of the engine until the time between fuel injection and ignition is 2.407 ms. The resulting cetane number is then calculated by determining which mixture of cetane (hexadecane) and isocetane (2,2,4,4,6,8,8-heptamethylnonane) will result in the same ignition delay. Ignition Quality Tester (IQT) Another reliable method of measuring the derived cetane number (DCN) of diesel fuel is the Ignition Quality Tester (IQT). This instrument applies a simpler, more robust approach to CN measurement than the CFR. Fuel is injected into a constant volume combustion chamber at approximately 575 °C and . The time between the start of injection and the recovery of the combustion chamber pressure to is defined as the ignition delay. This measured ignition delay is then used to calculate the DCN of the fuel. The fuel's DCN is then calculated using an empirical inverse relationship to ignition delay. Because of the reproducibility, material cost, and speed of the IQT, this has been the definitive source for DCN measurements of fuels since the late 2000s. Fuel ignition tester Another reliable method of measuring the derived cetane number of diesel fuel is the Fuel Ignition Tester (FIT). This instrument applies a simpler, more robust approach to CN measurement than the CFR. Fuel is injected into a constant volume combustion chamber in which the ambient temperature is approximately 575 °C. The fuel combusts, and the high rate of pressure change within the chamber defines the start of combustion. The ignition delay of the fuel can then be calculated as the time elapsed between the start of fuel injection and the start of combustion. The fuel's derived cetane number can then be calculated using an empirical inverse relationship to ignition delay. Cetane index Another method that fuel-users control quality is by using the cetane index (CI), which is a calculated number based on the density and a distillation range of the fuel. There are various versions of this, depending on whether metric or Imperial units are used, and the number of distillation points are used. These days most oil companies use the '4-point method', ASTM D4737, based on density, 10% 50% and 90% recovery temperatures. The '2-point method' is defined in ASTM D976 and uses just density and the 50% recovery temperature. This 2-point method tends to overestimate the cetane index and is not recommended. Cetane index calculations can not account for cetane improver additives and therefore do not measure the total