power factor which shows that the power transmitted may be less than the apparent power obtained by multiplying the readings of a voltmeter and ammeter in the same circuit. Electronic Electronic wattmeters are used for direct, small power measurements or for power measurements at frequencies beyond the range of electrodynamometer-type instruments. Digital A modern digital wattmeter samples the voltage and current thousands of times a second. For each sample, the voltage is multiplied by the current at the same instant; the average over at least one cycle is the real power. The real power divided by the apparent volt-amperes (VA) is the power factor. A computer circuit uses the sampled values to calculate RMS voltage, RMS current, VA, power (watts), power factor, and kilowatt-hours. The readings may be displayed on the device, retained to provide a log and calculate averages, or transmitted to other equipment for further use. Wattmeters vary considerably in correctly calculating energy consumption, especially when real power is much lower than VA (highly reactive loads, e.g. electric motors). Simple meters may be calibrated to meet specified accuracy only for sinusoidal waveforms. Waveforms for switched-mode power supplies as used for much electronic equipment may be very far from sinusoidal, leading to unknown and possibly large errors at any power. This may not be specified in the meter's manual. Precision and accuracy There are limitations to measuring power with inexpensive wattmeters, or indeed with any meters not designed for low-power measurements. This particularly affects low power (e.g. under 10 watts), as used in standby; readings may be so inaccurate as to be useless (although they do confirm that standby power is low, rather than high). The difficulty is largely due to difficulty in accurate measurement of the alternating current, rather than voltage, and the relatively little need for low-power measurements. The specification for the meter should specify the reading error for different situations. For a typical plug-in meter the error in wattage is stated as ±5% of measured value ±10 W (e.g., a measured value of 100 W may be wrong by 5% of 100 W plus 10 W, i.e., ±15 W, or 85–115 W); and the error in kW·h is stated as ±5% of measured value ±0.1 kW·h. If a laptop computer in sleep mode consumes 5 W, the meter may read anything from 0 to 15.25 W, without taking into account errors due to non-sinusoidal waveform. In