which prevents AC & DC ranges being selected simultaneously. For example, none of the DC ranges, current or voltage, can be connected unless the AC switch is set to its "DC" position. On a Model 8, this is the position with the AC switch arrow vertical. Similarly, to use the AC ranges, the DC switch must be set to its "AC" position. With the DC switch set to its "AC" position and the AC switch set to "DC", no current can flow through the instrument. However whenever any moving coil instrument is likely to be subjected to heavy shock in transit, it is good practice to damp the movement by short circuiting the moving coil using a heavy gauge wire connected across the terminals. On earlier Avometers, this may be done by short-circuiting the input terminals and selecting the most sensitive direct current range. The Model 8 Mark V, 6 & 7 were provided with an "OFF" position on the DC switch which both disconnected the meter's terminals and short-circuited the moving coil. AVOmeters designed from 1936 onwards were fitted with an overload cut-out operated by the moving coil frame hitting either forward or reverse sprung end stops. The Model 7 was the first type to use the end stop cut-out and it also featured an acceleration trip which, in the event of heavy overloads, could open the cut-out before the pointer had reached two-thirds of full scale. The acceleration cut-out was not however used in the Model 8. From the Mark III version, the Model 8 had further protection by a fuse on its resistance ranges and fuse protection was provided on all ranges of the Model 8 Marks 6 & 7. AVO multimeters were almost ubiquitous in British manufacturing and service industry, research and development and higher and further education. They were also widely used by utilities, government agencies and the British armed forces. A number of special versions were produced to British Admiralty and Air Ministry specifications and for other customers. The Model 8 Marks V, 6 & 7 were designed to meet a NATO specification and were standard issue to NATO services. Many commercial and military service manuals specified that values for measurements of current or voltage had been made with a Model 7 or Model 8 AVOmeter. Advertisements of the late 1930s compared the utility of the AVOmeter to the slide rule. Even nowadays it