Search

# HP-9150 Oscilloscope Probe 150 MHz Selectable X1/Ref/X10

 The HP-9150 is a passive high impedance oscilloscope probe designed and calibrated for use with instruments having an input impedance of 1MΩ. Switch Selectable x1/ x10/ Ref Probe.

The HP-9150 is a passive high impedance oscilloscope probe designed and calibrated for use with instruments having an input impedance of 1MΩ.

The probe incorporates a two position slide switch in the head which selects attenuation of x1 and x10.

X1 position:

• Input Impedance: R= 1Mohm, C= 68pf
• Bandwidth= 6MHz
• Max Voltage In(Vin)= 600V

X10 position:

• Input Impedance: R=10Mohm, C=18pf
• Bandwidth= 150MHz
• Max Voltage In(Vin)=600V

Back to the section
{"ft_lang_label":"__label__en","ft_lang_prob":0.7397078,"math_prob":0.79195875,"size":369,"snap":"2022-40-2023-06","text_gpt3_token_len":93,"char_repetition_ratio":0.10958904,"word_repetition_ratio":0.0,"special_character_ratio":0.24932249,"punctuation_ratio":0.060606062,"nsfw_num_words":0,"has_unicode_error":false,"math_prob_llama3":0.9542903,"pos_list":[0,1,2],"im_url_duplicate_count":[null,6,null],"WARC_HEADER":"{\"WARC-Type\":\"response\",\"WARC-Date\":\"2022-09-30T18:17:15Z\",\"WARC-Record-ID\":\"<urn:uuid:1fd27550-b31a-484d-b429-2231d9738f85>\",\"Content-Length\":\"24550\",\"Content-Type\":\"application/http; msgtype=response\",\"WARC-Warcinfo-ID\":\"<urn:uuid:1396f89d-d181-42ee-a0cb-30ee5abed0eb>\",\"WARC-Concurrent-To\":\"<urn:uuid:e81592d4-e6f2-437e-97b3-47df14107a63>\",\"WARC-IP-Address\":\"172.98.77.80\",\"WARC-Target-URI\":\"https://www.aktakom.com/products/index.php?SECTION_ID=415&ELEMENT_ID=29262\",\"WARC-Payload-Digest\":\"sha1:VMDFBFDEHTMQRFU7SHSW7WKEL7ZHSWVV\",\"WARC-Block-Digest\":\"sha1:V25K4IO3CJBQBAB44CHMLYRTU7NANHEG\",\"WARC-Identified-Payload-Type\":\"text/html\",\"warc_filename\":\"/cc_download/warc_2022/CC-MAIN-2022-40/CC-MAIN-2022-40_segments_1664030335504.22_warc_CC-MAIN-20220930181143-20220930211143-00769.warc.gz\"}"}