seen without the instruments these two resistors would be considered series resistors in this circuit. Use in current measuring An ammeter shunt allows the measurement of current values too large to be directly measured by a particular ammeter. In this case, a separate shunt, a resistor of very low but accurately known resistance, is placed in parallel with a voltmeter, so that virtually all of the current to be measured will flow through the shunt (provided the internal resistance of the voltmeter takes such a low portion of the current that it is negligible). The resistance is chosen so that the resultant voltage drop is measurable but low enough not to disrupt the circuit. The voltage across the shunt is proportional to the current flowing through it, and so the measured voltage can be scaled to directly display the current value. Shunts are rated by maximum current and voltage drop at that current. For example, a 500 A, 75 mV shunt would have a resistance of , a maximum allowable current of 500 amps and at that current the voltage drop would be 75 millivolts. By convention, most shunts are designed to drop 50 mV, 75 mV or 100 mV when operating at their full rated current and most ammeters consist of a shunt and a voltmeter with full-scale deflections of 50, 75, or 100 mV. All shunts have a derating factor for continuous (more than 2 minutes) use, 66% being the most common, so the example shunt should not be operated above 330 A (and 50 mV drop) longer than that. This limitation is due to thermal limits at which a shunt will no longer operate correctly. For manganin, a common shunt material, at 80 °C thermal drift begins to occur, at 120 °C thermal drift is a significant problem where error, depending on the design of the shunt, can be several percent and at 140 °C the manganin alloy becomes permanently damaged due to annealing resulting in the resistance value drifting up or down. If the current being measured is also at a high voltage potential this voltage will be present in the connecting leads too and in the reading instrument itself. Sometimes, the shunt is inserted in the return leg (grounded side) to avoid this problem. Some alternatives to shunts can provide isolation from the high voltage by not directly connecting the meter to the high voltage