Specific rotation is calculated using either of two equations, depending on whether the sample is a pure chemical to be tested or that chemical dissolved in solution. For pure liquids This equation is used: In this equation, α (Greek letter "alpha") is the measured rotation in degrees, l is the path length in decimeters, and ρ (Greek letter "rho") is the density of the liquid in g/mL, for a sample at a temperature T (given in degrees Celsius) and wavelength λ (in nanometers). If the wavelength of the light used is 589 nanometers (the sodium D line), the symbol “D” is used. The sign of the rotation (+ or −) is always given. ° For solutions For solutions, a slightly different equation is used: In this equation, α (Greek letter "alpha") is the measured rotation in degrees, l is the path length in decimeters, c is the concentration in g/mL, T is the temperature at which the measurement was taken (in degrees Celsius), and λ is the wavelength in nanometers. For practical and historical reasons, concentrations are often reported in units of g/100mL. In this case, a correction factor in the numerator is necessary: When using this equation, the concentration and the solvent may be provided in parentheses after the rotation. The rotation is reported using degrees, and no units of concentration are given (it is assumed to be g/100mL). The sign of the rotation (+ or −) is always given. If the wavelength of the light used is 589 nanometer (the sodium D line), the symbol “D” is used. If the temperature is omitted, it is assumed to be at standard room temperature (20 °C). For example, the specific rotation of a compound would be reported in the scientific literature as: (c 1.00, EtOH) Dealing with large and small rotations If a compound has a very large specific rotation or a sample is very concentrated, the actual rotation of the sample may be larger than 180°, and so a single polarimeter measurement cannot detect when this has happened (for example, the values +270° and −90° are not distinguishable, nor are the values 361° and 1°). In these cases, measuring the rotation at several different concentrations allows one to determine the true value. Another method would be to use shorter path-lengths to perform the measurements. In cases of very small or very large angles, one can also use the variation