are typically more expensive, and that the applications have to be compatible with the use of these very specific lanthanide dyes. The main use of TRF is found in drug screening applications, under a form called TR-FRET (time-resolved fluorescence energy transfer). TR-FRET assays are very robust (limited sensitivity to several types of assay interference) and are easily miniaturized. Robustness, the ability to automate and miniaturize are features that are highly attractive in a screening laboratory. Fluorescence polarization Fluorescence polarization measurement is also very close to FI detection. The difference is that the optical system includes polarizing filters on the light path: the samples in the microplate are excited using polarized light (instead of non-polarized light in FI and TRF modes). Depending on the mobility of the fluorescent molecules found in the wells, the light emitted will either be polarized or not. For example, large molecules (e.g. proteins) in solution, which rotate relatively slowly because of their size, will emit polarized light when excited with polarized light. On the other hand, the fast rotation of smaller molecules will result in a depolarization of the signal. The emission system of the plate reader uses polarizing filters to analyze the polarity of the emitted light. A low level of polarization indicates that small fluorescent molecules move freely in the sample. A high level of polarization indicates that fluorescent is attached to a larger molecular complex. As a result, one of the basic applications of FP detection is molecular binding assays, since they allow to detect if a small fluorescent molecule binds (or not) to a larger, non-fluorescent molecule: binding results in a slower rotation speed of the fluorescent molecule, and in an increase in the polarization of the signal. Light scattering and nephelometry Light scattering and nephelometry are methods for the determination of the cloudiness of a solution (i.e.: insoluble particles in a solution). A light beam passes through the sample and the light is scattered by the suspended particles. The measured forward scattered light indicates the amount of the insoluble particles present in solution. Common nephelometry/light scattering applications include automated HTS drug solubility screening, long-term microbial growth kinetics, flocculation, aggregation and the monitoring of polymerization and precipitation, including immunoprecipitation. Instruments and assays Many of the detection modes (absorbance, fluorescence intensity, luminescence, time-resolved fluorescence, and fluorescence polarization) are available stand-alone in dedicated plate readers, but are very often found today combined into one