devices, risk management is a process for identifying, evaluating and mitigating risks associated with harm to people and damage to property or the environment. Risk management is an integral part of medical device design and development, production processes and evaluation of field experience, and is applicable to all types of medical devices. The evidence of its application is required by most regulatory bodies such as the US FDA. The management of risks for medical devices is described by the International Organization for Standardization (ISO) in ISO 14971:2019, Medical Devices—The application of risk management to medical devices, a product safety standard. The standard provides a process framework and associated requirements for management responsibilities, risk analysis and evaluation, risk controls and lifecycle risk management. Guidance on the application of the standard is available via ISO/TR 24971:2020. The European version of the risk management standard was updated in 2009 and again in 2012 to refer to the Medical Devices Directive (MDD) and Active Implantable Medical Device Directive (AIMDD) revision in 2007, as well as the In Vitro Medical Device Directive (IVDD). The requirements of EN 14971:2012 are nearly identical to ISO 14971:2007. The differences include three "(informative)" Z Annexes that refer to the new MDD, AIMDD, and IVDD. These annexes indicate content deviations that include the requirement for risks to be reduced as far as possible, and the requirement that risks be mitigated by design and not by labeling on the medical device (i.e., labeling can no longer be used to mitigate risk). Typical risk analysis and evaluation techniques adopted by the medical device industry include hazard analysis, fault tree analysis (FTA), failure mode and effects analysis (FMEA), hazard and operability study (HAZOP), and risk traceability analysis for ensuring risk controls are implemented and effective (i.e. tracking risks identified to product requirements, design specifications, verification and validation results etc.). FTA analysis requires diagramming software. FMEA analysis can be done using a spreadsheet program. There are also integrated medical device risk management solutions. Through a draft guidance, the FDA has introduced another method named "Safety Assurance Case" for medical device safety assurance analysis. The safety assurance case is structured argument reasoning about systems appropriate for scientists and engineers, supported by a body of evidence, that provides a compelling, comprehensible and valid case that a system is safe for a given application in a given environment. With the guidance, a safety assurance case is expected