Gauge blocks (also known as gage blocks, Johansson gauges, slip gauges, or Jo blocks) are a system for producing precision lengths. The individual gauge block is a metal or ceramic block that has been precision ground and lapped to a specific thickness. Gauge blocks come in sets of blocks with a range of standard lengths. In use, the blocks are stacked to make up a desired length (or height). Gauge blocks were invented in 1896 by Swedish machinist Carl Edvard Johansson. They are used as a reference for the calibration of measuring equipment used in machine shops, such as micrometers, sine bars, calipers, and dial indicators (when used in an inspection role). Gauge blocks are the main means of length standardization used by industry. An important feature of gauge blocks is that they can be joined together with very little dimensional uncertainty. The blocks are joined by a sliding process called wringing, which causes their ultra-flat surfaces to cling together. A small number of gauge blocks can be used to create accurate lengths within a wide range. By using three blocks at a time taken from a set of 30 blocks, one may create any of the 1000 lengths from 3.000 to 3.999 mm in 0.001 mm steps (or .3000 to .3999 inches in 0.0001 inch steps). Description A gauge block is a block of metal or ceramic with two opposing faces ground precisely flat and parallel, a precise distance apart. Standard grade blocks are made of a hardened steel alloy, while calibration grade blocks are often made of tungsten carbide, chromium carbide or ceramic because they are harder and wear less. Gauge blocks come in sets of blocks of various lengths, along with additional wear blocks, to allow a wide variety of standard lengths to be made up by stacking them. The length of each block is actually slightly shorter than the nominal length stamped on it, because the stamped length includes the length of one wring film, a film of lubricant which separates adjacent block faces in normal use. The thickness of the wring film is about . The gauge's nominal length is also known as the interferometric length. In use, the blocks are removed from the set, cleaned of their protective coating (petroleum jelly or oil) and wrung together to form a stack of the required dimension. Gauge blocks are calibrated to be accurate at and should