allowed to take place for a set time before it is quenched. The deuteration pattern of a molecule that has undergone H/D exchange can be maintained in aprotic environments. However, some methods of deuteration analysis for molecules such as proteins, are performed in aqueous solution, which means that exchange will continue at a slow rate even after the reaction is quenched. Undesired deuterium-hydrogen exchange is referred to as back-exchange and various methods have been devised to correct for this. Detection H–D exchange was measured originally by the father of hydrogen exchange Kaj Ulrik Linderstrøm-Lang using density gradient tubes. In modern times, H–D exchange has primarily been monitored by the methods: NMR spectroscopy, mass spectrometry and neutron crystallography. Each of these methods have their advantages and drawbacks. NMR spectroscopy Hydrogen and deuterium nuclei are grossly different in their magnetic properties. Thus it is possible to distinguish between them by NMR spectroscopy. Deuterons will not be observed in a 1H NMR spectrum and conversely, protons will not be observed in a 2H NMR spectrum. Where small signals are observed in a 1H NMR spectrum of a highly deuterated sample, these are referred to as residual signals. They can be used to calculate the level of deuteration in a molecule. Analogous signals are not observed in 2H NMR spectra because of the low sensitivity of this technique compared to the 1H analysis. Deuterons typically exhibit very similar chemical shifts to their analogous protons. Analysis via 13C NMR spectroscopy is also possible: the different spin values of hydrogen (1/2) and deuterium (1) gives rise to different splitting multiplicities. NMR spectroscopy can be used to determine site-specific deuteration of molecules. Another method uses HSQC spectra. Typically HSQC spectra are recorded at a series of timepoints while the hydrogen is exchanging with the deuterium. Since the HSQC experiment is specific for hydrogen, the signal will decay exponentially as the hydrogen exchanges. It is then possible to fit an exponential function to the data, and obtain the exchange constant. This method gives residue-specific information for all the residues in the protein simultaneously The major drawback is that it requires a prior assignment of the spectrum for the protein in question. This can be very labor-intensive, and usually limits the method to proteins smaller than 25 kDa. Because it takes minutes to hours to record a HSQC spectrum, amides that exchange quickly must be measured using other pulse