Analytical ultracentrifugation is an analytical technique which combines an ultracentrifuge with optical monitoring systems. In an analytical ultracentrifuge (commonly abbreviated as AUC), a sample’s sedimentation profile is monitored in real time by an optical detection system. The sample is detected via ultraviolet light absorption and/or interference optical refractive index sensitive system, monitored by light-sensitive diode array or by film in the older machines. The operator can thus observe the change of sample concentration versus the axis of the rotation profile with time as a result of the applied centrifugal field. With modern instrumentation, these observations are electronically digitized and stored for further mathematical analysis. The information that can be obtained from an analytical ultracentrifuge includes the gross shape of macromolecules, conformational changes in macromolecules, and size distributions of macromolecules. With AUC it is possible to gain information on the number and subunit stoichiometry of non-covalent complexes and equilibrium constants of macromolecules such as proteins, DNA, nanoparticles or other assemblies from different molecule classes. The simplest measurement to be obtained is the sedimentation coefficient, which depends upon the size of the molecules being sedimented. This is the ratio of a particle's sedimentation velocity to the applied acceleration causing the sedimentation. Analytical ultracentrifugation has recently seen a rise in use because of increased ease of analysis with modern computers and the development of software, including a National Institutes of Health supported software package, SedFit. History Instrumentation An analytical ultracentrifuge has a light source and optical detectors. To allow the light to pass through the analyte during the ultracentrifuge run, specialized cells are required which have to meet high optical standards as well as to resist the centrifugal forces. Each cell consists of a housing, two windows made from optically pure quartz glass, and a centrepiece with one or two sectors and filling holes for the sector(s), closed with a screw plug in the housing. These cell are placed into a rotor cavity with a continuous bore, with a collar at the bottom to retain the cell. Theory Types of experiments By applying specific equipment and adapting measurement parameters several types of experiments can be performed. Most common AUC experiments are sedimentation velocity and sedimentation equilibrium experiments. Sedimentation velocity Sedimentation velocity experiments render the shape and molar mass of the analytes, as well as their size-distribution. The size resolution of this method scales approximately with the square of the particle radii, and by