Laboratory in Washington, D.C. and manufactured by ramé-hart (now ramé-hart instrument company), New Jersey, USA. The original manual contact angle goniometer used an eyepiece with a microscope. Today's contact angle goniometer uses a camera and software to capture and analyze the drop shape, and is better suited for dynamic and advanced studies. Surface tension Contact angle goniometers (q.v.)can also determine the surface tension for any liquid in gas or the interfacial tension between any two liquids. If the difference in densities between the two fluids is known, the surface tension or interfacial tension can be calculated by the pendant drop method. An advanced instrument often called a goniometer / tensiometer includes software tools that measure surface tension and interfacial tension using the pendant drop, inverted pendant drop, and sessile drop methods, in addition to contact angle. A centrifugal adhesion balance relates the contact angles to the adhesion of the drop to the surface. A gonioreflectometer measures the reflectivity of a surface at a number of angles. Positioning A positioning goniometer or goniometric stage is a device that rotates an object precisely about a fixed axis in space. It is similar to a linear stage— however, rather than move linearly relative to its base, the stage platform rotates partially about a fixed axis above the mounting surface of the platform. Positioning goniometers typically use a worm drive with a partial worm wheel fixed to the underside of the stage platform meshing with a worm in the base. The worm gear may be rotated manually, or by a motor in automated positioning systems. Knife and blade cutting edge angle measurement The included cutting angles of all kinds of sharp edge blades are measured using a laser reflecting goniometer. Developed by the Cutlery and Allied Trades Research Association (CATRA) in the UK, a range of devices can accurately determine the cutting edge profile, including a rounding of the tip to ½°. The included angle of a blade is important in controlling its cutting ability and edge strength—i.e., a low angle makes a thin sharp edge optimized for cutting softer materials, while a large angle makes a thick edge that is less sharp but stronger, which may be better for cutting harder materials. Doctor blade inspection Used doctor blades, from gravure and other printing and coating processes, can be inspected with a goniometer, typically with a built-in light source, to examine the blade edge