than chemical oxidation methods capable of digesting as much as 10 times more sample. Also, the salt content of the samples do not combust, and so therefore, gradually build a residue inside the combustion tube eventually clogging the catalyst resulting in poor peak shapes, and degraded accuracy or precision, unless appropriate maintenance procedures are followed. The catalyst should be regenerated or replaced as needed. To avoid this problem the manufacturing industry has developed several concepts, such as matrix separation, ceramic reactors, better process control or methods without catalysts. Photo-oxidation (ultraviolet light) In this oxidation scheme, ultraviolet light alone oxidizes the carbon within the sample to produce CO2. The UV oxidation method offers the most reliable, low maintenance method of analyzing TOC in ultra-pure waters. Ultraviolet/persulfate oxidation Like the photo-oxidation method, UV light is the oxidizer but the oxidation power of the reaction is magnified by the addition of a chemical oxidizer, which is usually a persulfate compound. The mechanisms of the reactions are as follows: Free radical oxidants formed: Excitation of organics: Oxidation of organics: The UV–chemical oxidation method offers a relatively low maintenance, high sensitivity method for a wide range of applications. However, there are oxidation limitations of this method. Limitations include the inaccuracies associated with the addition of any foreign substance into the analyte and samples with high amounts of particulates. Performing "system blank" analysis, which is to analyze then subtract the amount of carbon contributed by the chemical additive, inaccuracies are lowered. However, analyses of levels below 200 ppb TOC are still difficult. Thermochemical persulfate oxidation Also known as heated persulfate, the method utilizes the same free radical formation as UV persulfate oxidation except uses heat to magnify the oxidizing power of persulfate. Chemical oxidation of carbon with a strong oxidizer, such as persulfate, is highly efficient, and unlike UV, is not susceptible to lower recoveries caused by turbidity in samples. The analysis of system blanks, necessary in all chemical procedures, is especially necessary with heated persulfate TOC methods because the method is so sensitive that reagents cannot be prepared with carbon contents low enough to not be detected. Persulfate methods are used in the analysis of wastewater, drinking water, and pharmaceutical waters. When used in conjunction with sensitive NDIR detectors heated persulfate TOC instruments readily measure TOC at single digit parts per billion (ppb) up to hundreds of parts per million (ppm) depending on sample volumes.