disrupted by other charges. Despite this selectivity, excess amounts of an ion with a lower selectivity introduced to the column would cause the lesser ion to bind more to the stationary phase as the selectivity coefficient allows fluctuations in the binding reaction that takes place during ion exchange chromatography. Following table shows the commonly used ion exchangers Typical technique A sample is introduced, either manually or with an autosampler, into a sample loop of known volume. A buffered aqueous solution known as the mobile phase carries the sample from the loop onto a column that contains some form of stationary phase material. This is typically a resin or gel matrix consisting of agarose or cellulose beads with covalently bonded charged functional groups. Equilibration of the stationary phase is needed in order to obtain the desired charge of the column. If the column is not properly equilibrated the desired molecule may not bind strongly to the column. The target analytes (anions or cations) are retained on the stationary phase but can be eluted by increasing the concentration of a similarly charged species that displaces the analyte ions from the stationary phase. For example, in cation exchange chromatography, the positively charged analyte can be displaced by adding positively charged sodium ions. The analytes of interest must then be detected by some means, typically by conductivity or UV/visible light absorbance. Control an IC system usually requires a chromatography data system (CDS). In addition to IC systems, some of these CDSs can also control gas chromatography (GC) and HPLC. Membrane exchange chromatography A type of ion exchange chromatography, membrane exchange is a relatively new method of purification designed to overcome limitations of using columns packed with beads. Membrane Chromatographic devices are cheap to mass-produce and disposable unlike other chromatography devices that require maintenance and time to revalidate. There are three types of membrane absorbers that are typically used when separating substances. The three types are flat sheet, hollow fibre, and radial flow. The most common absorber and best suited for membrane chromatography is multiple flat sheets because it has more absorbent volume. It can be used to overcome mass transfer limitations and pressure drop, making it especially advantageous for isolating and purifying viruses, plasmid DNA, and other large macromolecules. The column is packed with microporous membranes with internal pores which contain adsorptive moieties that can bind the target protein. Adsorptive membranes are available in