in the avalanche induce a positive current signal on the pad plane. The readout is performed by the 557 568 pads that form the cathode plane of the multi-wire proportional chambers (MWPC) located at the end plates. This gives the radial distance to the beam and the azimuth. The last coordinate, z along the beam direction, is given by the drift time. Since energy-loss fluctuations can be considerable, in general many pulse-height measurements are performed along the particle track in order to optimize the resolution of the ionization measurement. Almost all of the TPC's volume is sensitive to the traversing charged particles, but it features a minimum material budget. The straightforward pattern recognition (continuous tracks) make TPCs the perfect choice for high-multiplicity environments, such as in heavy-ion collisions, where thousands of particles have to be tracked simultaneously. Inside the ALICE TPC, the ionization strength of all tracks is sampled up to 159 times, resulting in a resolution of the ionization measurement as good as 5%. Transition radiation detector Electrons and positrons can be discriminated from other charged particles using the emission of transition radiation, X-rays emitted when the particles cross many layers of thin materials. The identification of electrons and positrons is achieved using a transition radiation detector (TRD). In a similar manner to the muon spectrometer, this system enables detailed studies of the production of vector-meson resonances, but with extended coverage down to the light vector-meson ρ and in a different rapidity region. Below 1 GeV/c, electrons can be identified via a combination of particle identification detector (PID) measurements in the TPC and time of flight (TOF). In the momentum range 1–10 GeV/c, the fact that electrons may create TR when travelling through a dedicated "radiator" can be exploited. Inside such a radiator, fast charged particles cross the boundaries between materials with different dielectric constants, which can lead to the emission of TR photons with energies in the X-ray range. The effect is tiny and the radiator has to provide many hundreds of material boundaries to achieve a high enough probability to produce at least one photon. In the ALICE TRD, the TR photons are detected just behind the radiator using MWPCs filled with a xenon-based gas mixture, where they deposit their energy on top of the ionization signals from the particle's track. The ALICE TRD was designed to derive a fast trigger for charged particles with high momentum