil, which operate at high voltages and frequencies. The coil shields 417 and 425 therefore reduce the leakage currents that can be induced in the patient during charging, telemetry and ZVR operations.
[0058] The flexible chair pad cable 420 with connector 423 may be permanently attached to the chair pad, with the connector 423 placed distal end of the cable. The base station housing can have a complementary female receptacle for connection to the chair pad. The chair pad cable connector 423 can also contain a loop back connection that is used to immediately indicate whether the chair pad is connected to the base station.
[0059] Through the connector 423 and chair pad cable 420, the base station can directly access the chair pad temperature sensor, the current sensing resistor of the charging coil, the voltage sensing circuitry of the booster coil, the relay for the FSK antenna operation of the charging coil, the chair pad connector feedback loop, chair pad cable shield and chair pad coil shield. In addition, the connector provides indirect access, through the particular tuning circuitries, to the charging coil and booster coil.
[0060] The chair pad cable 420 is shielded and may contain six twisted conductor pairs. In a preferred embodiment, five conductor pairs may be 26 AWG copper stranded construction containing 65 strands of 44 AWG tinned copper wire while the remaining conductor pair may be 24 AWG copper stranded construction containing 105 strands of 44 AWG tinned copper wires. The cable can be shielded with a circumferential braid of 40 AWG tinned copper strands with 90% minimum coverage. The cable jacket insulation may be a thermoplastic elastomer (TPE).
[0061] In the recharge coil assembly 427, the dimensions of the charging coil contained within the assembly may be optimized to deliver inductive coupled power to a microstimulator that is implanted deep in body tissue, for example, at an implant depth of between 12 to 15 centimeters and using a preferred frequency of about 127 KHz. To achieve these targets, the charging coil 34 preferably has 24 turns of multi-stranded Litz wire wrapped around the 200 mm inside diameter coil spool (�recharge coil spool�). The resulting diameter of charging coil is about 10 to 12 centimeters. The quality or Q of the resonant circuit is preferably about 160.
[0062] Referring to FIG. 2 (2-1; 2-2), in operation, the chair pad may be used to charge the rechargeable battery in the implanted stimulator. For normal microstimulator battery charging, i.e., where the rechargeable microstimulator battery has not be depleted down to zero volts, the base station 50 may power the charging coil 34 inside the chair pad 32, which generates a magnetic field and thereby inductively recharges the battery in the microstimulator.
[0063] To implement forward and reverse radio-frequency, FSK telemetry communication with the microstimulator, the charging coil 34 is used as an RF antenna. As an emergency back-up, forward OOK telemetry is also provided wherein the charging coil also operates as an antenna. The data transfer rate with OOK, however, occurs at a much lower rate than with FSK communication because, with OOK, the information is carried by on-off amplitude modulation.
[0064] Referring to FIG. 3A, in accordance with the present invention, the charging coil 34 can be used to generate a magnetic field of sufficient strength to inductively charge the microstimulator battery at a carrier frequency of 127 KHz. The maximum magnetic flux required along the microstimulator axis to achieve battery recharging at a rate of C/2 (half the maximum charge) is estimated to be 1.61 Gauss (rms) at 127 KHz. To achieve a recharging rate of C/8, the magnetic flux required is about 0.4 Gauss (rms). To achieve satisfactory charging at a microstimulator implant depth of between 10-15 centimeters inside a patient's body, the charging coil may be driven with a 4.25 Ampere (rms) current. The resulting magnetic flux of 0.93 Gauss (rms) can be produced at a point 15 cm vertically from the center plane of the charging coil, 5 cm radially away from its axis (maximum charging distance) and up to a 30� off-axis misalignment along the microstimulator, which magnetic flux can satisfactorily recharge the battery in the implanted microstimulator at the target charging distance. When a 4.25 Ampere (rms) current is applied to the charging coil, this can generate a magnetic flux of 1.81 Gauss (rms) at a point 10 cm vertically away from the center plane of the charging coil and 5 cm radially away from charging coil axis with up to a 30� off-axis misalignment. The resulting field can recharge the battery in approximately 2 hours.
[0065] The antenna/charging coil 34 may also be used for forward OOK telemetry, employing a narrow bandwidth around 333 Hz. When the antenna/charging coil is used, however, as an FSK antenna, it is necessary to increase the bandwidth. Bandwidth can be increased by connecting a damping resistor 420 (as shown in FIG. 3A) in parallel to the charging coil 34. By using the charging coil for RF and OOK telemetry communication, the base station can communicate with an implanted microstimulator. During such communication, the microstimulator with its internal coil may be centered at distances of 15 cm away from the chair pad charging coil.
[0066] An important function of the charging/communication system is the ability to recover the microstimulator when the rechargeable battery is completely dead. During such zero-volt recovery (ZVR), the microstimulator is defaulted to a depletion mode. A short duration, high amplitude magnetic field at a frequency of approximately 1.2 MHz is delivered to the microstimulator in order to set the microstimulator charging circuitry into a regular charging mode. The chair pad contains a booster coil 419 which is used to generate the short duration (less than 1 sec) magnetic field in the 1.2 MHz range.
[0067] In the event, however, that the microstimulator battery voltage should drop to a complete depletion level or �zero volt� mode, the microstimulator circuitry that controls the charging frequency will default to a state that causes the resonant frequency of the microstimulator circuitry and receiver coil to shift to about 1.2 MHz which is the Zero Volt Recovery (ZVR) frequency. Before normal charging of the microstimulator battery can begin, the base station temporarily operates in ZVR mode. In this ZVR mode, the booster coil is driven at the ZVR frequency (1.2 MHz), which resets the battery charging circuitry in the microstimulator to 127 KHz, by activating the front-end switches of the microstimulator setting the microstimulator to a trickle charge mode.
[0068] The booster coil 419 may have 6 turns of multi-stranded Litz wire in 2 layers of 3 turns each, wrapped around the charging coil inside the coil spool 424. The Litz wire used in the booster coil 419 can be the same kind used for the charging coil 34. The minimum magnetic flux required along the microstimulator longitudinal axis for ZVR is estimated to be 175 mGauss (rms) at 1.22 MHz.
[0069] Driving the booster coil at 2.9 Amperes (rms) can produce a magnetic flux of 175 mGauss (rms) at a point 15 cm vertically from the center plane of the booster coil, 5 cm radially away from the booster coil axis (the maximum ZVR distance) with up to 300 of axis misalignment relative to the microstimulator. This satisfies the ZVR distance between the chair pad and microstimulator. The chairpad can operate in this mode for more than 1 second.
[0070] As shown in FIG. 3A, the chair pad PCB 416 contains the circuitry required to interface with the coils 34, 419 and coil shield 417 and with the base. station which provides the power to drive the coils via the chair pad cable. The PCB 416 may contain the following circuits: temperature sensor 418 for the chair pad, a tuning circuitry for the charging coil, a tuning circuitry for the booster coil, a frequency damping system for the FSK telemetry antenna, a current sensing circuitry for the charging coil and a voltage sensing circuitry for the booster coil. Additionally, the PCB may contain electrostatic discharge (ESD) protection circuits 471, 472, 473 and 474 that are only active when electrostatic discharge is present.
[0071] Temperature sensing of the chair pad PCB is accomplished by employing a sensor such as an IC mounted thermocouple 418 to sense PCB temperatures in a range of between about 0� to 100� C. The temperature sensor 418 can monitor temperatures inside the PCB cavity that is, formed by the nest and cover. The base station is directly linked to this temperature sensor (Temp_Out) through the chair pad cable. (Temp_VCC indicates the VDC voltage supplied to the temperature sensor.)
[0072] As shown in FIG. 3A, the tuning components for the charging coil are placed on the chair pad PCB 416. The tuning components consist of two capacitors, a parallel capacitor (Cp-tune) 421 and a series capacitor (Cs-tune) 422 relative to the charging coil 34. These tuning capacitors are medical-grade, high voltage capacitors rated to operate above 1000 Volts (rms) and 4.25 Amperes (rms) at 127 KHz. These capacitors resonate the antenna/charging coil 34 at 127 KHz +/−6 KHz, when the coil is operated during battery charging, FSK telemetry or OOK telemetry. In addition to tuning the charging coil, the tuning capacitor circuitry provides an impedance matching network of 50 Ohms to the base station amplifier, when the coil is operated in inductive charging mode, thereby improving the efficiency of power transfer. In addition the impedance matching networks also contribute to reducing the voltage and current present in the charging coil to safer levels of 25 Volts at 0.5 Amperes, as compared to in the base station, where 1000 Volts at 4.5 Amperes are seen. In this way, the use of a separate base station and external interface device, in this instance the chair pad, contributes to a safer system, since higher voltages and currents are isolated to the separate base station.
[0073] Similarly, tuning components for the booster coil are placed on the chair pad PCB. The tuning components consist of two capacitors, a parallel capacitor (Cp-tune) 423 and a series capacitor (Cs-tune) 424, relative to the booster coil 419. The tuning capacitors are preferably medical-grade, high voltage capacitors rated to operate above 400 Volts (rms) and 3 Amperes (rms) at 1.2 MHz. These capacitors resonate the booster coil 419 at 1.2 MHz +/−0.1 MHz, when the coil is operated during ZVR. In addition to tuning the booster coil, the tuning capacitor circuitry provides an impedance matching network of 50 Ohms to the base station amplifier.
[0074] A damping system is required to broaden the available bandwidth, when the antenna/charging coil 34 operates as an FSK telemetry antenna. To accomplish this, a 2 KOhm damping resistor 420 may be connected in parallel to the charging coil 34 using a relay switch 431 rated for 2 Amperes. When the damping resistor 420 is connected in the charging coil circuit, the bandwidth of the charging coil circuit is increased to 12.37 KHz. The base station has direct control of the relay switch 431 and closes the circuit only during FSK telemetry operations. When the relay switch 431 is �open,� the chair pad is in the charging mode.
[0075] Because there is no outside electrical access points to the charging coil or the booster coil, the system has means for measuring the coil currents or voltage (and, hence, the power) in the charging coil 34. In the charging mode, the current in the charging coil 34 is measured using the voltage across a 0.01 Ohm current sensing resistor 432 that is connected in series with the charging coil 34. The voltage across this resistor 432 is directly readable by the base station through the chair pad cable connector.
[0076] The voltage across the booster coil 419 is measured using a resistor voltage divider and rectifier circuit shown in FIG. 3A comprising diodes 436, 437 and resistors 434 and 433. The output signal of this voltage sensing circuitry is directly readable by the base station via the chair pad cable connector (at BCOIL_V).
[0077]FIG. 5 depicts, in accordance with the present invention, a block diagram of the analog section of the battery charger in the base station, which includes three subsections, a power converter section 600, a shut-down circuit section 610 and an audio section 660. The shut-down circuitry represented by block 602 uses a comparator circuitry that is designed to trigger when a measured parameter (such as predetermined temperature level) exceeds 10%. A thermocouple 611 may be used as a temperature sensor. The audio circuitry section 660 employs a speaker 661 that can emit an audible tone or sound to signal a significant system event such as the end of charging.
[0078] In a preferred embodiment, the power converter section 600 generates several D.C. voltages from an input +24 VDC supplied by an external AC adapter. The +24 VDC input supply voltage passes through a power switch and through a 50 VDC, 2 ampere slow-blow type fuse 601. All charging electronics in the base station are connected after this slow blow fuse. The AC adapter voltage following the fuse 601 may be monitored by software such that when the voltage rises or falls 20% above or below the nominal +24 VDC, it is sensed by the over voltage/current shut-down comparators shown as block 602, thereby shutting down the coil driver amplifiers and power supply. The power converter section 600 generates a +3.3. VDC for the digital section of the battery charger, a +12 VDC for the analog section of the charger and telemetry analog sections (from linear regulator 606), and a high power variable output that ranges from +7 to +20 VDC for the coil driver power supply from variable output power source 612.
[0079] In order to minimize power consumption and reduce generation of heat, a switching regulator is preferably used for the digital circuitry. However, during communication with the microstimulator, the switching frequency may interfere with the telemetry reception on the telemetry PCB in the base station. Because of this, a +6.0 VDC is generated by a 200 KHz switching regulator 603 followed by a linear regulator 604 which produces +3.3 VDC. A parallel linear regulator 605 produces a +4.5 VDC which output is OR connected with the +6.0 VDC output. Linear regulator 605 is normally shut off when there is no communication. During normal operation, when there is no communication, both regulators are turned on. Since the switching regulator 603 voltage of +6.0 VDC is higher than the linear regulator 605 of the +4.5 VDC, the +6.0 VDC output provides most of the power, thereby providing more power more efficiently. During communication with the stimulator, which may occur once every minute, the +6.0 VDC switching regulator is turned off, allowing the +4.5 VDC linear regulator to provide power momentarily during the communication interval, which interval is typically less than a few seconds. This provides a less efficient, but a lower noise power generation during communication. Any loss of power efficiency during communication is not critical since it is only for a short duration.
[0080] Because the chair pad may not be located in exactly the same placement relative to the microstimulator during each inductive charging, the base station charging circuitry may be designed to automatically tune the charging circuit in order maximize power to the charging coil. This is accomplished by (a) maximizing the current in the charging coil and (b) maximizing the efficiency of the charging coil amplifier-within the base station section 600 by tuning the frequency.
[0081] The efficiency of an amplifier is defined as output power divided by input power. Input power to the amplifier is measured by the product of the input supply voltage multiplied by the input supply current. Both the input supply voltage and input supply current can be measured. The current to the charging coil may be measured by measuring a current sampling resistor that is placed in series with the charging coil. Since the current through, the sampling resistor is measured and known, the power dissipated by the charging coil can be obtained from the relation: power equals the square of the input current multiplied by the known coil resistor value. The tuning frequency may be software controlled.
[0082] During inductive charging of the rechargeable battery, it may be necessary to reduce the charge field strength caused by induced eddy currents in the stimulator case to prevent overheating. To accomplish this, the variable output power supply 612 may be used. In the example shown, a +7 to +20 VDC may be selectable in 100 step increments.
[0083]FIG. 6 shows, in accordance with the present invention, a block diagram of the battery charger PCB analog section for the coil drivers located in the base station. A charging coil driver is required to provide forward and reverse FSK telemetry to the microstimulator within a carrier frequency, preferable in the range of 121 to 133 KHz, and a modulating signal frequency range of +/−4 KHz. To obtain FSK telemetry, an amplifier in the base station must drive the antenna/charging coil via connection 638. To increase the bandwidth, the Q of the charging coil and amplifier filter inductance is reduced to approximately about 10. During FSK communication (both transmit and receive), a software-controlled switch relay 431 is used to connect a damping resistor 420 in parallel to the charging coil and the amplifier inductor 636. Reverse FSK telemetry is also received via the damped charging coil 636. The receiver on the telemetry PCB connector 640 interfaces with a 50 Ohm impedance matching network 635.
[0084] For OOK operation, the battery charger is required to amplitude modulate the charge field at a maximum rate of 166 beats per second. This can be implemented by enabling and disabling the amplifier at the required data rate. Because the amplifier is impedance matched with the chair pad coil while the amplifier is active, power can be efficiently transferred to the coil. However, when the amplifier is turned off, the impedance matching is poor and can result in a sub-optimal power transfer out of the coil and a slow fall time. To improve the data transfer rate of the amplifier when the amplifier is turned off, the impedance matching network 635 is switched on using a transistor at the input of the class E network 637. This restores the 50 Ohm impedance matching between the amplifier network and charging coil while the amplifier is off and produces maximum power transfer out of the coil at an increased data rate. While the amplifier is turned on, the OOK network is switched out of the Class E network and is not operational during that time.
[0085] The booster coil amplifier is required to drive the chair pad booster coil via connection 641 with sufficient power to re-initialize the battery charger tuning circuitry in the stimulator to 127 KHz during ZVR, while the charging coil in the chair pad is at a maximum vertical separation distance of 15 cm from the microstimulator with a maximum horizontal offset from the center axis of the booster coil of 5 cm. The estimated power to the coil necessary to achieve re-initialization is approximately 11.25 Watts at 1.2 MHz. The booster coil amplifier may be designed to drive the booster coil with a maximum power of 20 Watts at 1.2 MHz. The booster coil amplifier power can be adjusted using the same variable DC/DC converter used for the charging coil amplifier.
[0086] Automatic tuning (impedance matching) is used to maximize the efficiency and power to the booster coil in the same manner as used for the charging coil. A 50 Ohm impedance matching network 634 may be used in the booster coil amplifier circuit that is placed serial to a class E amplifier network 642. The identical method that is used to measure input power to the charging coil amplifier is also used for the booster coil amplifier. The booster coil voltage is measured using a simple voltage divider and rectifier circuit in the chair pad. This voltage is monitored and used to calculate output power to the booster coil.
[0087] It can be appreciated that the charger and communications system comprising the chair pad and base station have been discussed in the context of charging and communicating with a microstimulator such as a BPB microstimulator. However, such a charging and communication system may be employed with any implanted stimulator that is larger in size, than a microstimulator. In general, a BPB microstimulator does not exceed a maximum lengthwise dimension of about 3.5 cm and has a width that is appreciably smaller, e.g., less than about 5 mm.
[0088] Furthermore, the charging and communication system of the present invention may also be adapted for use for other external device configurations. For example, instead of a chair pad, the electronic components may be used in an external charging/communications device (an �external interface� device) that is to be placed over other parts of the body, such as the back, over a limb, such as an arm or leg, or over the surface of the head. When used for the head, the chair pad may, instead, be configured and used as a pillow. The general concepts employed and components used inside the chair pad may be readily be adapted for use in a pillow or other external charging/communications device.
[0089] In yet a further embodiment of the present invention, the external charging/communications component may integrate those components included in the chair pad, e.g., the booster coil, the antenna/charging coil and chair pad PCB with components that are in the base station, such as the power converter, shut-off safety circuitry and the microcontroller, thereby eliminating a separate base station device in the system.
[0090] While the invention herein disclosed has been described by means of specific embodiments and applications thereof, numerous modifications and variations could be made thereto by those skilled in the art without departing from the scope of the invention set forth in the claims.
Referenced by
Legal Events
Original Image


text:
You are a fan of fast halal ? Come to discover the best forums fast halal on the internet and share your experience with a community fan of fast halal . You can also build a forum fast halal and create your online discussion.
The Home Of All Things 4 Wheeled And Fast
Make fast easy money in no time with our guides and staff ready to help!
flipchat101.tk, #fast, easy, money, guides, kinds, ready, help

subset: commoncrawl


text: Quick View Undiet: Eat Your Way To Vibrant Health by Meghan Telpner April 2, 2013 | Trade Paperback In stock online from $15.67 A toolbox for a complete lifestyle transformation -- from a holistic nutritionist with an inspiring and practical message. Is it possible to look and feel your absolute best while eating the most delicious food and avoiding the trap of diet deprivation and calorie counting? Meghan… See Details $16.49 / $15.67 lowest price*, or used from $20.49was $24.99


text:
Ticket Request for Presidential Panel on Climate Change
March 24, 2015
Cambridge, Mass.
Universities and the Challenge of Climate Change
March 17, 2015
Tsinghua University Global Vision Lecture Series, Beijing, China
President Faust at Morning Prayers
March 6, 2015
Cambridge, Mass.
Support for seven from president's climate fund
February 11, 2015
Harvard Gazette
Cambridge, Mass.


text:
A short about the Letterpress.
Thanks to AUCB Visual Communications.
Shot on 550D.
Loading more stuff…
Hmm…it looks like things are taking a while to load. Try again?
TM + © 2015 Vimeo, LLC. All rights reserved.
Made with k in NYC.
Your content preferences apply to any Vimeo videos you view. Learn more about content preferences in our FAQ.


text:
Economy strong in rural parts of 10 states
OMAHA, Neb. (AP) -- The economy in rural areas of 10 Midwest and Plains states remains strong as healthy rainfall this spring helps crops get established.
The overall economic index for the region grew to 60.5 in June from May's 58.8. That suggests growth in the months ahead.
Get the full story
Subscribe or Log In
Are you a newspaper subscriber but you don't have a Digital Access account yet? Activate your account.
You will need your subscription account number and phone number. Not sure if you have an account? Email us or call (605) 996-5514 and we can help you.


text:
100 Cummings Center, Suite 419-E
available to Lease
in Beverly, MA
posted on April 23, 2013
Dave Ryder
Peter Abair
Director of Economic Development
Property Description
Lab includes 11 fume hoods, benches and four lab sinks with backflow preventers and acid neutralization tanks. Skylights throughout. Free covered parking nearby. Adjacent to 10,000-lb freight lift serving priamry loading area. Easy access to highways 128/95/1.
$22.95 per square foot
Listing Details
|Property Use
|This property has parking available
|Square feet available
|Property Owner
|Cummings Properties
Images (Click on images to view larger size)
About MassBio
MassBio is a not-for-profit organization committed to advancing the development of critical new science, technology and medicines that benefit people worldwide.
Founded in 1985, MassBio represents over 600 biotechnology companies, academic institutions, research hospitals, and service organizations involved in life sciences and health care, and works to advance policy and promote education, while providing member programs and events, industry information, and services.
Latest Updates
- March 23, 2015 UMass Medical School's Dr. Collins to Receive MassBio's Leading Impact Award
- March 18, 2015 Third Rock Co-Founder Mark Levin to Receive MassBio's Innovative Leadership Award
- March 12, 2015 Top 10 U.S. Biopharma Clusters: GEN - 3/9/15
- March 12, 2015 Boston-Cambridge biopharma ranked first in U.S.: Boston Herald - 3/12/15
- March 9, 2015 2015 Women to Watch in Science & Tech Nominations Deadline Approaching


text:
£695,000 Offers Over
Offers Over £695,000
5 bedroom detached house for sale 14 Brunstane Mill Road,Joppa,Edinburgh,EH15 2LYThis stylish and substantial family home offers the perfect balance of tranquil privacy and city living. More details ›
£695,000 Offers Over
Offers Over £695,000
5 bedroom semi-detached house for sale Abercorn Terrace, Edinburgh, MidlothianAn immaculately presented five bedroom, semi-detached Victorian villa situated in the popular residential area of Joppa. The property is arranged over three floors, has been fully refurbished to an exceptional standard and benefits from fantastic views extending out over the Firth (contd...) More details ›
£575,000 Offers in Excess of
Offers in Excess of £575,000
6 bedroom detached house for sale 20 Milton Road East, EdinburghAn imposing detached, Victorian villa set on a prime corner plot in Joppa. Drawing room/bedroom 6, family room, dining room,conservatory, kitchen, utility room, store room, 5 double bedrooms,two bathrooms, separate WC, garden, garage & extensive off road parking. More details ›
£550,000 Offers in Region of
Offers in Region of £550,000
4 bedroom semi-detached house for sale 67 Brunstane Road, Brunstane, Edinburgh, EH15 2QSWell-presented and exceptionally-spacious, four-bedroom, stone-built period property, located in the Brunstane area, east of Edinburgh city centre. An impressive family home with generous room sizes, there are also private front and rear gardens, multi-car driveway, and a double garage with an in... More details ›
Added on 04/03/2015 by MOV8 Real Estate, Scotland, Head Office. Call: 0131 291 0068 Local call rate
£499,000 Offers Over
Offers Over £499,000
4 bedroom terraced house for sale 3 Ormelie Terrace, Joppa, Edinburgh, EH15 2EXImpressive 4 bedroom house with lovely front and rear private gardens with stunning sea views. More details ›
£490,000 Offers Over
Offers Over £490,000
4 bedroom terraced house for sale 16 Seaview Terrace, Joppa, Edinburgh, EH15 2HD16 Seaview Terrace is a stunning four-bedroom terraced house with breathtaking, uninterrupted views of the Firth of Forth in the sought-after and prestigious Joppa area of ... More details ›
£475,000 Offers Over
Offers Over £475,000
4 bedroom terraced house for sale 7 Pittville Street, Portobello, Edinburgh, EH15 2BZImpressive, rarely available 4/5 bedroom terraced Georgian house with garden and garage More details ›
£415,000 Offers Over
Offers Over £415,000
3 bedroom cottage for sale 43 Regent Street, Portobello, Edinburgh, EH15 2AYStunning three bedroom semi detached cottage in the prestigious Portobello district of Edinburgh More details ›
£375,000 Offers Over
Offers Over £375,000
4 bedroom terraced house for sale 36 Esplanade Terrace, Joppa, Edinburgh, EH15 2ESA wonderful opportunity has arisen to purchase a fantastic and immaculately presented four bedroom terraced house with stunning views stretching across to Fife over the Firth of ... More details ›
£375,000 Fixed Price
Fixed Price £375,000
3 bedroom terraced house for sale 76 Brunstane Road, Edinburgh, EH15 2QRExceptional and well presented traditional stone fronted terraced villa with front garden and enclosed rear garden, located in the desirable residential area of Joppa. The property offers spacious and flexible family accommodation with some modern features complemented perfectly with many origina... More details ›
Thank you.
Your information has been sent to 0 out of 11 agents who have property to sell in the area you specified.


text:
Written on: 10/09/2005 by Rex Limm (1 review written)
Good Points
Great phone.
Bad Points
I don't have any problems with it.
General Comments
I have a Sharp GX32 mobile phone, and people know it's different, but let me tell you, it's the best phone I have ever used. Well for one, I used the Sharp GX10 with Vodafone live settings and pxt settings; I rang up Vodafone and they enabled the pxt settings "I said it was a Sharp GX30". Ha, now I have Vodafone live, and I can download the same as the normal Vodafone live phones. I got a larger battery for mine as well. This was $100 cheaper. 10/10. This phone is great!
As rated by real users
" I got this phone more then 8 years " Read More
Written by davidsilva746
" Best phone I've ever had " Read More
Written by AliceWayte
" This Nokia 9210 mobile phone can be a real boon as it can give you ... " Read More
Written by stuart63
" Johnk777 " Read More
" I bought this nice looking phone in Nairobi-Kenya in July 2003 from... " Read More
Written by Peter Njoroge Mwangi


text:
Tape Media
Tape Drives
Removable Disk Storage
Imation RDX
Imation RDX Bundles
Imation RDX Media Secure
Quantum RDX
Quantum RDX 8000 Library
HP RDX Removable Disk
Tandberg Data RDX QuikStor
Software Backup Novastor
Qualstar Tape Libraries
Overland NEO Tape Libraries
Overland NEOs StorageLoader
Overland NEOs T24 Loader
Overland NEOs T48 Library
Overland NEO XL Series
Tape Drive Autoloaders
HP StoreEver Tape Libraries
Tandberg Data Storage Library
Quantum SuperLoaders
Quantum Scalar i40
Overland Storage NEO S
NAS Storage
SnapServer NAS
SnapServer XSD 40 NAS
SnapServer XSR NAS Series
SnapServer XSR 40 NAS
SnapServer XSR 120 NAS
SnapServer DX DynamicRAID
SnapServer DX1/DX2 Bundles
SnapScale X2 Clustered NAS
SnapScale X4 Scalable NAS
Snap Server Support
Snap Server Software
ioSafe Disaster Proof NAS
LenovoEMC StorCenter NAS
LenovoEMC px4-400d/ px4-400r
LenovoEMC px12-400r NAS
LenovoEMC px12-450r NAS
Pegasus RAID Storage for MAC
Pegasus2 RAID Thunderbolt
SnapSAN S1000 Storage Array
Nexsan Storage
Adaptec SCSI HBA Cards
ATTO ExpressSAS RAID
ATTO 10GbE NIC Cards
iSCSI/ FC HBA Cards
SATA/ SAS HBA Cards
Cables & Terminators
Barcode Labels
Turtle Storage Cases
HP Toner Cartridges
Imation Defender Flash Security
Repair Services
Reconditioned Tape Drives
Quick Comparision Table
How to Determine the Apropriate Solution
To decide whether LTO or RDX drives are right for you, you
should consider the following:
In General, RDX is most suitable for clients who want to
backup smaller datasets and don't need to archive lots of data for an extended
period. LTO is a better option for clients with large datasets to backup,
and / or who want to archive data for extended periods. Basically for a
smaller data set, say 100GB daily backup / 400GB weekly with 6 tapes /
cartridges in rotation the RDX solution will provide a lower cost solution.
However let's say you're backing up 400GB daily / 1.3TB weekly with 12 tapes /
cartridges, then the LTO Tape Solution is going to provide a lower cost
Feel free to contact a BackupWorks Account Rep for a free
consultation and determine what best fits your needs. Call us at 866 801


text:
Learn more about TreeHouse Interactive's innovative solutions; Reseller View™, Marketing View™, and Sales View™.
View Data Sheets
Download white papers developed by analysts, industry experts, and other innovators that detail best practices for marketers and channel managers to achieve their business goals.
View White Papers
Read first-hand accounts of how TreeHouse Interactive customers deployed our solutions to grow their partner programs and increase the effectiveness of their marketing automation campaigns.
View Case Studies
Join industry experts, analysts, customers, product managers, and technology innovators as they discuss the latest technologies and product advancements made in partner relationship management (PRM) and marketing automation platform (MAP) solutions.
Channel Webinars Marketing Webinars
Channel WebinarsMarketing Webinars


text:
The Slow Journey From Junk Room To Playroom
About a month ago we were walking around the park and Clara pointed to a dogwood tree and said "wow, that's a big big tree in a big big world." John and I cracked up. So now when I think of Clara's playroom/big girl room, I also think of the big big tree and the big big world of ours. Hilarious. Anyway, back to the room that we have affectionately (and ironically) called the playroom. Remember that it looked like this when we moved in?
And then it got all cluttered up with junk and looked like this?
And then we organized it with the help of my mom and it looked like this?
And then in a Groundhog-Day-esque turn of events a year later it looked like this again (thanks to months of book insanity)?
So we cleaned it one last time and had a big yard sale and auctioned a bunch of book projects off for charity and finally we were left with this:
I may or may not have done a victory dance around the daybed that included some very painful-to-watch moonwalking. Gaining any actual use of this room will be ten times better than the junked up storage room that it has been for the past three Christmases (how are we on year three of living here without reclaiming this room sooner?!). So as we've mentioned a few times, it will serve as a nice little playroom for Clara and eventually will be her big girl room once she graduates from sleeping in her crib.
Why switch rooms? Well, the nursery is so small that adding a daybed basically sucks up all the floor space, so moving her into the larger spare room and keeping the nursery for a possible future kiddo makes the most sense for us). Of course we can always update the nursery for a boy bean if one should come our way someday (heck, kids rooms are our favorite things, so we'd jump at the chance). In fact here's an old mood board we did a while back to show how Clara's room could be adapted for a little dude:
Once all the extra stuff was cleared out and remaining items were pushed to the center of the room, it was time to paint (we had used the room as a secret-book-project backdrop so it was a hot mess). We'd repainted sections of the walls around ten times in there to create different colored backdrops for shooting – and had even painted the trim around one window a moody gray color and stenciled a good portion of another wall. Nothing was safe from the madness of book shots.
Here are a few other oddly-painted angles:
So yes, paint was definitely necessary. First I painted the book-project-window-shade from gray back to glossy white:
Before we pick a wall color we like to think about the other things that'll be going on in the room. So we thought about that for a hot second or two. We actually wanted Clara to have some input, so we asked her what colors she wanted to see in her big girl room. Not surprisingly, she said "pink and red!" (her faves used to be pink and orange, but she moved to pink and red about a month ago).
It'll be nice to have another room in the house with warmer touches in those red and pink tones (you know we love us some green & blue, so that's prevalent in a lot of other areas). It was even cool with John since, in his words "as long as we're not going hot pink all over the living room, I'm down with Clara getting to pick the colors that we use in her big girl room." Mah man. He can build a deck and he's cool with pink.
So just to give you an idea where we're thinking of bringing in those pinks and reds (along with other hits of colors/textures that make a room feel layered and not-too-matchy-matchy) here's a quick picture of some items we've been saving for Clara's big girl room in a little floor-pile together. Floor-piles are the new mood board. Just kidding.
We don't want the room to be red/pink overkill, so using tones like white and navy – along with hits of other colors like yellow and teal – to break things up is definitely on the agenda. And since we wanted the items in the photo above to stand out (instead of being the same color as the walls) we opted for a soft and pretty neutral that we know will be just as workable with any other accent colors down the line. Let's face it, kids can change their favorite colors on a dime, so we don't want our walls to be forever dependent on a red & pink scheme, ya know?
And this time instead of going with a gray-ish neutral (which we've done in a few other rooms) we dipped back into the warmer neutral palette and went for a super soft tan tone. Sort of like tea when you put in too much milk. Mmmm. It definitely still works within our whole house palette, but it's a slightly warmer than some of the other grays.
We were between two extremely similar colors, so naturally, we picked the one with the better name. Totally reasonable tie-breaker move, right? The winner: Alaskan Skies by Benjamin Moore (we bought a gallon of their no-VOC Natura paint in an eggshell finish). The reason it won: we honeymooned in Alaska and took approximately 650,367 photos of the sky while we were there. The sweet little coincidence is that the paint in Clara's nursery is Proposal (also by BM), so I guess we're subconsciously drawn towards wedding/honeymoon related names.
Since I'm the cutter-inner and John's the roller in this relationship (those sound like weird euphemisms) I was totally jealous about John's big erase-the-past roller moves. Meanwhile I'm sitting on the floor painting above the baseboards. Look at me go! Just kidding. And now for a montage of John rolling over all of the randomly painted portions of the room (I wish it was a montage, it's just this picture and the one above).
Not gonna lie, it felt pretty good to watch him go over all of those random colors and stencils on the walls and finally move towards a room that's all one color again. Meanwhile I was inching my way around the room cutting in around the windows, doors and the ceiling. Speaking of which, we didn't paint the ceiling (just left it white) but we're debating adding some soft tone up there someday down the line after the room comes together more. Stay tuned…
I think we're in love with the wall color though. Not too pink, gray, or tan. And not too white, so the window trim, molding (and future crown molding that we'd love to install) pop nicely against it. Just like extra milky tea against a clean white mug. Yum.
So we're currently waiting for the paint to dry so we can move all of the furniture back into semi-normal spots.
Then we'll share more photos along with planning/details/projects as we go! Back in 2010 Clara's first nursery evolved over the course of about 20 posts and two or three months. Can't wait to watch this bare little room take shape…
So that's what we've been up to. What did you guys do this weekend?
If you enjoyed this post, please leave a comment or subscribe to the feed and get future articles delivered to your feed reader.
336 Comments
We used Alaskan Skies in our living room and dining room! I love how the color changes in the different light through the day and night. IMO it's the perfect neutral!!
I love the wall colour you chose! Can't wait to see this room all finished up with that awesome colour scheme!
Woohoo pink and red! Love that it's against a neutral backdrop. And I'm totes jealous of Clara's moose lamp!
My weekend was busy running after my new pit bull puppy. She's a lot of work and makes me realize I'm NOT ready for kids. Haha! But she sure is cute: http://instagram.com/p/UpqcCqEXDb/.
As we speak, I'm also going through the painful process of sorting, clearing, emptying, painting, bringing everything back in our home office room (a.k.a junk room) so I have very vivid in my mind the feeling of despair first and excitement for getting it ready later that you might be experiencing. Can't wait to see the end result!
Exciting. I am so interested on how she takes to her big girl room. My little man will play in his but then at night avoids it like the plague. Scared of the "monsters". I am trying everything but so far it is a no go.
AWESOME!!! Definately going to try that tonight. Thank's for the tip. Breaks my heart to see him scared at night so I cave and let him climb in bed with us which doesn't help matters of course. :-)
Thank you again – I'll let you know how it works tomorrow.
Just be careful- a friend of mine tried that and all it did was convince her daughter even more that the monsters were real. I guess it's all about knowing your kid :)
Ohhhhh yayyyyy….I have been anxiously awaiting Clara's big girl room!!!!
This weekend marked the end of our live (and super dry) Christmas tree. It had already been emptied of all things Christmas but we lacked the time and ability to get it to our city dump. Then my dad took it home with him to be put on his burn pile and we'll light it up when it gets warmer (it's below freezing here). Gotta love bonfires! Along with the tree, we're starting the same process as you (but have a much longer way to go) of taking our super messy room and turning it into my daughter's big girl room. But not only do we have a ton of things to empty out, we then have an entire remodel ahead of us – ripping everything out, new windows, insulation, drywall, paint, everything but the hardwood floors. We'll reuse the trim too, but it'll be painted bright white not baby pink (yes, the room is currently a glittery chocolate brown, wallpaper, and pink trim and doors… NOT done by us!). You can see it here.. http://thebrickbungalow.blogspot.com/2013/01/2013.html I wish we were almost done like you guys!!! Can't wait to see the finished room.. it'll be insipration for us!
Aren't you sad to abandon Clara's cool closet?
I have to confess, the idea of turning the nursery over to a new baby because the room is small for a twin bed confuses me. Won't you eventually put a bed in there when the baby isn't a baby anymore anyway?
Love the idea of a loft bed in the will-be-old-one-day nursery.
We currently live in a small 2 BR home, and the kids room has a loft bed (for the teen boy) and crib (for the wee girl). Under the loft bed is a dresser for the boy and a glider for the adults plus wee girl.
The room also doubles as a fitness room with free weights, mats, etc. stashed in the closet.
It's crazy how much diversity a small room can hold with the foundation of love, family, and creativity!
Love the color, I can't wait to see the big reveal!!
I spent the weekend painting my basement family room, I went with BM moonshine and I love it!!!
Strange question,how do you get the paint out of your hair, do you have any tips?? haha
Love the direction you guys are going in for her new room. The colors are really fun. Definitely seems like it will allow you to update and add other colors as she changes her mind later on. And hey, I think a pile on the floor works just as well as a mood board.
So great to see project come together. Iknow the room will be beautiful! Clara is such a doll and says the cutest things. And she will make an awesome big sister when the time comes!
Ooo nice color pick. I have a color like that in a few rooms in our house and it really is a great backdrop. I am all about the pops of color in a room, so leaving the walls neutral definitely helps it not be explosively in your face. Can't wait to see the rest of it come together!
What will you do with the ceiling fan in that room? Do you like it enough that you will keep it? update to a different light fixture? We have an old one in our room – we will probably do a cute light fixture of some sort :)
Alaskan skies or as we call it in Canada edgecomb gray haha! I love that colour its nice to see it in a room too very good greige :) cannot wait to see the final project !
Edgecomb Gray BM (color matched to a different paint) is the color of our kitchen and it's really nice. It's a little creamy though (is that a weird way of explaining paint?). It works though. My favorite is like a step down from that and it's called Revere Pewter. It's so pretty and a great neutral.
It's amazing and works really well in different light too. It's *hopefully* going to be a mainstain in our house when we repaint and may end up our bedroom color. I'm a little unsure what to do about colors though downstairs since it's open and the biggest focal point will be the cherry cabinets. Everyone puts yellow with cherry but since it's so open I want most of our downstairs to be primarily the same color. I'm almost ready to just stick with a nice white and then use arts/accessories for the bursts of color.
This weekend has consisted of watching my Ravens make it to the Superbowl (ahhh!!) and finally (I think) committing to a new paint color for our master bedroom. I've been staring at the samples for about two weeks now, I think it's time to make some moves. We already painted this room once about 4 months ago, so let's hope this new color is the winner!!
LOVE the floor-pile as mood board!
My 4-year-old Lily has a very pink and red room. It was my solution to both girls wanting pink rooms. Sophia got an antiquey and romantic room in lavender with hot pink touches and Lily got a modern pop-y room in seashell pink with red touches.
You can see some pics (and the shocking before and after) on my blog Sue at Home.
I love that Clara uses repetitive adjectives. Big, big. Pretty, pretty. You know what I say, Clara P? That's cute, cute!
Ooh, I love the mood board…fresh and happy! Inspired by you guys and all of your DIY greatness, my husband and I made a tree for our son's toddler room. It's probably my favorite project we've ever conquered.
3 layers of freehanded plywood, rounded edges, sanded, stained, tacked to the wall with a nail gun…
A little more modern outline…and maybe some birdies…could be fun for Clara too!
Great color. Love it!!
We are in the process of transitioning our just turned 10 years old daughter into her "tween" room. She wanted to move into a spare bedroom that is in the back of our house, vs. her current room which is in the front. Her current room is bright pink, like the Barbie Dream House threw up in it pink. After weeks of picking out every bright, vivid paint swatch at Home Depot, she finally decided neutral was the way to go in the new room. She picked "Gentle Fawn" (kind of a caramel color from Glidden). She got a new white bed and wardrobe from Ikea and in a surprising twist, picked blue and white toile bedding from Ikea with some other patterns mixed in for texture (her words!). She picked out a crystal chandelier, too. I love the room so much I may move into it as well. Still need to paint her closet doors and replace the hardware but almost done! This has been one of my very favorite projects ever in our house.
We are working on our Kenley's big girl room and I was so excited when I read the beginning of this post when you mentioned the "big big tree" because we are planning on painting a tree in one of the corners of her room and hanging birdhouses on the "branches." And I thought maybe you guys were thinking something "tree-related" too. Maybe you are? :)
Such a pretty, warm color! We've been tackling closets, getting rid of clothes and clutter. Our mantra for 2013 is simplify! I never realize how many things I hold onto until I closet dive. I found the cutest maternity jeans hat I must've messed putting away and would've been nice to have during this last pregnancy. Oops!
Can't wait to see the finished product! We are starting to brainstorm for our daughter and we are thinking a pinky gray with pink and light teal accents….we will see!
YAY! I have been dying to see some posts about Clara's big girl room! Any ideas on when she'll be moving in? Maybe as her big 3rd birthday present?! Speaking of 3rd birthdays, my favorite guy's was yesterday :)
New paint always makes a room feel so fresh!
I'm kind of a little sad that you no longer are going to have a "junk" room, since I just started piling stuff in our spare bedroom while we finish up our living room and get started on our bedroom. I really wanted to work on turning the spare room into my sewing room as my next project, but got outvoted by my hubby. Our bedroom has been THE junk room since we moved in here 15 years ago, so I guess it really IS time to clear it out and turn it into more of a sanctuary than a storeroom!
This weekend I cut and stained the wood to make a desk for my 27 year old son who just got a huge new computer with two large monitors for Christmas that won't fit on his little bitty computer desk. This new desk is 8 ft long with bookcases at each end, somewhat along the lines of the desk in your home office. Today I'm putting the bookcases together before hauling the pieces over to his apartment.
After living in my house for almost three years, I am finally figuring out the way it works for me the best. So… this weekend I moved the dining room into the living room, the tv into what was an underused office so it can become a den, and I'm turning the dining room into awhat I'm calling a project room. (Mostly because I think I am too cool for a craft room. Ha. ) So today it's off to IKEA for some shelving, maybe a couch, a tv stand. Wish me luck!
This comment section is currently closed.


text:
More Videos...
World Racing 2 mine garage 2
mine garage 2
Corrado VR6 Turbo
200 m. sprint at Club GOLF Danmark Summer Meet, June 2011 , wheel spin and
flaming Exhaust. Super sound!
Club Golf Danmark træf 2011 Ellested.
my super corrado VR6 turbo
CORRADO VR6 turbo FOREVER...
My Super Corrado vr6 Kompressor - Part 2
VW Corrado G60 Supercharged Volkswagen Karmann Golf Rallye Car Manual Hatchback Tuner G 60
VW Corrado G60 Supercharged Volkswagen Karmann Drift Car Manual Hatchback
Tuner G 60 http://www.1ownercarguy.com Here is a very nice VW Corrado G60
that has the supercharged engine and 5-Speed Manual Transmission. This car
is fast and a blast to drive. This car is a 1 Owner and is completely
factory so you know no one has been messing around with it. The exterior is
in excellent shape. The paint looks great and except for a few minor chips
here and there there is not much to say bad about it. The interior is very
clean and shows hardly any wear at all. You can really tell how good the
car was taken care of by the condition of the interior. This is a great car
that is a blast to drive and gets great gas mileage.
Here is the Soda Steam link Free Shipping and $10 off
Make sure and check Out My channel I have Lots of 1 & 2 Owner Cars Trucks
and SUV's on there and I sell them Cheap Also If You Subscribe You won't
Miss out..;-) I Have a few Channels You can Access them Via My Main Channel
page or also you can See the Links I post Below I will Also Post links to
My websites and Facebook so you don't Miss any of the action Sign Up
Nathan Wratislaw AKA 1 Owner Car Guy
Make sure and visit Some of My Websites Below I do All Sorts of Unique
Also Make Sure And ADD Me or Subscribe on Facebook & Youtube
And For Youtube i have 2 channels you may want to add
Send Me Hate Mail or Goodies & Stuff to show on Youtube
1 Owner Car Guy (Nathan Wratislaw)
PO Box 2505
Lakeside, Ca 92040
No Ticking items please..;-)
From Wikipedia:
The Volkswagen Corrado is a sports car developed by German automaker
Volkswagen, designed by Herbert Schaefer [1] and built by Karmann in
Osnabrück, Germany, between 1 September 1988 and 31 July 1995. In 1990,
the Corrado went on sale in the United States.
Originally conceived as a possible successor to the Porsche 944, the
Corrado is a three-door hatchback coupé with a 2+2 seating layout. The
car's floorpan is based on the A2 platform (i.e. MK2 Golf/Jetta) and, with
the exception of VR6 models, all versions use the subframes, suspension,
steering and braking components from the A2 model range. The VR6 uses
suspension components from the A3 model range, including the rear axle
assembly and some parts of the A3's 'plus' type front axle assembly. The
subsequent wider front wheel-track of the Corrado VR6 necessitated the
fitting of new front wings with wider wheel arches and liners along with a
new front bumper assembly. Together with a new raised-style bonnet to
accommodate the VR6 engine, these body improvements were carried across the
model range.
Launched in late 1988, (three years before the end of Scirocco production)
All Corrados were front-wheel drive and featured petrol engines. It debuted
with two engine choices: a 1.8 litre, 16-valve, inline-four with 136 bhp
(101 kW; 138 PS),[5] and a supercharged 1.8 litre, eight-valve inline-four,
marketed as the G60, and delivering 160 bhp (120 kW; 160 PS). The G60 is
named for the G-Lader with which it is equipped, a scroll Supercharger whose interior resembles the letter
G60 refers to an inline-four cylinder automobile petrol engine, which uses
a specific method of forced induction - by way of a specific type of Supercharger. The G60 engine was formerly
manufactured by the German carmaker Volkswagen Group, and was installed in
a limited number and range of 'hot hatch' cars from their Volkswagen
Passenger Cars marque from August 1988 to July 1993.[1]
"Volkswagen Corrado G60" G60 "Volkswagen Carrado" Karmann "VW Corrado"
Tuning Tuner Corrado Supercharged Supercharger Video Review Performance Drift1.8L
Engine Motor "5-Speed Manual" Hatchback "Karmann Ghia" Golf Jetta "Nathan
Wratislaw" "1 Owner Car Guy" Volkswagen Manual "Used Volkswagen" Buy Sell
"For Sale" "Volkswagen Corrado For Sale" Youtube Videos "Used Car Review"
"Test Drive" "Start Up" Options "Used Cars For Sale"
volkswagen corrado r32
60fps CORRADO VR6 TURBO GT3076RHF 1.6 bar 460 HP CRUISING
Powered by X-Parts
restauration corrado g60
forum www.les-arsouilleurs.com
Drag 2011 II etapas @VW corrado 2.0T 4wd fastest Lithuanian car ET 8,99 /273km/h
vw corrado 2.0T 4wd fastest Lithuanian car ET 8,99 /273km/h
World Racing 2 - VW Multivan Run
car by kodiak982
map by TomWin
VW Corrado VR6 Turbo
My VW Corrado
Exhaust Selfmade
World Racing 2 - VW Golf IV R32 Test Drive
car by Mario_13 map by TomWin
Old Top Gear Ford Probe
Old Top Gear from 1994
Jeremy Clarkson roadtests the Ford Probe
Including the famous line "...and on top of all that, it's the first Ford
that I can remember which looks good enough to snap knicker elastic at 50
This clip was extracted from the episode that was first broadcast sometime
in 1994
VW Corrado G60 corby & fz cars Tuning 2012
http://www.grafica-corby.com http://www.fz-cars.com
Corrado con motore completamente cromato, a specchio, cerchi scomponibili
da 12" con gomme 285/40/17, allargamenti in lamiera (no stucco!), scarichi
sottoporta, interni in pelle rossa ecc ecc
song credits:
MGMT - Time To Pretend (licenza SME)
World Racing 2 - Supra run
car and map by TomWin
sorry i forgot to put the music :(
Dodge Ram Crushes GT-R in 1/4-Mile Pass
Drag Racing Down Under — 1500HP VW
Wheelstand Wednesday — Compilation Video
All Show No Go Blown/Turbocharged 1969 Camaro
Which car is faster?
Similar 1/4 mile timeslips to browse:
1993 Volkswagen Corrado V8: 9.214 @ 145.610
Spencer Tramm, Engine: 468ci Chevrolet,
1993 Volkswagen Corrado SLC: 11.394 @ 128.590
Zachary Martin, Engine: VR6 - stock blcok, Supercharger: N/A Turbos: PT5862 Tires: 24.5
1990 Volkswagen Corrado G60: 12.260 @ 118.810
Don Fritsche, Engine: 2.ol ABA 16v, Turbos: GT3076R Tires: 24.5 slicks
1993 Volkswagen Corrado SLC VR6: 14.301 @ 98.060
Joey Langlois, Engine: 2.8L VR6 w/ 10.5:1 compression, Supercharger: - Turbos: - Tires: Khumo Ecsta ASX
1993 Volkswagen Corrado slc: 15.342 @ 92.060
neil bell, Engine: 2.8l vr6, Tires: toyo proxes 4
1992 Volkswagen Corrado SLC VR6 Ecu Tune: 15.574 @ 91.340
©2015 DragTimes - Disclaimer - Contact Us


text:
German - English
Closed - This job posting has been filled and work has been completed.
Job Description
We have an eBook, basically 14.500 Words, what we want to translate from German to English.
Basically it is very easy German, here qn example:
Klicken Sie auf OK wenn Sie fertig sind = Click OK, when you are done.
So I guess with medium German skills and / or Google translator, you should get that work, but you should have good English language skills.
Payment, I write here $500, but I think it should be rather less - because I guess it's a work load of 10 - 15h - what is important is, we need that soon back, so most perfect deal would be we make it this weekend - possible
Other open jobs by this client
- Fixed-Price – Taxominio
- Fixed-Price – Taxominio - 2