were found in roughly semi-circular regions on the wafers. The silicon detector was replaced for CLEO-c because of its poor performance, the reduced need for vertexing capabilities, and the desire to minimize the material near the beampipe. Calorimetry CLEO I had three separate calorimeters. All used layers of proportional tubes interleaved with sheets of lead. The octant shower detectors were outside the time-of-flight detectors in each of the octants. Each octant detector had 44 layers of proportional tubes, alternating parallel and perpendicular to the beampipe. Wires were ganged together to reduce the number of readout channels for a total of 774 gangs. The octant end shower detectors were sixteen layer devices placed at either end of the dE/dx chambers. The layers followed an azimuthal, positive stereo, azimuthal, negative stereo pattern. The stereo wires were parallel to the slanted sides of the detector. The layers were ganged in a similar fashion as the octant shower detectors. The pole tip shower detector was placed between the ends of the drift chamber and the pole tips of the magnet flux return. The pole tip shower detector had 21 layers, with seven groups of vertical, +120°, -120° layers. The shower detector on each side was built in two halves to allow access to the beampipe. The calorimetry was significantly improved during the CLEO II upgrade. The new electromagnetic calorimeter used 7784 CsI crystals doped with thallium. Each crystal was roughly 30 cm deep and had a 5 cm × 5 cm face. The central region of the calorimeter was a cylinder placed between the drift chamber and the solenoid magnet, and two endcap calorimeters were placed at either end of the drift chamber. The crystals in the endcap were oriented parallel to the beam line. The crystals in the central calorimeter faced a point displaced from the interaction point both longitudinally and transversely by a few centimeters to avoid inefficiencies from particles passing between neighboring crystals. The calorimeter primarily measured the energy of photons or electrons, however it was also used to detect antineutrons. All versions of the detector from CLEO-II through CLEO-c used the CsI calorimeter. Particle identification Five types of long-lived, charged particles are produced at CLEO: electrons, pions, muons, kaons and protons. Proper identification of each of these types significantly improves the capabilities of the detector. Particle identification was done by both dedicated subdetectors and by the calorimeter and drift chamber.