assesses the joint before performing an intervention, and continues to use the tool to monitor progress. The therapist can take these range of motion measurements at any joint. They typically require knowledge about the anatomy of the body, particularly bony landmarks. For example, when measuring the knee joint, the therapist places the axis (point of rotation) on the lateral epicondyle of the femur, and lines up the stationary arm with the greater trochanter of the femur. Finally, the therapist lines up the moveable arm of the goniometer with the lateral malleolus of the fibula, and records a measurement using the degree scale on the circular portion of the tool. Reading accuracy is sometimes a problem with goniometers. Issues with the intra-measure (between measures) and inter-tester (between clinicians) reliability may increase as the experience of the examiner decreases. Some studies suggest that these errors can be anywhere between 5 and 10 degrees. These goniometers come in different forms that some argue increase reliability. The universal standard goniometer is a plastic or metal tool with 1 degree increments. The arms are usually not longer than 12-inches, so it can be hard to accurately pinpoint the exact landmark for measurement. The telescopic-armed goniometer is more reliable—with a plastic circular axis like a classic goniometer, but with arms that extend to as long as two feet in either direction. More recently in the twenty-first century, smartphone application developers have created mobile applications that provide the functions of a goniometer. These applications (such as Knee Goniometer and Goniometer Pro) use the accelerometers in phones to calculate joint angles. Recent research supports these applications and their devices as reliable and valid tools with as much accuracy as a universal goniometer. Modern rehabilitative therapy motion capture systems perform goniometry at the very least measuring active range of motion. While in some cases accuracy may be inferior to a goniometer, measuring angles with a motion capture system is superior at measuring during dynamic, as opposed to static situations. Furthermore, using a traditional goniometer takes valuable time. In the clinical context, performing manual measurements takes valuable time and may not be practical. Surface science Contact angle goniometer In surface science, an instrument called a contact angle goniometer or tensiometer measures the static contact angle, advancing and receding contact angles, and sometimes surface tension. The first contact angle goniometer was designed by William Zisman of the United States Naval Research