Temperature programmed desorption (TPD) is the method of observing desorbed molecules from a surface when the surface temperature is increased. When experiments are performed using well-defined surfaces of single-crystalline samples in a continuously pumped ultra-high vacuum (UHV) chamber, then this experimental technique is often also referred to as thermal desorption spectroscopy or thermal desorption spectrometry (TDS). Desorption When molecules or atoms come in contact with a surface, they adsorb onto it, minimizing their energy by forming a bond with the surface. The binding energy varies with the combination of the adsorbate and surface. If the surface is heated, at one point, the energy transferred to the adsorbed species will cause it to desorb. The temperature at which this happens is known as the desorption temperature. Thus TPD shows information on the binding energy. Measurement Since TPD observes the mass of desorbed molecules, it shows what molecules are adsorbed on the surface. Moreover, TPD recognizes the different adsorption conditions of the same molecule from the differences between the desorption temperatures of molecules desorbing different sites at the surface, e.g. terraces vs. steps. TPD also obtains the amounts of adsorbed molecules on the surface from the intensity of the peaks of the TPD spectrum, and the total amount of adsorbed species is shown by the integral of the spectrum. To measure TPD, one needs a mass spectrometer, such as a quadrupole mass spectrometer or a time-of-flight (TOF) mass spectrometer, under ultrahigh vacuum (UHV) conditions. The amount of adsorbed molecules is measured by increasing the temperature at a heating rate of typically 2 K/s to 10 K/s. Several masses may be simultaneously measured by the mass spectrometer, and the intensity of each mass as a function of temperature is obtained as a TDS spectrum. The heating procedure is often controlled by the PID control algorithm, with the controller being either a computer or specialised equipment such as a Eurotherm. Other methods of measuring desorption are Thermal Gravimetric Analysis (TGA) or using infrared detectors, thermal conductivity detectors etc. Quantitative interpretation of TPD data TDS spectrum 1 and 2 are typical examples of a TPD measurement. Both are examples of NO desorbing from a single crystal in high vacuum. The crystal was mounted on a titanium filament and heated with current. The desorbing NO was measured using a mass spectrometer monitoring the atomic mass of 30. Before 1990 analysis of a TPD spectrum was usually