dry- packing methods, wet slurry packing, in which particles that are suspended in an appropriate solvent are delivered into a column under pressure, shows significant improvement. Three different approaches can be employed in performing wet slurry packing: the balanced density method (solvent’s density is about that of porous silica particles), the high viscosity method (a solvent of high viscosity is used), and the low viscosity slurry method (performed with low viscosity solvents). Polystyrene is used as a medium for ion- exchange. It is made from the polymerization of styrene with the use of divinylbenzene and benzoyl peroxide. Such exchangers form hydrophobic interactions with proteins which can be irreversible. Due to this property, polystyrene ion exchangers are not suitable for protein separation. They are used on the other hand for the separation of small molecules in amino acid separation and removal of salt from water. Polystyrene ion exchangers with large pores can be used for the separation of protein but must be coated with a hydrophilic substance. Cellulose based medium can be used for the separation of large molecules as they contain large pores. Protein binding in this medium is high and has low hydrophobic character. DEAE is an anion exchange matrix that is produced from a positive side group of diethylaminoethyl bound to cellulose or Sephadex. Agarose gel based medium contain large pores as well but their substitution ability is lower in comparison to dextrans. The ability of the medium to swell in liquid is based on the cross-linking of these substances, the pH and the ion concentrations of the buffers used. Incorporation of high temperature and pressure allows a significant increase in the efficiency of ion chromatography, along with a decrease in time. Temperature has an influence of selectivity due to its effects on retention properties. The retention factor (k = (tRg − tMg)/(tMg − text)) increases with temperature for small ions, and the opposite trend is observed for larger ions. Despite ion selectivity in different mediums, further research is being done to perform ion exchange chromatography through the range of 40–175 °C. An appropriate solvent can be chosen based on observations of how column particles behave in a solvent. Using an optical microscope, one can easily distinguish a desirable dispersed state of slurry from aggregated particles. Weak and strong ion exchangers A "strong" ion exchanger will not lose the charge on its matrix once the column is equilibrated