label compounds in biological samples. Developments within SIMS: Some chemical modifications have been made within SIMS to increase the efficiency of the process. There are currently two separate techniques being used to help increase the overall efficiency by increasing the sensitivity of SIMS measurements: matrix-enhanced SIMS (ME-SIMS) - This has the same sample preparation as MALDI does as this simulates the chemical ionization properties of MALDI. ME-SIMS does not sample nearly as much material. However, if the analyte being tested has a low mass value then it can produce a similar looking spectra to that of a MALDI spectra. ME-SIMS has been so effective that it has been able to detect low mass chemicals at sub cellular levels that was not possible prior to the development of the ME-SIMS technique. The second technique being used is called sample metallization (Meta-SIMS) - This is the process of gold or silver addition to the sample. This forms a layer of gold or silver around the sample and it is normally no more than 1-3 nm thick. Using this technique has resulted in an increase of sensitivity for larger mass samples. The addition of the metallic layer also allows for the conversion of insulating samples to conducting samples, thus charge compensation within SIMS experiments is no longer required. MALDI imaging Matrix-assisted laser desorption ionization can be used as a mass spectrometry imaging technique for relatively large molecules. It has recently been shown that the most effective type of matrix to use is an ionic matrix for MALDI imaging of tissue. In this version of the technique the sample, typically a thin tissue section, is moved in two dimensions while the mass spectrum is recorded. Although MALDI has the benefit of being able to record the spatial distribution of larger molecules, it comes at the cost of lower resolution than the SIMS technique. The limit for the lateral resolution for most of the modern instruments using MALDI is 20 m. MALDI experiments commonly use either an Nd:YAG (355 nm) or N2 (337 nm) laser for ionization. Pharmacodynamics and toxicodynamics in tissue have been studied by MALDI imaging. DESI imaging Desorption electrospray Ionization is a less destructive technique, which couples simplicity and rapid analysis of the sample. The sample is sprayed with an electrically charged solvent mist at an angle that causes the ionization and desorption of various molecular species. Then, two-dimensional maps of the