The Wide Range Gauge (WRG) family offers the capability of single port pressure measurement in the range atmosphere to 10-9 mbar with a linear output. Its a compact solution, halving the space and connectivity hardware requirement, which can be all important in many applications. The WRG has many novel features, including a new patented striker, pushbutton calibration and set point controls and comprehensive diagnostics. The WRG is a cost-effective vacuum management solution when used either with a Edwards controller or directly integrated into the system controls.
nWRG-S-NW25 - to głowica z wyjściem cyfrowym RS485. Karta katalogowa do pobrania poniżej.
Any vacuum system where there is a need to measure pressure over a wide range. The WRG with an AGD represents a very simple and cost effective means of achieving this.
The linear output and equation make WRG’s an attractive option for industrial OEM’s where the gauge may be directly integrated into the process controller.
The WRG is suitable for a wide range of HV and UHV applications, however if your process will spend a significant amount of time between 5x10-4 and 5x10-3mbar then Edwards recommend using independent APG100 Pirani and AIM Penning gauges, as this will improve gauge reliability for your application.
Power supply +14.5 to +36 V d.c.
Output signal 1.8 to 10.2 V d.c.
Maximum voltage 40 V d.c.
* Accuracy is reduced at the limits of the measuring range.