immobilization for tuberculosis of the knee. It is now commonly used for the immobilization of hip and thigh injuries. The modified Thomas splint adapted the original Thomas splint to include a traction screw and foot plate and limb support built into the splint body. The Hare traction splint is a further adaptation of the Thomas splint. Its length is adjustable via telescoping rods, and it has built-in straps to support the hip and leg at several points along its length. It also provides a more comfortable ankle strap and a small winch that makes it much easier to apply and adjust traction force. The Sager splint consists of a metallic splint that is placed between the person's legs. Some models may be placed on the side closest to the injury for bilateral femur fractures without pelvic trauma. Straps are then applied, first at the thigh and then at the ankle, to strap the injured leg to the pole and provide support. The pole is extended to supply the needed traction, and then both legs are wrapped with "tensor cravats" (strap-like bandages, in this application usually elastic, to supply tension.) The Kendrick traction device (KTD) eliminates the need for leg-raising and unnecessary rolling of the patient, and can be easily applied to both pediatric and adult applications. It consists of a round pole that can be located on the lateral aspect of the leg, with straps at the upper thigh and ankle for immediate placement, and three wider straps for immobilization. It is very light at 20 ounces. The KTD does not afford the rotational stability normally seen in long bone traction splints. The CT-6 was introduced in the 21st century and utilizes a 4:1 pulley system to achieve precise and powerful, when necessary, traction. This splint is built with carbon fiber tubing and weighs 500 grams. In 2003 the CT-6 was chosen as the splint of choice by the US military and currently has over 30,000 in the field. Its compact and light design, along with its greatly improved traction method, had greatly enhanced its popularity. The Donway traction splint is a pneumatic splint that can be applied to a patient in situ. Acting on the ankle and groin pressure is then applied via an integrated pump. The devices itself consists of a metal frame that surrounds the leg that is then strapped into place. The STS (Slishman Traction Splint)