could be seen that a smooth pressure gradient existed across the sample with the outermost portions of the sample acting as a kind of gasket. The sample was not evenly distributed across the diamond culet but localized in the center due to the "cupping" of the diamond at higher pressures. This cupping phenomenon is the elastic stretching of the edges of the diamond culet, commonly referred to as the "shoulder height". Many diamonds were broken during the first stages of producing a new cell or any time an experiment is pushed to higher pressure. The NBS group was in a unique position where almost endless supplies of diamonds were available to them. Customs officials occasionally confiscated diamonds from people attempting to smuggle them into the country. Disposing of such valuable confiscated materials could be problematic given rules and regulations. One solution was simply to make such materials available to people at other government agencies if they could make a convincing case for their use. This became an unrivaled resource as other teams at the University of Chicago, Harvard University, and General Electric entered the high pressure field. During the following decades DACs have been successively refined, the most important innovations being the use of gaskets and the ruby pressure calibration. The DAC evolved to be the most powerful lab device for generating static high pressure. The range of static pressure attainable today extends to 640 GPa, much higher than the estimated pressures at the Earth's center (~360 GPa). Components There are many different DAC designs but all have four main components: Force-generating device Relies on the operation of either a lever arm, tightening screws, or pneumatic or hydraulic pressure applied to a membrane. In all cases the force is uniaxial and is applied to the tables (bases) of the two anvils. Two opposing diamond anvils Made of high gem quality, flawless diamonds, usually with 16 facets, they typically weigh to carat (25 to 70 mg). The culet (tip) is ground and polished to a hexadecagonal surface parallel to the table. The culets of the two diamonds face one another, and must be perfectly parallel in order to produce uniform pressure and to prevent dangerous strains. Specially selected anvils are required for specific measurements – for example, low diamond absorption and luminescence is required in corresponding experiments. Gasket A gasket used in a diamond anvil cell experiment is a thin metal