Gamma-ray spectroscopy is the quantitative study of the energy spectra of gamma-ray sources, such as in the nuclear industry, geochemical investigation, and astrophysics. Most radioactive sources produce gamma rays, which are of various energies and intensities. When these emissions are detected and analyzed with a spectroscopy system, a gamma-ray energy spectrum can be produced. A detailed analysis of this spectrum is typically used to determine the identity and quantity of gamma emitters present in a gamma source, and is a vital tool in radiometric assay. The gamma spectrum is characteristic of the gamma-emitting nuclides contained in the source, just like in an optical spectrometer, the optical spectrum is characteristic of the material contained in a sample. Gamma ray characteristics Gamma rays are the highest-energy form of electromagnetic radiation, being physically the same as all other forms (e.g., X-rays, visible light, infrared, radio) but having (in general) higher photon energy due to their shorter wavelength. Because of this, the energy of gamma-ray photons can be resolved individually, and a gamma-ray spectrometer can measure and display the energies of the gamma-ray photons detected. Radioactive nuclei (radionuclides) commonly emit gamma rays in the energy range from a few keV to ~10 MeV, corresponding to the typical energy levels in nuclei with reasonably long lifetimes. Such sources typically produce gamma-ray "line spectra" (i.e., many photons emitted at discrete energies), whereas much higher energies (upwards of 1 TeV) may occur in the continuum spectra observed in astrophysics and elementary particle physics. The difference between gamma rays and X-rays is somewhat blurred. Gamma rays are always from electronic emission of atoms and are mono energetic, whereas X-rays are electrically generated (X-ray tube, linear accelerator) and have a broad energy range. Components of a gamma spectrometer The main components of a gamma spectrometer are the energy-sensitive radiation detector and the electronic devices that analyse the detector output signals, such as a pulse sorter (i.e., multichannel analyzer). Additional components may include signal amplifiers, rate meters, peak position stabilizers, and data handling devices. Detector Gamma spectroscopy detectors are passive materials that are able to interact with incoming gamma rays. The most important interaction mechanisms are the photoelectric effect, the Compton effect, and pair production. Through these processes, the energy of the gamma ray is absorbed and converted into a voltage signal by detecting the energy difference before and after the interaction (or, in a scintillation counter, the emitted photons