 To reduce the risk of rotor failures, centrifuge manufactures specify operating and maintenance procedures to ensure that rotors are regularly inspected and removed from service or derated (only operated at lower speeds) when they are past their expected lifetime.Another potential hazard is the aerosolization of hazardous samples during centrifugation.To prevent contamination of the laboratory, rotor lids with special aerosol-tight gaskets are available.