and principles used. Secondary electrons The conventional secondary electron detector of SEM (Everhart-Thornley detector) cannot be used in the presence of gas because of an electrical discharge (arcing) caused by the kilovolt bias associated with this detector. In lieu of this, the environmental gas itself has been used as a detector for imaging in this mode: Gaseous detection device In a simple form, the gaseous detection device (GDD) employs an electrode with a voltage up to several hundred volts to collect the secondary electrons in the ESEM. The principle of this SE detector is best described by considering two parallel plates at a distance d apart with a potential difference V generating a uniform electric field E = V/d, and is shown in the accompanying diagram of the GDD. Secondary electrons released from the specimen at the point of beam impingement are driven by the field force towards the anode electrode but the electrons also move radially due to thermal diffusion from collisions with the gas molecules. The variation of electron collection fraction R within anode radius r vs. r/d, for fixed values of anode bias V, at constant product of (pressure·distance) p·d = 1 Pa·m, is given by the accompanying characteristic curves of efficiency of the GDD. All of the secondary electrons are detected if the parameters of this device are properly designed. This clearly shows that practically 100% efficiency is possible within a small radius of collector electrode with only moderate bias. At these levels of bias, no catastrophic discharge takes place. Instead, a controlled proportional multiplication of electrons is generated as the electrons collide with gas molecules releasing new electrons on their way to the anode. This principle of avalanche amplification operates similarly to proportional counters used to detect high energy radiation. The signal thus picked up by the anode is further amplified and processed to modulate a display screen and form an image as in SEM. Notably, in this design and the associated gaseous electron amplification, the product p·d is an independent parameter, so that there is a wide range of values of pressure and electrode geometry which can be described by the same characteristics. The consequence of this analysis is that the secondary electrons are possible to detect in a gaseous environment even at high pressures, depending on the engineering efficacy of any given instrument. As a further characteristic of the GDD, a gaseous scintillation